Hanchen Ultrasonic Homogenizer 550w Sonicator Processor 0.1ml-600ml Lab Mixer Emulsifier Probe Diameter 6mm Chemistry Laboratory Equipment CE Certificate Two Year Warranty
Brand: Hanchen
Color: white and blue
Categories: Industrial & Scientific, Lab & Scientific Products, Lab Instruments & Equipment, Mixing & Blending Equipment, Homogenizer Equipment, Homogenizers

Hanchen Parameters Model: PZ-550LI Power supply: 110v/60HZ or 220V/50HZ optional (We will ship the model according to the common voltage of your billing country.) Power output: 550W Power regulation: 0%-100%, 5% increment Frequency: 19-25KHZ Net weight: 4.5KG Dimensions: 365×232×240(mm) Timer: 1s-99h adjustable Pulser: Digital, open loop, closed loop, 1s-99min optional Piezoelectric variable frequency energy converter CV33, PZT lead zirconate titanate piezoelectric ceramic Diameter: 63mm Length: 165mm Weight: 650g Cable length: 150cm Standard configuration horn Titanium alloy material, TI-64L-4V Head diameter: 6mm by default (Ф3mm optional. Before purchasing, please inform us the size you need.) Processing capacity: 0.1ml—600ml Total length: 139mm Weight: 370g Temperature Sensor: 0-300°(Overheat Protection) Things in package PZ-550LI Homogenizer* 1 Head diameter: 6mm by default (Ф3mm optional. Before purchasing, please inform us the size you need.) Soundproof box* 1 Stainless steel lifting platform* 1 Power cable* 1 Connection Cable* 1 Instruction manual* 1 One Year Warranty Card* 1 Fuse* 3 Note We provide TWO YEAR warranty for every our customer. If you have any questions, please feel free to contact us directly. We will give you a reply as soon as possible.

【Important Parameters】Head diameter: 6mm by default (Ф3mm optional. ASIN: B01MS3N0CH. Before purchasing, please inform us the size you need.) Power output: 550W；Processing capacity: 0.1ml—600ml.
【Easy operation】 The instrument adopts large screen display, and all functions are displayed on the large LCD screen. Automatic resonance point and power control, no need to adjust energy manually. Working time shows the cumulative status.
【Applications】 1. Extraction of cell contents, bacteria, viral tissue. 2. Dispersion and homogenization of the material particles. For example, nanomaterial dispersion (silicon dioxide, carbon nanotubes, graphene, etc.) 3. Accelerate the dissolution and accelerate the chemical reaction. For example for chemical synthesis.
【Ensure safety】Sample overload / over temperature protection; Frequency automatic tracking, automatic fault alarm. On/off pulse timer to protect sensitive samples under processing high-intensity, open and close cycles can be selected from 1 second to 99 hours. Automatic amplitude compensation to ensure that the amplitude of the probe does not change due to load changes during the ultrasonic process.
【CE Certificate and TWO YEAR Warranty】We provide TWO YEAR Warranty for every our customer.