Clinical Evaluation of Approved and Investigational Contact Lenses Using New UV Additive
Phase: NA
Conditions: Visual Acuity
Sponsor: Johnson & Johnson Vision Care, Inc.

 
 
Johnson & Johnson Vision Care 
7500 Centurion Parkway 
Jacksonville, FL 32256 
Protocol Number: CR-5638 
Version:  6.0, Amendment 5.0 
Date: 13 October 2016 
Clinical Study Protocol 
Long-term Evaluation of 
UV Blocker
Distribution 
Ms. Leilani Sonoda 
CONFIDENTIAL 
This document contains confidential information, which should not be copied, referred to, released or 
published without written approval from Johnson and Johnson Vision Care.  The information may not be 
disclosed to others except to the extent necessary to obtain Institutional Review Board/Independent Ethics 
Committee approval and informed consent, or as required by International, Federal and State Laws, as 
applicable. Persons to whom this information is disclosed must be informed that this information is privileged 
and confidential and that it should not be further disclosed without the written permission of Johnson and 
Johnson Vision Care.  Any supplemental information that may be added to this document is also confidential 
and proprietary to Johnson and Johnson Vision Care and must be kept in confidence in the same manner as 
the contents of this document. 
Key Words  
senofilcon 
Photochromic 
Physiological Response 
Subjective Performance 
Fitting Characteristics 
Dispensing 
Daily Wear 
 
 
 
 
 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 1 of 194

TABLE OF CONTENTS 
 
1.1 
PROTOCOL TITLE, NUMBER, DATE .............................................................................................................. 5 
1.2 
NAME AND ADDRESS OF SPONSOR ............................................................................................................ 5 
1.3 
AUTHORIZED SIGNATURES ......................................................................................................................... 5 
1.4 
MEDICAL MONITOR .................................................................................................................................... 6 
1.5 
INVESTIGATOR(S) SIGNATURE PAGE ........................................................................................................... 7 
1.6 
ESTIMATED REPORT DATE .......................................................................................................................... 8 
1.7 
CHANGE HISTORY ....................................................................................................................................... 8 
1.8   
PROTOCOL SYNOPSIS ................................................................................................................................. 8 
2.1 
NAME AND DESCRIPTION OF INVESTIGATIONAL PRODUCTS .................................................................... 13 
2.2 
SUMMARY OF FINDINGS FROM NONCLINICAL STUDIES ........................................................................... 14 
2.3 
SUMMARY OF KNOWN RISKS AND BENEFITS TO HUMAN SUBJECTS ........................................................ 14 
2.4 
DESCRIPTION OF TRIAL TREATMENTS ....................................................................................................... 14 
2.5 
STATEMENT OF COMPLIANCE TO PROTOCOL, GCP, AND APPLICABLE REGULATORY GUIDELINES ............ 14 
2.6 
DESCRIPTION OF POPULATION TO BE STUDIED, ENROLLMENT TARGETS, AND STUDY DURATION ........... 14 
2.7 
RELEVANT LITERATURE REFERENCES AND PRIOR DATA ............................................................................ 15 
3.1 
DESCRIPTION OF OBJECTIVES AND PURPOSE ........................................................................................... 15 
4.1 
PRIMARY AND SECONDARY ENDPOINTS .................................................................................................. 15 
4.1.1 
PRIMARY EFFICACY AND SAFETY ENDPOINTS: ..................................................................................................... 15 
4.1.2 
SECONDARY EFFICACY AND SAFETY ENDPOINTS: ................................................................................................. 15 
4.1.3 
PRIMARY HYPOTHESES: .................................................................................................................................. 15 
4.1.4 
SECONDARY HYPOTHESES: .............................................................................................................................. 16 
4.2 
INCLUSION CRITERIA ................................................................................................................................ 16 
4.3 
EXCLUSION CRITERIA ................................................................................................................................ 16 
4.4 
STUDY DESIGN, TIME AND EVENTS SCHEDULE, FLOWCHART .................................................................... 17 
4.4.1 
TIME AND EVENT SCHEDULE .................................................................................................................... 19 
4.4.2 
OPTIONAL FLOWCHART ........................................................................................................................... 20 
4.5 
RANDOMIZATION AND MASKING ............................................................................................................ 21 
4.6 
WEAR AND REPLACEMENT SCHEDULES, INCLUDING FORM, PACKAGING AND LABELING......................... 21 
4.7 
DETAILED STUDY PROCEDURES ................................................................................................................ 21 
4.7.1 
SEQUENCE OF EVENTS ............................................................................................................................. 21 
VISIT 1:  SCREENING ................................................................................................................................................. 21 
VISIT 1:  BASELINE INFORMATION ............................................................................................................................ 22 
VISIT 1:  LENS ASSIGNMENT #1 ...................................................................................................................................... 23 
VISIT 1:  FITTING #1 .................................................................................................................................................. 24 
VISIT 1:  TRIAL FITTING LENS MODIFICATION ........................................................................................................... 24 
VISIT 1:  FITTING #1, CONTINUED ............................................................................................................................. 25 
VISIT 1:  DISPENSING #1 ........................................................................................................................................... 26 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 2 of 194

VISIT 2:  7-DAY FOLLOW-UP ..................................................................................................................................... 26 
VISIT 3:  14-DAY FOLLOW-UP ................................................................................................................................... 28 
VISIT 3:  LENS ASSIGNMENT #2 ................................................................................................................................ 31 
VISIT 3:  FITTING #2 .................................................................................................................................................. 31 
VISIT 3:  TRIAL FITTING LENS MODIFICATION ........................................................................................................... 31 
VISIT 3:  FITTING #2, CONTINUED ............................................................................................................................. 32 
VISIT 3:  DISPENSING #2 ........................................................................................................................................... 33 
VISIT 4:  28-DAY FOLLOW-UP ................................................................................................................................... 34 
VISIT 4:  LENS ASSIGNMENT #3 ...................................................................................................................................... 36 
VISIT 4:  FITTING #3 .................................................................................................................................................. 36 
VISIT 4:  TRIAL FITTING LENS MODIFICATION ........................................................................................................... 36 
VISIT 4:  FITTING #3, CONTINUED ............................................................................................................................. 37 
VISIT 4:  DISPENSING #3 ........................................................................................................................................... 38 
VISIT 5:  56-DAY FOLLOW-UP ................................................................................................................................... 39 
VISIT 5:  LENS ASSIGNMENT #4 ...................................................................................................................................... 41 
VISIT 5:  FITTING #4 .................................................................................................................................................. 41 
VISIT 5:  TRIAL FITTING LENS MODIFICATION ........................................................................................................... 41 
VISIT 5:  FITTING #4, CONTINUED ............................................................................................................................. 42 
VISIT 5:  DISPENSING #4 ........................................................................................................................................... 43 
VISIT 6:  84-DAY FOLLOW-UP ................................................................................................................................... 44 
VISIT 6:  HABITUAL LENS ASSIGNMENT #1 ............................................................................................................... 46 
VISIT 6:  HABITUAL LENS FITTING #1......................................................................................................................... 46 
VISIT 6:  HABITUAL LENS RELEASE #1........................................................................................................................ 47 
VISIT 7:  7-DAY FOLLOW-UP  HABITUAL LENS ........................................................................................................... 48 
VISIT 8:  14-DAY FOLLOW-UP  HABITUAL LENS ......................................................................................................... 50 
VISIT 8:  FINAL EVALUATION ..................................................................................................................................... 51 
4.8 
DISCONTINUATION CRITERIA ................................................................................................................... 52 
4.9 
ACCOUNTABILITY PROCEDURES FOR STUDY ARTICLES ............................................................................. 52 
4.10 
PROCEDURES FOR MAINTAINING AND BREAKING RANDOMIZATION CODES ........................................... 53 
4.11  REPORTING  PRODUCT QUALITY COMPLAINTS ............................................................................................ 54 
5.1 
WITHDRAWAL CRITERIA ........................................................................................................................... 54 
6.1 
PRESTUDY AND CONCOMITANT THERAPY ................................................................................................ 55 
6.2 
MONITORING TREATMENT COMPLIANCE ................................................................................................. 55 
6.3 
UNSCHEDULED VISITS ............................................................................................................................... 56 
7.1 
EFFICACY PARAMETERS ............................................................................................................................ 56 
7.2 
METHODS FOR ASSESSING, RECORDING, AND ANALYZING EFFICACY ....................................................... 56 
8.1 
SAFETY PARAMETERS ............................................................................................................................... 56 
8.2 
ADVERSE EVENTS ..................................................................................................................................... 57 
8.3 
ADVERSE EVENT DEFINITIONS .................................................................................................................. 57 
8.4 
METHODS FOR ASSESSING, RECORDING AND ANALYZING SAFETY ........................................................... 59 
8.5 
ADVERSE EVENTS FOLLOW-UP ................................................................................................................. 61 
9.1 
STATISTICAL METHODS TO BE EMPLOYED ................................................................................................ 61 
9.2 
NUMBER OF SUBJECTS BY SITE AND JUSTIFICATION FOR SAMPLE SIZE .................................................... 64 
9.3 
LEVEL OF STATISTICAL SIGNIFICANCE ....................................................................................................... 64 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 3 of 194

9.4 
CRITERIA FOR STUDY TERMINATION ........................................................................................................ 64 
9.5 
PROCEDURE FOR ACCOUNTING FOR MISSING, UNUSED, AND SPURIOUS DATA ....................................... 64 
9.6 
PROCEDURE FOR REPORTING DEVIATIONS FROM STATISTICAL PLAN ...................................................... 64 
9.7 
EVALUABLE SUBJECTS............................................................................................................................... 64 
10.1 
ELECTRONIC CASE REPORT FORMS/DATA COLLECTION ............................................................................ 65 
10.2 
SOURCE DOCUMENTATION ...................................................................................................................... 65 
10.3 
ACCESS TO SOURCE DATA/DOCUMENTS .................................................................................................. 66 
10.4 
CONFIDENTIALITY OF INFORMATION ....................................................................................................... 66 
11.1 
DATA QUALITY ASSURANCE ..................................................................................................................... 66 
12.1 
STUDY-SPECIFIC DESIGN CONSIDERATIONS .............................................................................................. 66 
12.2 
INVESTIGATOR RESPONSIBILITY ............................................................................................................... 67 
12.3 
INDEPENDENT ETHICS COMMITTEE OR INSTITUTIONAL REVIEW BOARD (IEC/IRB) .................................. 67 
12.4 
INFORMED CONSENT ............................................................................................................................... 68 
12.5 
PRIVACY OF PERSONAL DATA ................................................................................................................... 68 
13.1 
DATA HANDLING AND RECORD KEEPING .................................................................................................. 69 
14.1 
FINANCIAL CONSIDERATIONS ................................................................................................................... 70 
15.1 
PUBLICATION ........................................................................................................................................... 70 
16.1 
PATIENT REPORTED OUTCOMES (STUDY QUESTIONNAIRES) .................................................................... 70 
16.2 
CLINICAL TECHNICAL PROCEDURES 
 ............................................................................................... 103 
 
LIMBAL & CONJUNCTIVAL (BULBAR) REDNESS ...................................................................................... 104 
 
EXPANDED SODIUM FLUORESCEIN CORNEAL STAINING ........................................................................... 112 
 
LENS FITTING CHARACTERISTICS ......................................................................................................... 117 
 
SUBJECT REPORTED OCULAR SYMPTOMS/PROBLEMS ............................................................................ 124 
 
FRONT AND BACK SURFACE LENS DEPOSIT GRADING PROCEDURE ............................................................ 126 
 
DETERMINATION OF DISTANCE SPHEROCYLINDRICAL REFRACTIONS ........................................................... 132 
 
BIOMICROSCOPY SCALE .................................................................................................................... 139 
 
DISTANCE AND NEAR VISUAL ACUITY EVALUATION ................................................................................ 145 
 
DISTANCE LOGMAR VISUAL ACUITY MEASUREMENT PROCEDURE ........................................................... 150 
 
WHITE LIGHT LENS SURFACE WETTABILITY .......................................................................................... 154 
16.3 
PATIENT INSTRUCTION GUIDE (INVESTIGATIONAL PRODUCT) ................................................................ 156 
16.4 
PACKAGE INSERT (APPROVED PRODUCT) ............................................................................................... 157 
16.5 
APPENDIX A:  GUIDE TO STARTING CONTACT LENS POWERS ................................................................. 186 
16.6 
APPENDIX B:  IRIS COLOR SCALE ............................................................................................................. 187 
16.7 
APPENDIX C:  TEST ARTICLE ACCOUNTABILITY IN THE EDC SYSTEM ........................................................ 188 
17.1 
LIST OF ABBREVIATIONS ......................................................................................................................... 194 
 
 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 4 of 194

1.1 
PROTOCOL TITLE, NUMBER, DATE 
TITLE:  Long-term Evaluation of 
 UV Blocker 
PROTOCOL NUMBER:  CR-5638 
VERSION:  6.0, Amendment 5.0 
DATE:  13 October 2016 
1.2 
NAME AND ADDRESS OF SPONSOR 
Johnson & Johnson Vision Care 
7500 Centurion Parkway, Jacksonville, FL 32256 
1.3 
AUTHORIZED SIGNATURES 
The signature below constitutes the approval of this protocol and the attachments, and provides the necessary 
assurances that this trial will be conducted according to all stipulations of the protocol, including all statements 
regarding confidentiality, and according to local legal and regulatory requirements and applicable U.S. federal 
regulations, ICH guidelines, and the Declaration of Helsinki. 
 
 
Author: 
See Electronic Signature Report 
 
 
 
Name:  John R. Buch, O.D., M.S., F.A.A.O. 
Title:  Principal Research Optometrist 
 
DATE 
 
 
Biostatistician: 
See Electronic Signature Report 
 
 
 
Name:  Jessica Cannon, M.S. 
Title:  Biostatistician II 
 
DATE 
 
 
R&D Platform Lead: 
See Electronic Signature Report 
 
 
 
Name:  Zohra Fadli, Ph.D. 
Title:  Principal Scientist, Platform Lead 
 
DATE 
 
 
Clinical Operations: 
See Electronic Signature Report 
 
 
 
Name:  Hongzhi Guo 
Title:  Clinical Operations Manager 
 
DATE 
 
 
Reviewer : 
See Electronic Signature Report 
 
 
 
Name:  Randy Paulk 
Title:  Clinical Project Manager, Data and System 
 
DATE 
 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 5 of 194

1.4 
MEDICAL MONITOR 
NAME:  John R. Buch, O.D., M.S., F.A.A.O. 
TITLE: Principal Research Optometrist 
ADDRESS: 7500 Centurion Parkway, Suite 100, Jacksonville, FL 32256 
24 HOUR CONTACT TELEPHONE #: 
 
 
The Medical Monitor should be notified by the clinical site in writing and by telephone within 24 hours of learning 
of a Serious Adverse Event.  The Medical Monitor may be contacted during business hours for adverse event 
questions.  General study related questions should be directed towards your assigned clinical research associate. 
 
The clinical site may be supplied an Adverse Event form to complete regarding the adverse event evaluation. 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 6 of 194

1.5 
INVESTIGATOR(S) SIGNATURE PAGE 
 
The Principal Investigator is responsible for ensuring that all study site personnel, including sub-investigators and 
other staff members, adhere to all ICH regulations and GCP guidelines regarding clinical trials during and after 
study completion. 
 
I have read and understand the protocol specified above and agree on its content.  I agree to conduct this study 
according to this protocol and GCP and ICH guidelines, the Declaration of Helsinki, and the pertinent individual 
country laws/regulations and to comply with its obligations, subject to ethical and safety considerations. I shall not 
disclose the information contained in this protocol or any results obtained from this study without written 
authorization. 
 
 
 
Principal Investigator: 
 
 
 
 
Signature 
 
 
 
Date 
 
Name (Printed) 
 
 
 
 
 
Institution Name: 
 
 
 
 
 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 7 of 194

Randomization and Dispensing:  Subjects will be randomly fit with either the investigational lenses on both eyes, 
or the control lens on both eyes.  After three months of wearing the study lenses, all subjects will return to their 
habitual lenses for two weeks.  
 
