GreatNorthCollective/greatnorth-canada-federal-laws-text huggingface dataset:
Electricity and Gas Inspection Regulations — 36
36 (1) The pressure multiplier for any pressure converting meter shall be established by using the following equation, namely, P m = P b /P s where (a) P m is the pressure multiplier; (b) P b is the pressure expressed on the absolute pressure scale and used as the base pressure for the calibration of the pressure converting meter; and (c) P s is the standard pressure expressed on the absolute pressure scale. (2) The pressure multiplier for any meter other than a pressure correcting meter shall be established by using the following equation, namely, P m = (P ga + P a )/P s where (a) P m is the pressure multiplier; (b) P ga is the average meter pressure expressed on the gauge pressure scale; (c) P a is (i) the mean atmospheric pressure at the meter location, expressed to at least four significant digits, (ii) the atmospheric pressure calculated for the meter location pursuant to subsection 37(1), expressed to at least four significant digits, (iii) the atmospheric pressure declared for the meter location pursuant to subsection 37(2), expressed to at least four significant digits, or (iv) the measured atmospheric pressure at the meter location, expressed to at least four significant digits; and (d) P s is the standard pressure expressed on the absolute pressure scale. SOR/95-532, s. 3
Electricity and Gas Inspection Regulations — 37
37 (1) The atmospheric pressure shall be calculated for a meter location by using the following equation, namely, (a) in the International system of units, P a = 101.560 - (0.0113 × M) kPa, or (b) in the Imperial system of units, P a = 14.73 - (0.0005 × F) psia where (c) P a is the calculated atmospheric pressure; and (d) M is the number of metres of elevation of the meter location above mean sea level and F is the number of feet of elevation of the meter location above mean sea level, in either case as determined by a survey carried out by a registered surveyor or as determined from the most recent applicable topographical maps published by the Department of Energy, Mines and Resources. (2) The atmospheric pressure declared for a meter location shall not differ from the atmospheric pressure calculated for that meter location pursuant to subsection (1) by more than (a) in the International system of units, 0.621 + (0.001 × P ga ) kPa, or (b) in the Imperial system of units, 0.090 + (0.001 × P ga ) psi where P ga is the average meter pressure expressed on the gauge pressure scale.
Electricity and Gas Inspection Regulations — 38
38 (1) Subject to section 39, the temperature multiplier for any temperature converting meter shall be established by using the following equation, namely, (a) in the International system of units, T m = 288.15/(T b + 273.15), or (b) in the Imperial system of units, T m = 519.67/(T b + 459.67) where (c) T m is the temperature multiplier; and (d) T b is the temperature established by the contractor and used as the base temperature for the calibration of the temperature converting meter. (2) Subject to section 39, the temperature multiplier of any meter other than a temperature converting meter shall be established by using the following equation, namely, (a) in the International system of units, T m = 288.15/(T ga + 273.15), or (b) in the Imperial system of units, T m = 519.67/(T ga + 459.67) where (c) T m is the temperature multiplier; and (d) T ga is the average flowing gas temperature. SOR/95-532, s. 3
Electricity and Gas Inspection Regulations — 39
39 A temperature multiplier of one may be used for (a) meters that are purchased prior to December 31, 1986 and (i) that have a capacity that is less than 14 standard cubic metres per hour at a differential pressure of 0.125 kPa, or (ii) that have a capacity that is less than 500 standard cubic feet per hour at a differential pressure of 0.5 in. of water column; (b) meters that are installed prior to December 31, 1986, where the average meter pressure exceeds the atmospheric pressure by more than 3.5 kPa in the International system of units or 0.5 psi in the Imperial system of units; and (c) meters that are installed where the average meter pressure does not exceed the atmospheric pressure by more than 3.5 kPa in the International system of units or 0.5 psi in the Imperial system of units and that (i) were most recently sealed prior to December 31, 1986, and (ii) have a capacity that is greater than that stated in subparagraph (a)(i) or (ii), whichever is applicable.
