NOYITO USB Logic Analyzer Device with USB Cable 24MHz 8CH 24MHz 8 Channel UART IIC SPI Debug
Brand: NOYITO
Categories: Industrial & Scientific, Test, Measure & Inspect, Electrical Testing, Logic Analyzers

Description Sampling rate up to: 24 MHz , can be 24MHz. 16MHz, 12MHz, 8MHz, 4MHz, 2MHz, 1MHz, 500KHz, 250KHz, 200KHz, 100KHz, 50KHz, 25KHz; The logic for each channel sampling rate of 24M/s. General applications around 10M, enough to cope with a variety ofoccasions; 8-channel; Supports the latest version: Version 1.1.15 Voltage range: Input voltage range: -0.5V to 5.25V Input Low Voltage: -0.5V to 0.8V Input High Voltage: 2.0V to 5.25V Works with 5V, 3.3V, 2.5V, 2.0V systems. May work with 1.8V but not recommended. Input Impedance: 1Mohm || 10pF (typical, approximate) Crystal: ±20ppm, 24MHz Error/Accuracy: pulse-width measurement: ±42ns (at 24MHz). Packing Included: 1xLogic Analyser 1xUSB Cable 1xDupont Cable Download Address: https://www.amazon.com/clouddrive/share/NFh0JRc8M2JpCevtrtNEl2WQ2Bblp5Uo3HnLFdsSjEA

The logic for each channel sampling rate of 24M/s. General applications around 10M, enough to cope with a variety ofoccasions; 8-channel.
Sampling rate up to: 24 MHz , can be 24MHz. 16MHz, 12MHz, 8MHz, 4MHz, 2MHz, 1MHz, 500KHz, 250KHz, 200KHz, 100KHz, 50KHz, 25KHz;
The logic for each channel sampling rate of 24M/s. General applications around 10M, enough to cope with a variety ofoccasions;
Input voltage range: -0.5V to 5.25V; Input Low Voltage: -0.5V to 0.8V; Input High Voltage: 2.0V to 5.25V
Input Impedance: 1Mohm || 10pF (typical, approximate); Crystal: ±20ppm, 24MHz.

Package Dimensions: 8.07 x 4.49 x 1.14 inches
Item Weight: 1.76 ounces
Manufacturer: NOYITO
Is Discontinued By Manufacturer: No
Date First Available: October 13, 2018