Electron microscope - wikipedia Electron microscope A modern transmission electron microscope Diagram of a transmission electron microscope Electron microscope constructed by Ernst Ruska in 1933 An electron microscope is a microscope that uses a beam of accelerated electrons as a source of illumination. As the wavelength of an electron can be up to 100,000 times shorter than that of visible light photons, electron microscopes have a higher resolving power than light microscopes and can reveal the structure of smaller objects. A scanning transmission electron microscope has achieved better than 50 pm resolution in annular dark - field imaging mode and magnifications of up to about 10,000,000 x whereas most light microscopes are limited by diffraction to about 200 nm resolution and useful magnifications below 2000x. Electron microscopes have electron optical lens systems that are analogous to the glass lenses of an optical light microscope. Electron microscopes are used to investigate the ultrastructure of a wide range of biological and inorganic specimens including microorganisms, cells, large molecules, biopsy samples, metals, and crystals. Industrially, electron microscopes are often used for quality control and failure analysis. Modern electron microscopes produce electron micrographs using specialized digital cameras and frame grabbers to capture the image. Contents 1 History 2 Types 2.1 Transmission electron microscope ( TEM ) 2.2 Scanning electron microscope ( SEM ) 2.3 Color 2.4 Reflection electron microscope ( REM ) 2.5 Scanning transmission electron microscope ( STEM ) 3 Sample preparation 4 Disadvantages 5 Applications 6 See also 7 References 8 External links 8.1 General 8.2 History 8.3 Other History Diagram illustrating the phenomena resulting from the interaction of highly energetic electrons with matter. The first electromagnetic lens was developed in 1926 by Hans Busch. According to Dennis Gabor, the physicist Leó Szilárd tried in 1928 to convince Busch to build an electron microscope, for which he had filed a patent. The physicist Ernst Ruska and the electrical engineer Max Knoll constructed the prototype electron microscope in 1931, capable of four - hundred - power magnification ; the apparatus was the first demonstration of the principles of electron microscopy. Two years later, in 1933, Ruska built an electron microscope that exceeded the resolution attainable with an optical ( light ) microscope. Moreover, Reinhold Rudenberg, the scientific director of Siemens - Schuckertwerke, obtained the patent for the electron microscope in May 1931. In 1932, Ernst Lubcke of Siemens & Halske built and obtained images from a prototype electron microscope, applying concepts described in the Rudenberg patent applications. Five years later ( 1937 ), the firm financed the work of Ernst Ruska and Bodo von Borries, and employed Helmut Ruska ( Ernst 's brother ) to develop applications for the microscope, especially with biological specimens. Also in 1937, Manfred von Ardenne pioneered the scanning electron microscope. The first commercial electron microscope was produced in 1938 by Siemens. The first North American electron microscope was constructed in 1938, at the University of Toronto, by Eli Franklin Burton and students Cecil Hall, James Hillier, and Albert Prebus ; and Siemens produced a transmission electron microscope ( TEM ) in 1939. Although contemporary transmission electron microscopes are capable of two million - power magnification, as scientific instruments, they remain based upon Ruska 's prototype. Types Transmission electron microscope ( TEM ) Main article : Transmission electron microscope Play media Operating principe of a Transmission Electron Microscope The original form of electron microscope, the transmission electron microscope ( TEM ) uses a high voltage electron beam to illuminate the specimen and create an image. The electron beam is produced by an electron gun, commonly fitted with a tungsten filament cathode as the electron source. The electron beam is accelerated by an anode typically at + 100 k eV ( 40 to 400 keV ) with respect to the cathode, focused by electrostatic and electromagnetic lenses, and transmitted through the specimen that is in part transparent to electrons and in part scatters them out of the beam. When it emerges from the specimen, the electron beam carries information about the structure of the specimen that is magnified by the objective lens system of the microscope. The spatial variation in this information ( the `` image '' ) may be viewed by projecting the magnified electron image onto a fluorescent viewing screen coated with a phosphor or scintillator material such as zinc sulfide. Alternatively, the image can be photographically recorded by exposing a photographic film or plate directly to the electron beam, or a high - resolution phosphor may be coupled by means of a lens optical system or a fibre optic light - guide to the sensor of a digital camera. The image detected by the digital camera may be displayed on a monitor or computer. The resolution of TEMs is limited primarily by spherical aberration, but a new generation of aberration correctors have been able to partially