Bunsen burner - wikipedia Bunsen burner Bunsen burner A Bunsen burner with needle valve. The hose barb for the gas tube is on the left and the needle valve for gas flow adjustment is on the opposite side. The air inlet on this particular model is adjusted by rotating the barrel, thus opening or closing the vertical baffles at the base. Uses Heating Sterilization Combustion Related items Hot plate Heating mantle Meker - Fisher burner Teclu burner A Bunsen burner, named after Robert Bunsen, is a common piece of laboratory equipment that produces a single open gas flame, which is used for heating, sterilization, and combustion. The gas can be natural gas ( which is mainly methane ) or a liquefied petroleum gas, such as propane, butane, or a mixture of both. Contents 1 History 2 Operation 3 Variants 4 See also 5 References 6 External links History In 1852 the University of Heidelberg hired Bunsen and promised him a new laboratory building. The city of Heidelberg had begun to install coal - gas street lighting, and so the university laid gas lines to the new laboratory. The designers of the building intended to use the gas not just for illumination, but also in burners for laboratory operations. For any burner lamp, it was desirable to maximize the temperature and minimize luminosity. However, existing laboratory burner lamps left much to be desired not just in terms of the heat of the flame, but also regarding economy and simplicity. While the building was still under construction in late 1854, Bunsen suggested certain design principles to the university 's mechanic, Peter Desaga, and asked him to construct a prototype. Similar principles had been used in an earlier burner design by Michael Faraday, as well as in a device patented in 1856 by the gas engineer R.W. Elsner. The Bunsen / Desaga design succeeded in generating a hot, sootless, non-luminous flame by mixing the gas with air in a controlled fashion before combustion. Desaga created adjustable slits for air at the bottom of the cylindrical burner, with the flame igniting at the top. By the time the building opened early in 1855 Desaga had made 50 burners for Bunsen 's students. Two years later Bunsen published a description, and many of his colleagues soon adopted the design. Bunsen burners are now used in laboratories all around the world. Operation Bunsen burner flames depend on air flow in the throat holes ( on the burner side, not the needle valve for gas flow ) : 1. air hole closed ( safety flame used for lighting or default ), 2. air hole slightly open, 3. air hole half - open, 4. air hole fully open ( roaring blue flame ). The device in use today safely burns a continuous stream of a flammable gas such as natural gas ( which is principally methane ) or a liquefied petroleum gas such as propane, butane, or a mixture of both. The hose barb is connected to a gas nozzle on the laboratory bench with rubber tubing. Most laboratory benches are equipped with multiple gas nozzles connected to a central gas source, as well as vacuum, nitrogen, and steam nozzles. The gas then flows up through the base through a small hole at the bottom of the barrel and is directed upward. There are open slots in the side of the tube bottom to admit air into the stream using the Venturi effect, and the gas burns at the top of the tube once ignited by a flame or spark. The most common methods of lighting the burner are using a match or a spark lighter. The amount of air mixed with the gas stream affects the completeness of the combustion reaction. Less air yields an incomplete and thus cooler reaction, while a gas stream well mixed with air provides oxygen in a stoichiometric amount and thus a complete and hotter reaction. The air flow can be controlled by opening or closing the slot openings at the base of the barrel, similar in function to the choke in a carburettor. A Bunsen burner situated below a tripod If the collar at the bottom of the tube is adjusted so more air can mix with the gas before combustion, the flame will burn hotter, appearing blue as a result. If the holes are closed, the gas will only mix with ambient air at the point of combustion, that is, only after