Electromyography - wikipedia Electromyography Not to be confused with other types of electrography. Electromyography EMG from gait termination, bottom left is the raw EMG, right is the rectified pattern ICD - 9 - CM 93.08 MeSH D004576 ( edit on Wikidata ) Electromyography ( EMG ) is an electrodiagnostic medicine technique for evaluating and recording the electrical activity produced by skeletal muscles. EMG is performed using an instrument called an electromyograph to produce a record called an electromyogram. An electromyograph detects the electric potential generated by muscle cells when these cells are electrically or neurologically activated. The signals can be analyzed to detect medical abnormalities, activation level, or recruitment order, or to analyze the biomechanics of human or animal movement. Contents 1 Medical uses 2 Technique 2.1 Skin preparation and risks 2.2 Surface and intramuscular EMG recording electrodes 2.3 Maximal voluntary contraction 2.4 Other measurements 2.5 EMG signal decomposition 2.6 EMG signal processing 2.7 Limitations 2.8 Electrical characteristics 3 Procedure outcomes 3.1 Normal results 3.2 Abnormal results 4 History 5 Research 6 See also 7 References 8 Further reading 9 External links Medical uses EMG testing has a variety of clinical and biomedical applications. EMG is used as a diagnostics tool for identifying neuromuscular diseases, or as a research tool for studying kinesiology, and disorders of motor control. EMG signals are sometimes used to guide botulinum toxin or phenol injections into muscles. EMG signals are also used as a control signal for prosthetic devices such as prosthetic hands, arms, and lower limbs. EMG then an acceleromyograph may be used for neuromuscular monitoring in general anesthesia with neuromuscular - blocking drugs, in order to avoid postoperative residual curarization ( PORC ). Except in the case of some purely primary myopathic conditions EMG is usually performed with another electrodiagnostic medicine test that measures the conducting function of nerves. This is called a nerve conduction studies ( NCS ). Needle EMG and NCSs are typically indicated when there is pain in the limbs, weakness from spinal nerve compression, or concern about some other neurologic injury or disorder. Spinal nerve injury does not cause neck, mid back pain or low back pain, and for this reason, evidence has not shown EMG or NCS to be helpful in diagnosing causes of axial lumbar pain, thoracic pain, or cervical spine pain. Needle EMG may aid with the diagnosis of nerve compression or injury ( such as carpal tunnel syndrome ), nerve root injury ( such as sciatica ), and with other problems of the muscles or nerves. Less common medical conditions include amyotrophic lateral sclerosis, myasthenia gravis, and muscular dystrophy. Technique Skin preparation and risks The first step before insertion of the needle electrode is skin preparation. This typically involves simply cleaning the skin with an alcohol pad. The actual placement of the needle electrode can be difficult and depends on a number of factors, such as specific muscle selection and the size of that muscle. Proper needle EMG placement is very important for accurate representation of the muscle of interest, although EMG is more effective on superficial muscles as it is unable to bypass the action potentials of superficial muscles and detect deeper muscles. Also, the more body fat an individual has, the weaker the EMG signal. When placing the EMG sensor, the ideal location is at the belly of the muscle : the longitudinal midline. The belly of the muscle can also be thought of as in - between the motor point ( middle ) of the muscle and the tendonus insertion point. Cardiac pacemakers and implanted cardiac defibrillators ( ICDs ) are used increasingly in clinical practice, and no evidence exists indicating that performing routine electrodiagnostic studies on patients with these devices pose a safety hazard. However, there are theoretical concerns that electrical impulses of nerve conduction studies ( NCS ) could be erroneously sensed by devices and result in unintended inhibition or triggering of output or reprogramming of the device. In general, the closer the stimulation site is to the pacemaker and pacing leads, the greater the chance for inducing a voltage of sufficient amplitude to inhibit the pacemaker. Despite such concerns, no immediate or delayed