Carbon Dioxide CO2 Detector Monitor Meter Tester Sensor Temperature Humidity Indoor 9999ppm NDIR Air Quality IAQ Handheld Portable Digital Wet Bulb
Brand: TEKCOPLUS
Color: Hand Held CO2 Meter
Categories: Industrial & Scientific, Test, Measure & Inspect, Airflow & Air Quality, Indoor Air Quality Meters

Shows CO2 level, temp. (°C/°F), Humidity, Dew Point & Wet Bulb. Auto/Manual power off, PC connection loggin and data analysis. Backlight, Hold data, Min/Max Features: 1.Triple displays of CO2 level, temperature (°C/°F) and Humidity 2.Stable NDIR sensor for CO2 detection Statistics of weighted averages TWA (8 hours weighted average) STEL (15 minutes-weighted average) 3.Updates reading every second on the meter, output to PC every 2 seconds Specifications: Measuring Range CO2: 0~9999ppm (2001~9999 out of scale) Temperature : -10~60°C (14~140°F) Humidity : 0.1% ~ 99.9%RH Dew Point : -20.0~59.9°C Wet Bulb : -5.0~59.9°C Resolution : 1ppm,0.1°C/°F, 0.1%RH Accuracy : C02 : ±50ppm ±5% of reading (0~2000ppm) other ranges are not specified Temp : ±0.6°C / ±0.9°F Humidity : ±3%RH (at 25°C, 10~90% RH); ±5%RH (at 25°C, others) Warm-up time : 30 seconds Response time CO2 : <30 seconds (90% step change) Temp : <2 mins (90% step change) RH : <10 mins (90% step change) NO RELAY Operating condition : 0~50°C, 0~95%RH (avoid condensation) Storage Condition : -20~50°C, 0~95%RH (avoid condensation) Set Includes: 1 x Portable CO2 meter 4 x AA batteries 1 x Power Adapter English Operation Manual Hard Carrying Case

MONITORING & ANALYSIS Allows continuous monitoring of CO2, RH & Temperature Level. Data output to PC every 2 seconds.
MULTIPLE MEASUREMENT DISPLAYS AT ONCE- CO2, Temperature, and Relative Humidity readings are all shown on the LCD display at the same time
NORMAL/ MAX/ MIN/ AVERAGE (15mins/ 8hrs) MODE- It can show not only the normal, maximum and minimum CO2 readings but also the weighted average CO2 readings in the past 15mins (STEL mode) and 8 hours (TWA mode).
AUDIBLE CO2 SOUND- The alarm beeps when the CO2 Level exceeds the preset value. The alarm threshold is adjustable.
NDIR SENSOR for CO2 DETECTION- Non-dispersive infrared technology is used to ensure the reliability and long-term stability. Measures 0~9999ppm while 0~2000ppm can maintain a high accuracy at +/- 50ppm

Brand: TEKCOPLUS
Style: Digital
Power Source: Battery Powered
Color: Hand Held CO2 Meter
Item Dimensions LxWxH: 13.78 x 2.76 x 8.66 inches
Item Weight: 200 Grams
Alarm: Audible
Operating Humidity: 95 percent
Upper Temperature Rating: 50 Degrees Celsius
Sensor Type: Non Dispersive Infrared
Manufacturer: Tekcoplus Ltd
Part Number: COTK-57
Product Dimensions: 13.78 x 2.76 x 8.66 inches
Item model number: COTK-57
Batteries: 3 AAA batteries required.
Is Discontinued By Manufacturer: No
Item Package Quantity: 1
Number Of Pieces: 1
Batteries Included?: No
Batteries Required?: No
Date First Available: August 22, 2016