Electricity and Gas Inspection Regulations — 40
40 For the purpose of paragraph 35(e), (a) subject to paragraphs (b) and (c), the supercompressibility factor for any meter shall be established based on (i) the pressure exerted by the gas within the meter and the temperature of the gas flowing through the meter as determined by continuous monitoring at the meter location, (ii) the relative density, the carbon dioxide content and the nitrogen content of the gas being measured, as determined by (A) a gas sampling process, if the process gives results that are representative of the gas at the meter location, or (B) continuous monitoring at the meter location, and (iii) if the parameters of the gas referred to in subparagraphs (i) and (ii) vary, either the time weighted or the volume weighted averages of those parameters during the period that the volume of the gas is measured by the meter; (b) the supercompressibility factor for any meter for which the pressure exerted by the gas within the meter expressed on the gauge pressure scale does not exceed 207 kPa in the International system of units, or 30 psi in the Imperial system of units, may be established based on (i) the pressure exerted by the gas within the meter, established and maintained through the use of a pressure factor metering regulator that has been approved and verified under section 9 of the Act, (ii) an assumed flowing gas temperature of 59°F or 15°C, and (iii) the relative density, the carbon dioxide content and the nitrogen content of the gas being measured as determined by a gas sampling process, if the process gives results that are representative of the gas at the meter location; or (c) a supercompressibility factor of one may be used if the pressure exerted by the gas within the meter expressed on the gauge pressure scale does not exceed 700 kPa in the International system of units or 100 psi in the Imperial system of units. SOR/2006-76, s. 2
Electricity and Gas Inspection Regulations — 41
41 Where gas is sold by energy units, the number of energy units shall be the product obtained by multiplying (a) the calorific power of a unit of mass of the gas by the number of the said units of mass of the gas; or (b) the calorific power of a unit of volume of the gas at a certain temperature and certain pressure by the number of the said units of volume of the gas at the same certain temperature and same certain pressure.
Electricity and Gas Inspection Regulations — 42
42 The calorific power of gas shall be determined by measurement or by calculation based on gas sampling and shall be expressed in terms of one of the following units: (a) joules or multiples thereof, per standard cubic metre of gas, free of water vapour; (b) Btu (60.5) , per standard cubic foot of gas, free of water vapour; or (c) joules or multiples thereof, per kilogram of gas, free of water vapour.
Electricity and Gas Inspection Regulations — 43
43 In order to obtain the calorific power of gas in Btu (60.5) per cubic foot of gas free of water vapour at a pressure P expressed on the absolute pressure scale and at standard temperature, where the gas has been tested using a method that gives results in Btu (60.5) per cubic foot of gas saturated with water vapour at the pressure P and standard temperature, the results shall be multiplied by a factor equal to P/(P - 0.25611)
Electricity and Gas Inspection Regulations — 44
44 The director shall set out specifications, subject to sections 42 and 43, for determination of the calorific power of gas.
Electricity and Gas Inspection Regulations — 45
45 The director shall set out specifications, subject to sections 35 to 40, for determination of the quantity of gas. SOR/89-317, s. 7(F)
Electricity and Gas Inspection Regulations — 46
46 The limits of error referred to in paragraph 33(k) of the Act are 3% of the amount of electricity or gas supplied. SOR/95-532, s. 3; SOR/2009-76, s. 7
Electricity and Gas Inspection Regulations — 47
 program manual, the fee set out for that type of meter in column III of (i) Part I of Schedule 1, for electricity meters, and (ii) Part IV of Schedule 1, for gas meters; (b) for each meter verified or reverified under inspection level 2 as set out in the quality monitoring meter verification program manual, two thirds of the fee set out for that type of meter in column III of (i) Part I of Schedule 1, for electricity meters, and (ii) Part IV of Schedule 1, for gas meters; and (c) for each meter verified or reverified under inspection level 3 as set out in the quality monitoring meter verification program manual, one third of the fee set out for that type of meter in column III of (i) Part I of Schedule 1, for electricity meters, and (ii) Part IV of Schedule 1, for gas meters. (13) Notwithstanding subsections (1), (3) and (5), no fee or charge is payable with respect to any investigation by an inspector pursuant to subsection 29(1). SOR/87-212, s. 5; SOR/89-425, s. 1; SOR/95-333, s. 2; SOR/2009-76, s. 8; SOR/2014-113, s. 7
Electricity and Gas Inspection Regulations — 48
48 The contravention of a provision of the Act or these Regulations set out in column 1 of Part 1 or 2, respectively, of Schedule 2 may be proceeded with as a violation in accordance with sections 29.11 to 29.29 of the Act. SOR/2014-113, s. 4
Electricity and Gas Inspection Regulations — 49
49 The violation of each provision set out in column 1 of Part 1 or 2 of Schedule 2 is classified as a minor, serious or very serious violation in accordance with the classification set out in column 2. SOR/2014-113, s. 4
Electricity and Gas Inspection Regulations — 50