overcome spherical aberration to increase resolution. Hardware correction of spherical aberration for the high - resolution transmission electron microscopy ( HRTEM ) has allowed the production of images with resolution below 0.5 angstrom ( 50 picometres ) and magnifications above 50 million times. The ability to determine the positions of atoms within materials has made the HRTEM an important tool for nano - technologies research and development. Transmission electron microscopes are often used in electron diffraction mode. The advantages of electron diffraction over X-ray crystallography are that the specimen need not be a single crystal or even a polycrystalline powder, and also that the Fourier transform reconstruction of the object 's magnified structure occurs physically and thus avoids the need for solving the phase problem faced by the X-ray crystallographers after obtaining their X-ray diffraction patterns of a single crystal or polycrystalline powder. One major disadvantage of the transmission electron microscope is the need for extremely thin sections of the specimens, typically about 100 nanometers. Creating these thin sections for biological and materials specimens is technically very challenging. Semiconductor thin sections can be made using a focused ion beam. Biological tissue specimens are chemically fixed, dehydrated and embedded in a polymer resin to stabilize them sufficiently to allow ultrathin sectioning. Sections of biological specimens, organic polymers and similar materials may require staining with heavy atom labels in order to achieve the required image contrast. Scanning electron microscope ( SEM ) Play media Operating principe of a Scanning Electron Microscope Main article : Scanning electron microscope Image of bacillus subtilis taken with a 1960s electron microscope. The SEM produces images by probing the specimen with a focused electron beam that is scanned across a rectangular area of the specimen ( raster scanning ). When the electron beam interacts with the specimen, it loses energy by a variety of mechanisms. The lost energy is converted into alternative forms such as heat, emission of low - energy secondary electrons and high - energy backscattered electrons, light emission ( cathodoluminescence ) or X-ray emission, all of which provide signals carrying information about the properties of the specimen surface, such as its topography and composition. The image displayed by an SEM maps the varying intensity of any of these signals into the image in a position corresponding to the position of the beam on the specimen when the signal was generated. In the SEM image of an ant shown below and to the right, the image was constructed from signals produced by a secondary electron detector, the normal or conventional imaging mode in most SEMs. Generally, the image resolution of an SEM is lower than that of a TEM. However, because the SEM images the surface of a sample rather than its interior, the electrons do not have to travel through the sample. This reduces the need for extensive sample preparation to thin the specimen to electron transparency. The SEM is able to image bulk samples that can fit on its stage and still be maneuvered, including a height less than the working distance being used, often 4 millimeters for high resolution images. The SEM also has a great depth of field, and so can produce images that are good representations of the three - dimensional surface shape of the sample. Another advantage of SEMs comes with environmental scanning electron microscopes ( ESEM ) that can produce images of good quality and resolution with hydrated samples or in low, rather than high, vacuum or under chamber gases. This facilitates imaging unfixed biological samples that are unstable in the high vacuum of conventional electron microscopes. An image of an ant in a scanning electron microscope Color In their most common configurations, electron microscopes produce images with a single brightness value per pixel, with the results usually rendered in grayscale. However, often these images are then colorized through the use of feature - detection software, or simply by hand - editing using a graphics editor. This may be done to clarify structure or for aesthetic effect and generally does not add new information about the specimen. In some configurations information about several specimen properties is gathered per pixel, usually by the use of multiple detectors. In SEM, the attributes of topography and material contrast can be obtained by a pair of backscattered electron detectors and such attributes can be superimposed in a single color image by assigning a different primary color to each attribute. Similarly, a combination of backscattered and secondary electron signals can be assigned to different colors and superimposed on a single color micrograph displaying simultaneously the properties of the specimen. Some types of detectors used in SEM have analytical capabilities, and can provide several items of data at each pixel. Examples are the Energy - dispersive X-ray spectroscopy ( EDS ) detectors used in elemental analysis and Cathodoluminescence microscope ( CL ) systems that analyse the intensity