pilot study of cyclaplex implant, an enhanced suture and button implant device in hallux valgus deformity

Conditions: Hallux Valgus

Summary: single center, open-label, single arm, pilot study designed to evaluate safety, tolerability and efficacy of cyclaplex device, a minimal invasive enhanced suture and button implant and surgery tools for correction of inter-metatarsal angle in subjects suffering from mild to moderate hallux valgus deformity

investigational therapy cycla orthopedics ltd has developed the cyclaplex , a minimally invasive enhanced type of suture and button implant. it is intended to: (a) reduce the intermetatarsal angle (ima) between 1mt and 2mt without osteotomy osteotomy b) reduce the risk of 2mt fractures (d) perform the implantation procedure in an easy and safe way. the cyclaplex predicated device is the "suture and button" type minitightrope. it is based on mini tightrope clinical experience, however with additional features to increase safety and functionality. the cyclaplex device comprise of two bone holders ('anchors ') connected by a metal cord. the cord is connected to the buttons by crimp method as part of the surgery procedures the bone holders are t shape screws that are screwed into the bones after drilling holes across the 2 metatarsals. the gap in the bone as a result of the drilling is filled by the enhanced t shape buttons (screws). a nut is residing internally as part of the 1st met anchor structure. the ima is reduced using specially designed tensioning tool cyclaplex device is indicated for reducing intermetatarsal angles in cases where ima is less than 200 eliminating the need for the traditional 1st metatarsal osteotomy. it used in conjunction with traditional release of the lateral mtp joint capsular structures and removal of the medial eminence as well as regular hva reduction procedures such as akin. the 2nd metatarsal fracture risk is reduced due multiple/ accumulated factors: (i) the hole drilled at 2nd metatarsal had been reduced significantly (ii) the hole is filled by conical screw and head. this firm anchoring method enable to distribute the horizontal forces applied during gate. (iii) the metal screw type device enables bone growth over it and strengthen the bone at that area. (iv) the connecting cord is housed inside the screw and has no direct touch with the bone, in both, 2nd and 1st metatarsals. the device is delivered sterilized ( steam sterilization) double bag residing in cardboard box. dedicated assistive tools for multiple use are provided as well housed in a special tray intended for surgery room steam sterilization. the tools comprise of aiming drilling guide,, screw driver for anchors insertion, tensioning tool for reduction of the ima, crimping tool, cutter, single use drill bits

Eligibility criteria:
inclusion criteria * adult males and female subjects, aged 20 to 75 years old, inclusive, at screening visit * subject weighing \<100 kg and body mass index (bmi) \<32 kg/m2 * subjects with confirm diagnosis of hv deformity based on radiography assessment of ima \> 12 º and less or equal to 20degree. * subjects who would need osteotomy of the 1st mt as part of their surgery. * able and willing to comply with the requirements of the protocol. * able to understand and sign written informed consent to participate in the study exclusion criteria * contraindication to implantation procedure * history of: diabetes mellitus,claudication vascular disease or known peripheral vascular disease,rheumatoid disease no palpable pulse (dp or pt),- gout or any systemic inflammatory arthropathy,osteoarthritis of first metatarsal,neurological conditions associated with spasticity,no prior operation at 1st and 2nd metatarsals * osteoporosis * secondary metatarsal complications including:fracture (past or present), aseptic necrosis or any deformity of 2nd mth,subject currently enrolled in or has not yet completed at least 30 days since ending * any condition that may jeopardize study participation (e.g. abnormal clinical or laboratory finding), the interpretation of study results or may impede the ability to obtain informed consent (e.g., mental condition)