50 (1) The amount of the penalty in respect of each violation is, subject to subsection (2), (a) $250, for a minor violation; (b) $500, for a serious violation; and (c) $1,000, for a very serious violation. (2) The amount of the penalty is to be adjusted in accordance with the calculation set out in column 2 of Schedule 3 that corresponds to the history set out in column 1 of the person who has committed the violation. (3) For the purposes of determining a person’s history, the points set out in column 3 of Part 1 or 2 of Schedule 2 are assigned for each violation of a provision set out in column 1. SOR/2014-113, s. 4
Electricity and Gas Inspection Regulations — 51
51 The notice of violation is to set out that an amount equal to 50% of the penalty may be paid as complete satisfaction of the penalty if it is paid within the period referred to in subsection 53(1) and in the manner set out in the notice of violation in accordance with subsection 53(6). SOR/2014-113, s. 4
Electricity and Gas Inspection Regulations — 52
52 If the Minister enters into a compliance agreement under subsection 29.14(1) of the Act, the conditions of which include the making of expenditures by a person, the amount of the penalty set out in the notice of violation is to be reduced by an amount equal to one half of the expenditures made in fulfilment of the agreement, with a maximum reduction to nil. SOR/2014-113, s. 4
Electricity and Gas Inspection Regulations — 53
53 (1) For the purposes of paragraph 29.12(2)(e) of the Act, payment is to be made within 15 days after the day on which the person is provided with the notice of violation. (2) For the purposes of subsection 29.13(1) of the Act, payment of the penalty set out in the notice of violation is to be made within 30 days after the day on which the person is provided with the notice. (3) For the purposes of paragraph 29.14(4)(a) of the Act, payment is to be made within 15 days after the day on which the person is provided with the notice of default. (4) For the purposes of subsection 29.15(1) of the Act, payment is to be made within 15 days after the day on which the person is provided with the Minister’s notice. (5) For the purposes of subsection 29.16(3) of the Act, payment is to be made within 15 days after the day on which the person is provided with notice of the Minister’s decision under subsection 29.16(1) or (2) of the Act. (6) A payment referred to in any of subsections (1) to (5) is to be made in one of the following manners, as set out in the applicable notice: (a) by means of a certified cheque or money order made payable to the Receiver General for Canada and submitted to a Measurement Canada office (i) [Repealed, SOR/2018-252, s. 2] (ii) by registered mail, or (iii) by courier; or (b) by electronic means. SOR/2014-113, s. 4; SOR/2018-252, s. 2
Electricity and Gas Inspection Regulations — 54
54 (1) A request under paragraph 29.13(2)(a) or (b) of the Act shall be made, as set out in the notice, by submitting the following information in writing to a Measurement Canada office in person, by registered mail or courier or by electronic means within 30 days after the day on which the person is provided with the notice of violation: (a) the notice of violation number, as set out in the notice; (b) the person’s name, mailing address, email address, telephone number, facsimile number and contact person, if any; (c) an indication of whether the person prefers to communicate in English or French; (d) in the case of a request under paragraph 29.13(2)(a) of the Act, a proposal detailing the corrective action that will be taken to ensure the person’s compliance with the provision to which the violation relates; and (e) in the case of a request made under 29.13(2)(b) of the Act, the person’s reasons for requesting the review. (2) [Repealed, SOR/2018-252, s. 3] (3) The date of the request is (a) the day on which it is submitted, if it is submitted in person; or (b) the day on which it is sent, if it is sent by registered mail, courier or electronic means. SOR/2014-113, s. 4; SOR/2018-252, s. 3
Electricity and Gas Inspection Regulations — (b)
(b) single-rate demand type, per function not included in item 2
Electricity and Gas Inspection Regulations — (iii)
(iii) power line, radio or other frequency-activated type, per rate
Electricity and Gas Inspection Regulations — (d)
(d) multi-rate, microprocessor-based, time-of-use type
Electricity and Gas Inspection Regulations — (b)
(b) electronic type, per billing function
Electricity and Gas Inspection Regulations — (b)
(b) greater than 283 m 3 /h (10 000 ft 3 /h), for each additional 283 m 3 /h (10 000 ft 3 /h) or part thereof
Electricity and Gas Inspection Regulations — (c)
(c) multi-rate, microprocessor-based, time-of-use type,
Electricity and Gas Inspection Regulations — (i)
(i) input method 1 set out in the specifications made under section 45
Electricity and Gas Inspection Regulations — (ii)
(ii) input method 2 set out in the specifications made under section 45
Electricity and Gas Inspection Regulations — (iii)
(iii) a user-programmable input method, per input range
Electricity and Gas Inspection Regulations — (i)
(i) during inspector’s normal working hours
Electricity and Gas Inspection Regulations — (ii)
(ii) outside inspector’s normal working hours
Atlantic Fishery Regulations, 1985 — *
Her Excellency the Governor General in Council, on the recommendation of the Minister of Fisheries and Oceans, pursuant to sections 8, 34 * , 34.3 ** , 51 *** and 69 **** of the Fisheries Act , is pleased hereby to revoke the Atlantic Coast Marine Plant Regulations , C.R.C., c. 805; the Atlantic Crab Fishery Regulations , C.R.C., c. 806; the Atlantic Fishery Regulations , C.R.C., c. 807; the Atlantic Fishing Registration and Licensing Regulations , C.R.C., c. 808; the Atlantic Herring Fishery Regulations made by P.C. 1983-23 of 13 January, 1983 ***** ; the Fishing Gear Marking Regulations , C.R.C., c. 813; and the Lobster Fishery Regulations , C.R.C., c. 817, and to make the annexed Regulations respecting the management and allocation of fishery resources on the Atlantic coast of Canada , in substitution therefor, effective January 8, 1986. * R.S.C. 1970, c. 17 (1st supp.), s. 4 ** S.C. 1976-77, c. 35, s. 11 *** S.C. 1976-77, c. 35, s. 14 **** S.C. 1976-77, c. 35, s. 19 ***** SOR/83-82, 1983 Canada Gazette Part II, p. 437