Principal Investigators:  TBD 
 
 
Study Sites: 
 External, U.S. 
Microbiology or Other Testing Laboratory:  NA 
Phase or Type of Study:  Feasibility B 
Primary Endpoint:  
The primary endpoints are the safety and efficacy of the investigational product measured every 2 or 4 weeks over 
a 12 week period. 
x
Safety:  Biomicroscopy findings 
x
Efficacy:  Distance monocular logMAR acuity 
x
Efficacy: Eyestrain caused by glare 
 
 
Secondary Endpoint:  
The secondary endpoints are the safety and efficacy of the investigational product measured every 2 or 4 weeks 
over a 12 week period 
x
Safety:  Ocular symptoms 
x
Efficacy:  Wearing time 
 
Other Observations: 
x
Subjective performance as measured by the 
 and lens Preference 
Study Design: This study is a randomized, 8-visit, partially subject-masked (Control arm only), 14-weeks dispensing 
trial.  The study  lenses will be worn as daily wear (DW) for a period of 12 weeks, with follow-up visits occurring 
after 1, 2, 4, 8 and 12 weeks.  Afterwards, the habitual lenses will be worn for a period of two weeks with weekly 
visits.  The lenses are expected to be worn at least five (5) days per week for at least six (6) hours per day worn. 
Sample Size:  Approximately 60 test subjects and approximately 60 control subjects will be enrolled with the intent 
of completing with 50 cohorts in each group.  A replacement subject may be enrolled if a subject discontinues from 
the study prematurely; the decision whether to enroll replacement subjects will be made by the joint agreement of 
the Investigator and Sponsor.   
Screening:  Healthy male and femaůĞǀŽůƵŶƚĞĞƌƐ;шϭϴǇĞĂƌƐ) will be screened as per criteria outlined below. 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 9 of 194

Study Population Characteristics:  Participant subjects will be at least 18 years of age or older who fulfill the 
following inclusion/exclusion criteria: Subjects should meet all study Inclusion Criteria as outlined below: 
1. The subject must read and sign the Informed Consent form. 
2. The subject must appear able and willing to adhere to the instructions set forth in this clinical protocol. 
3. Healthy adult males or females age t 18 years of age with signed informed consent.  Eligible presbyopes 
will be those that wear full distance contact lenses in both eyes, then wear reading glasses over them. 
4. The subject’s optimal vertexed spherical equivalent distance correction must be between -1.00 and  
– 6.00D. 
5. The subject’s refractive cylinder must be ч 1.00D in each eye. 
6. The subject must have visual acuity best correctable to 20/25+3 or better for each eye. 
7. Subjects should own a wearable pair of spectacles.  
8. The subject is a current spherical soft contact lens wearer (defined as a minimum of 6 hours of DW per 
day, at least 5 days per week, for a minimum of 1 month prior to the study) and willing to wear the study 
lenses on a similar basis. 
9. Subjects must be able and willing to wear the study lenses at least 6 hours a day, a minimum of 5 days per 
week 
10. The subject must have normal eyes (i.e., no ocular medications or infections of any type). 
 
 
Subjects meeting any of the following Exclusion Criteria will not be eligible to participate in the study:  
1. Currently pregnant or lactating (subjects who become pregnant during the study will be discontinued). 
2. Any systemic disease, autoimmune disease, or use of medication, which may interfere with contact lens 
wear. This may include, but not be limited to, diabetes, hyperthyroidism, recurrent herpes simplex/zoster, 
Sjögren’s syndrome, xerophthalmia, acne rosacea, Stevens-Johnson syndrome, and immunosuppressive 
diseases or any infectious diseases (e.g. hepatitis, tuberculosis). 
3. Use of any of the following medications within 1 week prior to enrollment: oral retinoid isotretinoin (e.g. 
Accutane), oral tetracyclines, topical scopolamine, oral (e.g. Seldane, Chlor-Trimeton, and Benadryl) and 
ophthalmic antihistamines, oral phenothiazines (e.g., Haldol, Mellaril, Thorazine, Elavil, Pamelor, 
Compazine), oral and ophthalmic Beta-adrenergic blockers (e.g., Propranolol, Timolol, and Practolol), 
systemic steroids, and any prescribed or over the counter (OTC) ocular medication. 
4. Entropion, ectropion, extrusions, chalazia, recurrent styes, glaucoma, history of recurrent corneal 
erosions, aphakia, or moderate or above corneal distortion by keratometry. 
5. Any previous, or planned, ocular or interocular surgery (e.g.., radial keratotomy, PRK, LASIK, etc.). 
6. Any grade 2 or greater slit lamp findings (e.g.., edema, corneal neovascularization, corneal staining, tarsal 
abnormalities, conjunctival injection) on the FDA classification scale, any previous history or signs of a 
contact lens-related corneal inflammatory event (e.g.., past peripheral ulcer or round peripheral scar), or 
any other ocular abnormality that may contraindicate contact lens wear. 
7. Any known hypersensitivity or allergic reaction to Optifree®Puremoist® multi-purpose care solution or 
Eye-Cept® rewetting drop solution 
8. Any ocular infection, allergy or clinically significant ocular disease (e.g. corneal edema, uveitis, severe 
keratoconjunctivitis sicca, ocular hypertension), or ocular conditions (e.g. strabismus), which might 
interfere with the study. 
9. Any corneal distortion resulting from previous hard or rigid gas permeable contact lens wear. 
10. Toric, extended wear, monovision or multi-focal contact lens correction. 
11. Participation in any contact lens or lens care product clinical trial within 30 days prior to study enrollment. 
12. History of binocular vision abnormality or strabismus. 
13. Employee or relative of employees of sponsor or investigational clinic (e.g., Investigator, Coordinator, 
Technician) 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 11 of 194

Schedule of Events: 
 
Procedure 
Baseline 
Trial Fit & Dispense 
Follow-up  
1, 2, 3, 4, 5, 6, 7 
Unsched 
Exit 
Visit 
1 
1 
2, 3, 4, 5, 6, 7, 8 
PRN 
8 
Informed consent 
3 
- 
- 
- 
- 
Eligibility screening 
3 
- 
- 
- 
- 
PRO Baseline Questionnaires 
3 
- 
- 
- 
- 
Subject demographics 
3 
- 
- 
- 
- 
General health and medication history 
3 
- 
- 
- 
- 
Subject’s own contact lens information 
3 
- 
- 
- 
- 
Habitual lens care 
3 
- 
- 
- 
- 
Visual acuity (Snellen) 
3 
- 
- 
- 
3 
Spherocylindrical refraction and BVA 
3 
- 
- 
3 
3 
Slit lamp biomicroscopy 
3 
- 
3 
3 
- 
Keratometry 
3 
- 
- 
- 
3 
Trial fitting lens information 
- 
3 
- 
- 
- 
Lens Damage 
- 
3 
- 
- 
- 
Distance spherical over-refraction 
- 
3 
3 
- 
- 
Lens modification 
- 
3 
- 
- 
- 
Symptoms 
- 
3 
3 
3 
- 
PRO dispensing questionnaires 
- 
3 
- 
- 
- 
Visual acuity (logMAR) 
- 
3 
3 
3 
- 
Lens fitting assessment 
- 
3 
3 
* 
- 
Lens dispensing information and criteria 
- 
3 
- 
- 
- 
Patient instructions 
- 
3 
- 
- 
- 
Lens information 
- 
- 
3 
3 
- 
Compliance 
- 
- 
3 
3 
- 
Wearing times 
- 
- 
3 
3 
- 
PRO follow-up questionnaires 
- 
- 
3 
* 
- 
Surface characteristics 
- 
- 
3 
* 
- 
Chief complaint, diagnosis, treatment 
- 
- 
- 
3 
- 
 
* If wearing lenses 
 
Disallowed Medications: 
Use of any prescription or over-the-counter (OTC) medications that may affect contact lens wear from 24 hours 
prior to receiving the study product through the study period of a14 weeks.  Habitual medications taken by 
successful soft contact lens wearers are considered acceptable.  Note that habitual medications should be taken 
throughout the study period. 
Stopping Rules:   
The occurrence of one or more Unanticipated Adverse Device Effect (UADE), or any SAE where relationship to 
study agent cannot be ruled out, will result in prompt investigation and possible stopping of further dispensing 
investigational product.  In the event of a UADE or SAE, the Sponsor Medical Monitor may unmask the treatment 
regimen of subject(s) and may discuss this with the Investigator before any further subjects are enrolled. 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 12 of 194

Clinical Safety: 
Assessments by the Investigator will occur at all study visits and will include review of Adverse Events and 
Concomitant Medications.  A Safety Review Board (SRB) will be formed in association with this study.  The SRB will 
include the medical monitor and clinical research manager (CRM).  The medical monitor and CRM will review and 
evaluate all events and will determine if the remaining subjects assigned to that treatment group may proceed 
with treatment. 
Laboratory Review: 
The collection of samples is not required in this study.  
Safety Review Board: 
A Safety Review Board (SRB) will be formed in association with this study.  The SRB will include the medical 
monitor and clinical research manager (CRM).  The medical monitor and CRM will review and evaluate all events 
and will determine if the remaining subjects assigned to that treatment group may proceed with treatment.  
Statistical Methods: 
All data summaries and statistical analyses will be performed using the SAS software Version 9.4 or higher (SAS 
Institute, Cary, NC). Throughout the analysis of data, the results for each subject/eye will be used when available 
for summarization and statistical analysis. Unscheduled visits will be summarized separately and excluded from the 
analysis.  
2.1 
NAME AND DESCRIPTION OF INVESTIGATIONAL PRODUCTS 
 The following contact lenses will be used in this study: 
 
 
Control (Oasys) 
Test 
Manufacturer 
JJVCI 
JJVCI 
Lens Design 
PHN130 
ECL100 
Production Run Number 
Varies 
 
Packaging Form 
Sterile blister pack 
Sterile blister pack 
Nominal Water Content 
38 
38 
Nominal Dk (edge corrected) 
103 
103 
Nominal Modulus 
101 
101 
Inversion Indicator 
123 
123 
Nominal Base Curve/Diameter @ 22 °C (mm) 
8.4 / 14.0 
8.4 / 14.0 
Nominal Center Thickness @ -3.00 D (mm) 
0.070 
0.085 
Nominal Powers (D)
-1.00 to -6.00 in 0.25 steps 
-1.00 to -6.00 in 0.25 steps 
Norbloc UV Blocker 
Yes 
Yes 
 UV Blocker 
No 
Yes 
Example of Foil 
 
 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 13 of 194

lenses for approximately 12 weeks on a daily wear (DW) basis, then wear their habitual lenses for approximately 2 
weeks, for a total study duration of approximately 14 weeks (98 days) per subject. 
2.7 
RELEVANT LITERATURE REFERENCES AND PRIOR DATA 
See 
 
3.1 
DESCRIPTION OF OBJECTIVES AND PURPOSE 
The purpose of this investigation is to evaluate the safety and efficacy of the JJVCI Investigational Contact Lens 
with new UV blocker by comparison with the otherwise similar ACUVUE® OASYS® when worn for as daily wear 
with lens replacement every two weeks for 12 weeks. 
 
This is a pilot study and all primary and secondary hypotheses are exploratory in nature. 
4.1 
PRIMARY AND SECONDARY ENDPOINTS  
4.1.1 
Primary Efficacy and Safety Endpoints: 
The co-primary endpoints for efficacy evaluations are: 
 
1.
Eyestrain caused by glare score. This will be assessed using eyestrain patient reported outcomes. 
2.
Monocular distance visual acuity on LogMAR scale using ETDRS charts.  
 
The primary endpoint for safety evaluation is physiological findings (grade 3 or higher) using slit lamp.  The 
measures include edema, corneal neovascularization, corneal staining, conjunctival injection, tarsal abnormalities, 
and other complications using the FDA slit lamp grading scale 0 to 4, where 0 = none, 1 = trace, 2 = Mild, 3 = 
moderate and 4 = severe.   
4.1.2 
Secondary Efficacy and Safety Endpoints: 
The secondary endpoints are average daily wear time (in hours) and presence of ocular symptom (yes/no). 
 
Both primary and secondary endpoints will be evaluated at each follow-up visit. 
 
The subject’s adverse events, keratormety changes, lens deposit, reasons for discontinuation, lens damage, 
number and reasons for unscheduled lens replacement,  overall comfort, overall vision and overall handling will be 
monitored at the follow-up visits.

4.1.3 
Primary Hypotheses: 
1. Efficacy (Eyestrain caused by glare): Subjects wearing the JJVCI Investigational Test contact lenses on a 
daily wear basis report eyestrain caused by glare that is, on average, significantly lower to that of subjects 
wearing ACUVUE® OASYS® contact lenses on a daily wear basis.  The assessments will be made at the 2-, 
4-, 8- and 12-week follow-up visits. 
 
2. Efficacy (Visual Acuity): Subjects wearing the JJVCI Investigational Test contact lenses on a daily wear basis 
report an average distance monocular logMAR visual acuity that is statistically no different to that of 
subjects wearing ACUVUE® OASYS® contact lenses on a daily wear basis.  The assessments will be made at 
the 2-, 4-, 8- and 12-week follow-up visits. 
 
3. Safety (Slit Lamp Findings): Subjects wearing the JJVCI Investigational Test contact lenses on a daily wear 
basis have a percentage of grade 3 or higher slit lamp findings that is statistically no different to that of  
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 15 of 194

subjects wearing ACUVUE® OASYS® Contact Lenses on a daily wear basis. The assessments will be made at 
each follow-up visit. 
4.1.4 
Secondary Hypotheses: 
1. Efficacy (Average Wear Time): Subjects wearing the JJVCI Investigational Test contact lenses on a daily 
wear basis report an average daily wear time that  is statistically no different to that of subjects wearing 
ACUVUE® OASYS® contact lenses on a daily wear basis.  The assessments will be made at the 2-, 4-, 8- and 
12-week follow-up visits. 
 
2. Safety (Symptoms): Subjects wearing the JJVCI Investigational Test contact lenses on a daily wear basis 
have a percentage of reported symptoms, problems, or complaints (Yes or No) that is statistically no 
different to that of subjects wearing ACUVUE® OASYS® contact lenses on a daily wear basis.  The 
assessments will be made at each follow-up visit. 
4.2 
INCLUSION CRITERIA 
Potential subjects must satisfy all of the following criteria to be enrolled in the study: 
Subjects should meet all study Inclusion Criteria as outlined below: 
1. The subject must read and sign the Informed Consent form. 
2. The subject must appear able and willing to adhere to the instructions set forth in this clinical protocol. 
3. Healthy adult males or females age t 18 years of age with signed informed consent.  Eligible presbyopes 
will be those that wear full distance contact lenses in both eyes, then wear reading glasses over them. 
4. The subject’s optimal vertexed spherical equivalent distance correction must be between -1.00 and  
– 6.00D. 
5. The subject’s refractive cylinder must be ч 1.00D in each eye. 
6. The subject must have visual acuity best correctable to 20/25+3 or better for each eye. 
7. Subjects should own a wearable pair of spectacles.  
8. The subject is a current spherical soft contact lens wearer (defined as a minimum of 6 hours of DW per 
day, at least 5 days per week, for a minimum of 1 month prior to the study) and willing to wear the study 
lenses on a similar basis. 
9. Subjects must be able and willing to wear the study lenses at least 6 hours a day, a minimum of 5 days per 
week 
10. The subject must have normal eyes (i.e., no ocular medications or infections of any type). 
4.3 
EXCLUSION CRITERIA 
Potential subjects who meet any of the following criteria will be excluded from participating in the study:  
 
1. Currently pregnant or lactating (subjects who become pregnant during the study will be discontinued). 
2. Any systemic disease, autoimmune disease, or use of medication, which may interfere with contact lens 
wear. This may include, but not be limited to, diabetes, hyperthyroidism, recurrent herpes simplex/zoster, 
Sjögren’s syndrome, xerophthalmia, acne rosacea, Stevens-Johnson syndrome, and immunosuppressive 
diseases or any infectious diseases (e.g. hepatitis, tuberculosis). 
3. Use of any of the following medications within 1 week prior to enrollment: oral retinoid isotretinoin (e.g. 
Accutane), oral tetracyclines, topical scopolamine, oral (e.g. Seldane, Chlor-Trimeton, and Benadryl) and 
ophthalmic antihistamines, oral phenothiazines (e.g., Haldol, Mellaril, Thorazine, Elavil, Pamelor, 
Compazine), oral and ophthalmic Beta-adrenergic blockers (e.g., Propranolol, Timolol, and Practolol), 
systemic steroids, and any prescribed or over the counter (OTC) ocular medication. 
4. Entropion, ectropion, extrusions, chalazia, recurrent styes, glaucoma, history of recurrent corneal 
erosions, aphakia, or moderate or above corneal distortion by keratometry. 
5. Any previous, or planned, ocular or interocular surgery (e.g.., radial keratotomy, PRK, LASIK, etc.). 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 16 of 194

4.4.1 
TIME AND EVENT SCHEDULE 
 
Procedure 
Baseline 
Trial Fit & Dispense 
Follow-up  
1, 2, 3, 4, 5, 6, 7 
Unsched 
Exit 
Visit 
1 
1 
2, 3, 4, 5, 6, 7, 8 
PRN 
8 
Informed consent 
3 
- 
- 
- 
- 
Eligibility screening 
3 
- 
- 
- 
- 
PRO Baseline Questionnaires 
3 
- 
- 
- 
- 
Subject demographics 
3 
- 
- 
- 
- 
General health and medication history 
3 
- 
- 
- 
- 
Subject’s own contact lens information 
3 
- 
- 
- 
- 
Habitual lens care 
3 
- 
- 
- 
- 
Visual acuity (Snellen) 
3 
- 
- 
- 
3 
Spherocylindrical refraction and BVA 
3 
- 
- 
3 
3 
Slit lamp biomicroscopy 
3 
- 
3 
3 
- 
Keratometry 
3 
- 
- 
- 
3 
Trial fitting lens information 
- 
3 
- 
- 
- 
Lens Damage 
- 
3 
- 
- 
- 
Distance spherical over-refraction 
- 
3 
3 
- 
- 
Lens modification 
- 
3 
- 
- 
- 
Symptoms 
- 
3 
3 
3 
- 
PRO dispensing questionnaires 
- 
3 
- 
- 
- 
Visual acuity (logMAR) 
- 
3 
3 
3 
- 
Lens fitting assessment 
- 
3 
3 
* 
- 
Lens dispensing information and criteria 
- 
3 
- 
- 
- 
Patient instructions 
- 
3 
- 
- 
- 
Lens information 
- 
- 
3 
3 
- 
Compliance 
- 
- 
3 
3 
- 
Wearing times 
- 
- 
3 
3 
- 
PRO follow-up questionnaires 
- 
- 
3 
* 
- 
Surface characteristics 
- 
- 
3 
* 
- 
Chief complaint, diagnosis, treatment 
- 
- 
- 
3 
- 
 
* If wearing lenses 
Each lens type must be worn for approximately 14 days.  ENTER “P” FOR STUDY PROCEDURE  DATA SUPPORTING 
PRIMARY ENDPOINT.  
 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 19 of 194

4.4.2 
OPTIONAL FLOWCHART 
 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 20 of 194

2. The designee will record the subject’s number on the appropriate line of the randomization scheme. 
The designee will pull the appropriate test articles from the study supply.  All test articles that were opened, 
whether dispensed or not, must be recorded on the Test Article Accountability Log in the “Dispensed” section.  
4.11  REPORTING  PRODUCT QUALITY COMPLAINTS 
A Product Quality Complaint (PQC) refers to any written, electronic, or oral communication that alleges 
deficiencies related to the identity, quality, durability, reliability, safety, effectiveness or performance of test 
articles after they have been released for clinical trial use.  A PQC) is associated with any investigational product 
(i.e. product manufactured or supplied specifically for a clinical trial).  
Complaint Handling 
Once site personnel have become aware that a PQC has occurred, it shall then be recorded in the EDC system, 
which triggers an automatic email notification to the appropriate COM/CRA and Clinical QA representative. In 
cases where the EDC system in use is not configured to send automatic notifications, then the COM/CRA is 
responsible for notifying Clinical QA upon discovery that a PQC has occurred.   
Upon receipt of the EDC notification the COM/CRA shall complete the applicable sections of the 
 
 or electronic equivalent).  
 