adverse effects have been reported with routine NCS. No known contraindications exist for performing needle EMG or NCS on pregnant patients. Additionally, no complications from these procedures have been reported in the literature. Evoked potential testing, likewise, has not been reported to cause any problems when it is performed during pregnancy. Patients with lymphedema or patients at risk for lymphedema are routinely cautioned to avoid percutaneous procedures in the affected extremity, namely venipuncture, to prevent development or worsening of lymphedema or cellulitis. Despite the potential risk, the evidence for such complications subsequent to venipuncture is limited. No published reports exist of cellulitis, infection, or other complications related to EMG performed in the setting of lymphedema or prior lymph node dissection. However, given the unknown risk of cellulitis in patients with lymphedema, reasonable caution should be exercised in performing needle examinations in lymphedematous regions to avoid complications. In patients with gross edema and taut skin, skin puncture by needle electrodes may result in chronic weeping of serous fluid. The potential bacterial media of such serous fluid and the violation of skin integrity may increase the risk of cellulitis. Before proceeding, the physician should weigh the potential risks of performing the study with the need to obtain the information gained. Surface and intramuscular EMG recording electrodes There are two kinds of EMG : surface EMG and intramuscular EMG. Surface EMG assesses muscle function by recording muscle activity from the surface above the muscle on the skin. Surface electrodes are able to provide only a limited assessment of the muscle activity. Surface EMG can be recorded by a pair of electrodes or by a more complex array of multiple electrodes. More than one electrode is needed because EMG recordings display the potential difference ( voltage difference ) between two separate electrodes. Limitations of this approach are the fact that surface electrode recordings are restricted to superficial muscles, are influenced by the depth of the subcutaneous tissue at the site of the recording which can be highly variable depending of the weight of a patient, and can not reliably discriminate between the discharges of adjacent muscles. Intramuscular EMG can be performed using a variety of different types of recording electrodes. The simplest approach is a monopolar needle electrode. This can be a fine wire inserted into a muscle with a surface electrode as a reference ; or two fine wires inserted into muscle referenced to each other. Most commonly fine wire recordings are for research or kinesiology studies. Diagnostic monopolar EMG electrodes are typically insulated and stiff enough to penetrate skin, with only the tip exposed using a surface electrode for reference. Needles for injecting therapeutic botulinum toxin or phenol are typically monopolar electrodes that use a surface reference, in this case, however, the metal shaft of a hypodermic needle, insulated so that only the tip is exposed, is used both to record signals and to inject. Slightly more complex in design is the concentric needle electrode. These needles have a fine wire, embedded in a layer of insulation that fills the barrel of a hypodermic needle, that has an exposed shaft, and the shaft serves as the reference electrode. The exposed tip of the fine wire serves as the active electrode. As a result of this configuration, signals tend to be smaller when recorded from a concentric electrode than when recorded from a monopolar electrode and they are more resistant to electrical artifacts from tissue and measurements tend to be somewhat more reliable. However, because the shaft is exposed throughout its length, superficial muscle activity can contaminate the recording of deeper muscles. Single fiber EMG needle electrodes are designed to have very tiny recording areas, and allow for the discharges of individual muscle fibers to be discriminated. To perform intramuscular EMG, typically either a monopolar or concentric needle electrode is inserted through the skin into the muscle tissue. The needle is then moved to multiple spots within a relaxed muscle to evaluate both insertional activity and resting activity in the muscle. Normal muscles exhibit a brief burst of muscle fiber activation when stimulated by needle movement, but