and spectrum of electron - induced luminescence in ( for example ) geological specimens. In SEM systems using these detectors it is common to color code the signals and superimpose them in a single color image, so that differences in the distribution of the various components of the specimen can be seen clearly and compared. Optionally, the standard secondary electron image can be merged with the one or more compositional channels, so that the specimen 's structure and composition can be compared. Such images can be made while maintaining the full integrity of the original signal, which is not modified in any way. Reflection electron microscope ( REM ) In the reflection electron microscope ( REM ) as in the TEM, an electron beam is incident on a surface but instead of using the transmission ( TEM ) or secondary electrons ( SEM ), the reflected beam of elastically scattered electrons is detected. This technique is typically coupled with reflection high energy electron diffraction ( RHEED ) and reflection high - energy loss spectroscopy ( RHELS ). Another variation is spin - polarized low - energy electron microscopy ( SPLEEM ), which is used for looking at the microstructure of magnetic domains. Scanning transmission electron microscope ( STEM ) Main article : Scanning transmission electron microscopy The STEM rasters a focused incident probe across a specimen that ( as with the TEM ) has been thinned to facilitate detection of electrons scattered through the specimen. The high resolution of the TEM is thus possible in STEM. The focusing action ( and aberrations ) occur before the electrons hit the specimen in the STEM, but afterward in the TEM. The STEMs use of SEM - like beam rastering simplifies annular dark - field imaging, and other analytical techniques, but also means that image data is acquired in serial rather than in parallel fashion. Often TEM can be equipped with the scanning option and then it can function both as TEM and STEM. Sample preparation An insect coated in gold for viewing with a scanning electron microscope Materials to be viewed under an electron microscope may require processing to produce a suitable sample. The technique required varies depending on the specimen and the analysis required : Chemical fixation -- for biological specimens aims to stabilize the specimen 's mobile macromolecular structure by chemical crosslinking of proteins with aldehydes such as formaldehyde and glutaraldehyde, and lipids with osmium tetroxide. Negative stain -- suspensions containing nanoparticles or fine biological material ( such as viruses and bacteria ) are briefly mixed with a dilute solution of an electron - opaque solution such as ammonium molybdate, uranyl acetate ( or formate ), or phosphotungstic acid. This mixture is applied to a suitably coated EM grid, blotted, then allowed to dry. Viewing of this preparation in the TEM should be carried out without delay for best results. The method is important in microbiology for fast but crude morphological identification, but can also be used as the basis for high resolution 3D reconstruction using EM tomography methodology when carbon films are used for support. Negative staining is also used for observation of nanoparticles. Cryofixation -- freezing a specimen so rapidly, in liquid ethane, and maintained at liquid nitrogen or even liquid helium temperatures, so that the water forms vitreous ( non-crystalline ) ice. This preserves the specimen in a snapshot of its solution state. An entire field called cryo - electron microscopy has branched from this technique. With the development of cryo - electron microscopy of vitreous sections ( CEMOVIS ), it is now possible to observe samples from virtually any biological specimen close to its native state. Dehydration -- or replacement of water with organic solvents such as ethanol or acetone, followed by critical point drying or infiltration with embedding resins. Also freeze drying. Embedding, biological specimens -- after dehydration, tissue for observation in the transmission electron microscope is embedded so it can be sectioned ready for viewing. To do this the tissue is passed through a ' transition solvent ' such as propylene oxide ( epoxypropane ) or acetone and then infiltrated with an epoxy resin such as Araldite, Epon, or Durcupan ; tissues may also be embedded directly in water - miscible acrylic resin. After the resin has been polymerized ( hardened ) the sample is thin sectioned ( ultrathin sections ) and stained -- it is then ready for viewing. Embedding, materials -- after embedding in resin, the specimen is usually ground and polished to a mirror - like finish using ultra-fine abrasives. The polishing process must be performed carefully to minimize scratches and other polishing artifacts that reduce image quality. Metal shadowing -- Metal ( e.g. platinum ) is evaporated from an overhead electrode and applied to the surface of a biological sample at an angle. The surface topography results in variations in the thickness of the metal that are seen as variations in brightness and contrast in the electron microscope image. Replication -- A surface shadowed with metal ( e.g. platinum, or a mixture of carbon