Strain gauge - wikipedia Strain gauge Typical foil strain gauge ; the blue region is conductive and resistance is measured from one large blue pad to the other. The gauge is far more sensitive to strain in the vertical direction than in the horizontal direction. The markings outside the active area help to align the gauge during installation. A strain gauge is a device used to measure strain on an object. Invented by Edward E. Simmons and Arthur C. Ruge in 1938, the most common type of strain gauge consists of an insulating flexible backing which supports a metallic foil pattern. The gauge is attached to the object by a suitable adhesive, such as cyanoacrylate. As the object is deformed, the foil is deformed, causing its electrical resistance to change. This resistance change, usually measured using a Wheatstone bridge, is related to the strain by the quantity known as the gauge factor. Contents 1 Physical operation 2 Gauge factor 3 In practice 3.1 Variations in temperature 4 Errors and compensation 5 Other types 6 Mechanical types 7 See also 8 References Physical operation An unmounted resistive foil strain gauge. A strain gauge takes advantage of the physical property of electrical conductance and its dependence on the conductor 's geometry. When an electrical conductor is stretched within the limits of its elasticity such that it does not break or permanently deform, it will become narrower and longer, changes that increase its electrical resistance end - to - end. Conversely, when a conductor is compressed such that it does not buckle, it will broaden and shorten, changes that decrease its electrical resistance end - to - end. From the measured electrical resistance of the strain gauge, the amount of induced stress may be inferred. A typical strain gauge arranges a long, thin conductive strip in a zig - zag pattern of parallel lines. This does not increase the sensitivity, since the percentage change in resistance for a given strain for the entire zig - zag is the same as for any single trace. However, a single linear trace would have to be extremely thin and hence liable to overheating ( which would both change its resistance and cause it to expand ), or would have to be operated at a much lower voltage, making it harder to measure resistance changes accurately. Gauge factor The gauge factor G F ( displaystyle GF ) is defined as : G F = Δ R / R G ε ( displaystyle GF = ( \frac ( Delta R / R_ ( G ) ) ( epsilon ) ) ) where Δ R ( displaystyle Delta R ) is the change in resistance caused by strain, R G ( displaystyle R_ ( G ) ) is the resistance of the undeformed gauge, and ε ( displaystyle epsilon ) is strain. For common metallic foil gauges, the gauge factor is usually a little over 2. For a single active gauge and three dummy resistors of the same resistance about the active gauge in a Wheatstone bridge configuration, the output v ( displaystyle v ) from the bridge is approximately : v = B V ⋅ G F ⋅ ε 4 ( displaystyle v = ( \frac ( BV cdot GF cdot epsilon ) ( 4 ) ) ) where B V ( displaystyle BV ) is the bridge excitation voltage. Foil gauges typically have active areas of about 2 -- 10 mm in size. With careful installation, the correct gauge, and the correct adhesive, strains up to at least 10 % can be measured. In practice Visualization of the working concept behind the strain gauge on a beam under exaggerated bending. An excitation voltage is applied to input leads of the gauge network, and a voltage reading is taken from the output leads. Typical input voltages are 5 V or 12 V and typical output readings are in millivolts. Foil strain gauges are used in many situations. Different applications place different requirements on the gauge. In most cases the orientation of the strain gauge is significant. Gauges attached to a load cell would normally be expected to remain stable over a period of years, if not decades ; while those used to measure response in a dynamic experiment may only need to remain attached to the object for a few days, be energized for less than an hour, and operate for less than a second. Strain gauges are attached to the substrate with a special glue. The type of glue depends on the required lifetime of the measurement