a randomized multicenter clinical study on the high vacuum body cavity drainage system following open heart surgery

Conditions: Coronary Artery Disease; Heart Valve Disease

Summary: the purpose of this study is to compare a new high vacuum pressure chest drainage system (no-numo™) with the standard low vacuum pressure drainage system already in use in cardiac surgeries. the new system uses smaller diameter drainage tubing to remove blood from the chest after open heart surgery. this is less painful to the patient and should help the patient to breathe better after operation. preliminary data suggests that the high vacuum suction is effective and practical, and could actually reduce the amount of chest tube bleeding after surgery without compromising heart function. this new high vacuum drainage system was approved by the fda in 2002. subsequent testing at university of california, irvine indicated that this unique system may allow the surgeon to predict excessive postoperative bleeding before the patient is transferred out of the operating room, thus adding safety, convenience, and cost effectiveness to their use.

high vacuum chest drainage was once considered inadvisable because of a possible suction injury to soft tissues, such as the lungs or mediastinum. wakabayashi showed that excessive suction force generated by a high vacuum pressure was related to the surface area of the suction applied to the tissues and not the vacuum pressure itself1\&2. he postulated that suction injury to the lung would not occur if the suction force of a chest tube is below the capillary blood pressure of 30 - 35 torr2. based on this theory, wakabayashi developed a double-lumen chest drainage tube3 (trade name no-numo™) and its clinical use was approved by fda4 in 2002. a prototype 7-5fr double-lumen chest tube was successfully tested in laboratory animals and three patients1\&2 and the safety of high vacuum chest drainage was proven in a large number of patients undergoing uncomplicated thoracoscopic surgeries2. the unique design of the new drainage tube is the small diameter double lumen tube with hundreds of tiny holes that allows rapid drainage without damaging effects of high suction over a limited surface area. high vacuum suction is available on wall units already in place in the or and icu beds. to adopt this system for clinical use, a high vacuum gauge was needed (most don't go beyond 200 torr) to monitor suction, and a high pressure battery powered vacuum pump for transport was necessary, so that drainage was maintained and blood clotting in the tube avoided during the period of prolonged transport from the or and the setup in the icu. several brands of portable vacuum pumps on the market were evaluated before concluding that the vario™ pump (medela® inc.) was superior. the maximal vacuum pressure that vario pump can generate is 630 torr or 84 kpa which is equivalent to -857 cmh2o. the no-numo™ high vacuum body cavity drainage system consist of; (1) disposable no-numo™ body cavity drainage tubes, (2) disposable vario™ fluid management canisters and (3) vario™ portable vacuum pump. we evaluated this new system, using a 13fr triple-tube set for the mediastinal drainage, a 13fr single-tube for the left pleural drainage and two 13fr no-numo tubes for the drainage of the lower extremity subcutaneous wounds following the harvesting of the saphenous veins. the tubing was connected to a disposable liner of a medi-vac™ (cardinal health, inc.) suction canisters, which was connected to a vario pump at 300 torr (= 40 kpa) in patients following open heart surgeries.

Eligibility criteria:
inclusion criteria: 1. male or female 18 year-old or older 2. patients undergoing open heart surgery by means of a standard cardiopulmonary bypass 3. the surgical procedures accepted include: 1. coronary-artery-bypass grafting 2. valve replacement or repair, or a combination of both 3. excision of a left ventricular aneurysm 4. bentall procedure is included but we do not include those undergoing replacement of the aortic arch aneurysm, utilizing deep hypothermic circulatory arrest. 5. we include both first time operation and re-do operations. 6. patients with insulin dependent diabetes mellitus are included. those with chronic renal failure who are on a hemodialysis treatment will be included. 7. emergency cabg for unstable angina following acute myocardial infarction 4. patients must agreed to participate and sign an informed consent form before the surgical procedure exclusion criteria: 1. transmyocardial laser revascularization 2. open heart surgery for congenital heart diseases.

Keywords: open heart surgery cardiopulmonary bypass coronary artery bypass grafting chest tube valve replacement valve repair