and platinum ) at an angle is coated with pure carbon evaporated from carbon electrodes at right angles to the surface. This is followed by removal of the specimen material ( e.g. in an acid bath, using enzymes or by mechanical separation ) to produce a surface replica that records the surface ultrastructure and can be examined using transmission electron microscopy. Sectioning -- produces thin slices of specimen, semitransparent to electrons. These can be cut on an ultramicrotome with a diamond knife to produce ultra-thin sections about 60 -- 90 nm thick. Disposable glass knives are also used because they can be made in the lab and are much cheaper. Staining -- uses heavy metals such as lead, uranium or tungsten to scatter imaging electrons and thus give contrast between different structures, since many ( especially biological ) materials are nearly `` transparent '' to electrons ( weak phase objects ). In biology, specimens can be stained `` en bloc '' before embedding and also later after sectioning. Typically thin sections are stained for several minutes with an aqueous or alcoholic solution of uranyl acetate followed by aqueous lead citrate. Freeze - fracture or freeze - etch -- a preparation method particularly useful for examining lipid membranes and their incorporated proteins in `` face on '' view. The fresh tissue or cell suspension is frozen rapidly ( cryofixation ), then fractured by breaking or by using a microtome while maintained at liquid nitrogen temperature. The cold fractured surface ( sometimes `` etched '' by increasing the temperature to about − 100 ° C for several minutes to let some ice sublime ) is then shadowed with evaporated platinum or gold at an average angle of 45 ° in a high vacuum evaporator. A second coat of carbon, evaporated perpendicular to the average surface plane is often performed to improve stability of the replica coating. The specimen is returned to room temperature and pressure, then the extremely fragile `` pre-shadowed '' metal replica of the fracture surface is released from the underlying biological material by careful chemical digestion with acids, hypochlorite solution or SDS detergent. The still - floating replica is thoroughly washed free from residual chemicals, carefully fished up on fine grids, dried then viewed in the TEM. Freeze - fracture replica immunogold labelling ( FRIL ) -- the freeze - fracture method has been modified to allow the identification of the components of the fracture face by immunogold labeling. Instead of removing all the underlying tissue of the thawed replica as the final step before viewing in the microscope the tissue thickness is minimized during or after the fracture process. The thin layer of tissue remains bound to the metal replica so it can be immunogold labeled with antibodies to the structures of choice. The thin layer of the original specimen on the replica with gold attached allows the identification of structures in the fracture plane. There are also related methods which label the surface of etched cells and other replica labeling variations. Ion beam milling -- thins samples until they are transparent to electrons by firing ions ( typically argon ) at the surface from an angle and sputtering material from the surface. A subclass of this is focused ion beam milling, where gallium ions are used to produce an electron transparent membrane in a specific region of the sample, for example through a device within a microprocessor. Ion beam milling may also be used for cross-section polishing prior to SEM analysis of materials that are difficult to prepare using mechanical polishing. Conductive coating -- an ultrathin coating of electrically conducting material, deposited either by high vacuum evaporation or by low vacuum sputter coating of the sample. This is done to prevent the accumulation of static electric fields at the specimen due to the electron irradiation required during imaging. The coating materials include gold, gold / palladium, platinum, tungsten, graphite, etc. Earthing -- to avoid electrical charge accumulation on a conductive coated sample, it is usually electrically connected to the metal sample holder. Often an electrically conductive adhesive is used for this purpose. Disadvantages Electron microscopes are expensive to build and maintain, but the capital and running costs of confocal light microscope systems now overlaps with those of basic electron microscopes. Microscopes designed to achieve high resolutions must be housed in stable buildings ( sometimes underground ) with special services such as magnetic field cancelling systems. The samples largely have to be viewed in vacuum, as the molecules that make up air would scatter the electrons. An exception is liquid - phase electron microscopy using either a closed liquid cell or an environmental chamber, for example, in the environmental scanning electron microscope, which allows hydrated samples to be viewed in a low - pressure ( up to 20 Torr or 2.7 kPa ) wet environment. Various techniques for in situ electron microscopy of gaseous samples have been developed as well. Scanning electron microscopes operating in conventional high - vacuum mode usually image conductive