system. For short term measurements ( up to some weeks ) cyanoacrylate glue is appropriate, for long lasting installation epoxy glue is required. Usually epoxy glue requires high temperature curing ( at about 80 - 100 ° C ). The preparation of the surface where the strain gauge is to be glued is of the utmost importance. The surface must be smoothed ( e.g. with very fine sand paper ), deoiled with solvents, the solvent traces must then be removed and the strain gauge must be glued immediately after this to avoid oxidation or pollution of the prepared area. If these steps are not followed the strain gauge binding to the surface may be unreliable and unpredictable measurement errors may be generated. Strain gauge based technology is utilized commonly in the manufacture of pressure sensors. The gauges used in pressure sensors themselves are commonly made from silicon, polysilicon, metal film, thick film, and bonded foil. Variations in temperature Variations in temperature will cause a multitude of effects. The object will change in size by thermal expansion, which will be detected as a strain by the gauge. Resistance of the gauge will change, and resistance of the connecting wires will change. Most strain gauges are made from a constantan alloy. Various constantan alloys and Karma alloys have been designed so that the temperature effects on the resistance of the strain gauge itself cancel out the resistance change of the gauge due to the thermal expansion of the object under test. Because different materials have different amounts of thermal expansion, self - temperature compensation ( STC ) requires selecting a particular alloy matched to the material of the object under test. Strain gauges that are not self - temperature - compensated ( such as isoelastic alloy ) can be temperature compensated by use of the dummy gauge technique. A dummy gauge ( identical to the active strain gauge ) is installed on an unstrained sample of the same material as the test specimen. The sample with the dummy gauge is placed in thermal contact with the test specimen, adjacent to the active gauge. The dummy gauge is wired into a Wheatstone bridge on an adjacent arm to the active gauge so that the temperature effects on the active and dummy gauges cancel each other. ( Murphy 's Law was originally coined in response to a set of gauges being incorrectly wired into a Wheatstone bridge. ) Temperature effects on the lead wires can be cancelled by using a `` 3 - wire bridge '' or a `` 4 - wire ohm circuit '' ( also called a `` 4 - wire Kelvin connection '' ). In any case it is a good engineering practice to keep the Wheatstone bridge voltage drive low enough to avoid the self heating of the strain gauge. The self heating of the strain gauge depends on its mechanical characteristic ( large strain gauges are less prone to self heating ). Low voltage drive levels of the bridge reduce the sensitivity of the overall system. Errors and compensation Zero Offset - If the impedance of the four gauge arms are not exactly the same after bonding the gauge to the force collector, there will be a zero offset which can be compensated by introducing a parallel resistor to one or more of the gauge arms. Temperature coefficient of gauge factor ( TCGF ) is the change of sensitivity of the device to strain with change in temperature. This is generally compensated for by the introduction of a fixed resistance in the input leg, whereby the effective supplied voltage will decrease with a temperature increase, compensating for the increase in sensitivity with the temperature increase. This is known as modulus compensation in transducer circuits. As the temperature rises the load cell element becomes more elastic and therefore under a constant load will deform more and lead to an increase in output ; but the load is still the same. The clever bit in all this is that the resistor in the bridge supply must be a temperature sensitive resistor that is matched to both the material to which the gauge is bonded and also to the gauge element material. The value of that resistor is dependent on both of those values and can be calculated. In simple terms if the output increases then the resistor value also increase