真空采血管

title: Principles.
content: Vacutainer needle are double ended, with one side being encased in a thin rubber coating for safety. When the needle is screwed into the translucent plastic needle holder, the rubber needle is inside the holder, and the exposed needle will be inserted into the vein. When a Vacutainer tube is inserted into the holder, its rubber cap is punctured by the inner needle and the vacuum in the tube pulls blood through the needle and into the tube. The filled tube is then removed and another can be inserted and filled the same way. The amount of air evacuated from the tube predetermines how much blood will fill the tube before blood stops flowing.  Each tube is topped with a color-coded plastic or rubber cap. Tubes often include additives that mix with the blood when collected, and the color of each tube's plastic cap indicates which additives it contains.  Blood collection tubes expire because over time the vacuum is lost and blood will not be drawn into the tube when the needle punctures the cap.
title: Types of tubes.
content: Vacutainer tubes may contain additional substances that preserve blood for processing in a medical laboratory. Using the wrong tube may make the blood sample unusable for the intended purpose. These additives are typically thin film coatings applied using an ultrasonic nozzle.  The additives may include anticoagulants (EDTA, sodium citrate, heparin) or a gel with density between those of blood cells and blood plasma. Additionally, some tubes contain additives that preserve certain components of or substances within the blood, such as glucose. When a tube is centrifuged, the materials within are separated by density, with the blood cells sinking to the bottom and the plasma or serum accumulating at the top. Tubes containing gel can be easily handled and transported after centrifugation without the blood cells and serum mixing. The meanings of the various colors are standardized across manufacturers.  The term "order of draw" refers to the sequence in which tubes should be filled. The needle which pierces the tubes can carry additives from one tube into the next, so the sequence is standardized so that any cross-contamination of additives will not affect laboratory results.
title: History.
title: History.
content: Vacutainer technology was developed in 1947 by Joseph Kleiner, and is currently marketed by Becton Dickinson (B-D).  The plastic tube version, known as Vacutainer PLUS, was developed at B-D in the early 1990s by E. Vogler, D. Montgomery and G. Harper amongst others of the Surface Science Group as US patents 5344611, 5326535, 5320812, 5257633 and 5246666.  Vacutainers are widely used in phlebotomy in developed countries due to safety and ease of use. Vacutainers have the advantage of being prepared with additives, allowing easy multi-tube draws, and having a lower chance of hemolysis. In developing countries, it is still common to draw blood using a syringe or syringes.

Summary: 真空采血管(Vacutainer)是一种真空无菌玻璃或塑料管封闭的血液收集管,管内的真空促进预定的液体量疏入。这种真空管通常用来静脉采血,也有的用来收集尿液样本。真空管管壁有时会有一层添加剂来帮助稳定和保存样本。

Vacutainer: Vacutainer est une marque déposée de Becton-Dickinson. Elle a été développée en 1947 par Joseph Kleiner et est apposée sur une série de tubes sous-vide utilisés lors d'une ponction veineuse, ainsi que sur un support appelé "tulipe" permettant l'adaptation d'une aiguille d'un côté et de ces tubes de l'autre.