For each complaint, the following minimum information shall be recorded by the CRA/COM on the 
 
: 
 
x
Date the complaint was received/recorded in the EDC System (Date of Sponsor Awareness) 
x
Who received the complaint 
x
Study number 
x
Investigational site information (contact name, site ID, telephone number) 
x
Lot number(s) 
x
Unique Subject Identifier(s) 
x
Indication of who first observed complaint (site personnel or subject) 
x
OD/OS indication, along with whether or not the lens was inserted 
x
Any related AE number if applicable 
x
Detailed complaint description (scheduled/unscheduled visit, wear time, symptoms, resolution of 
symptoms, etc.) 
x
Eye Care Provider objective (slit lamp) findings if applicable 
x
Confirmation of product availability for return (and tracking information, if available), or rationale if 
product is not available for return (Refer to 
 
 for test article return instructions) 
 
Clinical QA will assign a unique number to the PQC. Complaint numbering is assigned as follows: 
 
 
 
5.1 
WITHDRAWAL CRITERIA 
A subject may be withdrawn from the study for any of the following reasons: 
 
x
Lost to follow-up 
x
Withdrawal of consent 
x
Death 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 54 of 194

x
Subject was randomized but did not satisfactorily complete the fitting process, 
x
A scheduled visit is missed, 
x
Subject no longer met eligibility criteria (i.e. subject became pregnant during the study), 
x
Discontinuation of study treatment as a result of the investigator’s belief that for safety reasons (e.g., 
adverse event) it is in the best interest of the subject to stop treatment. 
x
Non-compliance to the protocol. 
 
For discontinued subjects, the Investigator will: 
 
x
Update the enrollment log to document reason for discontinuation and provide to CRA 
x
Complete the “last” Follow-up Visit form (scheduled or unscheduled) 
x
Complete the Final Evaluation form, indicating the reason that the subject was discontinued from the 
study 
x
Record the spherocylindrical refraction with best corrected distance visual acuity  
x
Collect used study lenses and test articles (worn or brought to the visit) from the subject and discard them 
x
Collect all unused study lenses and test articles from the subject 
 
Subjects becoming pregnant during the study will be discontinued.  Once discontinued, pregnant participants and 
their fetuses will not be monitored for study related purposes.  At the Investigators’ discretion, the study 
participant may be followed by the Investigator through delivery. However, this data will not be collected as part 
of the clinical study database.  Pregnant participants are not discontinued from contact lens or solution related 
studies for safety concerns, but due to general concerns relating to pregnancy and contact lens use.  Specifically, 
pregnant women are discontinued due to fluctuations in refractive error and/or visual acuity that occur secondary 
to systemic hormonal changes, and not due to unforeseen health risks to the mother or fetus. 
 
Investigators may be asked to replace discontinued subjects if the minimum target number of total subjects across 
all sites may otherwise not be met. 
 
In cases where a subject is lost to follow-up, every possible effort must be made to contact the subject and 
determine the reason for discontinuation / withdrawal.  The measures taken to follow up must be documented 
including two written attempts and a certified letter as the final attempt.  
6.1 
PRESTUDY AND CONCOMITANT THERAPY 
Concomitant medications will be documented during screening and during the study.  Disallowed medications for 
this study include those medications new to the subject within the past month and pose a threat to safe and 
comfortable contact lens wear. 
6.2 
MONITORING TREATMENT COMPLIANCE 
Johnson & Johnson Vision Care, Inc. representatives or designees will monitor the study in a manner consistent 
with ICH GCP E6.  The study monitors will maintain close contact with the Principal Investigator and the 
Investigator’s designated staff.  The monitor’s responsibilities will include: 
 
x
Ensuring that the investigation is being conducted according to the protocol 
x
Ensuring the rights and wellbeing of subjects are protected 
x
Ensuring that protocol deviations are documented with corrective action plans, as applicable 
x
Ensuring that the site has sufficient test article and supplies 
x
Clarifying questions regarding the study 
x
Resolving study issues or problems that may arise 
x
Reviewing the study records to ensure completeness and accuracy 
x
Study and subject source document records reviewed will include: 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 55 of 194

x
The Information and Consent Form per 21CFR Parts 50 and 56 and the HIPAA documents 
x
Source documentation including consenting and HIPAA process, medical history, concomitant 
medications, and adverse event information as applicable.  The source document should be initialed and 
dated by the study investigator/s. 
x
Investigational product shipping, dispensing, accountability, and return/destruction records 
x
Study related Regulatory documents as per ICH E3 section 8 
 
A separate monitoring plan will be provided.  
6.3 
UNSCHEDULED VISITS 
If, during the investigation, a subject experiences any investigational device-related difficulties and/or problems 
requiring an unscheduled visit to the clinic, the following information will be collected at a minimum: 
 
x
Chief complaint prompting the visit.  If the reason is an adverse event, the applicable eCRF for the adverse 
event should be completed and source documentation completed as appropriate. 
x
Date and time of the visit and all procedures completed at the unscheduled visit 
x
Review of adverse event and concomitant medications 
x
Documentation of any investigational product dispensed or collected from the subject. 
x
Slit lamp findings (using the Slit Lamp Classification Scale) 
 
If the Investigator withdraws a subject from the study, the final study visit case report forms must be completed 
indicating the reason(s) why the subject was withdrawn.  The subject enrollment log should be completed 
documenting the date and primary reason for withdrawal and the study CRA notified. 
 
Any investigational device-related difficulties and/or problems that are ongoing at the time of the final study visit 
will be followed by the Investigator, within licensure, until they have returned to pre-treatment status, stabilized, 
or been satisfactorily explained.  If further treatment (i.e., beyond licensure) is required, the subject will be 
referred to the appropriate health care provider. 
7.1 
EFFICACY PARAMETERS 
The primary safety parameters in this study are Biomicroscopy findings and the primary efficacy parameters are 
visual acuity (LogMAR) and Eyestrain caused by glare. The secondary safety parameter in this study is ocular 
symptoms and the efficacy parameter is average wear time. Other observations are subjective performance 
measured by GSI, CLUE and Lens Preference.  
7.2 
METHODS FOR ASSESSING, RECORDING, AND ANALYZING EFFICACY 
Refer to Section 4.1 (Primary and Secondary Endpoints) and Section 9 (Statistical Analysis). 
8.1 
SAFETY PARAMETERS 
All subjects randomized and treated will be in the safety analyses.  All reported AEs will be summarized.  All safety 
data will be listed by subject.  The following safety parameters will be monitored and evaluated: 
 
x
Adverse events 
x
Ocular physiology (corneal staining,  limbal and bulbar conjunctival redness) 
x
Ocular symptoms 
x
Slit Lamp Findings 
x
Lens Deposits 
x
Reasons for discontinuation 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 56 of 194

x
Lens damage 
x
Reason for unplanned lens replacement 
x
Snellen VA 
8.2 
ADVERSE EVENTS 
Non-Serious Adverse Events: 
All non-serious adverse events, including non-serious adverse device effects, will be reported to the sponsor by the 
Investigator no later than 48 hours from discovery for review by the Medical Monitor. 
 
Serious Adverse Events: 
The Investigator will inform the sponsor of all serious adverse events occurring during the study period as soon as 
possible by e-mail, fax, or telephone, but no later than 24 hours following discovery of the event.  The investigator 
is obligated to pursue and obtain information requested by the Sponsor in addition to that information reported 
on the CRF. All subjects experiencing a serious adverse event must be followed up and all outcomes must be 
reported. 
When medically necessary, the investigator may break the randomization code to determine the identity of the 
treatment that the subject received.  The Sponsor and study monitor should be notified prior to unmasking the 
test articles. 
 
In the event of a serious adverse event, the investigator must: 
 
x
Notify the Sponsor immediately 
x
Obtain and maintain in the subject’s file all pertinent medical records, information and medical judgment 
for colleagues who assisted in the treatment and follow-up of the subject 
x
Provide the Sponsor with a complete case history which includes a statement as to whether the event was 
or was not related to the use of the investigational test article 
x
Notify the IRB/IEC as required by the IRB/IEC reporting procedure according to national regulations 
 
Unanticipated (Serious) Adverse Device Effect (UADE): 
In the event of an Unanticipated (Serious) Adverse Device Effect (UADE), the Investigator will submit a report of 
the UADE to the Sponsor and IRB/IEC as soon as possible, but no later than 24 hours after the Investigator first 
learns of the effect.  This report is in addition to the immediate notification mentioned above. 
 
The Sponsor must conduct an evaluation of the UADE and must report the results of the evaluation to FDA, the 
IRB/IEC and participating investigators within 10 working days after the Sponsor first receives notification of the 
effect. 
8.3 
ADVERSE EVENT DEFINITIONS 
Adverse Event (AE) – An AE is any untoward (unwanted) medical occurrence in a patient or clinical investigation 
subject administered a test article whether or not caused by the test article or treatment.  An AE can therefore be 
any unfavorable or unintended sign (including an abnormal finding), symptom, or disease temporally associated 
with the use of the test article whether or not related to the test article. 
 
An AE includes any condition (including a pre-existing condition) that:  1) was not present prior to study treatment, 
but appeared or reappeared following initiation of study treatment; or 2) was present prior to study treatment, 
but worsened during study treatment.  This would include any condition resulting from concomitant illnesses, 
reactions to concomitant medications, or progression of disease states.  Pregnancy should be documented as an 
adverse event and should be reported to the clinical monitor and to the Sponsor immediately upon learning of the 
event. 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 57 of 194

Serious Adverse Event (SAE) – An SAE is any untoward medical occurrence that: 
x
Results in death 
x
Is life threatening 
x
Requires in-patient hospitalization or prolongation of existing hospitalization 
x
Results in persistent or significant disability/incapacity (i.e., a significant persistent or permanent change, 
impairment, damage, or disruption to the subject’s body) 
x
Is a congenital anomaly/birth defect, or 
x
Requires intervention to prevent permanent damage (the use of the investigational product resulting in a 
condition which requires medical or surgical intervention to preclude permanent impairment of the body 
structure or a body function).  Important medical events that may not result in death, be life-threatening, 
or require hospitalization may be considered an SAE when, based upon appropriate medical judgment, 
they may jeopardize the patient or subject and may require medical or surgical intervention to prevent 
one of the outcomes listed in the above definition. 
 
Diagnoses and conditions that are considered Serious Adverse Events include: 
 
x
Microbial Keratitis (MK) 
x
Iritis 
x
Permanent decrease in best spectacle corrected visual acuity equivalent to 2 acuity lines or greater 
x
Central Corneal Opacity 
x
Central Corneal Neovascularization 
x
Uveitis 
x
Endophthalmitis 
x
Hypopyon 
x
Hyphemia 
x
Penetration of Bowman’s Membrane 
x
Persistent Epithelial Defect 
x
Limbal cell Damage leading to Conjunctivalization 
 
Significant Adverse Events – Those events that are usually symptomatic and warrant discontinuation (temporary 
or permanent) of the test article (excluding Serious Adverse Events).  Diagnoses and conditions that are considered 
Significant Adverse Events include the following: 
 
x
Contact Lens Induced Peripheral Ulcer (CLPU) 
x
Significant Infiltrative Events (SIE) 
x
Superior Epithelial Arcuate Lesions (SEALs) 
x
Any Temporary Loss of > 2 Lines of BSCVA 
x
Other grade 3 or higher corneal findings, such as abrasions or edema 
x
Any corneal event which necessitates temporary lens discontinuation > 2 weeks 
x
Non-contact lens related corneal events - e.g. EKC (Epidemic Keratoconjunctivitis) 
x
Asymptomatic Corneal Scar 
 
Non-Significant Adverse Events – Those conditions that are usually asymptomatic and usually do not warrant 
discontinuation (temporary or permanent) of the test article.  However, the Investigator may choose to treat as a 
precautionary measure.  Diagnoses and conditions that are considered Non-Significant Adverse Events include the 
following: 
 
x
Non-significant Infiltrative Event 
x
Contact Lens Papillary Conjunctivitis 
x
Superficial Punctate Keratitis 
x
Conjunctivitis: Bacterial, Viral, Allergic 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 58 of 194

x
Blepharitis 
x
Meibomianitis 
x
Contact Dermatitis 
x
Localized Allergic Reactions 
x
Any corneal event not explicitly defined as serious or significant adverse event, which necessitates 
temporary lens discontinuation < 2 weeks 
 
Adverse Device Effect (ADE) – A sub-set of AEs, and include only those adverse events that are cause by or related 
to the investigational device or study procedure. 
 
Unanticipated Adverse Device Effect (UADE) – Any serious adverse effect on health or safety or any life-
threatening problem or death caused by, or associated with, the test article, if that effect, problem, or death was 
not previously identified in nature, severity, or degree of incidence in the investigational plan, investigator’s 
brochure or protocol, or any other unanticipated serious problem associated with the test article that relates to 
the rights, safety and welfare of subjects. 
Causality Assessment – A determination of the relationship between an adverse event and the test article. The 
test article relationship for each adverse event should be determined by the investigator using these explanations: 
 
x
Not Related- An adverse event that is not related to the use of the test article. 
x
Doubtful – An adverse event for which an alternative explanation is more likely, e.g. concomitant 
treatment, concomitant disease(s), or the relationship of time suggests that a causal relationship is not 
likely. 
x
Possible – An adverse event that might be due to the use of the test article.  An alternative explanation, 
e.g. concomitant treatment, concomitant disease(s), is inconclusive.  The relationship in time is 
reasonable.  Therefore, the causal relationship cannot be excluded. 
x
Probable – An adverse event that might be due to the use of the test article.  The relationship in time is 
suggestive (e.g. confirmed by de-challenge).  An alternative explanation is less likely, e.g. concomitant 
treatment or concomitant disease(s). 
x
Very Likely – An adverse event that is listed as a possible adverse effect (device) or adverse reaction (drug) 
and cannot be reasonably explained by an alternative explanation, e.g. concomitant treatment of 
concomitant disease(s).  The relationship in time is very suggestive, e.g. it is confirmed by de-challenge 
and re-challenge.  
 
Severity Assessment – A qualitative assessment of the degree of intensity of an adverse event as determined by 
the investigator or reported to him/her by the subject.  The assessment of severity is made irrespective of test 
article relationship or seriousness of the event and should be evaluated according to the following scale: 
 
x
Mild – Event is noticeable to the subject, but is easily tolerated and does not interfere with the subject’s 
daily activities. 
x
Moderate – Event is bothersome, possible requiring additional therapy, and may interfere with the 
subject’s daily activities. 
x
Severe – Event is intolerable, necessitates additional therapy or alteration of therapy and interferes with 
the subject’s daily activities. 
8.4 
METHODS FOR ASSESSING, RECORDING AND ANALYZING SAFETY 
The recording and documenting of adverse events (ocular and non-ocular) begin when the subjects are exposed to 
the test article or study treatment.  Adverse events reported before the use of test article or start of study 
treatment should be recorded as medical history.  Untoward medical events reported after the subject’s exit from 
the study will be recorded as adverse events at the discretion of the Investigator. 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 59 of 194

All adverse events observed by the Investigator; reported by the subject spontaneously; or in response to direct 
questioning; will be recorded in the source document.  Such documentation will include a description of the 
adverse event, time of onset, duration of event, treatment regimen instituted, any referral to another health care 
provider (if needed), any new concomitant medications, outcome, ocular damage (if any), and likely etiology.  Best 
Corrected Visual Acuity (BCVA) should be recorded prior to the report of an adverse event (as part of the baseline 
evaluation), upon report of the subject’s report of the adverse event, and after the adverse event has resolved.  All 
adverse events will be followed in accordance with licensing requirements. 
 