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Capelin Fishing Area means a Capelin Fishing Area illustrated and enumerated in Schedule IV; ( zone de pêche du capelan )
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class A vessel [Repealed, SOR/88-141, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class A-1 vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class A-2 vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class B vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class C vessel [Repealed, SOR/88-141, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class C-1 vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class C-2 vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class D vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class E vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class F vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class G vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class H vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class I vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class J vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class K vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class L vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class M vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class N vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class O vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class P vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class Q vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class R vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class S vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class Sc-1 vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class Sc-2 vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class Sc-3 vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class Sc-4 vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class Sc-5 vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class Sc-6 vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class Sc-7 vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class Sc-8 vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class Sc-9 vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class T vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class U vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Class V vessel [Repealed, SOR/89-584, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
cod-end means a bag-like extension that is attached to the after end of the belly of an otter trawl and is used to retain the catch; ( cul de chalut )
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Convention means the Convention on Future Multilateral Cooperation in the Northwest Atlantic Fisheries that was ratified by Canada on November 30, 1978 and came into force on January 1, 1979; ( Convention )
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Convention Area means the waters of the Northwest Atlantic Ocean north of 35°00′ north latitude and west of a line extending due north from 35°00′ north latitude and 42°00′ west longitude to 59°00′ north latitude, thence due west to 44°00′ west longitude, and thence due north to the coast of Greenland, and the waters of the Gulf of St. Lawrence, Davis Strait and Baffin Bay south of 78°10′ north latitude; ( zone de la Convention )
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Crab Fishing Area means a Crab Fishing Area illustrated and enumerated in Schedule XI; ( zone de pêche du crabe )
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Department means the Department of Fisheries and Oceans; ( ministère )
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
dip net means a bag-like net that is fixed to a frame to which a handle is attached and is used to catch fish without enmeshing them; ( épuisette )
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Division means a division described in Schedule III; ( division )
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
drift net means a gill net that drifts in the water whether or not it is attached to a vessel and includes a gill net that is not anchored at both ends; ( filet dérivant )
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
fishing trip means a voyage that commences at the time a fishing vessel leaves a port to engage in fishing and terminates at the time fish caught during that period are offloaded; ( expédition de pêche )
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
fixed gear means fishing gear other than mobile gear, angling gear, a drift net, a crab trap, a lobster trap or a dip net; ( engin fixe )
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
foreign fishing vessel [Repealed, SOR/93-61, s. 1]
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
gill net means a net by which fish are caught by being enmeshed but that does not enclose an area of water; ( filet maillant )
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
groundfish means any species of fish set out in Part II of Schedule I; ( poisson de fond )
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
Herring Fishing Area means a Herring Fishing Area illustrated and enumerated in Schedule VI; ( zone de pêche du hareng )
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
hoop net means a box-shaped or funnel-shaped apparatus that is constructed of netting or wire mesh fastened to a hoop or frame and that is used to catch fish without enmeshing them, and includes any fishing gear commonly referred to as a hand bow, hand trap or hand pot; ( verveux )
Atlantic Fishery Regulations, 1985 — Atlantic Fishery Regulations, 1985
incidental catch [Repealed, SOR/94-292, s. 1]