specimens ; therefore non-conductive materials require conductive coating ( gold / palladium alloy, carbon, osmium, etc. ). The low - voltage mode of modern microscopes makes possible the observation of non-conductive specimens without coating. Non-conductive materials can be imaged also by a variable pressure ( or environmental ) scanning electron microscope. Small, stable specimens such as carbon nanotubes, diatom frustules and small mineral crystals ( asbestos fibres, for example ) require no special treatment before being examined in the electron microscope. Samples of hydrated materials, including almost all biological specimens have to be prepared in various ways to stabilize them, reduce their thickness ( ultrathin sectioning ) and increase their electron optical contrast ( staining ). These processes may result in artifacts, but these can usually be identified by comparing the results obtained by using radically different specimen preparation methods. Since the 1980s, analysis of cryofixed, vitrified specimens has also become increasingly used by scientists, further confirming the validity of this technique. Applications Semiconductor and data storage Circuit edit Defect analysis Failure analysis Biology and life sciences Cryobiology Cryo - electron microscopy Diagnostic electron microscopy Drug research ( e.g. antibiotics ) Electron tomography Particle analysis Particle detection Protein localization Structural biology Tissue imaging Toxicology Virology ( e.g. viral load monitoring ) Materials research Device testing and characterization Dynamic materials experiments Electron beam - induced deposition Materials qualification Medical research Nanometrology Nanoprototyping Industry Chemical / Petrochemical Direct beam - writing fabrication Food science Forensics Fractography Micro-characterization Mining ( mineral liberation analysis ) Pharmaceutical QC See also Acronyms in microscopy Electron diffraction Electron energy loss spectroscopy ( EELS ) Electron microscope images Energy filtered transmission electron microscopy ( EFTEM ) Environmental scanning electron microscope ( ESEM ) Field emission microscope HiRISE In situ electron microscopy Microscope image processing Microscopy Nanoscience Nanotechnology Neutron microscope Scanning confocal electron microscopy Scanning electron microscope ( SEM ) Scanning tunneling microscope Surface science Transmission Electron Aberration - Corrected Microscope X-ray diffraction X-ray microscope References ^ Jump up to : Erni, Rolf ; Rossell, MD ; Kisielowski, C ; Dahmen, U ( 2009 ). `` Atomic - Resolution Imaging with a Sub-50 - pm Electron Probe ''. Physical Review Letters. 102 ( 9 ) : 096101. Bibcode : 2009PhRvL. 102i6101E. doi : 10.1103 / PhysRevLett. 102.096101. PMID 19392535. Jump up ^ Mathys, Daniel, Zentrum für Mikroskopie, University of Basel : Die Entwicklung der Elektronenmikroskopie vom Bild über die Analyse zum Nanolabor, p. 8 Jump up ^ Dannen, Gene ( 1998 ) Leo Szilard the Inventor : A Slideshow ( 1998, Budapest, conference talk ). dannen.com ^ Jump up to : Ruska, Ernst ( 1986 ). `` Ernst Ruska Autobiography ''. Nobel Foundation. Retrieved 2010 - 01 - 31. Jump up ^ Rudenberg, H Gunther ; Rudenberg, Paul G ( 2010 ). `` Chapter 6 -- Origin and Background of the Invention of the Electron Microscope : Commentary and Expanded Notes on Memoir of Reinhold Rüdenberg ''. Advances in Imaging and Electron Physics. 160. Elsevier. doi : 10.1016 / S1076 - 5670 ( 10 ) 60006 - 7. ISBN 978 - 0 - 12 - 381017 - 5. Jump up ^ Kruger DH ; Schneck P ; Gelderblom HR ( May 2000 ). `` Helmut Ruska the visualisation of viruses ''. Lancet. 355 ( 9216 ) : 1713 -- 7. doi : 10.1016 / S0140 - 6736 ( 00 ) 02250 - 9. PMID 10905259. Jump up ^ von Ardenne, M ; Beischer, D ( 1940 ). `` Untersuchung von metalloxyd - rauchen mit dem universal - elektronenmikroskop ''. Zeitschrift Electrochemie ( in German ). 46 : 270 -- 277. doi : 10.1002 / bbpc. 19400460406 ( inactive 2017 - 10 - 14 ). Jump up ^ History of electron microscopy, 1931 -- 2000. Authors.library.caltech.edu ( 2002 - 12 - 10 ). Retrieved on 2017 - 04 - 29. Jump up ^ `` James Hillier ''. Inventor of the Week : Archive. 2003 - 05 - 01. Retrieved 2010 - 01 - 31. Jump up ^ `` The Scale of Things ''. Office of Basic Energy Sciences, U.S. Department of Energy. 2006 - 05 - 26. Archived from the original on 2010 - 02 - 01. Retrieved 2010 - 01 - 31. Jump up ^ O'Keefe MA ; Allard LF ( 2004 - 01 - 18 ). `` Sub-Ångstrom Electron Microscopy for Sub-Ångstrom Nano - Metrology '' ( PDF ). Information Bridge : DOE Scientific and Technical Information -- Sponsored by OSTI. Jump up ^ Burgess, Jeremy ( 1987 ). Under the Microscope : A Hidden World Revealed. CUP Archive. p. 11. ISBN 0 - 521 - 39940 - 8. Jump up ^ `` Introduction to Electron Microscopy '' ( PDF ). FEI Company. p. 15. Retrieved 12 December 2012. Jump up ^ Antonovsky, A. ( 1984 ). `` The application of colour to sem imaging for increased definition ''. Micron and Microscopica Acta. 15 ( 2 ) : 77 -- 84. doi : 10.1016 / 0739 - 6260 ( 84 ) 90005 - 4. Jump up ^ Danilatos, G.D. ( 1986 ). `` Colour micrographs for backscattered electron signals in the SEM ''. Scanning. 