Pevor Portable Rechargeable 12 inch-Resolution Color Screen 6 Parameter Monitors
Brand: Pevor

Specification: E--G: Lead Mode:5 Leads ( R, L, F, N, C or RA, LA, LL, RL,V) Lead selection:I, II, III, avR, avL, avF, V, Waveform:2 ch Lead mode:3 Leads ( R, L, F or RA, LA, LL) Lead selection:I, II, III, Waveform:1 ch Gain:´2.5mm/mV, ´5.0mm/mV, ´10mm/mV, ´20mm/mV, auto HR and Alarm Range: Adult:15 ~ 300 bpm Neo/Ped:15 ~ 350 bpm Accuracy:± 1% or ± 1bpm,which great Resolution:1 bpm Sensitivity:> 200 (uV P-P) Differential Input Impedance:> 5 MΩ RESPARATION: Method:Impedance between R-F(RA-LL) Differential Input Impedance:>2.5 MΩ Measuring Impedance Range:0.3~5.0Ω Base line Impedance Range:0 – 2.5 KΩ Bandwidth:0.3 ~ 2.5 Hz R-esp. Rate Measuring and Alarm Range Adult:0 ~ 120 rpm Neo/Ped:0 ~ 150 rpm Resolution:1 rpm Accuracy:±2 rpm Apean Alarm:10 ~ 40 S Pulse Rate: Measuring and Alarm Range:0~254bpm Resolution:1bpm Accuracy:±2bpm I.5TEMP measurement: Channel:1 Measuring and Alarm Range:0 ~ 50 °C Resolution:0.1°C Accuracy:±0.1°C Actualization interval: about 1 Sec. Average Time Constant: < 10 Sec. Sp(0)2: Measuring Range:0 ~ 100 % Alarm Range:0 ~ 100 % Resolution:1 % Accuracy: 70% ~ 100% ±2 %; 0% ~ 69% unspecified Actualization interval:about 1 Sec. Alarm Delay:10 Sec. About N(1)BP: Method:Oscillometric Mode:Manual, Auto, STAT Measuring Interval in AUTO Mode: 1, 2, 3, 4, 5, 10, 15, 30, 60, 90, 120, 180, 240,480 (Min) Measuring Period in STAT Mode:5 Min Pulse Rate Range:40 ~ 240 bpm Alarm Type:SYS, DIA, MEAN

12 inch large LCD screen for convenient to read. Dr-ug solubility calculation function, ArrhythmiA analysis function.
High quality and easy to operate, comes with manual. The back of the machine is designed with a storage box.
Can be arbitrarily set the alarm limit, automatic sound and light alarm.
Built-in lithium battery, AC and DC, can provide long-term monitoring after power failure.
100%GUARANTEE-We provide a 30days no reason refund policy, make you buy with confidence! Order now!