All adverse events will be documented in the appropriate section of the subject’s Case Report Form (CRF).  
In conjunction with the medical monitor, the Investigator will evaluate adverse events to ensure the events are 
categorized correctly.  Elements of categorization will include: 
 
x
Seriousness (see definition in Section 8.3) 
x
Expectedness – i.e. if the event was unexpected or unanticipated in that it was not previously identified in 
nature, severity, or degree of incidence (see definition in Section 8.3) 
x
Causality or Relatedness – i.e. the relationship between the test article and the adverse event (not 
related; doubtful; possible; probable; very likely - see definition in Section 8.3) 
x
Adverse Event Intensity or Classification – Adverse event intensity is used to assess the degree of intensity 
of the adverse event (mild, moderate, severe for all  events).  In addition Adverse event Classification is 
used to assess the severity of ocular adverse events (AE not requiring treatment, non-significant or 
significant see definition in Section 8.3). 
x
Outcome – Fatal, not resolved, resolved, resolved with sequelae, resolving and unknown.   
x
Actions Taken – None, temporarily discontinued, permanently discontinued 
x
Other action taken 
 
Upon finding an adverse event, the Principal Investigator will document the condition on the follow-up visit 
worksheet source document and in the CRF’s using photos or drawings (where appropriate) that detail size, 
location, and depth.  He will also complete the Adverse Event Classification (AEC) Discovery form / eCRF.  In 
addition, if an infiltrate(s) is present, he will complete the Corneal Infiltrate Assessment Form / eCRF.  Where 
necessary, a culture of the corneal lesion will be collected to determine if the infection is microbial in nature.  If 
cultures are collected, a source document note should be completed specifying the date of culture collection and 
laboratory utilized.  An eCRF documenting this should be completed in a comment or unscheduled visit. 
 
Complete description of all adverse events must be available in the source documents. All Adverse Events including 
local and systemic reactions not meeting the criteria for “serious adverse events” should be captured on the 
appropriate case report form or electronic data system.  Information to be recorded, based on above assessment 
criteria, includes date site notified, event description, date and time of onset, investigator assessment of severity, 
relationship to Study Agent(s)/Intervention(s), and time of resolution/stabilization of the event.  All adverse events 
occurring while on study must be documented appropriately regardless of relationship.  Define a timeframe for 
CRF completion and entry of the adverse event information into the database, as applicable. 
 
Any medical condition that is present at the time that the patient is screened should be considered as baseline and 
not recorded as an AE.  However, if the condition deteriorates at any time during the study it should be recorded 
and reported as an AE.  
 
Changes in the severity of an AE should be documented to allow an assessment of the duration of the event at 
each level of intensity to be performed.  Adverse events characterized as intermittent require documentation of 
the onset and duration of each episode. Changes in the assessment of relationship to the Study 
Agent(s)/Interventions should also be clearly documented. 
 
The investigator will notify the Sponsor of an adverse event by e-mail, facsimile, or telephone as soon as possible 
and no later than 24 hours from discovery for a serious / significant adverse event, and 2 days from discovery for a 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 60 of 194

non-significant adverse event.  In addition, a written report will be submitted by the Principal Investigator to the 
IRB/IEC according to their requirements (Section 1212.3).  Such a report should comment whether or not the 
adverse event was considered to be related to the test article. 
8.5 
ADVERSE EVENTS FOLLOW-UP 
It is the Investigator’s responsibility to maintain documentation of each reported adverse event.  All adverse 
events will be followed in accordance with applicable licensing requirements.  Such documentation will include the 
following: 
 
x
Adverse Event (Diagnosis not symptom) 
x
Detailed drawings or photographs, when appropriate 
x
Date and time of onset 
x
Date and time of resolution 
x
Adverse event intensity and classification, as applicable 
x
Treatment regimen instituted, including concomitant medications prescribed, in accordance with 
applicable licensing requirements 
x
Any referral to another health care provider if needed 
x
Outcome, ocular damage (if any) 
x
Likely etiology. 
x
Best corrected visual acuity at the discovery of the event and upon conclusion of the event 
 
In addition, if an infiltrate(s) is present, the Investigator will complete the Corneal Infiltrate Assessment Form / 
eCRF. 
 
Photographs or video recordings may be collected at the Investigator’s discretion for purposes of documenting 
adverse event findings. 
 
Visual acuity (best corrected) should be recorded prior to the report of an adverse event (as part of the Baseline 
Evaluation), upon the subject’s report of the adverse event, and after the adverse event has resolved.   
 
Subjects who present with an adverse events should be followed by the Investigator, within licensure, until all 
signs and symptoms have returned to pre-treatment status, stabilized, or been satisfactorily resolved.  If further 
treatment (i.e. beyond licensure) is required, the patient will be referred to the appropriate health care provider.  
The Investigator should use his/her clinical judgment as to whether or not a subject (eye) reporting with an 
adverse event should continue in the study.  If a subject is discontinued from the study, it will be the responsibility 
of the Investigator to record the reason for discontinuation.  The investigator will also complete the Adverse Event 
Classification (AEC) Outcome form / eCRF.  Any subjects with ongoing adverse events related to the test article as 
of the final study visit date should be followed to resolution of the adverse event or until referral to an appropriate 
health care provider, as recommended by the Investigator. 
9.1 
STATISTICAL METHODS TO BE EMPLOYED 
This section is a general outline of the statistical methods that will be implemented in this clinical trial. For more 
details, refer to the stand-alone Statistical Analysis Plan (SAP) of this clinical study. 
 
General Considerations: 
Statistical programming and analyses will be performed using SAS (SAS Institute, Cary, NC)5 Version 9.4 or higher.  
Throughout the analysis of data, the results for each eye will be used when available for summarization and 
statistical analysis. 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 61 of 194

Descriptive statistics will be reported at baseline and each post fit time period for all safety and efficacy variables 
as appropriate. Unless otherwise noted, continuous variables will be summarized using descriptive statistics: 
number of non-missing values (N), mean (Mean), standard deviation (SD), median (Median), minimum (Min), and 
maximum (Max); and categorical variables will be summarized using the frequency count (N) and the percentage 
(%) of subjects or eyes in each category. 
 
Summaries will be presented by study article type (Test or Control) and will be performed separately by 
completion status (completed or discontinued).  See Section 9.7 for additional details regarding subject completion 
status. The denominator for percentages of counts will be the same for each visit on the summary tables and will 
equal the final number of subjects or eyes in the group under consideration.  Unscheduled visits will be 
summarized separately and will be excluded from efficacy analysis. 
 
Analysis Population Sets: 
 
Efficacy analyses will be performed on all randomized subjects who completed the study and did not substantially 
deviate from the protocol as determined by the trial cohort review committee prior to database hard lock (Per-
Protocol Population). Justification of excluding subjects with protocol deviations in the per-protocol population set 
will be documented in a memo to file.Additional post-hoc analyses may be conducted by including all randomized 
subjects who have been successfully dispensed and have at least one follow-up visit. Refer to section 9.7 Evaluable 
Subjects for more details.
 
Safety analyses will be performed on the safety population, which will be comprised of all randomized subjects 
who have been successfully dispensed at least one study lens. 
 
Unless otherwise specified, all statistical tests will be 2-sided and conducted at the 0.05 level of significance. 
 
Efficacy Analysis: 
Visual Acuity 
Contact lens monocular distance best-corrected visual acuity (BCVA) on logMAR scale will be analyzed using a 
linear mixed model to test for the differences between the Test and Control group at each follow-up visit.  Lens 
group, time  (1-, 2-, 4-, 8- and 12-week)  and the lens group by time interaction will be included as fixed effects in 
the model; and investigator site, subject and eye nested within subject as random effects when appropriate. 
The covariance between residuals from the same eye and subject at different time points will be selected based on 
the finite-sample Corrected Akaike’s Information Criterion. Covariance structures considered include Compound 
Symmetry (CS), Heterogeneous Compound Symmetry (CSH), Spatial Power (SP(POW)), Ante-dependence (ANTE(1)) 
and unstructured (UN). For ANTE(1) and SP(POW) structures, site, subject and eye nested within subject will be 
included as random effects. For the remaining structures only site and subjects will be included as random effects. 
The covariance structure that returns the lowest Akaike Information Criteria Corrected (AICC) will be selected as 
the structure that best fit the data. Heterogeneous residuals covariance structures (R-side) across lens groups will 
be considered when appropriate. The log-likelihood ratio test will be used to test for the homogeneity between 
the residual covariance structures. The Kenward and Roger method will be used for the denominator degree of 
freedom.  
Comparison between the Test and the Control groups will be conducted at each time point using a t-test on least-
square means from the repeated measure analysis. Adjustment for multiple comparisons across time will be 
performed using Bonferroni’s method with alpha equal to 0.01 (0.05/5).  The corresponding simultaneous 
confidence intervals of least-square means differences will be calculated with 99% confidence. 
 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 62 of 194

Average Wear Time (AWT) 
AWT (in hours) will be analyzed using a linear mixed model to test for difference between the Test and the Control 
groups.  Lens Group,  Time  (1-, 2-, 4-, 8- and 12-week) and the interaction term the lens group by time will be 
included as fixed effects in the model; and site as random effect. 
The covariance between residuals from the same subject at different time points will be selected based on the 
finite-sample corrected Akaike’s Information Criterion. Covariance structures considered include homogenous 
compound symmetry (CS), heterogeneous Compound symmetry (CSH), Spatial Power (SP(POW)), Ante-
dependence (ANTE(1)) and unstructured (UN). For ANTE(1) and SP(POW) structures, site and subject will be 
included as random effects. For the remaining structures only site will be included as random effect. The 
covariance structure that returns the lowest AICC will be selected as the structure that best fit the data. 
Heterogeneous residuals covariance structures (R-side) across lens groups will be considered when appropriate. 
The log-likelihood ratio test will be used to test for the homogeneity between the residual covariance structures. 
The Kenward and Roger method will be used for the denominator degree of freedom.  
Comparison between the Test and the Control groups will be conducted at each time point using a t-test on least-
square means from the repeated measure analysis.  Adjustment for multiple comparisons across time will be 
performed using Bonferroni’s method with alpha equal to 0.01 (0.05/5).  The corresponding simultaneous 
confidence intervals of least-square means differences will be calculated with 99% confidence. 
Safety Analysis: 
Slit Lamp Findings (SLF) 
SLF responses will be categorized into a binary outcome as 0 if no Grade 3 or higher SLF or 1 if any Grade 3 or 
higher SLF.  Events occurring during an unscheduled visit will be counted in the subsequent scheduled visit. For 
example, if a SLF grade 3 or higher occurs between Visit 1 and Visit 2, it will be counted in Visit 2.  If there are 
multiple events at a given time point for one eye, it will be counted only once.  The safety endpoint will be 
analyzed using a Generalized Estimating Equation (GEE) model (Zeger and Liang; 1986)2 with a binomial 
distribution and logit link function. The regression model will include terms for lens group, time (1-, 2-, 4-, 8- and 
12-week) and the lens group by time interaction (group*time).  Appropriate working correlation matrix will be 
selected to take on consideration the correlation between measurements within each subject using Quasi-
likelihood Information Criterion (QIC). Working correlation structures to be considered will include independence 
(IND), exchangeable (Exch), autoregressive AR(1) and unstructured (UN).  The covariance structure that returns the 
lowest QIC will be selected as the structure that best fit the data.  If  all model fails to converge, a reduced model 
will be considered by removing group*time and time from the model. 
 
The odds ratio of outcome 1 between the Test and the Control groups will be calculated with a 95% 
confidence interval to test for the difference between the Test and Control groups.  
 
If there are no sufficient SLF findings of Grade 3 or higher, analysis will be performed on any SLF grade 2 or higher 
SLF. Further analyses by subject, visit and/or slit lamp findings category will also be considered if necessary. 
 
Ocular Symptoms, problems or complaints 
Reported ocular symptoms, problems or complaints (0 = No, 1 = Yes) at any visit will be analyzed using a GEE 
model for clustered binary outcome to assess the difference between the Test and Control groups. The regression 
model will include terms for lens group, time (1-, 2-, 4-, 8- and 12-week) and the lens group by time interaction.  
Appropriate working correlation matrix will be selected to take on consideration the correlation between 
measurements within each subject using Quasi-likelihood Information Criterion (QIC). Working correlation 
structures to be considered will include exchangeable, autoregressive AR(1) and unstructured (UN). Odds ratio of 
outcome 1 between the Test and the Control groups will be calculated with a 95% confidence interval.  
 
If the interaction lens by time is significant at the 0.15 significance level, the difference between the Test and 
Control will be assessed for each follow-up visit. 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 63 of 194

9.2 
NUMBER OF SUBJECTS BY SITE AND JUSTIFICATION FOR SAMPLE SIZE 
The plan is to enroll 60 eligible subjects per arm with a target completion of 50 subjects per arm. This is a pilot 
study and the sample size was not based on  any empirical calculation. A statistical power analysis for different 
scenarios of effect size and variance will be provided in the stand alone SAP document.  
 
During the enrollment period, the subject dropout rate will be closely monitored, if unexpectedly high dropout 
rate is observed in certain arm (s), the targeted total enrollment number will be increased accordingly in order to 
ensure a minimum of 50 subjects per group to complete the 3-month follow-up. Please refer to the stand alone 
Statisitical Analysis Plan (SAP). 
9.3 
LEVEL OF STATISTICAL SIGNIFICANCE 
In addition to the unadjusted estimates, adjusted estimates with 5% significance level (type I error rate) will be 
provided. 
9.4 
CRITERIA FOR STUDY TERMINATION 
In addition to the stopping rules defined in section 5.1, the occurrence of one or more Serious Unanticipated 
Adverse Device Effect (USADE), or any SAE where the relationship to study agent cannot be ruled out, may result in 
stopping further dispensing of investigational product.  In the event of a USADE or SAE, the Sponsor may unmask 
the treatment regimen for the subject(s) and may discuss this with the Investigator before any further subjects are 
enrolled. 
 
The sponsor may determine when a study will be stopped.  The principal investigator always has the discretion to 
initiate stopping the study based on patient safety or if information indicates the study’s results may be 
compromised. 
9.5 
PROCEDURE FOR ACCOUNTING FOR MISSING, UNUSED, AND SPURIOUS DATA 
Missing or spurious values will not be imputed. The count of missing values will be included in the summary tables 
and listings. 
9.6 
PROCEDURE FOR REPORTING DEVIATIONS FROM STATISTICAL PLAN 
The analysis will be conducted according to section 9.1.  There are no known reasons for which it is planned to 
deviate from these analysis methods.  If for any reason a change is made, the change will be documented in the 
study report along with a justification for the change. 
9.7 
EVALUABLE SUBJECTS 
 
Subjects will be allocated to one of four mutually exclusive groups: 
 
1. Enrolled, Not Randomized:  Subjects are considered to be Enrolled Not Randomized if they were (i) 
enrolled to the study (provided informed consent) but failed to satisfy the eligibility criteria 
(inclusion/exclusion criteria) or (ii) were not randomized to study lens for any reason. 
 
2. Randomized, Not Successfully Dispensed:  Subjects are considered to be Randomized Not Successfully 
Dispensed if they are randomized but did not satisfactorily complete the dispensing process. This includes 
subjects who were dispensed a different type of study lens for each eye or who were dispensed same 
study lens for both eyes but did not  return for any follow-up visits. 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 64 of 194

3. Successfully Dispensed, Discontinued Subjects:  Subjects are considered to have discontinued from the 
study if they are (i) randomized (ii) Successfully Dispensed and (iii) discontinued because of one of the 
reasons described in section 5.1. 
 
4. Successfully Dispensed, Completed Subjects:  Subjects are considered to have completed the study if they 
have completed all required visits through Visit 8 (14-week Follow-up Visit). 
 
Safety and efficacy analysis sets will be defined as subsets of successfully dispensed subjects (completed + 
discontinued). 
10.1 
ELECTRONIC CASE REPORT FORMS/DATA COLLECTION 
The data for this study will be captured on electronic case report forms (eCRFs) using an EDC system (BioClinica), 
when possible. Designated study site personnel will enter study data into the electronic CRFs (eCRFs) using the EDC 
system (BioClinica). Data collected on equipment that is not possible to be captured in EDC will be formatted to 
the specification of the JJVCI database manager and sent to JJVCI for analysis.  Data generated from post hoc 
measurements (e.g. Compositional characteristics of contact lens lipid uptake, Measured tear film protective 
capability, Measured contact lens surface dehydration rate) will be collected on specific Microsoft Office Excel 
format worksheets at the clinical site and at the completion of the analysis transferred to JJVCI biostatistician for 
data analysis in such format. 
 