9 ( 3 ) : 8 -- 18. doi : 10.1111 / j. 1365 - 2818.1986. tb04287. x. Jump up ^ Danilatos, G.D. ( 1986 ). `` Environmental scanning electron microscopy in colour ''. J. Microscopy. 142 : 317 -- 325. doi : 10.1002 / sca. 4950080104. Jump up ^ `` SPLEEM ''. National Center for Electron Microscopy ( NCEM ). Archived from the original on 2010 - 05 - 29. Retrieved 2010 - 01 - 31. Jump up ^ Luft, J.H. ( 1961 ). `` Improvements in epoxy resin embedding methods ''. The Journal of Biophysical and Biochemical Cytology. 9 ( 2 ). p. 409. PMC 2224998. PMID 13764136. Jump up ^ Williams, R.C. ; Wyckoff, R.W. ( 1945 - 06 - 08 ). `` Electron shadow micrography of the tobacco mosaic virus protein ''. Science. 101 ( 2632 ) : 594 -- 596. Bibcode : 1945Sci... 101... 594W. doi : 10.1126 / science. 101.2632. 594. PMID 17739444. Jump up ^ Juniper, B.E. ; Bradley, D.E. ( 1958 ). `` The carbon replica technique in the study of the ultrastructure of leaf surfaces ''. Journal of ultrastructure research. 2 ( 1 ) : 16 -- 27. doi : 10.1016 / s0022 - 5320 ( 58 ) 90045 - 5. Jump up ^ Reynolds, E.S. ( 1963 ). `` The use of lead citrate at high pH as an electron - opaque stain in electron microscopy ''. Journal of Cell Biology. 17 : 208 -- 212. doi : 10.1083 / jcb. 17.1. 208. PMC 2106263. PMID 13986422. Jump up ^ Meryman H.T. and Kafig E. ( 1955 ). The study of frozen specimens, ice crystals and ices crystal growth by electron microscopy. Naval Med. Res. Ints. Rept NM 000 018.01. 09 Vol. 13 pp 529 -- 544 Jump up ^ Steere, Russell L. ( 1957 - 01 - 25 ). `` Electron microscopy of structural detail in frozen biological specimens ''. The Journal of Biophysical and Biochemical Cytology. 3 ( 1 ) : 45 -- 60. PMC 2224015. PMID 13416310. Jump up ^ Moor H, Mühlethaler K. Fine structure in frozen - etched yeast cells. The Journal of Cell Biology. 1963 ; 17 ( 3 ) : 609 -- 628. Jump up ^ Bullivant, Stanley ; Ames, Adelbert ( 1966 - 06 - 01 ). `` A simple freeze - fracture replication method for electron microscopy ''. The Journal of Cell Biology. 29 ( 3 ) : 435 -- 447. PMC 2106967. PMID 5962938. Jump up ^ Gruijters, W.T. ; Kistler, J ; Bullivant, S ; Goodenough, D.A. ( 1987 - 03 - 01 ). `` Immunolocalization of MP70 in lens fiber 16 - 17 - nm intercellular junctions ''. The Journal of Cell Biology. 104 ( 3 ) : 565 -- 572. PMC 2114558. PMID 3818793. Jump up ^ da Silva, Pedro Pinto ; Branton, Daniel ( 1970 - 06 - 01 ). `` Membrane splitting in freeze - etching ''. The Journal of Cell Biology. 45 ( 3 ) : 598 -- 605. PMC 2107921. PMID 4918216. Jump up ^ Rash, J.E. ; Johnson, T.J. ; Hudson, C.S. ; Giddings, F.D. ; Graham, W.F. ; Eldefrawi, M.E. ( 1982 - 11 - 01 ). `` Labelled - replica techniques : post-shadow labelling of intramembrane particles in freeze - fracture replicas ''. Journal of Microscopy. 128 ( Pt 2 ) : 121 -- 138. PMID 6184475. Jump up ^ de Jonge, N. ; Ross, F.M. ( 2011 ). `` Electron microscopy of specimens in liquid ''. Nature Nanotechnology. 6 ( 8 ) : 695 -- 704. Bibcode : 2003NatMa... 2... 532W. doi : 10.1038 / nmat944. Jump up ^ Gai, P.L. ; Boyes, E.D. ( 2009 ). `` Advances in atomic resolution in situ environmental transmission electron microscopy and 1A aberration corrected in situ electron microscopy ''. Microsc Res Tech. 72 ( 3 ) : 153 -- 164. arXiv : 1705.05754. doi : 10.1002 / jemt. 20668. Jump up ^ Adrian, Marc ; Dubochet, Jacques ; Lepault, Jean ; McDowall, Alasdair W. ( 1984 ). `` Cryo - electron microscopy of viruses ''. Nature. 308 ( 5954 ) : 32 -- 36. Bibcode : 1984Natur. 308... 32A. doi : 10.1038 / 308032a0. PMID 6322001. Jump up ^ Sabanay, I. ; Arad, T. ; Weiner, S. ; Geiger, B. ( 1991 ). `` Study of vitrified, unstained frozen tissue sections by cryoimmunoelectron microscopy ''. Journal of Cell Science. 100 ( 1 ) : 227 -- 236. PMID 1795028. Jump up ^ Kasas, S. ; Dumas, G. ; Dietler, G. ; Catsicas, S. ; Adrian, M. ( 2003 ). `` Vitrification of cryoelectron microscopy specimens revealed by high - speed photographic imaging ''. Journal of Microscopy. 211 ( 1 ) : 48 -- 53. doi : 10.1046 / j. 1365 - 2818.2003. 01193. x. Jump up ^ Boehme, L. ; Bresin, M. ; Botman, A. ; Ranney, J. ; Hastings, J.T. ( 2015 ). `` Focused electron beam induced etching of copper in sulfuric acid solutions ''. Nanotechnology. 26 ( 49 ) : 495301. Bibcode : 2015Nanot... 26W5301B. doi : 10.1088 / 0957 - 4484 / 26 / 49 / 495301. PMID 26567988. Jump up ^ Kacher, J. ; Cui, B. ; Robertson, I.M. ( 2015 ). `` In situ and tomographic characterization of damage and dislocation processes in irradiated metallic alloys by transmission electron microscopy ''. Journal of Materials Research. 30 ( 9 ) : 1202 -- 1213. Bibcode : 2015JMatR... 30.1202 K. doi : 10.1557 / jmr. 2015.14. Jump up ^ Rai, R.S. ; Subramanian, S. ( 2009 ). `` Role of transmission electron microscopy in the semiconductor industry for process development and failure analysis ''. Progress in crystal growth and characterization of materials. 55 ( 3 -- 4 ) : 63 -- 97. doi : 10.1016 / j. pcrysgrow. 2009.09. 002. Jump up ^ Morris, G.J. ; Goodrich, M. ; Acton, E. ; Fonseca, F. ( 2006 ). `` The high viscosity encountered during freezing in glycerol solutions : Effects on cryopreservation ''. Cryobiology. 52 ( 3 ) : 323 -- 334. doi : 10.1016 / j. cryobiol. 