this rarely lasts more than 100ms. The two most common pathologic types of resting activity in muscle are fasciculation and fibrillation potentials. A fasciculation potential is an involuntary activation of a motor unit within the muscle, sometimes visible with the naked eye as a muscle twitch or by surface electrodes. Fibrillations, however, are only detected by needle EMG, and represent the isolated activation of individual muscle fibers, usually as the result of nerve or muscle disease. Often, fibrillations are triggered by needle movement ( insertional activity ) and persist for several seconds or more after the movement ceases. After assessing resting and insertional activity, the electromyographer assess the activity of muscle during voluntary contraction. The shape, size, and frequency of the resulting electrical signals are judged. Then the electrode is retracted a few millimetres, and again the activity is analyzed. This is repeated, sometimes until data on 10 -- 20 motor units have been collected in order to draw conclusions about motor unit function. Each electrode track gives only a very local picture of the activity of the whole muscle. Because skeletal muscles differ in the inner structure, the electrode has to be placed at various locations to obtain an accurate study. Single fiber electromyography assesses the delay between the contractions of individual muscle fibers within a motor unit and is a sensitive test for dysfunction of the neuromuscular junction caused by drugs, poisons, or diseases such as myasthenia gravis. The technique is complicated and typically only performed by individuals with special advanced training. Surface EMG is used in a number of settings ; for example, in the physiotherapy clinic, muscle activation is monitored using surface EMG and patients have an auditory or visual stimulus to help them know when they are activating the muscle ( biofeedback ). A review of the literature on surface EMG published in 2008, concluded that surface EMG may be useful to detect the presence of neuromuscular disease ( level C rating, class III data ), but there are insufficient data to support its utility for distinguishing between neuropathic and myopathic conditions or for the diagnosis of specific neuromuscular diseases. EMGs may be useful for additional study of fatigue associated with post-poliomyelitis syndrome and electromechanical function in myotonic dystrophy ( level C rating, class III data ). Certain US states limit the performance of needle EMG by nonphysicians. New Jersey declared that it can not be delegated to a physician 's assistant. Michigan has passed legislation saying needle EMG is the practice of medicine. Special training in diagnosing medical diseases with EMG is required only in residency and fellowship programs in neurology, clinical neurophysiology, neuromuscular medicine, and physical medicine and rehabilitation. There are certain subspecialists in otolaryngology who have had selective training in performing EMG of the laryngeal muscles, and subspecialists in urology, obstetrics and gynecology who have had selective training in performing EMG of muscles controlling bowel and bladder function. Maximal voluntary contraction One basic function of EMG is to see how well a muscle can be activated. The most common way that can be determined is by performing a maximal voluntary contraction ( MVC ) of the muscle that is being tested. Muscle force, which is measured mechanically, typically correlates highly with measures of EMG activation of muscle. Most commonly this is assessed with surface electrodes, but it should be recognized that these typically only record from muscle fibers in close approximation to the surface. Several analytical methods for determining muscle activation are commonly used depending on the application. The use of mean EMG activation or the peak contraction value is a debated topic. Most studies commonly use the maximal voluntary contraction as a means of analyzing peak force and force generated by target muscles. According to the article, Peak and average rectified EMG measures : Which method of data reduction should be used for assessing core training exercises ?, concluded that the `` average rectified EMG data ( ARV ) is significantly less variable when measuring the muscle activity of the core musculature compared to the peak EMG variable. '' Therefore, these researchers would