4-20mA Generator, DROK Adjustable Signal Generator, DC 0-10V 0 4-20mA Current Voltage Analog Simulator for Value Adjusting PLC Panel LED Testing
Brand: DROK
Categories: Industrial & Scientific, Test, Measure & Inspect, Electrical Testing, Function Generators

Product Features: By rotating a multi-cycle precise potentiometer, this product can generate 0/4 - 20mA current signal and 0-10V voltage signal, which can used as signal generator, valve adjustment, inverter control, PLC debugging, panel debugging, LED testing, simulating the output of transmitter. Product Parameters: Working voltage: DC24V Input signal: Potentiometer Output signal: 0/4-20mA or 0-10V Display accuracy: Current: 0.1mA; Voltage: 0.01V Dimension: 100*68*38mm Debugging Procedures: A. Current display calibration Clockwise rotating the multi-cycle potentiometer to the maximum, switching the display switch to the right and then using a multimeter to test the actual value of the output current. At this time, you can adjust the current display adjusting potentiometer to make the display value equal to the actual output value. B. Voltage display calibration Clockwise rotating the multi-cycle potentiometer to the maximum, switching the display switch to the left and then using a multimeter to test the actual value of the output voltage. At this time, you can adjust the voltage display adjusting potentiometer to make the display value equal to the actual output value. Package Includes: 1* Signal Generator

AMP & VOLT DUAL SIGNAL GENERATOR: Multifunctional DROK signal generator kit is able to generate ampere & voltage by one module, which generate 0/4 - 20mA current signal and 0-10V voltage signal just by rotating a multi-cycle precise potentiometer, simple and handy.
CLEAR LED DISPLAY: Our 4-20ma signal generator has bright and clear 3-digit LED digital display screen, which can be used to monitor the output current and voltage. You just need to push the switch below the monitor easy to switch the display.
AMP & VOLT DISPLAY CALIBRATION: There are current and voltage display calibrating potentiometer on the back of the module. If the display is accurate, you can calibrate the display value of current voltage by yourself.
WIDE APPLICATION: Our signal adapter can be used as signal generator, valve adjustment, inverter control, PLC debugging, panel debugging, LED testing, simulating the output of transmitter.

Is Discontinued By Manufacturer: No
Package Dimensions: 5.31 x 3.9 x 2.36 inches; 4.66 Ounces
Item model number: 300482
Date First Available: May 1, 2017
Manufacturer: DROK