2006.01. 003. PMID 16499898. ^ Jump up to : von Appen, A. ; Beck, M. ( 2016 ). `` Structure determination of the nuclear pore complex with three - dimensional cryo electron microscopy '' ( PDF ). Journal of Molecular Biology. 428 ( 10A ) : 2001 -- 2010. doi : 10.1016 / j. jmb. 2016.01. 004. PMID 26791760. Jump up ^ Florian, P.E. ; Rouillé, Y. ; Ruta, S. ; Nichita, N. ; Roseanu, A. ( 2016 ). `` Recent advances in human viruses imaging studies ''. Journal of Basic Microbiology. 56 ( 6 ) : 591 -- 607. doi : 10.1002 / jobm. 201500575. PMID 27059598. ^ Jump up to : Cushnie, T.P. ; O'Driscoll, N.H. ; Lamb, A.J. ( 2016 ). `` Morphological and ultrastructural changes in bacterial cells as an indicator of antibacterial mechanism of action ''. Cellular and Molecular Life Sciences. 73 ( 23 ) : 4471 -- 4492. doi : 10.1007 / s00018 - 016 - 2302 - 2. PMID 27392605. Jump up ^ Li, M. - H. ; Yang, Y. - Q. ; Huang, B. ; Luo, X. ; Zhang, W. ; Han, M. ; Ru, J. - G. ( 2014 ). `` Development of advanced electron tomography in materials science based on TEM and STEM ''. Transactions of Nonferrous Metals Society of China. 24 ( 10 ) : 3031 -- 3050. doi : 10.1016 / S1003 - 6326 ( 14 ) 63441 - 5. Jump up ^ Li, W.J. ; Shao, L.Y. ; Zhang, D.Z. ; Ro, C.U. ; Hu, M. ; Bi, X.H. ; Geng, H. ; Matsuki, A. ; Niu, H.Y. ; Chen, J.M. ( 2016 ). `` A review of single aerosol particle studies in the atmosphere of East Asia : morphology, mixing state, source, and heterogeneous reactions ''. Journal of Cleaner Production. 112 ( 2 ) : 1330 -- 1349. doi : 10.1016 / j. jclepro2015. 04.050 ( inactive 2017 - 10 - 14 ). Jump up ^ Sousa, R.G. ; Esteves, T. ; Rocha, S. ; Figueiredo, F. ; Quelhas, P. ; Silva, L.M. ( 2015 ). `` Automatic detection of immunogold particles from electron microscopy images ''. Image Analysis and Recognition. Lecture Notes in Computer Science. 9164 : 377 -- 384. doi : 10.1007 / 978 - 3 - 319 - 20801 - 5_41. ISBN 978 - 3 - 319 - 20800 - 8. Jump up ^ Perkins, G.A. ( 2014 ). `` The use of miniSOG in the localization of mitochondrial proteins ''. Methods in Enzymology. Methods in Enzymology. 547 : 165 -- 179. doi : 10.1016 / B978 - 0 - 12 - 801415 - 8.00010 - 2. ISBN 9780128014158. PMID 25416358. Jump up ^ Chen, X.D. ; Ren, L.Q. ; Zheng, B. ; Liu, H. ( 2013 ). `` Physics and engineering aspects of cell and tissue imaging systems : microscopic devices and computer assisted diagnosis ''. Biophotonics in Pathology : Pathology at the Crossroads. 185 : 1 -- 22. doi : 10.3233 / 978 - 1 - 61499 - 234 - 9 - 1 ( inactive 2017 - 04 - 29 ). PMID 23542929. Jump up ^ Fagerland, J.A. ; Wall, H.G. ; Pandher, K. ; LeRoy, B.E. ; Gagne, G.D. ( 2012 ). `` Ultrastructural analysis in preclinical safety evaluation ''. Toxicologic Pathology. 40 ( 2 ) : 391 -- 402. doi : 10.1177 / 0192623311430239. PMID 22215513. Jump up ^ Heider, S. ; Metzner, C. ( 2014 ). `` Quantitative real - time single particle analysis of virions ''. Virology. 462 -- 463 : 199 -- 206. doi : 10.1016 / j. virol. 2014.06. 005. PMC 4139191. PMID 24999044. Jump up ^ Tsekouras, G. ; Mozer, A.J. ; Wallace, G.G. ( 2008 ). `` Enhanced performance of dye sensitized solar cells utilizing platinum electrodeposit counter electrodes ''. Journal of the Electrochemical Society. 155 ( 7 ) : K124 -- K128. doi : 10.1149 / 1.2919107. Jump up ^ Besenius, P. ; Portale, G. ; Bomans, P.H.H. ; Janssen, H.M. ; Palmans, A.R.A. ; Meijer, E.W. ( 2010 ). `` Controlling the growth and shape of chiral supramolecular polymers in water '' ( PDF ). Proceedings of the National Academy of Sciences of the United States of America. 107 ( 42 ) : 17888 -- 17893. Bibcode : 2010PNAS... 10717888B. doi : 10.1073 / pnas. 1009592107. Jump up ^ Furuya, K. ( 2008 ). `` Nanofabrication by advanced electron microscopy using intense and focused beam ''. Science and Technology of Advanced Materials. 9 ( 1 ) : Article 014110. Bibcode : 2008STAdM... 9a4110F. doi : 10.1088 / 1468 - 6996 / 9 / 1 / 014110. Jump up ^ Maloy, S.A. ; Sommer, W.F. ; James, M.R. ; Romero, T.J. ; Lopez, M.R. ; Zimmermann, E. ; Ledbetter, J.M. ( 2000 ). `` The accelerator production of tritium materials test program ''. Nuclear Technology. 132 ( 1 ) : 103 -- 114. ISSN 0029 - 5450. Jump up ^ Ukraintsev, V. ; Banke, B. ( 2012 ). `` Review of reference metrology for nanotechnology : significance, challenges, and solutions ''. Journal of Micro / Nanolithography, MEMS, and MOEMS. 11 ( 1 ) : 011010. doi : 10.1117 / 1. JMM. 11.1. 011010. Jump up ^ Wilhelmi, O. ; Roussel, L. ; Faber, P. ; Reyntjens, S. ; Daniel, G. ( 2010 ). `` Focussed ion beam fabrication of large and complex nanopatterns ''. Journal of Experimental Nanoscience. 5 ( 3 ) : 244 -- 250. Bibcode : 2010JENan... 5... 244W. doi : 10.1080 / 17458080903487448. Jump up ^ Vogt, E.T.C. ; Whiting, G.T. ; Chowdhury, A.D. ; Weckhuysen, B.M. ( 2015 ). `` Zeolites and zeotypes for oil and gas conversion ''. Advances in Catalysis. Advances in Catalysis. 58 : 143 -- 314. doi : 10.1016 / bs. acat. 2015.10. 001. ISBN 9780128021262. Jump up ^ Lai, S.E. ; Hong, Y.J. ; Chen, Y.T. ; Kang, Y.T. ; Chang, P. ; Yew, T.R. ( 2015 ). `` Direct - writing of Cu nano - patterns with an electron beam ''. Microscopy and Microanalysis. 21 ( 6 ) : 1639 -- 1643. Bibcode : 2015MiMic... 