suggest that `` ARV EMG data should be recorded alongside the peak EMG measure when assessing core exercises. '' Providing the reader with both sets of data would result in enhanced validity of the study and potentially eradicate the contradictions within the research. Other measurements EMG can also be used for indicating the amount of fatigue in a muscle. The following changes in the EMG signal can signify muscle fatigue : an increase in the mean absolute value of the signal, increase in the amplitude and duration of the muscle action potential and an overall shift to lower frequencies. Monitoring the changes of different frequency changes the most common way of using EMG to determine levels of fatigue. The lower conduction velocities enable the slower motor neurons to remain active. A motor unit is defined as one motor neuron and all of the muscle fibers it innervates. When a motor unit fires, the impulse ( called an action potential ) is carried down the motor neuron to the muscle. The area where the nerve contacts the muscle is called the neuromuscular junction, or the motor end plate. After the action potential is transmitted across the neuromuscular junction, an action potential is elicited in all of the innervated muscle fibers of that particular motor unit. The sum of all this electrical activity is known as a motor unit action potential ( MUAP ). This electrophysiologic activity from multiple motor units is the signal typically evaluated during an EMG. The composition of the motor unit, the number of muscle fibres per motor unit, the metabolic type of muscle fibres and many other factors affect the shape of the motor unit potentials in the myogram. Nerve conduction testing is also often done at the same time as an EMG to diagnose neurological diseases. Some patients can find the procedure somewhat painful, whereas others experience only a small amount of discomfort when the needle is inserted. The muscle or muscles being tested may be slightly sore for a day or two after the procedure. EMG signal decomposition EMG signals are essentially made up of superimposed motor unit action potentials ( MUAPs ) from several motor units. For a thorough analysis, the measured EMG signals can be decomposed into their constituent MUAPs. MUAPs from different motor units tend to have different characteristic shapes, while MUAPs recorded by the same electrode from the same motor unit are typically similar. Notably MUAP size and shape depend on where the electrode is located with respect to the fibers and so can appear to be different if the electrode moves position. EMG decomposition is non-trivial, although many methods have been proposed. EMG signal processing Rectification is the translation of the raw EMG signal to a signal with a single polarity, usually positive. The purpose of rectifying the signal is to ensure the signal does not average to zero, due to the raw EMG signal having positive and negative components. Two types of rectification are used : full - wave and half - wave rectification. Full - wave rectification adds the EMG signal below the baseline to the signal above the baseline to make a conditioned signal that is all positive. If the baseline is zero, this is equivalent to taking the absolute value of the signal. This is the preferred method of rectification because it conserves all of the signal energy for analysis. Half - wave rectification discards the portion of the EMG signal that is below the baseline. In doing so, the average of the data is no longer zero therefore it can be used in statistical analyses. Limitations Needle EMG used in clinical settings has practical applications such as helping to discover disease. Needle EMG has limitations, however, in that it does involve voluntary activation of muscle, and as such is less informative in patients unwilling or unable to cooperate, children and infants, and in individuals with paralysis., Surface EMG can have limited applications due to inherent problems associated with surface EMG. Adipose tissue ( fat ) can affect EMG recordings. Studies show that as adipose tissue increased the active muscle directly below the surface decreased. As adipose tissue increased, the amplitude of the surface EMG signal directly above the center of the active muscle decreased. EMG signal recordings are typically more accurate with individuals who have lower body fat, and more compliant skin, such as young