Elitech WJL-6000 HVAC Leak Detector Halogen Gas Tester HVAC Refrigerant AC Sniffer R22 R410A R134A
Brand: Elitech
Color: Halogen Sensor WJL-6000
Categories: Automotive, Tools & Equipment, Air Conditioning Tools & Equipment, Air Conditioning Line Repair Tools

General descriptions: Incorporated with advanced leak detect technology. Elitech WJL-6000 is our new product with the stable function and small appearance; It’s an ideal and economic halogen leak detector. Under normal working circumstance, the detector will beep when it is turned on. It is different with the rapid sound of warning alarm. Applications: The WJL-6000 may also be used to: Detect leaks in other systems and storage/recovery containers. It will respond to all halogenated (contains Chlorine of Fluorine) refrigerants. This includes, but is not limited to: CFCs e.g. R12, R11, R500, R503 etc.. . HCFCs e.g. R22, R123, R124, R502 etc... HFCs e.g. R134a, R404s, R125 etc... Blends such as AZ-50. HP62. MP39 etc... Detect Ethylene Oxide gas leaks in hospital sterilizing equipment (it will detect the halogenated carrier gas). Detect SF-6 in high voltage circuit breakers. Detect most gases that contain chlorine, fluorine and bromine (halogen gases). Detect cleaning agents used in dry cleaning applications such as per chloroethylene. Detect halogen gases in fire-extinguishing system. Technical parameters: Operating temperature: 0~52℃ (30℉ to 125℉) Maximum sensitivity: 6 gr/yr, for all halogenated refrigerants Battery life: Approximately 50 hours normal use Response time: Instantaneous Work mode: continuous, no limitation Fixed probe length: 20cm Unit dimensions: 1.37 x 1.37 x 5.90 in Warm-up time: approximately 6 seconds Reset time: two to ten seconds Power supply: 6V DC four cell 7# excellent batteries

Freon leak detector triggers on R22, R134A, R404a, R410A and all halogenated refrigerants including HFCs, CFCs,HCFCs and blends.
Response within 3 seconds; Warm up in 6 seconds.
7 Adjustable Sensitivity; Maximum sensitivity is 0.21oz/yr.
Audible and visual gas leakage indicators and dual battery indicator.
A flexible probe around 7.9-inch allows detections in difficult and hard-to-reach areas. 24/7 US Technician Support via Email and Phone.

Operating Humidity: 95 percent
Power Source: Battery Powered
Upper Temperature Rating: 52 Degrees Celsius
Alarm: Audible, Visual
Item Weight: ‎7.2 ounces
Brand: ‎Elitech
Style: Modern
Color: Halogen Sensor WJL-6000
Item Dimensions LxWxH: 1.37 x 1.37 x 5.9 inches
Voltage: ‎220 Volts
Batteries: ‎4 Lithium Metal batteries required.
Product Dimensions: ‎1.37 x 1.37 x 5.9 inches
Manufacturer: ‎Elitech
Date First Available: September 16, 2015
Item model number: ‎WJL-6000
OEM Part Number: ‎WJL-6000
Customer Reviews: 4.2 out of 5 stars 4,283 ratings 4.2 out of 5 stars
Is Discontinued By Manufacturer: ‎No
Wattage: ‎1.5 watts
ASIN: B015FK1EH2
Manufacturer Part Number: ‎WJL-6000