it has exited the tube at the top. This reduced mixing produces an incomplete reaction, producing a cooler but brighter yellow, which is often called the `` safety flame '' or `` luminous flame ''. The yellow flame is luminous due to small soot particles in the flame, which are heated to incandescence. The yellow flame is considered `` dirty '' because it leaves a layer of carbon on whatever it is heating. When the burner is regulated to produce a hot, blue flame, it can be nearly invisible against some backgrounds. The hottest part of the flame is the tip of the inner flame, while the coolest is the whole inner flame. Increasing the amount of fuel gas flow through the tube by opening the needle valve will increase the size of the flame. However, unless the airflow is adjusted as well, the flame temperature will decrease because an increased amount of gas is now mixed with the same amount of air, starving the flame of oxygen. Generally, the burner is placed underneath a laboratory tripod, which supports a beaker or other container. The burner will often be placed on a suitable heatproof mat to protect the laboratory bench surface. A Bunsen burner is also regularly used in microbiology labs to work sterile. The burner is placed on the bench ( working surface in the lab ), where it creates an aseptic region, in which microbiologist can work Variants Other burners based on the same principle exist. The most important alternatives to the Bunsen burner are : Teclu burner -- The lower part of its tube is conical, with a round screw nut below its base. The gap, set by the distance between the nut and the end of the tube, regulates the influx of the air in a way similar to the open slots of the Bunsen burner. The Teclu burner provides better mixing of air and fuel and can achieve higher flame temperatures than the Bunsen burner. Meker burner -- The lower part of its tube has more openings with larger total cross-section, admitting more air and facilitating better mixing of air and gas. The tube is wider and its top is covered with a wire grid. The grid separates the flame into an array of smaller flames with a common external envelope, and also prevents flashback to the bottom of the tube, which is a risk at high air - to - fuel ratios and limits the maximum rate of air intake in a conventional Bunsen burner. Flame temperatures of up to 1,100 -- 1,200 ° C ( 2,000 -- 2,200 ° F ) are achievable if properly used. The flame also burns without noise, unlike the Bunsen or Teclu burners. Tirrill burner -- The base of the burner has a needle valve which allows the regulation of gas intake directly from the Burner, rather than from the gas source. Maximum temperature of flame can reach 1560 ° C. See also Alcohol burner Heating mantle Meker - Fisher burner References Jump up ^ Lockemann, G. ( 1956 ). `` The Centenary of the Bunsen Burner ''. J. Chem. Ed. 33 : 20 -- 21. Bibcode : 1956JChEd... 33... 20L. doi : 10.1021 / ed033p20. Jump up ^ Rocke, A.J. ( 2002 ). `` Bunsen Burner ''. Oxford Companion to the History of Modern Science. p. 114. Jump up ^ Jensen, William B. ( 2005 ). `` The Origin of the Bunsen Burner '' ( PDF ). J. Chem. Ed. 82 ( 4 ) : 518. Bibcode : 2005JChEd... 82... 518J. doi : 10.1021 / ed082p518. Archived from the original ( PDF ) on November 9, 2006. Jump up ^ Griffith, J.J. ( 1838 ). Chemical Reactions -- A compendium of experimental chemistry ( 8th ed. ). Glasgow : R Griffin and Co. Jump up ^ Kohn, Moritz ( 1950 ). `` Remarks on the history of laboratory burners ''. J. Chem. Educ. 27 ( 9 ) : 514. Bibcode : 1950JChEd... 27... 