21.1639 L. doi : 10.1017 / S1431927615015111. PMID 26381450. Jump up ^ Sicignano, A. ; Di Monaco, R. ; Masi, P. ; Cavella, S. ( 2015 ). `` From raw material to dish : pasta quality step by step ''. Journal of the Science of Food and Agriculture. 95 ( 13 ) : 2579 -- 2587. doi : 10.1002 / jsfa. 7176. PMID 25783568. Jump up ^ Brożek - Mucha, Z. ( 2014 ). `` Scanning electron microscopy and X-ray microanalysis for chemical and morphological characterisation of the inorganic component of gunshot residue : selected problems ''. Biomed Research International. 2014 : 428038. doi : 10.1155 / 2014 / 428038. PMC 4082842. PMID 25025050. Jump up ^ Carbonell - Verdu, A. ; Garcia - Sanoguera, D. ; Jorda - Vilaplana, A. ; Sanchez - Nacher, L. ; Balart, R. ( 2016 ). `` A new biobased plasticizer for poly ( vinyl chloride ) based on epoxidized cottonseed oil ''. Journal of Applied Polymer Science. 33 ( 27 ) : 43642. doi : 10.1002 / app. 43642. Jump up ^ Ding, J. ; Zhang, Z.M. ; Wang, J.Q. ; Han, E.H. ; Tang, W.B. ; Zhang, M.L. ; Sun, Z.Y. ( 2015 ). `` Micro-characterization of dissimilar metal weld joint for connecting the pipe - nozzle to the safe - end in generation III nuclear power plant ''. Acta Metallurgica Sinica. 51 ( 4 ) : 425 -- 439. doi : 10.11900 / 0412.1961. 2014.00299 ( inactive 2017 - 10 - 14 ). Jump up ^ Tsikouras, B. ; Pe - Piper, G. ; Piper, D.J.W. ; Schaffer, M. ( 2011 ). `` Varietal heavy mineral analysis of sediment provenance, Lower Cretaceous Scotian Basin, eastern Canada ''. Sedimentary Geology. 237 ( 3 -- 4 ) : 150 -- 165. Bibcode : 2011SedG... 237... 150T. doi : 10.1016 / j. sedgeo. 2011.02. 011. Jump up ^ Li, X. ; Jiang, C. ; Pan, L.L. ; Zhang, H.Y. ; Hu, L. ; Li, T.X. ; Yang, X.H. ( 2015 ). `` Effects of preparing techniques and aging on dissolution behavior of the solid dispersions of NF / Soluplus / Kollidon SR : identification and classification by a combined analysis by FT - IR spectroscopy and computational approaches ''. Drug Development and Industrial Pharmacy. 41 ( 1 ) : 2 -- 1. doi : 10.3109 / 03639045.2014. 938080. PMID 25026247. External links Library resources about Electron microscopy Resources in your library Resources in other libraries Wikimedia Commons has media related to Electron microscope images. An Introduction to Electron Microscopy : resources for teachers and students Cell Centered Database -- Electron microscopy data Science Aid : Electron Microscopy : high school ( GCSE, A Level ) resource General Animations and explanations of various types of microscopy including electron microscopy ( Université Paris Sud ) Environmental Scanning Electron Microscopy ( ESEM ) ETH Zurich website : graphics and images illustrating various procedures Eva Nogales 's seminar : `` Introduction to Electron Microscopy '' FEI Image Contest : FEI has had a microscopy image contest every year since 2008 Introduction to electron microscopy by David Szondy Nanohedron.com image gallery : images generated by electron microscopy X-ray element analysis in electron microscopy : information portal with X-ray microanalysis and EDX contents History John H.L. Watson 's recollections at the University of Toronto when he worked with Hillier and Prebus Rubin Borasky Electron Microscopy Collection, 1930 -- 1988 ( Archives Center, National Museum of American History, Smithsonian Institution ) Other The Royal Microscopical Society, Electron Microscopy Section ( UK ) Albert Lleal. Natural history subjects at Scanning Electron Microscope SEM EM Images GND : 4014327 - 2 NDL : 00561446 Retrieved from `` https://en.wikipedia.org/w/index.php?title=Electron_microscope&oldid=812370689 '' Categories : Electron microscopy Microscopes Accelerator physics Anatomical pathology Pathology Scientific techniques German inventions Protein imaging 1931 in science Hidden categories : Pages with DOIs inactive since 2017 CS1 German - language sources ( de ) All articles with unsourced statements Articles with unsourced statements from April 2017 Articles with unsourced statements from July 2008 Commons category with local link different than on Wikidata Wikipedia articles with GND identifiers Talk Contents About Wikipedia Afrikaans Azərbaycanca Беларуская Български Bosanski Català Čeština Cymraeg Dansk Deutsch Eesti Ελληνικά Español Esperanto Euskara فارسی Français Frysk Galego 한국어 Հայերեն हिन्दी Hrvatski Bahasa Indonesia Italiano עברית ქართული Қазақша Kreyòl ayisyen Кыргызча Latviešu Lietuvių Magyar Македонски മലയാളം Bahasa Melayu Nederlands 日本 語 Norsk Norsk nynorsk ਪੰਜਾਬੀ پنجابی Polski Português Română Русский Scots Seeltersk Sicilianu Simple English Slovenčina Српски / srpski Srpskohrvatski / српскохрватски Suomi Svenska தமிழ் తెలుగు ไทย Türkçe Українська Tiếng Việt Winaray 粵語 中文 Edit links This page was last edited on 27 November 2017, at 13 : 53. Text is available under the Creative Commons Attribution - ShareAlike License ; additional terms may apply. By using this site, you agree to the Terms of Use and Privacy Policy. Wikipedia ® is a registered trademark of the Wikimedia Foundation, Inc., a non-profit organization. About Wikipedia

Question: what did the electron microscope allow scientists to see
Answer: the ultrastructure