The clinical data will be recorded on dedicated electronic case report forms (eCRFs) specifically designed to match 
the testing routine for each visit.  Once completed, The CRFs will be reviewed for accuracy and completeness and 
signed by the investigator.  Unless otherwise stated, the eCRFs will be considered the source document. The 
sponsor or sponsor‘s representatives will be authorized to gain access to the source documentation for the 
purposes of monitoring and auditing the study.  
 
BioClinica is compliant with all relevant aspects of ICH/Good Clinical Practices, and 21 CFR Part 11 (Electronic 
Records & Electronic Signatures) regulations.  In addition, the Sponsor and BioClinica are Safe Harbor Certified.  
 
Edit checks, electronic queries, and audit trails are built into the system to ensure accurate and complete data 
collection. Data will be transmitted from the investigational site to a secure central database as forms are 
completed or updated, ensuring information accuracy, security, and confidentiality. After the final database lock, 
the Investigator will be provided with Individual Patient Profiles (IPP) including the full audit trail on electronic 
media in PDF format for all of the study data. The IPP should be retained in the study files as a certified copy of the 
source data for the study.  
 
The content and structure of the CRFs are compliant with ISO14155:2011 [3]. 
10.2 
SOURCE DOCUMENTATION 
At a minimum, source documentation should be available for the following to confirm data collected in the CRF: 
subject identification, eligibility, and study identification; study discussion, provision of and date of informed 
consent; visit dates; results of safety and efficacy parameters as required by the protocol; a record of all adverse 
events; follow-up of adverse events; concomitant medication; investigational product receipt / dispensing / return 
records; study investigational product administration information; date of study completion; reason for early 
discontinuation of investigational product or withdrawal from the study, if applicable. 
 
It is recommended that the author of an entry in the source documents be identifiable.  Adverse event notes 
should be reviewed and initialed by the Investigator. 
 
At a minimum, the type and level of detail of source data available for a study subject should be consistent with 
that commonly recorded at the site as a basis for standard medical care.  Specific details required as source data 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 65 of 194

for the study will be reviewed with the investigator before the study and will be described in the monitoring 
guidelines (or other equivalent documents). 
10.3 
ACCESS TO SOURCE DATA/DOCUMENTS 
The Investigator(s) / Institution(s) will permit trial-related monitoring, audits, IRB/IEC review and regulatory 
inspection(s) by providing direct access to source data / documents.  Should the clinical site be contacted for an 
audit by an IRB/IEC or regulatory authority, JJVCI should be contacted and notified in writing within 24 hours. 
10.4 
CONFIDENTIALITY OF INFORMATION 
Information concerning the investigational product and patent application processes, scientific data or other 
pertinent information is confidential and remains the property of JJVCI.  The Investigator may use this information 
for the purposes of the study only. It is understood by the Investigator that JJVCI will use information developed in 
this clinical study in connection with the development of the investigational product and therefore may disclose it 
as required to other clinical investigators and to regulatory agencies. In order to allow the use of the information 
derived from this clinical study, the Investigator understands that he/she has an obligation to provide complete 
test results and all data developed during this study to the Sponsor. 
 
The Investigator may not submit for publication or presentation the results of this study without first receiving 
written authorization from JJVCI.  JJVCI agrees that, before it publishes any results of the study, it shall provide the 
Investigator with at least 30 days for review of the pre-publication manuscript prior to the submission of the 
manuscript to the publisher. 
11.1 
DATA QUALITY ASSURANCE 
Steps to be taken to ensure the accuracy and reliability of data include the selection of qualified investigators and 
appropriate study sites and review of protocol procedures with the principal investigator.  The principle 
investigator, in turn, must ensure that all sub-investigators and study staff are familiar with the protocol and all 
study-specific procedures and have appropriate knowledge of the study article. 
 
Guidelines for case report form completion will be provided and reviewed with study personnel before the start of 
the study.  The sponsor, Johnson & Johnson Vision Care, Inc. will review case report forms for accuracy and 
completeness during monitoring visits and after transmission to data management.  Any data discrepancies will be 
resolved with the investigator or designee, as appropriate. 
 
Quality Assurance representatives from Johnson & Johnson Vision Care, Inc. may visit study sites to review data 
produced during the study and to access compliance with applicable regulations pertaining to the conduct of 
clinical trials.  The study sites will provide direct access to study-related source data/documents and reports for the 
purpose of monitoring and auditing by Johnson & Johnson Vision Care, Inc. and for inspection by local and 
regulatory authorities. 
12.1 
STUDY-SPECIFIC DESIGN CONSIDERATIONS 
Potential subjects will be fully informed of the risks and requirements of the study and, during the study, subjects 
will be given any new information that may affect their decision to continue participation.  They will be told that 
their consent to participate in the study is voluntary and may be withdrawn at any time with no reason given and 
without penalty or loss of benefits to which they would otherwise be entitled.  Only subjects who are fully able to 
understand the risks, benefits, and potential adverse events of the study, and provide their consent voluntarily will 
be enrolled. 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 66 of 194

12.2 
INVESTIGATOR RESPONSIBILITY 
The Investigator is responsible for ensuring that the clinical study is performed in accordance with the protocol, 
Section 4 of the ICH E6 guidelines on Good Clinical Practice (GCP), and applicable regulatory requirements.  GCP is 
an international ethical and scientific quality standard for designing, conducting, recording, and reporting studies 
that involve the participation of human subjects. Compliance with this standard provides public assurance that the 
rights, safety, and well-being of study subjects are protected, consistent with the principles of the Declaration of 
Helsinki 59th WMA General Assembly 2008 and that the clinical study data are credible.  The Investigator must 
maintain clinical study files in accordance with Section 8 of the ICH E6 guidelines on Good Clinical Practice (GCP), 
and applicable regulatory requirements. 
12.3 
INDEPENDENT ETHICS COMMITTEE OR INSTITUTIONAL REVIEW BOARD (IEC/IRB) 
Before the start of the study, the Investigator (or Sponsor when applicable) will provide the IEC/IRB with current 
and complete copies of the following documents (where applicable): 
 
x
final protocol and, if applicable, amendments 
x
Sponsor-approved informed consent form (and any other written materials to be provided to the subjects) 
x
Investigator’s Brochure (or equivalent information) and amendments 
x
Sponsor-approved subject recruitment materials 
x
information on compensation for study-related injuries or payment to subjects for participation in the 
study, if applicable 
x
Investigator’s curriculum vitae, clinical licenses, or equivalent information (unless not required, as 
documented by IEC/IRB) 
x
information regarding funding, name of the Sponsor, institutional affiliations, other potential conflicts of 
interest, and incentives for subjects 
x
any other documents that the IEC/IRB requests to fulfill its obligation 
 
This study will be undertaken only after IEC/IRB has given full approval of the final protocol, amendments (if any), 
the informed consent form, applicable recruiting materials, and subject compensation programs, and the Sponsor 
has received a copy of this approval. This approval letter must be dated and must clearly identify the documents 
being approved. 
 
During the study the Investigator (or Sponsor when applicable) will send the following documents to the IEC/IRB 
for their review and approval, where appropriate: 
 
x
protocol amendments 
x
revision(s) to informed consent form and any other written materials to be provided to subjects 
x
if applicable, new or revised subject recruiting materials approved by the Sponsor 
x
revisions to compensation for study-related injuries or payment to subjects for participation in the study, if 
applicable 
x
Investigator’s Brochure amendments or new edition(s) 
x
summaries of the status of the study (at least annually or at intervals stipulated in guidelines of the 
IEC/IRB) 
x
reports of adverse events that are serious, unanticipated, and associated with the investigational product, 
according to the IRB’s requirements 
x
new information that may adversely affect the safety of the subjects or the conduct of the study 
x
major protocol deviations as required by the IEC/IRB 
x
report of deaths of subjects under the Investigator's care 
x
notification if a new investigator is responsible for the study at the site 
x
any other requirements of the IEC/IRB 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 67 of 194

For protocol amendments that increase subject risk, the amendment and applicable informed consent form 
revisions must be submitted promptly to the IEC/IRB for review and approval before implementation of the 
change(s). 
 
At least once a year, the IEC/IRB will be asked to review and reapprove this clinical study. This request should be 
documented in writing. 
 
At the end of the study, the Investigator (or Sponsor where required) will notify the IEC/IRB about the study 
completion.  Documentation of this notification must be filed in the study Investigator binder and a copy provided 
to the CRO or Sponsor as applicable. 
12.4 
INFORMED CONSENT 
Each subject must give written consent according to local requirements after the nature of the study has been fully 
explained. The consent form must be signed before performance of any study-related activity. The consent form 
that is used must be approved by both the Sponsor and by the reviewing IEC/IRB. The informed consent should be 
in accordance with principles that originated in the Declaration of Helsinki, current ICH and GCP guidelines, 
applicable regulatory requirements, and sponsor policy. 
 
Before entry into the study, the Investigator or an authorized member of the investigational staff must explain to 
potential subject the aims, methods, reasonably anticipated benefits, and potential hazards of the study, and any 
discomfort it may entail. Subjects will be informed that their participation is voluntary and that they may withdraw 
consent to participate at any time. They will be informed that choosing not to participate will not affect the care 
the subject will receive.  Finally, they will be told that the Investigator will maintain a subject identification register 
for the purposes of long-term follow-up if needed and that their records may be accessed by health authorities 
and authorized sponsor staff without violating the confidentiality of the subject, to the extent permitted by the 
applicable law(s) or regulations. By signing the informed consent form the subject is authorizing such access, and 
agrees to be contacted after study completion, by health authorities and authorized sponsor staff, for the purpose 
of obtaining consent for additional safety evaluations if needed. 
 
The subject will be given sufficient time to read the informed consent form and the opportunity to ask questions. 
After this explanation and before entry into the study, consent should be appropriately recorded by means of the 
subject's dated signature. After having obtained the consent, a copy of the informed consent form must be given 
to the subject. 
 
In the event additional investigators / sites are added to the protocol, the informed consent will be modified to 
include the Investigator’s name, address, phone number and 24-hour emergency number.  
12.5 
PRIVACY OF PERSONAL DATA 
The collection, processing and disclosure of personal data and medical information related to the Study Subject, 
and personal data related to Principal Investigator and any investigational staff (e.g., name, clinic address and 
phone number, curriculum vitae) is subject to compliance with the Data Protection Act of 1998 and other 
applicable personal data protection and security laws and regulations.  Appropriate measures will be employed to 
safeguard these data, to maintain the confidentiality of the person’s related health and medical information, to 
properly inform the concerned persons about the collection and processing of their personal data, to grant them 
reasonable access to their personal data and to prevent access by unauthorized persons. 
 
All information obtained during the course of the investigation will be regarded as confidential. All personal data 
gathered in this trial will be treated in strictest confidence by investigators, monitors, Sponsors personnel and 
independent ethics committee.  No data will be disclosed to any third party without the express permission of the 
subject concerned, with the exception of sponsor personnel (monitor, auditor), independent ethics committee and 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 68 of 194

regulatory organizations in the context of their investigation related activities that, as part of the investigation will 
have access to the CRFs and source documents. 
 
The collection and processing of personal data from subjects enrolled in this study will be limited to those data 
that are necessary to investigate the efficacy, safety, quality, and utility of the investigational product(s) used in 
this study.  These data must be collected and processed with adequate precautions to ensure confidentiality and 
compliance with applicable data privacy protection laws and regulations.  The Sponsor ensures that the personal 
data will be: 
 
x
processed fairly and lawfully 
x
collected for specified, explicit, and legitimate purposes and not further processed in a way incompatible 
with these purposes 
x
adequate, relevant, and not excessive in relation to said purposes 
x
accurate and, where necessary, kept current 
 
Explicit consent for the processing of personal data will be obtained from the participating subject before 
collection of data. Such consent should also address the transfer of the data to other entities and to other 
countries. 
 
The subject has the right to request through the Investigator access to his personal data and the right to request 
rectification of any data that are not correct or complete. Reasonable steps should be taken to respond to such a 
request, taking into consideration the nature of the request, the conditions of the study, and the applicable laws 
and regulations. 
 
Appropriate technical and organizational measures to protect the personal data against unauthorized disclosures 
or access, accidental or unlawful destruction, or accidental loss or alteration must be put in place. Sponsor 
personnel whose responsibilities require access to personal data agree to keep the identity of study subjects 
confidential 
13.1 
DATA HANDLING AND RECORD KEEPING 
In compliance with the ICH/GCP guidelines, the Investigator / Institution will maintain all CRFs and all source 
documents that support the data collected from each subject, as well as all study documents as specified in 
ICH/GCP Section 8, Essential Documents for the Conduct of a Clinical Trial, and all study documents as specified by 
the applicable regulatory requirement(s). The Investigator / Institution will take measures to prevent accidental or 
premature destruction of these documents. 
 
Essential documents must be retained until at least 2 years after the last approval of a marketing application in an 
ICH region and until there are no pending or contemplated marketing applications in an ICH region or until at least 
2 years have elapsed since the formal discontinuation of clinical development of the investigational product. These 
documents will be retained for a longer period if required by the applicable regulatory requirements or by an 
agreement with the Sponsor. It is the responsibility of the Sponsor to inform the Investigator / Institution as to 
when these documents no longer need to be retained. 
 
If the responsible Investigator retires, relocates, or for other reasons withdraws from the responsibility of keeping 
the study records, custody must be transferred to a person who will accept the responsibility. The Sponsor must be 
notified in writing of the name and address of the new custodian. Under no circumstance shall the Investigator 
relocate or dispose of any study documents before having obtained written approval from the Sponsor. 
 
If it becomes necessary for the Sponsor or the appropriate regulatory authority to review any documentation 
relating to this study, the Investigator must permit access to such reports. 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 69 of 194

If the Investigator has a question regarding retention of study records, he should contact JJVCI Research and 
Development. 
14.1 
FINANCIAL CONSIDERATIONS 
Remuneration for study services and expenses will be set forth in detail in the Investigator’s Research Agreement.  
The Research Agreement will be signed by the Principal Investigator and a Johnson & Johnson Vision Care 
management representative prior to study initiation. 
 
Case Report Forms will be completed in real time according to the study procedures specified in the study 
protocol.  Case Report Forms should be completed and reviewed and signed as applicable by the Investigator 
within 3 days of visit completion.  Data queries must be addressed with complete responses within 3 days of 
generation.  Johnson & Johnson Vision Care reserves the right to withhold remuneration until these activities are 
addressed. 
 