thereby reducing the net voltage to the transducer. Get the resistor value right and you will see no change. Zero shift with temperature - If the TCGF of each gauge is not the same, there will be a zero shift with temperature. This is also caused by anomalies in the force collector. This is usually compensated for with one or more resistors strategically placed in the compensation network. Linearity is an error whereby the sensitivity changes across the pressure range. This is commonly a function of the force collection thickness selection for the intended pressure and the quality of the bonding. Hysteresis is an error of return to zero after pressure excursion. Repeatability - This error is sometimes tied - in with hysteresis but is across the pressure range. EMI induced errors - As strain gauges output voltage is in the mV range, even μV if the Wheatstone bridge voltage drive is kept low to avoid self heating of the element, special care must be taken in output signal amplification to avoid amplifying also the superimposed noise. A solution which is frequently adopted is to use `` carrier frequency '' amplifiers which convert the voltage variation into a frequency variation ( as in VCOs ) and have a narrow bandwidth thus reducing out of band EMI. Overloading -- If a strain gauge is loaded beyond its design limit ( measured in microstrain ) its performance degrades and can not be recovered. Normally good engineering practice suggests not to stress strain gauges beyond ± 3000 microstrain. Humidity -- If the wires connecting the strain gauge to the signal conditioner are not protected against humidity, such as bare wire, corrosion can occur, leading to parasitic resistance. This can allow currents to flow between the wires and the substrate to which the strain gauge is glued, or between the two wires directly, introducing an error which competes with the current flowing through the strain gauge. For this reason, high - current, low - resistance strain gauges ( 120 ohm ) are less prone to this type of error. To avoid this error it is sufficient to protect the strain gauges wires with insulating enamel ( e.g., epoxy or polyurethane type ). Strain gauges with unprotected wires may be used only in a dry laboratory environment but not in an industrial one. In some applications, strain gauges add mass and damping to the vibration profiles of the hardware they are intended to measure. In the turbomachinery industry, one used alternative to strain gauge technology in the measurement of vibrations on rotating hardware is the Non-Intrusive Stress Measurement System, which allows measurement of blade vibrations without any blade or disc - mounted hardware... Other types For measurements of small strain, semiconductor strain gauges, so called piezoresistors, are often preferred over foil gauges. A semiconductor gauge usually has a larger gauge factor than a foil gauge. Semiconductor gauges tend to be more expensive, more sensitive to temperature changes, and are more fragile than foil gauges. Nanoparticle - based strain gauges emerge as a new promising technology. These resistive sensors whose active area is made by an assembly of conductive nanoparticles, such as gold or carbon, combine a high gauge factor, a large deformation range and a small electrical consumption due to their high impedance. In biological measurements, especially blood flow and tissue swelling, a variant called mercury - in - rubber strain gauge is used. This kind of strain gauge consists of a small amount of liquid mercury enclosed in a small rubber tube, which is applied around e.g., a toe or leg. Swelling of the body part results in stretching of the tube, making it both longer and thinner, which increases electrical resistance. Fiber optic sensing can be employed to measure strain along an optical fiber. Measurements can be distributed along the fiber, or taken at predetermined points on the fiber. The 2010 America 's Cup boats Alinghi 5 and USA - 17 both employ embedded sensors of this type. Microscale strain gauges are widely used in microelectromechanical systems ( MEMS ) to measure strains such as those induced by force, acceleration, pressure or sound. As example, airbags in cars