people when compared to old. Muscle cross talk occurs when the EMG signal from one muscle interferes with that of another limiting reliability of the signal of the muscle being tested. Surface EMG is limited due to lack of deep muscles reliability. Deep muscles require intramuscular wires that are intrusive and painful in order to achieve an EMG signal. Surface EMG can only measure superficial muscles and even then it is hard to narrow down the signal to a single muscle. Electrical characteristics The electrical source is the muscle membrane potential of about -- 90 mV. Measured EMG potentials range between less than 50 μV and up to 20 to 30 mV, depending on the muscle under observation. Typical repetition rate of muscle motor unit firing is about 7 -- 20 Hz, depending on the size of the muscle ( eye muscles versus seat ( gluteal ) muscles ), previous axonal damage and other factors. Damage to motor units can be expected at ranges between 450 and 780 mV. Procedure outcomes Normal results Muscle tissue at rest is normally electrically inactive. After the electrical activity caused by the irritation of needle insertion subsides, the electromyograph should detect no abnormal spontaneous activity ( i.e., a muscle at rest should be electrically silent, with the exception of the area of the neuromuscular junction, which is, under normal circumstances, very spontaneously active ). When the muscle is voluntarily contracted, action potentials begin to appear. As the strength of the muscle contraction is increased, more and more muscle fibers produce action potentials. When the muscle is fully contracted, there should appear a disorderly group of action potentials of varying rates and amplitudes ( a complete recruitment and interference pattern ). Abnormal results EMG findings vary with the type of disorder, the duration of the problem, the age of the patient, the degree to which the patient can be cooperative, the type of needle electrode used to study the patient, and sampling error in terms of the number of areas studied within a single muscle and the number of muscles studied overall. Interpreting EMG findings is usually best done by an individual informed by a focused history and physical examination of the patient, and in conjunction with the results of other relevant diagnostic studies performed including most importantly, nerve conduction studies, but also, where appropriate, imaging studies such as MRI and ultrasound, muscle and nerve biopsy, muscle enzymes, and serologic studies. Abnormal results may be caused by the following medical conditions ( please note this is not an exhaustive list of conditions that can result in abnormal EMG studies ) : Disorders of Muscle : Inflammatory myopathies Polymyositis Dermatomyositis Inclusion body myopathy Myotonia Congenita Muscular dystrophies : Duchenne Muscular Dystrophy Becker muscular dystrophy Facioscapulohumeral dystrophy Limb girdle muscular dystrophy Centronuclear myopathy Myotonic dystrophy Mitochondrial myopathies Disorders of the neuromuscular junction : Myasthenia Gravis Lambert -- Eaton myasthenic syndrome Botulism poisoning Organophosphate poisoning Hypermagnesemia Hypocalcemia Disorders of Nerves : Carpal tunnel syndrome Ulnar neuropathy at the elbow Radial nerve palsy ( Saturday night palsy ) Peroneal ( fibular ) nerve palsy Diabetic neuropathy Alcohol related neuropathy Nutritional neuropathy Amyloid neuropathy AIDP CIDP Bell 's palsy Laryngeal neuropathy Pudendal neuropathy Femoral neuropathy Sciatic Neuropathy Tibial neuropathy Tarsal tunnel syndrome Charcot - Marie - Tooth syndrome Zoster neuropathy Oculomotor, Facial, vagal, trigeminal, glossopharyngeal, spinal accessory neuropathies Hemifacial spasm Multifocal motor neuropathy Axillary neuropathy Long thoracic neuropathy Suprascapular neuropathy Toxic neuropathies Drug - induced neuropathies Plexus disorders : Neuralgic Amyotrophy ( idiopathic brachial plexitis ) Traumatic brachial plexopathy Lumbosacral radiculopathy Hirayama disease Root disorders : Cervical, thoracic, lumbar, sacral radiculopathy Spinal stenosis Arachnoiditis Leptomeningeal disorders Motor neuron disease Amyotrophic lateral sclerosis West Nile virus Poliomyelitis Kennedy 's syndrome ( Spinobulbar muscular atrophy ). History The first documented experiments dealing with EMG started with Francesco Redi 's works in 1666. Redi discovered a highly specialized muscle of the