Johnson & Johnson Vision Care reserves the right to withhold remuneration for costs associated with protocol 
violations such as: 
 
x
Continuing an ineligible subject in the study 
x
Scheduling a study visit outside the subject’s acceptable visit range 
15.1 
PUBLICATION 
This study will be registered on ClinicalTrials.gov. 
16.1 
PATIENT REPORTED OUTCOMES (STUDY QUESTIONNAIRES) 
 
 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 70 of 194

 
Limbal & Conjunctival (Bulbar) Redness 
 
 
 
 
 
 
 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 104 of 194

 
Expanded Sodium Fluorescein Corneal Staining  
 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 112 of 194

 
Lens Fitting Characteristics  
 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 117 of 194

 
Subject Reported Ocular Symptoms/Problems 
 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 124 of 194

 
Front and Back Surface Lens Deposit Grading Procedure  
 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 126 of 194

 
Determination of Distance Spherocylindrical Refractions 
 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 132 of 194

 
Biomicroscopy Scale 
 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 139 of 194

 
Distance and Near Visual Acuity Evaluation  
 
 
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 145 of 194

 
Distance LogMAR Visual Acuity Measurement Procedure 
 
 
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 150 of 194

 
White Light Lens Surface Wettability   
 
 
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 154 of 194

16.3 
PATIENT INSTRUCTION GUIDE (INVESTIGATIONAL PRODUCT) 
The investigational patient instruction guide will be provided separately. 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
 
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 156 of 194

16.4 
PACKAGE INSERT (APPROVED PRODUCT) 
 
 
 
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 157 of 194

DESCRIPTION
The ACUVUE OASYS® Brand Contact Lenses, the ACUVUE OASYS® Brand Con-
tact Lenses for ASTIGMATISM, and the ACUVUE OASYS® Brand Contact Lenses 
for PRESBYOPIA are soft (hydrophilic) contact lenses available as spherical, toric, 
or multifocal lenses and include HYDRACLEAR® PLUS Technology. The lenses 
are made of a silicone hydrogel material containing an internal wetting agent with 
visibility tinted UV absorbing monomer.
These lenses are tinted blue using Reactive Blue Dye #4 to make the lenses more 
visible for handling. A benzotriazole UV absorbing monomer is used to block UV 
radiation.
The transmittance characteristics are less than 1% in the UVB range of 280 nm to 
315 nm and less than 10% in the UVA range of 316 nm to 380 nm for the entire 
power range.
Lens Properties:
• Speciﬁ c Gravity (calculated): 
0.98 – 1.12
• Refractive Index: 
1.42
• Light Transmittance:
85% minimum
• Surface Character:
Hydrophilic
• Water Content:
38%
• Oxygen Permeability:
VALUE
METHOD
103 x 10-11 (cm2/sec)
(ml O2/ml x mm Hg) @ 35°C  
Fatt (boundary corrected, edge corrected)
122 x 10-11 (cm2/sec)
(ml O2/ml x mm Hg) @ 35°C
Fatt (boundary corrected, non-edge 
corrected)
Lens Parameters:
• Diameter Range:
12.0 mm to 15.0 mm
• Center Thickness:
varies with power
• Base Curve Range:
7.85 mm to 10.00 mm
• Spherical Power Range:
Daily Wear:  -20.00D to +20.00D
Extended Wear:  -20.00D to +14.00D
• Multifocal ADD Power Range:
+0.25D to +4.00D
• Cylinder Power Range:
-0.25D to -10.00D
• Axis Range:
2.
80°
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 160 of 194

AVAILABLE LENS PARAMETERS
The ACUVUE OASYS® Brand Contact Lenses are hemispherical shells of the 
following dimensions:
Diameter:
14.0 mm
Center Thickness:
Minus Lens - varies with power (e.g. -4.00D: 0.070 mm)
Plus Lens - varies with power (e.g. +4.00D: 0.168 mm)
Base Curve:
8.4 mm, 8.8 mm
Powers:
-0.50D to -6.00D (in 0.25D increments)
-6.50D to -12.00D (in 0.50D increments)
+0.50D to +6.00D (in 0.25D increments)
+6.50D to +8.00D (in 0.50D increments)
The ACUVUE OASYS® Brand Contact Lenses for ASTIGMATISM are hemitoric 
shells of the following dimensions:
Diameter:
14.5 mm
Center Thickness:
Minus Lens - varies with power (e.g. -4.00D: 0.080 mm)
Plus Lens - varies with power (e.g. +4.00D: 0.172 mm)
Base Curve:
8.6 mm
Powers:
plano to -6.00D (in 0.25D increments)
-6.50D to -9.00D (in 0.50D increments)
+0.25D to +6.00D (in 0.25D increments)
Cylinder:  -0.75D, -1.25D, -1.75D, -2.25D, -2.75D
Axis:  10° to 180° (in 10° increments)
The ACUVUE OASYS® Brand Contact Lenses for PRESBYOPIA are hemispherical 
shells of the following dimensions:
Diameter:
14.3 mm
Center Thickness:
Minus Lens - varies with power (e.g. -4.00D: 0.070 mm)
Plus Lens - varies with power (e.g. +4.00D: 0.168 mm)
Base Curve:
8.4 mm
Powers:
-9.00D to +6.00D (in 0.25D increments)
ADD Powers:
+1.25 (LOW), +1.75 (MID), +2.50 (HGH)
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 161 of 194

ACUVUE OASYS® Brand Contact Lenses vs. 24 yr. old human cornea vs. 25 yr. old 
human crystalline lens
…….
ACUVUE OASYS® 
(senoﬁ lcon A)*
_____
24 YR. OLD 
HUMAN CORNEA1
- - - - - 
25 YR. OLD HUMAN 
CRYSTALLINE LENS2
*The data was obtained from measurements taken through the central 3-5 mm portion for the thinnest 
marketed lens (-1.00D lens, 0.070 mm center thickness).
1. Lerman, S., Radiant Energy and the Eye, MacMillan, New York, 1980, p.58, ﬁ gure 2-21
2. Waxler, M., Hitchins, V.M., Optical Radiation and Visual Health, CRC Press, Boca Raton, Florida,
 1986, p.19, ﬁ gure 5
WARNING: UV absorbing contact lenses are NOT substitutes for protec-
tive UV absorbing eyewear, such as UV absorbing goggles or sunglasses 
because they do not completely cover the eye and surrounding area. The 
patient should continue to use UV absorbing eyewear as directed.
ACTIONS
In its hydrated state, the contact lens, when placed on the cornea, acts as a 
refracting medium to focus light rays on the retina. When hydrated and placed on 
the cornea for therapeutic use, the contact lens acts as a bandage to protect the 
cornea.
The transmittance characteristics are less than 1% in the UVB range of 280 nm to 
315 nm and less than 10% in the UVA range of 316 nm to 380 nm for the entire 
power range.
NOTE: Long-term exposure to UV radiation is one of the risk factors 
associated with cataracts. Exposure is based on a number of factors 
such as environmental conditions (altitude, geography, cloud cover) and 
personal factors (extent and nature of outdoor activities). UV-Blocking 
contact lenses help provide protection against harmful UV radiation. 
TRANSMITTANCE CURVE
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 162 of 194

However, clinical studies have not been done to demonstrate that wearing 
UV-Blocking contact lenses reduces the risk of developing cataracts or 
other eye disorders. The Eye Care Professional should be consulted for 
more information.
INDICATIONS (USES)
The ACUVUE OASYS® Brand Contact Lens is indicated for the optical correction 
of refractive ametropia (myopia and hyperopia) in phakic or aphakic persons with 
non-diseased eyes who have 1.00D or less of astigmatism.
The ACUVUE OASYS® Brand Contact Lens for ASTIGMATISM is indicated for the 
optical correction of visual acuity in phakic or aphakic persons with non-diseased 
eyes that are hyperopic or myopic and may have 10.00D or less of astigmatism.
The ACUVUE OASYS® Brand Contact Lens for PRESBYOPIA is indicated for the 
optical correction of distance and near vision in presbyopic, phakic or aphakic 
persons with non-diseased eyes who may have 0.75D or less of astigmatism.
These lenses contain a UV Blocker to help protect against transmission of harmful 
UV radiation to the cornea and into the eye.
Eye Care Professionals may prescribe the lenses either for single-use dispos-
able wear or frequent/planned replacement wear with cleaning, disinfection and 
scheduled replacement (see “REPLACEMENT SCHEDULE”). When prescribed for 
frequent/planned replacement wear, the lenses may be cleaned and disinfected 
using a chemical disinfection system only.
These lenses have been approved for daily and extended wear for up to 
6 nights/7 days of continuous wear. It is recommended that the contact lens wearer 
ﬁ rst be evaluated on a daily wear schedule. If successful, then a gradual introduction 
of extended wear can be followed as determined by the prescribing Eye Care 
Professional.
These lenses are also indicated for therapeutic use as a bandage lens for the follow-
ing acute and chronic ocular conditions:
• For corneal protection in lid and corneal abnormalities such as entropion, 
trichiasis, tarsal scars, and recurrent corneal erosion. In addition, they are 
indicated for protection where sutures or ocular structure malformation, 
degeneration or paralysis may result in the need to protect the cornea from 
exposure or repeated irritation.
• For corneal pain relief in conditions such as bullous keratopathy, epithelial 
erosion and abrasion, ﬁ lamentary keratitis, and post-keratoplasty.
• For use as a barrier during the healing process of epithelial defects such 
as chronic epithelial defects, corneal ulcer, neurotrophic and neuroparalytic 
keratitis, and chemical burns.
• For post surgical conditions where bandage lens use is indicated such as 
post refractive surgery, lamellar grafts, corneal ﬂ aps, and additional ocular 
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 163 of 194

surgical conditions.
• For structural stability and protection in piggy back lens ﬁ tting where the cor-
nea and associated surfaces are too irregular to allow for corneal rigid gas 
permeable (RGP) lenses to be ﬁ t. In addition, the use of the lens can prevent 
irritation and abrasions in conditions where there are elevation differences in 
the host/graph junction or scar tissue. 
Lenses prescribed for therapeutic use may be worn for daily or extended wearing 
periods.
CONTRAINDICATIONS (REASONS NOT TO USE) 
When prescribing contact lens wear for REFRACTIVE AMETROPIA USE, DO NOT 
USE these lenses when any of the following conditions exist:
• Acute or subacute inﬂ ammation or infection of the anterior chamber of the 
eye
• Any eye disease, injury or abnormality that affects the cornea, conjunctiva 
or eyelids
• Severe insufﬁ ciency of lacrimal secretion (dry eye)
• Corneal hypoesthesia (reduced corneal sensitivity)
• Any systemic disease that may affect the eye or be exaggerated by wearing 
contact lenses
• Allergic reactions of ocular surfaces or adnexa that may be induced or exag-
gerated by wearing contact lenses or use of contact lens solutions
• Ocular irritation due to allergic reactions which may be caused by use of 
contact lens solutions (i.e., cleaning and disinfecting solutions, rewetting 
drops, etc.) that contain chemicals or preservatives (such as mercury or 
Thimerosal, etc.) to which some people may develop an allergic response
• Any active corneal infection (bacterial, fungal, protozoal or viral)
• If eyes become red or irritated
For THERAPEUTIC USE, the Eye Care Professional may prescribe these lenses to 
aid in the healing process of certain ocular conditions, which may include those 
cited above.
WARNINGS
Patients should be advised of the following warnings pertaining to contact 
lens wear:
EYE PROBLEMS, INCLUDING CORNEAL ULCERS, CAN DEVELOP RAPIDLY 
AND LEAD TO LOSS OF VISION; IF THE PATIENT EXPERIENCES:
• Eye Discomfort,
• Excessive Tearing,
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 164 of 194

• Vision Changes,
• Loss of Vision,
• Eye Redness,
• Or Other Eye Problems, 
THE PATIENT SHOULD BE INSTRUCTED TO IMMEDIATELY REMOVE THE 
LENSES AND PROMPTLY CONTACT THE EYE CARE PROFESSIONAL.
• When prescribed for daily wear, patients should be instructed not to wear 
lenses while sleeping. Clinical studies have shown that the risk of serious 
adverse reactions is increased when lenses are worn overnight, and that the risk 
of ulcerative keratitis is greater for extended wear contact lens users than for 
daily wear users.3
• Studies have shown that contact lens wearers who are smokers have a higher 
incidence of adverse reactions than nonsmokers.
• Problems with contact lenses or lens care products could result in serious injury 
to the eye. Patients should be cautioned that proper use and care of contact 
lenses and lens care products, including lens cases, are essential for the safe 
use of these products.
• The overall risk of ulcerative keratitis may be reduced by carefully following 
directions for lens care, including cleaning the lens case.
3 New England Journal of Medicine, September 21, 1989; 321 (12), pp. 773-783
Speciﬁ c Instructions for Use and Warnings:
• Water Activity
Instructions for Use
Do not expose contact lenses to water while wearing them.
WARNING:
Water can harbor microorganisms that can lead to severe infection, vision loss 
or blindness. If lenses have been submersed in water when participating in water 
sports or swimming in pools, hot tubs, lakes or oceans, the patient should be 
instructed to discard them and replace them with a new pair. The Eye Care 
Professional should be consulted for recommendations regarding wearing lenses 
during any activity involving water.
• Soaking and Storing Your Lenses
Instructions for Use
Use only fresh multi-purpose (contact lens disinfecting) solution each time the 
lenses are soaked (stored).
WARNING:
Do not reuse or “top off” old solution left in the lens case since solution reuse 
reduces effective lens disinfection and could lead to severe infection, vision loss, 
or blindness.
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 165 of 194

“Topping-Off” is the addition of fresh solution to solution that has been sitting 
the case.
• Discard Date on Multi-Purpose Solution Bottle
Instructions for Use
– Discard any remaining solution after the recommended time period indicated 
on the bottle of multi-purpose solution used for disinfecting and soaking the 
contact lenses.
– The Discard Date refers to the time that the patient can safely use contact 
lens care product after the bottle has been opened. It is not the same as the 
expiration date, which is the last date that the product is still effective before 
it is opened.
WARNING:
Using multi-purpose solution beyond the discard date could result in contami-
nation of the solution and can lead to severe infection, vision loss, or blindness.
– To avoid contamination, DO NOT touch tip of container to any surface. 
Replace cap after using.
– To avoid contaminating the solution, DO NOT transfer to other bottles or 
containers.
• Rub and Rinse Time
Instructions for Use
To adequately disinfect the lenses, the patient should rub and rinse the lenses 
according to the recommended lens rubbing and rinsing times in the labeling of 
the multi-purpose solution.
WARNING:
– Rub and rinse lenses for the recommended amount of time to help prevent 
serious eye infections.
– Never use water, saline solution, or rewetting drops to disinfect the lenses. 
These solutions will not disinfect the lenses. Not using the recommended 
disinfectant can lead to severe infection, vision loss, or blindness.
• Lens Case Care
Instructions for Use
– Empty and clean contact lens cases with digital rubbing using fresh, sterile 
disinfecting solutions/contact lens cleaner. Never use water. Cleaning should 
be followed by rinsing with fresh, sterile disinfecting solutions (never use 
water) and wiping the lens cases with fresh, clean tissue is recommended. 
Never air dry or recap the lens case lids after use without any additional 
cleaning methods. If air drying, be sure that no residual solution remains in 
the case before allowing it to air dry.
– Replace the lens case according to the directions provided by the Eye Care 
Professional or the manufacturer’s labeling that accompanies the case.
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 166 of 194

– Contact lens cases can be a source of bacterial growth.
WARNING:
Do not store lenses or rinse lens cases with water or any non-sterile solution. 
Only fresh multi-purpose solution should be used to prevent contamination of 
the lenses or lens case. Use of non-sterile solution can lead to severe infection, 
vision loss, or blindness.
PRECAUTIONS
Special Precautions for Eye Care Professionals:
• Due to the small number of patients enrolled in clinical investigation of lens-
es, all refractive powers, design conﬁ gurations, or lens parameters available 
in the lens material are not evaluated in signiﬁ cant numbers. Consequently, 
when selecting an appropriate lens design and parameters, the Eye Care 
Professional should consider all characteristics of the lens that can affect 
lens performance and ocular health, including oxygen permeability, wettabili-
ty, central and peripheral thickness, and optic zone diameter.
The potential impact of these factors on the patient’s ocular health should 
be carefully weighed against the patient’s need for refractive correction; 
therefore, the continuing ocular health of the patient and lens performance 
on the eye should be carefully monitored by the prescribing Eye Care 
Professional.
• Patients who wear these lenses to correct presbyopia using monovision 
may not achieve the best corrected visual acuity for either far or near vision. 
Visual requirements vary with the individual and should be considered when 
selecting the most appropriate type of lens for each patient.
• Fluorescein, a yellow dye, should not be used while the lenses are on 
the eyes. The lenses absorb this dye and become discolored. Whenever 
ﬂ uorescein is used in eyes, the eyes should be ﬂ ushed with a sterile saline 
solution that is recommended for in-eye use.
• Eye Care Professionals should instruct the patient to remove lenses immedi-
ately if the eyes become red or irritated. 
Eye Care Professionals should carefully instruct patients about the follow-
ing care regimen and safety precautions:
Handling Precautions:
• Before leaving the Eye Care Professional’s ofﬁ ce, the patient should be able 
to promptly remove the lenses or should have someone else available who 
can remove the lenses for him or her.
• DO NOT use if the sterile blister package is opened or damaged.
• Always wash and rinse hands 
ndling lenses. Do not get cosmetics, 
lotions, soaps, creams, deodo
or sprays in the eyes or on the lenses. It 
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 167 of 194

is best to put on lenses before putting on makeup. Water-based cosmetics 
are less likely to damage lenses than oil-based products.
• DO NOT touch contact lenses with the ﬁ ngers or hands if the hands are not 
free of foreign materials, as microscopic scratches of the lenses may occur, 
causing distorted vision and/or injury to the eye.
• Carefully follow the handling, insertion, removal, and wearing instructions in 
the Patient Instruction Guide for these lenses and those prescribed by the 
Eye Care Professional.
• Always handle lenses carefully and avoid dropping them.
• Never use tweezers or other tools to remove lenses from the lens container 
unless speciﬁ cally indicated for that use. Slide the lens up the side of the 
bowl until it is free of the container.
• Do not touch the lens with ﬁ ngernails.
• Close supervision is necessary for the Therapeutic use of these lenses. 
Ocular medications used during treatment with a bandage lens should be 
closely monitored by the Eye Care Professional. In certain ocular conditions, 
only the Eye Care Professional will insert and remove the lenses. In these 
cases, patients should be instructed not to handle the lenses themselves.
Lens Wearing Precautions:
• If the lens sticks (stops moving) on the eye, follow the recommended 
directions in “Care for Sticking (Non-Moving) Lenses”. The lens should move 
freely on the eye for the continued health of the eye. If non-movement of the 
lens continues, the patient should be instructed to immediately consult his 
or her Eye Care Professional.
• Never wear lenses beyond the period recommended by the Eye Care 
Professional.
• The patient should be advised to never allow anyone else to wear their 
lenses. They have been prescribed to ﬁ t their eyes and to correct their vision 
to the degree necessary. Sharing lenses greatly increases the chance of eye 
infections.
• If aerosol products, such as hair spray, are used while wearing lenses, 
exercise caution and keep eyes closed until the spray has settled.
• Avoid all harmful or irritating vapors and fumes while wearing lenses.
Lens Care Precautions:
• Different solutions cannot always be used together and not all solutions are 
safe for use with all lenses. Use only recommended solutions.
• Never use solutions recommended for conventional hard contact lenses 
only.
• Chemical disinfection solutions should not be used with heat unless 
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 168 of 194

speciﬁ cally indicated on product labeling for use in both heat and chemical 
disinfection.
• Always use fresh, unexpired lens care solutions and lenses.
• Do not change solution without consulting with your Eye Care Professional.
• Always follow directions in the package inserts for the use of contact lens 
solutions.
• Use only a chemical (not heat) lens care system. Use of a heat (thermal) care 
system can damage these lenses.
• Sterile unpreserved solutions, when used, should be discarded after the 
time speciﬁ ed in the directions.
• Do not use saliva or anything other than the recommended solutions for 
lubricating or wetting lenses.
• Always keep the lenses completely immersed in the recommended storage 
solution when the lenses are not being worn (stored). Prolonged periods of 
drying will reduce the ability of the lens surface to return to a wettable state. 
Follow the lens care directions in "Care For A Dried Out (Dehydrated) Lens" 
if lens surface does become dried out.
Other Topics to Discuss with Patients:
• Always contact the Eye Care Professional before using any medicine in the 
eyes.
• Certain medications, such as antihistamines, decongestants, diuretics, 
muscle relaxants, tranquilizers, and those for motion sickness may cause 
dryness of the eye, increased lens awareness, or blurred vision. Should such 
conditions exist, proper remedial measures should be prescribed. Depend-
ing on the severity, this could include the use of lubricating drops that are 
indicated for use with soft contact lenses or the temporary discontinuance 
of contact lens wear while such medication is being used.
• Oral contraceptive users could develop visual changes or changes in 
lens tolerance when using contact lenses. Patients should be cautioned 
accordingly.
• As with any contact lens, follow-up visits are necessary to assure the con-
tinuing health of the patient’s eyes. The patient should be instructed as to a 
recommended follow-up schedule.
Who Should Know That the Patient is Wearing Contact Lenses?
• Patients should inform all doctors (Health Care Professionals) about being a 
contact lens wearer.
• Patients should always inform their employer of being a contact lens wearer. 
Some jobs may require use of
ction equipment or may require that 
the patient not wear contact le
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 169 of 194