are often triggered with MEMS accelerometers. As alternative to piezo - resistant strain gauges, integrated optical ring resonators may be used to measure strain in Micro-Opto - Electro - Mechanical Systems ( MOEMS ). Capacitive strain gauges use a variable capacitor to indicate the level of mechanical deformation. Vibrating wire strain gauges are used in geotechnical and civil engineering applications. The gauge consists of a vibrating, tensioned wire. The strain is calculated by measuring the resonant frequency of the wire ( an increase in tension increases the resonant frequency ). Mechanical types Simple mechanical types are used in civil engineering to measure movement of buildings, foundations, and other structures. More sophisticated mechanical types incorporate dial indicators and mechanisms to compensate for temperature changes. These types can measure movements as small as 0.002 mm. See also Resistance thermometer References Jump up ^ Strain Gage : Materials Jump up ^ Strain Gage : Sensitivity Jump up ^ Constantan Alloy : Strain Gauge Selection Jump up ^ Shull, Larry C., `` Basic Circuits '', Hannah, R.L. and Reed, S.E. ( Eds. ) ( 1992 ). Strain Gage Users ' Manual, p. 122. Society for Experimental Mechanics. ISBN 0 - 912053 - 36 - 4. Jump up ^ Spark, N. ( 2006 ). A History of Murphy 's Law. Periscope Film. ISBN 978 - 0 - 9786388 - 9 - 4 Jump up ^ The Strain Gage Jump up ^ ( 1 ) Jump up ^ Bryzek, J. ; Roundy, S. ; Bircumshaw, B. ; Chung, C. ; Castellino, K. ; Stetter, J.R. ; Vestel, M. ( 10 April 2006 ). `` Marvelous MEMS ''. IEEE Circuits and Devices Magazine. 22 ( 2 ) : 8 -- 28. doi : 10.1109 / MCD. 2006.1615241. Jump up ^ Westerveld, W.J. ; Leinders, S.M. ; Muilwijk, P.M. ; Pozo, J. ; van den Dool, T.C. ; Verweij, M.D. ; Yousefi, M. ; Urbach, H.P. ( 10 January 2014 ). `` Characterization of Integrated Optical Strain Sensors Based on Silicon Waveguides ''. IEEE Journal of Selected Topics in Quantum Electronics. 20 ( 4 ). doi : 10.1109 / JSTQE. 2013.2289992. Jump up ^ Mastrad Quality and Test Systems web site Sensors Acoustic, sound, vibration Geophone Hydrophone Microphone Seismometer Automotive, transportation Air -- fuel ratio meter Blind spot monitor Crankshaft position sensor Curb feeler Defect detector Engine coolant temperature sensor Hall effect sensor MAP sensor Mass flow sensor Omniview technology Oxygen sensor Parking sensors Radar gun Speed sensor Speedometer Throttle position sensor Tire - pressure monitoring system Torque sensor Transmission fluid temperature sensor Turbine speed sensor Variable reluctance sensor Vehicle speed sensor Water sensor Wheel speed sensor Chemical Breathalyzer Carbon dioxide sensor Carbon monoxide detector Catalytic bead sensor Chemical field - effect transistor Electrochemical gas sensor Electrolyte -- insulator -- semiconductor sensor Electronic nose Fluorescent chloride sensors Holographic sensor Hydrocarbon dew point analyzer Hydrogen sensor Hydrogen sulfide sensor Infrared point sensor Ion selective electrode Microwave chemistry sensor Nitrogen oxide sensor Nondispersive infrared sensor Olfactometer Optode Oxygen sensor Pellistor pH glass electrode Potentiometric sensor Redox electrode Smoke detector Zinc oxide nanorod sensor Electric, magnetic, radio Current sensor Electroscope Galvanometer Hall effect sensor Hall probe Magnetic anomaly detector Magnetometer MEMS magnetic field sensor Metal detector Planar Hall sensor Radio direction finder Test light Environment, weather, moisture Actinometer Bedwetting alarm Ceilometer Dew warning Electrochemical gas sensor Fish counter Frequency domain sensor Gas detector Hook gauge evaporimeter Humistor Hygrometer Leaf sensor Psychrometer Pyranometer Pyrgeometer Rain gauge Rain sensor SNOTEL Snow gauge Soil moisture sensor Stream gauge Tide gauge Weather radar Flow, fluid velocity Air flow meter Anemometer Flow sensor Gas meter Mass flow sensor Water metering Ionising radiation, subatomic particles Bubble chamber Cloud chamber Geiger -- Müller tube Geiger counter Ionization chamber Neutron detection Particle detector Proportional counter Scintillation counter Semiconductor detector Scintillator Thermoluminescent dosimeter Wire chamber Navigation instruments