clinical trial on the accuracy of blood pressure measurement by ye990 medical automatic electronic sphygmomanometer

Conditions: Blood Pressure

Summary: this study was a single-center, open-ended, self-controlled clinical study . same armsequential method was used to measure the blood pressure of the subjects，inclusion of ≥85 subjects aged 12 years or older who met the enrollment criteria.in accordance with the "association for the advancement of medical instrumentation/european society of hypertension/international organization for standardization (aami/esh/iso) universal standard (iso 81060-2:2018/amd 1:2020)",the purpose of this study was to compare the blood pressure and pulse rate measured by an electronic sphygmomanometer with those measured by a reference sphygmomanometer (an upper-arm noninvasive auscultatory sphygmomanometer) and by palpation of the radial artery, and then to evaluate the accuracy and safety of the measurement of systolic blood pressure, diastolic blood pressure, and pulse rate by a fully automated electronic sphygmomanometer for medical use (model no. ye990), which is manufactured by jiangsu yuyue medical equipment co.

this study was a single-center, open-ended, self-controlled clinical study . same armsequential method was used to measure the blood pressure of the subjects，inclusion of ≥85 subjects aged 12 years or older who met the enrollment criteria.in accordance with the "aami/esh/iso universal standard (iso 81060-2:2018/amd 1:2020)",the purpose of this study was to compare the blood pressure and pulse rate measured by an electronic sphygmomanometer with those measured by a reference sphygmomanometer (an upper-arm noninvasive auscultatory sphygmomanometer) and by palpation of the radial artery, and then to evaluate the accuracy and safety of the measurement of systolic blood pressure, diastolic blood pressure, and pulse rate by a fully automated electronic sphygmomanometer for medical use (model no. ye990), which is manufactured by jiangsu yuyue medical equipment co. (one） criteria for admission inclusion criteria subjects should meet all of the following requirements for inclusion in the study: 1. both sexes; at least 30% of the subjects are male and at least 30% of the subjects are female. 2. subject's age \> 12 years. 3. arm circumference distribution: 1. at least 20% of the subjects should have arm circumference within each quarter interval of the total arm circumference range; 2. at least 10% of the subjects should have arm circumference within the upper one-eighth interval of the total arm circumference range; 3. at least 10% of the subjects shall have an arm circumference within the lower one-eighth interval of the total arm circumference range. 4. blood pressure distribution: 1. at least 5% of subjects shall have a systolic blood pressure ≤100 mmhg (13.33 kpa); 2. at least 5% of subjects shall have a systolic blood pressure ≥ 160 mmhg (21.33 kpa); 3. at least 20% of subjects should have a systolic blood pressure ≥ 140 mmhg (18.66 kpa); 4. diastolic blood pressure should be ≤60 mmhg (8.0 kpa) in at least 5% of subjects; 5. diastolic blood pressure should be ≥100 mmhg (13.33 kpa) in at least 5% of subjects; (f) at least 20% of subjects should have a diastolic blood pressure ≥85 mmhg (11.33 kpa); 5. subjects must be willing and able to comply with the study procedures by signing an informed consent form after being informed about the study. consent form after understanding the contents of the study. (6) good compliance and ability to complete the clinical trial. 2. exclusion criteria subjects who fulfill any of the following requirements cannot be included in the study: 1. mental illness, impaired consciousness, or other conditions that prevent them from cooperating with the study; 2. require hemodialysis; 3. taking anticoagulant drugs for cardiovascular diseases; 4. pregnant and lactating women; 5. patients with cardiac arrhythmia; 6. patients with peripheral arterial disease and diseases that may affect arterial compliance; 7. other conditions that are considered by the physician to be unsuitable for enrollment. （two）evaluation methods 2.1. evaluation of effectiveness according to the standard requirements of "aami/esh/iso universal standard (iso 81060-2:2018/amd 1:2020)", the relationship between the mean value of the error and the standard deviation, the consistency between the measured and reference blood pressure values, and the relationship between the arm circumference and the measurement error should be evaluated. 1. error mean and standard deviation calculate the mean and standard deviation of the difference (i.e. error) between the measured and reference blood pressure values, and the result shall meet the requirements of iso 81060-2:2018/amd 1:2020 5.4.1.2. 2. agreement between measured and reference blood pressure values (bland-altman analysis) for each measurement position, a scatter plot of the difference between the investigator and the device under test is obtained for each comparison of the investigator and the device under test, and the values of systolic and diastolic blood pressures shall be plotted separately. 3. relationship between arm circumference and measurement error (bland-altman analysis) the relationship between upper arm circumference and measurement error is shown by plotting a scatter plot. 4. relative pulse rate error descriptive statistics of relative error in pulse rate. 2.2. safety evaluation according to the "code for quality management of clinical trials of medical devices" (2022) iii, medical devices should be collected and recorded when conducting clinical trials. when conducting clinical trials of medical devices, the occurrence of adverse events and defects of the devices in the course of use by the subjects should be collected and recorded. 1. incidence of adverse events; 2. incidence of device defects. 2.3. evaluation of product performance clinical use performance evaluation is mainly for product stability evaluation, the specific evaluation criteria are as follows, the following items if evaluation if the evaluation of the following items is "yes", it is necessary to specifically describe the cause of the failure: evaluation item 1: electronic sphygmomanometer power on and off yes: cannot be turned on and/or off normally; no: can be turned on and off normally. evaluation item ii: whether the device automatically shuts off during the checking process yes: automatic shutdown during the inspection; no: there is no automatic shutdown during the inspection. evaluation item 3: whether there is any abnormal interruption due to the machine during the inspection process. yes: abnormal interruptions occurred during the inspection process due to the machine; no: no abnormal interruptions occurred during the inspection. evaluation item 4: whether inflation and exhaust failures occurred during the inspection process yes: inflation and/or exhaust failure occurred during the inspection; no: no inflation and exhaust failure occurred during the inspection. evaluation item 5: whether air leakage occurred during the inspection process yes: air leakage occurred; no: no air leakage occurred. （three）test medical device and control medical device 1. test medical equipment (blood pressure monitor) name of device: medical automatic electronic sphygmomanometer (model: ye990) registrant name: jiangsu yuyue medical equipment co. registration certificate no.: su meizhi quasi 20222071271 2. control medical devices (reference sphygmomanometer) name of device: sphygmomanometer - stethoscope health care box (model: a type) registrant name: jiangsu yuyue medical equipment co. certificate of registration no.: sumechiqi 20152070947 （four）statistical analysis 1. general principles descriptive statistical analysis: qualitative indicators are described by frequency tables and percentages; quantitative indicators are described by mean, standard deviation, minimum, median, maximum, upper quartile (q1) and lower quartile (q3). 2. content of statistical analysis 1. the number of subjects enrolled, dislodged, and completed cases, and a list of reasons why subjects withdrew from the study. 2. demographic and vital sign data of the subjects quantitative data were analyzed by describing their number of cases, mean, standard deviation, minimum, median, maximum, and upper quartile. maximum, upper quartile (q1), and lower quartile (q3); qualitative data were analyzed by describing the number of cases and their proportion of the total number of subjects. the qualitative data were analyzed by describing the number of cases and their percentages. 3. validity analysis error mean and standard deviation according to iso 81060-2:2018/amd 1:2020 5.4.1.2, the error mean and standard deviation deviation should meet the following evaluation criteria: criterion 1: calculate the mean and standard deviation of the errors of the 255 groups of measured and reference blood pressure values for 85 subjects. criterion 1: calculate the mean and standard deviation of the errors of 255 sets of measured and reference blood pressure values of 85 subjects, and compare the systolic and diastolic blood pressures separately, with the mean value of the error not exceeding ±5 mmhg and the standard deviation not exceeding 8 mmhg. criterion 2: comparison of the mean of the three sets of errors for each subject, comparing systolic and diastolic blood pressures separately. agreement between measured and reference blood pressure values (bland-altman analysis) according to iso 81060-2:2018/amd 1:2020, a scatter plot of the difference between the reference device and the device under test should be plotted for each occasion, and the difference between systolic and diastolic blood pressure should be plotted separately. x-axis indicates the mean value of the 2 methods of measurement for each subject, and y-axis indicates the difference between the 2 methods of measurement for each subject. relationship between arm circumference and measurement error (bland-altman analysis) plot the results of all subjects by arm circumference according to iso 81060-2:2018/amd 1:2020 5.1.4 d. the x-axis shows the actual arm circumference of the subject and the y-axis shows the difference between the measured blood pressure value and the reference blood pressure value. pulse rate error the relative error of the pulse rate measurement is counted.