ADVERSE REACTIONS
The patient should be informed that the following problems may occur 
when wearing contact lenses:
• The eye may burn, sting and/or itch.
• There may be less comfort than when the lens was ﬁ rst placed on the eye.
• There may be a feeling of something in the eye (foreign body, scratched 
area).
• There may be the potential for some temporary impairment due to 
peripheral inﬁ ltrates, peripheral corneal ulcers, and corneal erosion. There 
may be the potential for other physiological observations, such as local or 
generalized edema, corneal neovascularization, corneal staining, injection, 
tarsal abnormalities, iritis and conjunctivitis, some of which are clinically 
acceptable in low amounts.
• There may be excessive watering, unusual eye secretions or redness of 
the eye.
• Poor visual acuity, blurred vision, rainbows or halos around objects, photo-
phobia, or dry eyes may also occur if the lenses are worn continuously or 
for too long a time. 
• The patient should be instructed to conduct a simple 3-part self-examina-
tion at least once a day. They should ask themselves:
• How do the lenses feel on my eyes?
• How do my eyes look?
• Have I noticed a change in my vision?
If the patient reports any problems, he or she should be instructed to IMMEDIATE-
LY REMOVE THE LENS.  If the problem or discomfort stops, the patient should 
discard the lens and place a new fresh lens on the eye.
If after inserting the new lens, the problem continues, the patient should be 
directed to IMMEDIATELY REMOVE THE LENS AND CONTACT HIS OR HER EYE 
CARE PROFESSIONAL.
The patient should be instructed NOT to use a new lens as self-treatment for the 
problem. 
The patient should be advised that when any of the above symptoms occur, a 
serious condition such as infection, corneal ulcer, neovascularization or iritis may 
be present. He or she should be instructed to seek immediate professional identiﬁ -
cation of the problem and prompt treatment to avoid serious eye damage.
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 170 of 194

GENERAL FITTING GUIDELINES
A.  Patient Selection:
Patients selected to wear these lenses should be chosen based on:
• Motivation to wear lenses
• Ability to follow instructions regarding lens wear care
• General health
• Ability to adequately handle and care for the lenses
• Ability to understand the risk and beneﬁ ts of lens wear
Patients who do not meet the above criteria should not be provided with 
contact lenses.
B. Pre-ﬁ tting Examination:
Initial evaluation of the patient should begin with a thorough case history to 
determine if there are any contraindications to contact lens wear. During the 
case history, the patient’s visual needs and expectations should be determined 
as well as an assessment of their overall ocular, physical, and mental health.
Preceding the initial selection of trial contact lenses, a comprehensive ocular 
evaluation should be performed that includes, but is not limited to, the mea-
surement of distance and near visual acuity, distance and near refractive pre-
scription (including determining the preferred reading distance for presbyopes), 
keratometry and biomicroscopic evaluation.
Based on this evaluation, if it is determined that the patient is eligible to wear 
these lenses, the Eye Care Professional should proceed to the appropriate lens 
ﬁ tting instruction outlined below.
C. Initial Power Determination
A spectacle refraction should be performed to establish the patient’s baseline 
refractive status and to guide in the selection of the appropriate lens power. 
Remember to compensate for vertex distance if the refraction is greater than 
±4.00D.
D. Base Curve Selection (Trial Lens Fitting)
The following trial lenses should be selected for patients regardless of keratome-
try readings. However, corneal curvature measurements should be performed to 
establish the patient’s baseline ocular status. 
• ACUVUE OASYS®: 8.4 mm/14.0 mm
• ACUVUE OASYS® for ASTIGMATISM: 8.6 mm/14.5 mm
• ACUVUE OASYS® for PRESBYOPIA: 8.4 mm/14.3 mm
The trial lenses should be placed on each of the patient’s eyes and evaluated 
after the patient has adjusted to the lenses.
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 171 of 194

1. Criteria of a Properly Fit Lens
A properly ﬁ t lens will center and completely cover the cornea (i.e., no limbal ex-
posure), have sufﬁ cient movement to provide tear exchange under the contact 
lens with the blink, and be comfortable. The lens should move freely when 
manipulated digitally with the lower lid, and then return to its properly centered 
position when released. 
2. Criteria of a Flat Fitting Lens
A ﬂ at ﬁ tting lens may exhibit one or more of the following characteristics: 
decentration, incomplete corneal coverage (i.e., limbal exposure), excessive 
movement with the blink and/or edge standoff. If the lens is judged to be ﬂ at 
ﬁ tting, it should not be dispensed to the patient.
3. Criteria of a Steep Fitting Lens
A steep ﬁ tting lens may exhibit one or more of the following characteristics: 
insufﬁ cient movement with the blink, conjunctival indentation, and resistance 
when pushing the lens up digitally with the lower lid. If the lens is judged to be 
steep ﬁ tting, it should not be dispensed to the patient.
If the initial trial base curve is judged to be ﬂ at or steep ﬁ tting, the alternate base 
curve, if available, should be trial ﬁ t and evaluated after the patient has adjusted 
to the lens. The lens should move freely when manipulated digitally with lower 
lid, and then return to a properly centered position when released. If resistance 
is encountered when pushing the lens up, the lens is ﬁ tting tightly and should 
not be dispensed to the patient.
E. Final Lens Power (Spherical)
A spherical over-refraction should be performed to determine the ﬁ nal lens 
power after the lens ﬁ t is judged acceptable. The spherical over-refraction 
should be combined with the trial lens power to determine the ﬁ nal lens pre-
scription. The patient should experience good visual acuity with the correct lens 
power unless there is excessive residual astigmatism.
Example 1
Diagnostic lens:
-2.00D
Spherical over-refraction: 
-0.25D
Final lens power: 
-2.25D
Example 2
Diagnostic lens:
-2.00D
Spherical over-refraction: 
+0.25D
Final lens power: 
-1.75D
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 172 of 194

If vision is acceptable, perform a slit lamp examination to assess adequate ﬁ t 
(centration and movement). If ﬁ t is acceptable, dispense the lenses instructing 
the patient to return in one week for reassessment (see dispensing and follow-up 
information in PATIENT MANAGEMENT).
All patients should be supplied with a copy of the PATIENT INSTRUCTION 
GUIDE for these lenses. Copies are available for download at 
www.acuvue.com.
TORIC FITTING GUIDELINES
Although most aspects of the ﬁ tting procedure are identical for all types of soft 
contact lenses, including torics, there are some additional steps and/or rules to 
follow to assure the proper ﬁ t of toric lenses.
The only new steps you must follow in prescribing ACUVUE OASYS® for ASTIG-
MATISM contact lenses are that you must determine the stability, repeatability and 
drift angle of the lens axis so that you can prescribe the correct lens axis for your 
patient.
A.  How to Determine Lens Cylinder and Axis Orientation 
1. Locate the Orientation Marks
To help determine the proper orientation of the toric lens, you’ll ﬁ nd two primary 
marks about 1 mm from the lens edge representing the vertical position on 
opposite ends of the lens at 6 and 12 o’clock (Fig. 1). Because of the lens’ 
ballasting system, either mark can represent the vertical position – there is no 
“top” and “bottom” as in a prism-ballasted lens. You don’t need to view both 
marks to assess orientation; simply look for the 6 o’clock mark as you would 
with a prism-ballasted lens.
Figure 1
You’ll need a biomicroscope and a 1 mm or 2 mm parallelepiped beam to 
highlight the marks when the lens is ﬁ tted to the eye. There are a number of 
techniques you can use to improve the visibility of the 6 o’clock mark. Using a 
parallelepiped beam and medium magniﬁ cation (10x or 15x), slowly pan down 
the lens, looking just below the direct illumination at the retroilluminated area. 
Backlighting the mark this way should make it more visible. Sometimes manip-
ulating the lower lid may be necessary to uncover the mark.
2. Observe Lens Rotation and Stability
Observe the position and stability of the “bottom” mark. It usually stabilizes at 
the 6 o’clock position. If it does, calculation of the lens power will be straightfor-
ward. The 6 o’clock position is not a “must”; however, the absolute requirement 
is that the axis position be stable and repeatable.
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 173 of 194

The mark may stabilize somewhat left or right (drift) of the vertical meridian and 
still enable you to ﬁ t a toric lens for that eye, as long as the lens always returns 
to the same “drift axis” position after settling. The deviation can be compen-
sated for in the ﬁ nal prescription. Your objective is to ensure that whatever 
position the initial lens assumes near 6 o’clock, this position must be stable 
and repeatable. With full eye movement or heavy blink, you may see the marks 
swing away, but they must return quickly to the original stable position. If the 
lens does not return quickly, you may need to select a different lens.
Assessing Rotation
Imagine the eye as a clock dial and every hour represents a 30° interval. If the 
orientation mark of the initial lens stabilizes somewhat left or right of the vertical 
position, the ﬁ nal lens will orient on the eye with the same deviation. You can 
use an axis reticule in the slit lamp or use a line-scribed lens in a spectacle trial 
frame to measure or estimate the “drift angle” of the cylinder axis.
To compensate for this “drift”, measure or estimate the “drift”, then add or 
subtract it from the refractive axis to determine the correct cylinder axis. Use 
the LARS (Left Add, Right Subtract) method to determine which direction to 
compensate.
B.  How to Determine the Final Lens Power 
When the diagnostic lens has its axis aligned in the same meridian as the 
patient’s refractive axis, a spherocylindrical over-refraction may be performed 
and visual acuity determined. However, in the case of crossed axes, such as 
when the diagnostic lens axis is different from the patient’s refractive axis, it is 
not advisable to over-refract because of the difﬁ culty in computing the resultant 
power. In ﬁ tting contact lenses, it is customary to prescribe the full power in the 
sphere. In the cylinder, however, any lens rotation is visually disturbing to the 
patient, so it’s more practical to prescribe as weak a cylinder as possible.
So, here is how to determine the ﬁ nal lens power.
For the Sphere:
If sphere alone or combined sphere and cylinder Rx >±4.00D, compensate for 
vertex distance. If sphere alone or combined sphere and cylinder Rx ≤± 4.00D, 
vertex compensation is not necessary.
For the Cylinder:
Adjust the axis by the drift angle using LARS. Choose a cylinder that is
≤ 0.25D from the refractive cylinder.
Case Examples:
Example 1
Manifest (spectacle) refraction:
O.D. –2.50 -1.25 x 180 20/20
O.S. –2.00 -1.00 x 180 20/20
Choose a diagnostic lens for each eye with an axis as close to 180° as 
possible. Place the lens on each eye and allow a minimum of 3 minutes for 
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 174 of 194

it to equilibrate, based on the patient’s initial response to the lens. If the lens 
has not yet stabilized, recheck until stable.
Check the orientation of the axis mark. If the bottom axis mark is in the 6 
o’clock position on both eyes, choose the appropriate cylinder as listed 
previously. If the lens has not yet stabilized, recheck until stable.
Here is the Rx Prescribed:
O.D. -2.50 -1.25 x 180
O.S. –2.00 -0.75 x 180
Example 2
Manifest (spectacle) refraction:
O.D. –3.00 -1.00 x 90 20/20
O.S. –4.75 -2.00 x 90 20/20
Choose diagnostic lenses of –3.00 –0.75 x 90 for the right eye and –4.50 
-1.75 x 90 for the left eye, the nearest lenses available to the spherical 
power and axis needed. Place the lens on each eye and allow a minimum of 
3 minutes for it to equilibrate, based on the patient's initial response to the 
lens.  If the lens has not yet stabilized, recheck until stable. The orientation 
mark on the right lens rotates left from the 6 o’clock position by 10°.
The ﬁ tting indicates the following:
Right Eye:
Compensate the 10° axis drift by adding it to the manifest refraction axis.
Here is the Rx prescribed:
O.D. –3.00 –0.75 x 100
Left Eye
The lens on the left eye shows good centration, movement and a consistent  
tendency for the mark to drift right by 10° from the 6 o’clock position 
following a forced blink. Since the manifest refraction called for a power of 
–4.75D, adjust for the vertex distance and reduce the sphere by 0.25D and 
prescribe the –1.75D cylinder. Compensate for the 10° axis drift by subtract-
ing it from the manifest  refraction. 
Here is the Rx Prescribed: 
O.S. –4.50 –1.75 x 80
If vision is acceptable, perform a slit lamp examination to assess adequate 
ﬁ t (centration and movement). If ﬁ t is acceptable, dispense the lenses in-
structing the patient to return in one week for reassessment (see dispensing  
and follow up information in PATIENT MANAGEMENT).
All patients should be supplied with a copy of the PATIENT INSTRUCTION 
GUIDE for these lenses. Copies are available for download at 
www.acuvue.com.
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 175 of 194

MULTIFOCAL FITTING GUIDELINES
A. Presbyopic Needs Assessment & Patient Education
Multifocal contact lenses may produce compromise to vision under certain 
circumstances and the patient should understand that they might not ﬁ nd their 
vision acceptable in speciﬁ c situations (i.e., reading a menu in a dim restaurant, 
driving at night in rainy/foggy conditions, etc.). Therefore, caution should be 
exercised when the patient is wearing the correction for the ﬁ rst time until 
they are familiar with the vision provided in visually challenging environments. 
Occupational and environmental visual demands should be considered. If the 
patient requires critical visual acuity and stereopsis, it should be determined by 
trial whether this patient can function adequately with the ACUVUE OASYS® for 
PRESBYOPIA contact lenses. Wear may not be optimal for activities such as: 
1. visually demanding situations such as operating potentially dangerous
machinery or performing other potentially hazardous activities; and
2. driving automobiles (e.g., driving at night). Patients who cannot pass their 
state driver's license requirements with these lenses should be advised to 
not drive with this correction, OR may require that additional over-correction 
be prescribed.
These lenses are not recommended for patients who have –1.00D or greater 
of refractive cylinder as this level of uncorrected cylinder may lead to additional 
visual compromise.
These lenses are available in the following ADD powers:
• Lens “LOW” = “low” near ADD lens (Max +1.25 ADD)
• Lens “MID” = “medium” near ADD lens  (Max +1.75 ADD)
• Lens “HGH” = “high” near ADD lens  (Max +2.50 ADD)
B. Fitting Instructions
1. Determine the following:
• Eye dominance (the methods described in MONOVISION FITTING GUIDE-
LINES may be used)
• Spherical equivalent distance prescription (vertex corrected if necessary and 
rounded to less minus if between powers)
• Near ADD
2 . Select the initial trial lens as follows:
• For each eye select the trial lens distance power that is closest to the 
patient’s distance spherical equivalent
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 176 of 194