electric ray fish ( Electric Eel ) generated electricity. By 1773, Walsh had been able to demonstrate that the eel fish 's muscle tissue could generate a spark of electricity. In 1792, a publication entitled De Viribus Electricitatis in Motu Musculari Commentarius appeared, written by Luigi Galvani, in which the author demonstrated that electricity could initiate muscle contraction. Six decades later, in 1849, Emil du Bois - Reymond discovered that it was also possible to record electrical activity during a voluntary muscle contraction. The first actual recording of this activity was made by Marey in 1890, who also introduced the term electromyography. In 1922, Gasser and Erlanger used an oscilloscope to show the electrical signals from muscles. Because of the stochastic nature of the myoelectric signal, only rough information could be obtained from its observation. The capability of detecting electromyographic signals improved steadily from the 1930s through the 1950s, and researchers began to use improved electrodes more widely for the study of muscles. The AANEM was formed in 1953 as one of several currently active medical societies with a special interest in advancing the science and clinical use of the technique. Clinical use of surface EMG ( sEMG ) for the treatment of more specific disorders began in the 1960s. Hardyck and his researchers were the first ( 1966 ) practitioners to use sEMG. In the early 1980s, Cram and Steger introduced a clinical method for scanning a variety of muscles using an EMG sensing device. It was not until the middle of the 1980s that integration techniques in electrodes had sufficiently advanced to allow batch production of the required small and lightweight instrumentation and amplifiers. At present, a number of suitable amplifiers are commercially available. In the early 1980s, cables that produced signals in the desired microvolt range became available. Recent research has resulted in a better understanding of the properties of surface EMG recording. Surface electromyography is increasingly used for recording from superficial muscles in clinical or kinesiological protocols, where intramuscular electrodes are used for investigating deep muscles or localized muscle activity. There are many applications for the use of EMG. EMG is used clinically for the diagnosis of neurological and neuromuscular problems. It is used diagnostically by gait laboratories and by clinicians trained in the use of biofeedback or ergonomic assessment. EMG is also used in many types of research laboratories, including those involved in biomechanics, motor control, neuromuscular physiology, movement disorders, postural control, and physical therapy. Research EMG can be used to sense isometric muscular activity where no movement is produced. This enables definition of a class of subtle motionless gestures to control interfaces without being noticed and without disrupting the surrounding environment. These signals can be used to control a prosthesis or as a control signal for an electronic device such as a mobile phone or PDA. EMG signals have been targeted as control for flight systems. The Human Senses Group at the NASA Ames Research Center at Moffett Field, CA seeks to advance man - machine interfaces by directly connecting a person to a computer. In this project, an EMG signal is used to substitute for mechanical joysticks and keyboards. EMG has also been used in research towards a `` wearable cockpit, '' which employs EMG - based gestures to manipulate switches and control sticks necessary for flight in conjunction with a goggle - based display. Unvoiced speech recognition recognizes speech by observing the EMG activity of muscles associated with speech. It is targeted for use in noisy environments, and may be helpful for people without vocal cords and people with aphasia. EMG has also been used as a control signal for computers and other devices. An interface device based on EMG could be used to control moving objects, such as mobile robots or an electric wheelchair. This may be helpful for individuals that can not operate a joystick - controlled wheelchair. Surface EMG recordings may also be a suitable control signal for some interactive video games. In 1999 an EMG program called Echidna was used to enable a man with locked in syndrome to send a message to a computer. That program, now called NeuroSwitch, developed by Control Bionics enables