Airspeed indicator Machmeter Altimeter Attitude indicator Depth gauge Fluxgate compass Gyroscope Inertial navigation system Inertial reference unit Magnetic compass MHD sensor Ring laser gyroscope Turn coordinator Variometer Vibrating structure gyroscope Yaw - rate sensor Position, angle, displacement Accelerometer Angular rate sensor Auxanometer Capacitive displacement sensor Capacitive sensing Gravimeter Inclinometer Integrated circuit piezoelectric sensor Laser rangefinder Laser surface velocimeter Lidar Linear encoder Linear variable differential transformer Liquid capacitive inclinometers Odometer Photoelectric sensor Piezoelectric accelerometer Position sensor Rotary encoder Rotary variable differential transformer Selsyn Sudden Motion Sensor Tachometer Tilt sensor Ultrasonic thickness gauge Variable reluctance sensor Velocity receiver Optical, light, imaging Active pixel sensor Angle -- sensitive pixel Back - illuminated sensor Charge - coupled device Contact image sensor Electro - optical sensor Flame detector Infrared Kinetic inductance detector LED as light sensor Light - addressable potentiometric sensor Nichols radiometer Optical fiber Photodetector Photodiode Photoelectric sensor Photoionization detector Photomultiplier Photoresistor Photoswitch Phototransistor Phototube Position sensitive device Scintillometer Shack -- Hartmann wavefront sensor Single - photon avalanche diode Superconducting nanowire single - photon detector Transition edge sensor Tristimulus colorimeter Visible - light photon counter Wavefront sensor Pressure Barograph Barometer Boost gauge Bourdon gauge Hot - filament ionization gauge Ionization gauge McLeod gauge Oscillating U-tube Permanent Downhole Gauge Piezometer Pirani gauge Pressure gauge Pressure sensor Tactile sensor Time pressure gauge Force, density, level Bhangmeter Force gauge Hydrometer Level sensor Load cell Magnetic level gauge Nuclear density gauge Piezoelectric sensor Strain gauge Torque sensor Viscometer Thermal, heat, temperature Bimetallic strip Bolometer Calorimeter Exhaust gas temperature gauge Flame detection Gardon gauge Golay cell Heat flux sensor Infrared thermometer Microbolometer Microwave radiometer Net radiometer Quartz thermometer Resistance thermometer Silicon bandgap temperature sensor Special sensor microwave / imager Thermistor Thermocouple Thermometer Proximity, presence Alarm sensor Doppler radar Motion detector Occupancy sensor Passive infrared sensor Proximity sensor Reed switch Stud finder Touch switch Triangulation sensor Wired glove Sensor technology Active pixel sensor Back - illuminated sensor Biochip Biosensor Capacitance probe Carbon paste electrode Catadioptric sensor Digital sensors Displacement receiver Electromechanical film Electro - optical sensor Fabry -- Pérot interferometer Fisheries acoustics Image sensor Image sensor format Inductive sensor Intelligent sensor Lab - on - a-chip Leaf sensor Machine vision Microelectromechanical systems Photoelasticity Quantum sensor Radar Ground - penetrating radar Synthetic aperture radar Radar tracker Sensor array Sensor fusion Sensor grid Sensor node Soft sensor Sonar Staring array Transducer Ultrasonic sensor Video sensor technology Visual sensor network Wheatstone bridge Wireless sensor network Related List of sensors Retrieved from `` https://en.wikipedia.org/w/index.php?title=Strain_gauge&oldid=836743070 '' Categories : Sensors Elasticity ( physics ) Talk Contents About Wikipedia Alemannisch Català Čeština Dansk Deutsch Español فارسی Français हिन्दी Bahasa Indonesia Italiano Қазақша Magyar Bahasa Melayu Nederlands 日本 語 Oʻzbekcha / ўзбекча Polski Português Русский Suomi Svenska Татарча / tatarça Українська 中文 17 more Edit links This page was last edited on 16 April 2018, at 16 : 10. Text is available under the Creative Commons Attribution - ShareAlike License ; additional terms may apply. By using this site, you agree to the Terms of Use and Privacy Policy. Wikipedia ® is a registered trademark of the Wikimedia Foundation, Inc., a non-profit organization. About Wikipedia

Question: describe the structure of a metal wire strain gauge
Answer: arranges a long, thin conductive strip in a zig-zag pattern of parallel lines