• Select the near power of the lens based on the patient's ADD range as 
follows:
ADD:  +0.75 to +1.25 use a “LOW” near ADD lens on each eye
ADD:  +1.50 to +1.75 use a “MID” near ADD lens on each eye
ADD:  +2.00 to +2.50 use a “HGH” near ADD lens on each eye
3. Allow the lens to settle for a minimum 10 minutes.
4. Assess distance and near vision binocularly and monocularly.
• Demonstrate the vision under various lighting conditions (normal and de-
creased illumination) and at distance, intermediate and near.
• Make adjustments in power as necessary (see Multifocal Troubleshooting 
below). The use of hand-held trial lenses is recommended. 
• If distance and near vision are acceptable, perform a slit lamp examination to 
assess adequate ﬁ t (centration and movement). If ﬁ t is acceptable, dispense 
the lenses instructing the patient to return in one week for reassessment (see 
dispensing and follow-up information in PATIENT MANAGEMENT).
C. Multifocal Troubleshooting
Unacceptable Near Vision:
Determine the amount of additional plus, or less minus, over one or both eyes 
that is acceptable, while checking the effect on distance and near vision.  If 
vision is still not acceptable, change the non-dominant eye to the next highest 
ADD power. 
Unacceptable Distance Vision:
Determine the amount of additional minus, or less plus, over one or both eyes 
that is acceptable while checking the effect on distance and near vision.  If 
vision is still not acceptable, change the dominant eye to the next lowest ADD 
power.  If the patient is wearing two low ADD lenses, change the dominant 
eye to a sphere lens with a power equal to the spherical equivalent distance 
prescription.
Unacceptable Distance and Near Vision:
Determine the amount of additional plus and/or minus over one or both eyes 
that is acceptable while checking the effect on distance and near vision.  If 
additional plus and/or minus is not required, change the lens power in the dom-
inant eye to the next lowest ADD power and the lens power in the non-domi-
nant eye to the next highest ADD power, if applicable.
All patients should be supplied with a copy of the PATIENT INSTRUCTION 
GUIDE for these lenses. Copies are available for download at 
www.acuvue.com.
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 177 of 194

MONOVISION FITTING GUIDELINES
A. Patient Selection
Monovision Needs Assessment
For a good prognosis, the patient should have adequately corrected distance 
and near visual acuity in each eye. The amblyopic patient or the patient with 
signiﬁ cant astigmatism (greater than 1.00D) in one eye may not be a good 
candidate for monovision correction with these lenses.
Occupational and environmental visual demands should be considered. If the 
patient requires critical vision (visual acuity and stereopsis), it should be deter-
mined by trial whether this patient can function adequately with monovision 
correction. Monovision contact lens wear may not be optimal for activities such 
as:
1.  visually demanding situations such as operating potentially dangerous 
machinery or performing other potentially hazardous activities; and
2.  driving automobiles (e.g., driving at night). Patients who cannot pass their 
state driver's license requirements with monovision correction should be 
advised to not drive with this correction, OR may require that additional 
over-correction be prescribed.
Patient Education
All patients do not function equally well with monovision correction. Patients 
may not perform as well for certain tasks with this correction as they have with 
spectacles (multifocal, bifocal, trifocal, readers, progressives). Each patient 
should understand that monovision, as well as other presbyopic alternatives, 
can create a vision compromise that may reduce visual acuity and depth 
perception for distance and near tasks. Therefore, caution should be exercised 
when the patient is wearing the correction for the ﬁ rst time until they are familiar 
with the vision provided in visually challenging environments (e.g., reading 
a menu in a dim restaurant, driving at night in rainy/foggy conditions, etc.). 
During the ﬁ tting process, it is necessary for the patient to realize the disad-
vantages as well as the advantages of clear near vision and straight ahead and 
upward gaze that monovision contact lenses provide.
B. Eye Selection
Generally, the non-dominant eye is corrected for near vision. The following two 
methods for eye dominance can be used.
1. Ocular Preference Determination Methods
Method 1: Determine which eye is the “sighting eye.” Have the patient point 
to an object at the far end of the room. Cover one eye. If the 
patient is still pointing directly at the object, the eye being used is 
the dominant (sighti
e.
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 178 of 194

Allow the lenses to settle for about 20 minutes with the correct power lenses 
in place. Walk across the room and have the patient look at you. Assess the 
patient’s reaction to distance vision under these circumstances. Then have the 
patient look at familiar near objects such as a watch face or ﬁ ngernails. Again 
assess the reaction. As the patient continues to look around the room at both 
near and distance objects, observe the reactions. Only after these vision tests 
are completed should the patient be asked to read print. Evaluate the patient’s 
reaction to large print (e.g., typewritten copy) at ﬁ rst and then graduate to 
newsprint and ﬁ nally smaller type sizes.
After the patient’s performance under the above conditions is completed, tests 
of visual acuity and reading ability under conditions of moderately dim illumina-
tion should be attempted.
An initial unfavorable response in the ofﬁ ce, while indicative of a guarded prog-
nosis, should not immediately rule out a more extensive trial under the usual 
conditions in which a patient functions.
4. Adaptation
Visually demanding situations should be avoided during the initial wearing pe-
riod. A patient may at ﬁ rst experience some mild blurred vision, dizziness, head-
aches, and a feeling of slight imbalance. You should explain the adaptational 
symptoms to the patient. These symptoms may last for a brief minute or for 
several weeks. The longer these symptoms persist, the poorer the prognosis 
for successful adaptation.
To help in the adaptation process, the patient can be advised to ﬁ rst use the 
lenses in a comfortable familiar environment such as in the home.
Some patients feel that automobile driving performance may not be optimal 
during the adaptation process. This is particularly true when driving at night. 
Before driving a motor vehicle, it may be recommended that the patient be a 
passenger ﬁ rst to make sure that their vision is satisfactory for operating an au-
tomobile. During the ﬁ rst several weeks of wear (when adaptation is occurring), 
it may be advisable for the patient to only drive during optimal driving condi-
tions. After adaptation and success with these activities, the patient should be 
able to drive under other conditions with caution.
D. Other Suggestions
The success of the monovision technique may be further improved by having the 
patient follow the suggestions below:
• Have a third contact lens (distance power) to use when critical distance viewing 
is needed.
• Have a third contact lens (near power) to use when critical near viewing is 
needed.
• Having supplemental spectacles 
ver the monovision contact lenses 
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 180 of 194

for speciﬁ c visual tasks may improve the success of monovision correction. 
This is particularly applicable for those patients who cannot pass state drivers 
licensing requirements with monovision correction.
• Make use of proper illumination when carrying out visual tasks.
Monovision ﬁ tting success can be improved by the following suggestions:
• Reverse the distance and near eyes if a patient is having trouble adapting.
• Reﬁ ne the lens powers if there is trouble with adaptation. Accurate lens power 
is critical for presbyopic patients.
• Emphasize the beneﬁ ts of clear near vision and straight ahead and upward 
gaze with monovision.
The decision to ﬁ t a patient with a monovision correction is most appropriately left 
to the Eye Care Professional in conjunction with the patient after carefully consid-
ering the patient’s needs.
All patients should be supplied with a copy of the PATIENT INSTRUC-
TION GUIDE for these lenses. Copies are available for download at www.
acuvue.com.
PATIENT MANAGEMENT
Dispensing Visit
• PROVIDE THE PATIENT WITH A COPY OF THE PATIENT INSTRUCTION 
GUIDE FOR THESE LENSES. REVIEW THESE INSTRUCTIONS WITH THE 
PATIENT SO THAT HE OR SHE CLEARLY UNDERSTANDS THE PRESCRIBED 
WEARING AND REPLACEMENT SCHEDULE (DISPOSABLE OR FREQUENT 
REPLACEMENT).
• Recommend an appropriate cleaning and disinfecting system and provide the 
patient with instructions regarding proper lens care. Chemical or hydrogen 
peroxide disinfection is recommended.
• Schedule a follow-up examination.
Follow-up Examinations
• Follow-up care (necessary to ensure continued successful contact lens wear) 
should include routine periodic progress examinations, management of speciﬁ c 
problems, if any, and a review with the patient of the wear schedule, lens 
replacement schedule, and proper lens care and handling procedures.
• Recommended Follow-up Examination Schedule (complications and speciﬁ c 
problems should be managed on an individual patient basis):
1. One week from the initial lens dispensing to patient
2. One month post-dispensing
3. Every three to six months there
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 181 of 194

NOTE:  More frequent or additional follow-up visits may be recommended for 
patients on an extended wear schedule.
• Preferably, at the follow-up visits, lenses should be worn for at least six hours. 
If the lenses are being worn for continuous wear, the examination should be 
performed as early as possible on the morning following overnight wear.
• Recommended Procedures for Follow-Up Visits:
1. Solicit and record patient’s symptoms, if any.
2. Measure visual acuity monocularly and binocularly at distance and near with 
the contact lenses.
3. Perform an over-refraction at distance and near to check for residual 
refractive error.
4. With the biomicroscope, judge the lens ﬁ tting characteristics (as described 
in the GENERAL FITTING GUIDELINES) and evaluate the lens surface for 
deposits and damage.
5. Following lens removal, examine the cornea and conjunctiva with the biomi-
croscope and ﬂ uorescein (unless contraindicated).
• The presence of vertical corneal striae in the posterior central cornea and/
or corneal neovascularization is indicative of excessive corneal edema.
• The presence of corneal staining and/or limbal-conjunctival hyperemia 
can be indicative of an unclean lens, a reaction to solution preservatives, 
excessive lens wear, and/or a poorly ﬁ tting lens.
• Papillary conjunctival changes may be indicative of an unclean and/or 
damaged lens.
6. Periodically perform keratometry and spectacle refractions. The values 
should be recorded and compared to the baseline measurements.
If any observations are abnormal, use professional judgment to alleviate 
the problem and restore the eye to optimal conditions. If the criteria for 
successful ﬁ t are not satisﬁ ed during any follow-up examinations, repeat the 
patient’s trial ﬁ tting procedure and reﬁ t the patient.
The wearing and replacement schedules should be determined by the Eye Care 
Professional. Regular checkups, as determined by the Eye Care Professional, are 
also extremely important.
For Daily Wear:
Patients tend to overwear the lenses initially. The Eye Care Professional should 
emphasize the importance of adhering to the initial maximum wearing schedule.  
Maximum wearing time should be determined by the Eye Care Professional based 
upon the patient’s physiological eye 
because individual response to 
contact lenses varies. 
WEARING SCHEDULE
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 182 of 194

The maximum suggested wearing time for these lenses is:
DAY
HOURS
1
6-8
2
8-10
3
10-12
4
12-14
5 and after
all waking hours
For Extended Wear:
It is recommended that the contact lens wearer ﬁ rst be evaluated on a daily wear 
schedule.  If successful, then a gradual introduction of extended wear can be 
followed as determined by the prescribing Eye Care Professional. 
These lenses have been approved for extended wear up to 6 nights/7 days of 
continuous wear.  Not all patients can achieve the maximum wear time.
For Therapeutic lens wear, close supervision by the Eye Care Professional is 
necessary. These lenses can be worn for extended wear for up to 6 nights/7 days 
of continuous wear. The Eye Care Professional should determine the appropriate 
wearing time and provide speciﬁ c instructions to the patient regarding lens care, 
insertion, and removal.
REPLACEMENT SCHEDULE
For Lenses Prescribed for Frequent Replacement:
When prescribed for daily wear (frequent replacement), it is recommended that the 
lenses be discarded and replaced with a new lens every 2 weeks.  However, the Eye 
Care Professional is encouraged to determine an appropriate replacement schedule 
based upon the response of the patient.
For Lenses Prescribed for Disposable Wear: 
When prescribed for disposable wear, the replacement schedule should be deter-
mined by the Eye Care Professional based upon the patient’s history and their ocular 
examination, as well as the practitioner’s experience and clinical judgment.
Once removed, it is recommended that the lens remain out of the eye for a period 
of rest of overnight or longer and be discarded in accordance with the prescribed 
wearing schedule. The Eye Care Professional should examine the patient during the 
early stages of extended wear.
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 183 of 194

LENS CARE DIRECTIONS
When lenses are dispensed, the Eye Care Professional should provide the patient 
with appropriate and adequate warnings and instructions in accordance with the 
individual patient’s lens type and wearing schedule. The Eye Care Professional 
should recommend an appropriate care system tailored to the patient’s individual 
requirements.
For complete information concerning contact lens handling, care, cleaning, disin-
fecting and storage, refer to the Patient Instruction Guide for these lenses.  Copies 
are available for download at www.acuvue.com.
For Lenses Prescribed for Frequent Replacement Wear: 
The Eye Care Professional should review with the patient, lens care directions for 
cleaning, disinfecting and storing, including both basic lens care information and 
speciﬁ c instructions on the lens care regimen recommended for the patient.
For Lenses Prescribed for Disposable Wear: 
The Eye Care Professional should review with patients that no cleaning or disinfec-
tion is needed with disposable lenses. Patients should always dispose of lenses 
when they are removed and have replacement lenses or spectacles available. 
Lenses should only be cleaned, rinsed, and disinfected on an emergency basis 
when replacement lenses or spectacles are not available.
Care for a Dried Out (Dehydrated) Lens
If the frequent replacement lens is off the eye and exposed to air from 30 minutes 
to 1 hour or more, its surface will become dry and gradually become non-wetting. If 
this should occur, discard the lens and use a new one.
Care for Sticking (Non-Moving) Lenses
If the lens sticks (stops moving), the patient should be instructed to apply a 
few drops of the recommended lubricating or rewetting solution directly to the 
eye and wait until the lens begins to move freely on the eye before removing it. 
If non-movement of the lens continues after a few minutes, the patient should 
immediately contact the Eye Care Professional.
 
 
 
 
 
 
EMERGENCIES
The patient should be informed that if chemicals of any kind (household products, 
gardening solutions, laboratory chemicals, etc.) are splashed into the eyes, the pa-
tient should: FLUSH EYES IMMEDIATELY WITH TAP WATER AND IMMEDIATELY 
CONTACT THE EYE CARE PROFESSIONAL OR VISIT A HOSPITAL EMERGENCY 
ROOM WITHOUT DELAY.
C
5638 Protocol v6.0, Amend
JJVC Confidential
Page 184 of 194

16.6 
Appendix B:  IRIS COLOR SCALE 
 
 
 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 187 of 194

16.7 
Appendix C:  TEST ARTICLE ACCOUNTABILITY IN THE EDC SYSTEM 
 
Part 1 – Overall Test Article Accountability 
y
Once the test article shipped sheet is completed for the shipment the investigational site must log the 
shipment into the lens depot in the EDC system. 
1. Log into the Study Database in BioClinica Express 5.4 
 
 
2. Click on Site Level Data 
    
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 188 of 194

 
3. Click on Site Shipment Article Log 
 
x
All lenses shipped to the site are listed in the Site Shipment Article Log 
 
4. Click on to open each form to receive the lenses. 
*Lenses MUST be received before the First Subject is seen. 
5. Enter Total Quantity Received & save each eCRF.  
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 189 of 194

 
*If Quantity shipped is different from Total Quantity Received, please notify your regional CRA immediately (a 
query will populate if this occurs to be resolved)  
y
At the end of the study you will need to return all unused test article 
y
At your close out visit your monitor will help you complete a Test Article Return Worksheet, documenting 
how much test article is being returned.   
y
Once the Test Article Return Worksheet is completed, you will need to enter the quantity of lenses being 
returned into the lens depot in the EDC system. 
6. Enter Total Quantity Returned after the study is completed and lens accountability has 
been completed on paper. 
  
 
Part 2 – Individual Test Article Accountability 
y
Individual subject test article will need to be recorded for all lenses dispensed. 
y
All dispensed lenses will be recorded into the EDC system as well as on individual test article 
accountability logs. 
y
What will be recorded: 
y
Lot # 
y
Subject # 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 190 of 194

 
y
# Lenses dispensed 
y
Date dispensed 
y
Dispenser initials 
y
Lens count 
 
y
The Lens Information eCRF is within the subject’s eCRF and is used to record individual lens information. 
 
y
The Lot Number field is an Auto Fill field – start to type in the Lot # and the system will complete the field. 
y
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 191 of 194

 
x
Once “Yes” is selected and the form is saved an additional form will be generated to enter the 
replacement lens information. 
 
 
 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 193 of 194

17.1 
List of Abbreviations 
 
 
AE 
Adverse Event/Adverse Experience 
CFR 
Code of Federal Regulations 
CIB 
Clinical Investigator’s Brochure 
CRF 
Case Report Form 
CRO 
Contract Research Organization 
FDA 
Food and Drug Administration 
GCP 
Good Clinical Practice 
HIPAA 
Health Insurance Portability and Accountability Act 
IB 
Investigator’s Brochure 
ICF 
Informed Consent Form 
ICH 
International Conference on Harmonization 
IDE 
Investigational Device Exemption 
IEC 
Independent Ethics Committee 
IRB 
Institutional Review Board 
MedDRA © 
Medical Dictionary for Regulatory Activities 
MOP 
Manual of Procedures 
NIH 
National Institutes of Health 
OHRP 
Office for Human Research Protections 
OHSR 
Office for Human Subjects Research 
PHI 
Protected Health Information 
PI 
Principal Investigator 
QA 
Quality Assurance 
QC 
Quality Control 
SAE 
Serious Adverse Event/Serious Adverse Experience 
SOP 
Standard Operating Procedure 
UADE 
Unanticipated Adverse Device Effect 
USADE 
Unanticipated Serious Adverse Device Effect 
 
 
 
 
 
 
 
CR-5638 Protocol v6.0, Amendment 5.0
 
JJVC Confidential
Page 194 of 194