people with severe disabilities to communicate by text, email, SMS, computer - generated voice and to control computer games and programs, and - through the internet - Anybots telepresence robots. A joint project involving Microsoft, the University of Washington in Seattle, and the University of Toronto in Canada has explored using muscle signals from hand gestures as an interface device. A patent based on this research was submitted on June 26, 2008. See also Chronaxie Compound muscle action potential Electrical muscle stimulation Electrodiagnostic medicine Electromyoneurography Magnetomyography Nerve conduction study Neuromuscular ultrasound Phonomyography References Jump up ^ Kamen, Gary. Electromyographic Kinesiology. In Robertson, DGE et al. Research Methods in Biomechanics. Champaign, IL : Human Kinetics Publ., 2004. Jump up ^ Electromyography at the US National Library of Medicine Medical Subject Headings ( MeSH ) Jump up ^ Harvey AM, Masland RL : Actions of durarizing preparations in the human. Journal of Pharmacology And Experimental Therapeutics, Vol. 73, Issue 3, 304 - 311, 1941 Jump up ^ Botelho SY : Comparison of simultaneously recorded electrical and mechanical activity in myasthenia gravis patients and in partially curarized normal humans. Am J Med. 1955 Nov ; 19 ( 5 ) : 693 - 6. PMID 13268466 Jump up ^ Christie TH, Churchill - Davidson HC : The St. Thomas 's Hospital nerve stimulator in the diagnosis of prolonged apnoea. Lancet. 1958 Apr 12 ; 1 ( 7024 ) : 776. PMID 13526270 Jump up ^ Engbaek J, Ostergaard D, Viby - Mogensen J : Double burst stimulation ( DBS ) : a new pattern of nerve stimulation to identify residual neuromuscular block. Br J Anaesth. 1989 Mar ; 62 ( 3 ) : 274 - 8. PMID 2522790 ^ Jump up to : North American Spine Society ( February 2013 ), `` Five Things Physicians and Patients Should Question '', Choosing Wisely : an initiative of the ABIM Foundation, North American Spine Society, retrieved 25 March 2013, which cites Sandoval, AE ( Nov 2010 ). `` Electrodiagnostics for low back pain ''. Physical Medicine and Rehabilitation Clincs of North America. 21 ( 4 ) : 767 -- 76. doi : 10.1016 / j. pmr. 2010.06. 007. PMID 20977959. North American Spine Society ( 2011 ). `` Diagnosis and treatment of degenerative lumbar spinal stenosis ''. Burr Ridge, Illinois : Agency for Healthcare Research and Quality : 104. Jump up ^ https://www.delsys.com/Attachments_pdf/TN101%20-%20EMG%20Sensor%20Placement-web.pdf ^ Jump up to : http://www.aanem.org/getmedia/2034191e-583b-4c55-b725-fc38ea8262e2/risksinEDX.pdf.aspx. Jump up ^ Arthur C. Rothman, MD, v. Selective Insurance Company of America, Supreme Court of New Jersey, Jan. 19 Jump up ^ Texas Court of Appeals, Third District, at Austin, Cause No. 03 - 10 - 673 - CV. April 5, 2012 Jump up ^ Section 333.17018 Michigan Compiled Laws http://legislature.mi.gov/doc.aspx?mcl-333-17018 Jump up ^ Behm, D.G., Whittle, J., Button, D., & Power, K. ( 2002 ). Intermuscle differences in activation. Muscle and Nerve. 25 ( 2 ) ; 236 - 243. Jump up ^ Hibbs, A.E., Thompson, K.G., French, D.N., Hodgson, D., Spears, I.R. Peak and average rectified EMG measures : Which method of data reduction should be used for assessing core training exercises ? Journal of Electromyography and Kinesiology. 21 ( 1 ), 102 -- 111. 2011. Jump up ^ Buchanan, T.S., Lloyd, D.G., Manal, K., & Besier, T.F. ( 2004 ). Neuromusculoskeletal modeling : estimation of muscle forces and joint moments and movements from measurements of neural command. Journal of applied biomechanics, 20 ( 4 ), 367. Jump up ^ Halperin, I., Aboodarda, S.J., Button, D.C., Andersen, L.L., & Behm, D.G. ( 2014 ). ROLLER MASSAGER IMPROVES RANGE OF MOTION OF PLANTAR FLEXOR MUSCLES WITHOUT SUBSEQUENT DECREASES IN FORCE PARAMETERS. International journal of sports physical therapy, 9 ( 1 ), 92. Jump up ^ Cifrek, M., Medved, V., Tonković, S., & Ostojić, S. ( 2009 ). Surface EMG based muscle fatigue evaluation in biomechanics. Clinical Biomechanics, 24 ( 4 ), 327 - 340. Jump up ^ Raez, M.B.I. ; Hussain, M.S. ; Mohd - Yasin, F. ( Mar 23, 2006 ). `` Techniques of EMG signal analysis : detection, processing, classification and applications ''. Biol. Proced. Online ( 8 ) : 11 -- 35. doi : 10.1251 / bpo115. PMC 1455479. PMID 16799694. Jump up ^ Weir, JP ; Wagner, LL ; Housh, TJ ( 1992 ). `` Linearity and reliability of the IEMG v. torque relationship for the forearm flexors and leg extensors ''. American Journal of Physical Medicine & Rehabilitation. 