514K. doi : 10.1021 / ed027p514. Jump up ^ Ihde, Aaron John ( 1984 ). The development of modern chemistry. Courier Dover Publications. pp. 233 -- 236. ISBN 978 - 0 - 486 - 64235 - 2. Jump up ^ Youtube movie to show proper use of bunsen burner https://www.youtube.com/watch?v=bRadiLXkqoU. Jump up ^ PDF on how to work in the lab http://www.chemistry.ucla.edu/sites/default/files/safety/sop/SOP_Spreading_Bacterial_Cultures.pdf. Jump up ^ Teclu, Nicolae ( 1892 ). `` Ein neuer Laboratoriums - Brenner ''. J. Prakt. Chem. 45 ( 1 ) : 281 -- 286. doi : 10.1002 / prac. 18920450127. Jump up ^ Partha, Mandal Pratim & Mandal, B. ( 2002 - 01 - 01 ). A Text Book of Homoeopathic Pharmacy. Kolkata, India : New Central Book Agency. p. 46. ISBN 978 - 81 - 7381 - 009 - 1. Jump up ^ Hale, Charles W. ( 1915 ). Domestic Science, Volume 2. London : Cambridge University Press. p. 38. Jump up ^ Flinn scientific. http://www.flinnsci.com/store/Scripts/prodView.asp?idproduct=14010 External links Look up bunsen burner in Wiktionary, the free dictionary. Sella, Andrea ( 2007 ). `` Bunsen 's Burner ''. Classic Kit, Chemistry World. Royal Society of Chemistry. Poliakoff, Martyn ( 2011 ). `` Robert Bunsen and his Burner ''. The Periodic Table of Videos. University of Nottingham. Laboratory equipment General Heaters Dryers Alcohol burner Bunsen burner Desiccator Heating mantle Hot plate Lab oven Kiln Meker - Fisher burner Striker Teclu burner Water bath Vacuum dry box Mixers Shakers Chemostat Homogenizer Liquid whistle Magnetic stirrer Mortar and pestle Shaker Sonicator Static mixer Stirring rod Vortex mixer Wash bottle Stands Clamps Holders Beaker clamp Clamp holder Tripod Burette clamp Extension clamp Flask clamp Funnel support Iron ring Pinch clamp Retort stand Screw clamp Test tube holder Test tube rack Wire gauze Lab drying rack Containers Storage Agar plate Cryogenic storage dewar Incubator Laminar flow cabinet Microtiter plate Petri dish Picotiter plate Refrigerator Weighing boat Weighing dish Other items Aspirator Autoclave Balance brush Cork borer Crucible Filter paper File Forceps Centrifuge Microscope Pipeclay triangle Spectrophotometer Splint Stopper Scoopula Spatula Test tube brush Wire brush Inoculation needle Inoculation loop Glassware Apparatus Dean - Stark Soxhlet extractor Kipp 's Bottles Boston round Pycnometer Condensers Cold finger Liebig Dishes Evaporating Petri Watch glass Flasks Büchner Vacuum ( Dewar ) Erlenmeyer Fernbach Fleaker Florence Retort Round - bottom Schlenk Volumetric Funnels Büchner Hirsch Dropping Separatory Measuring devices Burette Conical measure Cuvette Eudiometer Graduated cylinder Ostwald viscometer Pipette Pipette ( dropper ) Tubes Boiling Cragie Ignition Nuclear magnetic resonance ( NMR ) Test Thiele Thistle Other items Beaker Gas syringe Vial Analytical chemistry Compositional AutoAnalyzer CHN analyzer Colorimeter Inductively coupled plasma ( ICP ) device Gas chromatograph ( GC ) Liquid Chromatograph ( LC ) Mass spectrometer ( MS ) pH indicator pH meter Microscopy Scanning electron microscope ( SEM ) Transmission electron microscope ( TEM ) Thermochemistry Calorimeter differential scanning Melting point apparatus Thermometer Thermogravimetric analyzer ( TGA ) Other items Analytical balance Colony counter Spiral plater Nuclear magnetic resonance ( NMR ) instrument Plate reader Electronics Ammeter Current source Function generator Galvanostat Multimeter Network analyzer Oscilloscope Pulse generator Potentiostat Spectrum analyzer Time - domain reflectometer Voltage source Voltmeter Safety Personal protective equipment ( PPE ) Lab coat Face shield Respirator Rubber apron Safety shower Eye and hand Acid - resistant gloves Eyewash station Glove box Medical gloves Nitrile gloves Safety glasses Safety goggles Other items Acid ( solvent ) cabinet Biosafety cabinet Fire blanket Fire extinguisher Fume hood Instruments used in medical laboratories Retrieved from `` https://en.wikipedia.org/w/index.php?title=Bunsen_burner&oldid=803586287 '' Categories : Burners Laboratory equipment Combustion engineering German inventions Hidden categories : Wikipedia pages semi-protected against vandalism Talk Variants View source Contents About Wikipedia বাংলা Български Català Čeština Dansk Deutsch Eesti Ελληνικά Español Esperanto Euskara فارسی Français Gaeilge Galego हिन्दी Hrvatski Bahasa Indonesia Italiano עברית Magyar മലയാളം Bahasa Melayu Nederlands 日本 語 Norsk پنجابی Polski Português Română Русский සිංහල Simple English Српски / srpski Srpskohrvatski / српскохрватски Suomi Svenska Tagalog தமிழ் ไทย Українська Tiếng Việt 中文 Edit links This page was last edited on 3 October 2017, at 11 : 33. About Wikipedia

Question: give the name of a substance that is used as a fuel in the bunsen burner
Answer: natural gas