Eligibility criteria:
inclusion criteria: subjects should meet all of the following requirements for inclusion in the study: 1. both sexes; at least 30% of the subjects are male and at least 30% of the subjects are female. 2. subject's age \> 12 years. 3. arm circumference distribution: 1. at least 20% of the subjects should have arm circumference within each quarter interval of the total arm circumference range; 2. at least 10% of the subjects should have arm circumference within the upper one-eighth interval of the total arm circumference range; 3. at least 10% of the subjects shall have an arm circumference within the lower one-eighth interval of the total arm circumference range. 4. blood pressure distribution: 1. at least 5% of subjects shall have a systolic blood pressure ≤100 mmhg (13.33 kpa); 2. at least 5% of subjects shall have a systolic blood pressure ≥ 160 mmhg (21.33 kpa); 3. at least 20% of subjects should have a systolic blood pressure ≥ 140 mmhg (18.66 kpa); 4. diastolic blood pressure should be ≤60 mmhg (8.0 kpa) in at least 5% of subjects; 5. diastolic blood pressure should be ≥100 mmhg (13.33 kpa) in at least 5% of subjects; (f) at least 20% of subjects should have a diastolic blood pressure ≥85 mmhg (11.33 kpa); 5. subjects must be willing and able to comply with the study procedures by signing an informed consent form after being informed about the study. consent form after understanding the contents of the study. 6. good compliance and ability to complete the clinical trial. exclusion criteria: subjects who fulfill any of the following requirements cannot be included in the study: (1) mental illness, impaired consciousness, or other conditions that prevent them from cooperating with the study; (2) require hemodialysis; (3) taking anticoagulant drugs for cardiovascular diseases; (4) pregnant and lactating women; (5) patients with cardiac arrhythmia; (6) patients with peripheral arterial disease and diseases that may affect arterial compliance; (7) other conditions that are considered by the physician to be unsuitable for enrollment. -

Keywords: accuracy systolic blood pressure diastolic blood pressure pulse rate 12 years of age or older automatic electronic sphygmomanometers