71 ( 5 ) : 283 -- 287. Jump up ^ Vrendenbregt, J ; Rau, G ; Housh ( 1973 ). `` Surface eletromyography in relation to force, muscle length and endurance ''. New developments in electromyography and clinical neurophysiology : 607 -- 622. Jump up ^ Kuiken, TA ; Lowery, Stoykob ( April 2003 ). `` The Effect of Subcutaneous Fat on myoelectric signal amplitude and cross talk ''. Prosthetics and Orthodontics International. 27 ( 1 ) : 48 -- 54. Jump up ^ Nigg B.M., & Herzog W., 1999. Biomechanics of the Musculo - Skeletal system. Wiley. Page : 349. Jump up ^ Patterson, John R. `` Fitwise ''. Castillo. Brian T. Retrieved 24 June 2009. Jump up ^ Cram, JR. ; Steger, JC. ( Jun 1983 ). `` EMG scanning in the diagnosis of chronic pain ''. Biofeedback Self Regul. 8 ( 2 ) : 229 -- 41. PMID 6227339. Jump up ^ Andreasen, DS. ; Gabbert DG, : EMG Switch Navigation of Power Wheelchairs, RESNA 2006. ( 1 ) Jump up ^ Park, DG. ; Kim, HC. Muscleman : Wireless input device for a fighting action game based on the EMG signal and acceleration of the human forearm. ( 2 ) Jump up ^ Hsu, Jeremy ( 2009 - 10 - 29 ). `` The Future of Video Game Input : Muscle Sensors ''. Live Science. Retrieved 2010 - 01 - 16. Jump up ^ `` Recognizing Gestures from Forearm EMG Signals ''. United States Patent and Trademark Office. 2008 - 06 - 26. Archived from the original on 2017 - 01 - 12. Retrieved 2010 - 01 - 16. Further reading Piper, H. : Elektrophysiologie menschlicher Muskeln. Berlin, J. Springer, 1912. External links Association of Electromyography Technologists of Canada ( AETC ) MedlinePlus entry on EMG describes EMG Neuromuscular.edu American Association of Neuromuscular & Electrodiagnostic Medicine EmedicineHealth page on EMG Risks in Electrodiagnostic Medicine Interactively design an EMG signal processing RMS envelope measurement application with the ASN Filter Designer ( ASN - AN024 ) Neurosurgical and other procedures ( ICD - 9 - CM V3 01 -- 05 + 89.1, ICD - 10 - PCS 00 -- 01 ) Skull Craniotomy Decompressive craniectomy Cranioplasty CNS Brain thalamus and globus pallidus : Thalamotomy Thalamic stimulator Pallidotomy ventricular system : Ventriculostomy Suboccipital puncture Intracranial pressure monitoring cerebrum : Psychosurgery Lobotomy Bilateral cingulotomy Hemispherectomy Anterior temporal lobectomy pituitary gland : Hypophysectomy hippocampus : Amygdalohippocampectomy Brain biopsy Cerebral meninges Meningeal biopsy Spinal cord and spinal canal Spinal cord and roots Cordotomy Rhizotomy Vertebrae and intervertebral discs : see Template : Bone, cartilage, and joint procedures Medical imaging Computed tomography of the head Cerebral angiography Pneumoencephalography Echoencephalography / Transcranial Doppler Magnetic resonance imaging of the brain Brain PET Myelography Diagnostic Electroencephalography Lumbar puncture Polysomnography CPRs Glasgow Coma Scale Mini -- mental state examination National Institutes of Health Stroke Scale CHADS2 score PNS Cranial and peripheral nerves Nerve block Vagotomy Sympathetic nerves or ganglia Ganglionectomy Sympathectomy Endoscopic thoracic sympathectomy Nerves ( general ) Axotomy Neurectomy Nerve biopsy Diagnostic Nerve conduction study Electromyography Medical imaging Magnetic resonance neurography Retrieved from `` https://en.wikipedia.org/w/index.php?title=Electromyography&oldid=837966430 '' Categories : Electrodiagnosis Electrophysiology Neurology procedures Neurophysiology Neurotechnology Hidden categories : All articles with unsourced statements Articles with unsourced statements from February 2015 Talk Contents About Wikipedia Afrikaans Azərbaycanca Català Čeština Dansk Deutsch Español Euskara فارسی Français Gaeilge ગુજરાતી 한국어 Bahasa Indonesia Italiano Қазақша Кыргызча Nederlands 日本 語 Norsk Polski Português Română Русский Shqip Српски / srpski Suomi Svenska Türkçe Українська 22 more Edit links This page was last edited on 24 April 2018, at 02 : 55. Text is available under the Creative Commons Attribution - ShareAlike License ; additional terms may apply. By using this site, you agree to the Terms of Use and Privacy Policy. Wikipedia ® is a registered trademark of the Wikimedia Foundation, Inc., a non-profit organization. About Wikipedia

Question: what features of muscle contraction can be determined from an emg (electromyogram)
Answer: electric potential generated