Graduated cylinder - wikipedia Graduated cylinder Different types of graduated cylinder : 10mL, 25mL, 50mL and 100mL graduated cylinder A graduated cylinder, measuring cylinder or mixing cylinder is a common piece of laboratory equipment used to measure the volume of a liquid. It has a narrow cylindrical shape. Each marked line on the graduated cylinder represents the amount of liquid that has been measured. Contents 1 Materials and structure 2 Common uses 3 Scales and accuracy 4 Measurement 5 Additional images 6 References Materials and structure Large graduated cylinders are usually made up of polypropylene for its excellent chemical resistance or polymethylpentene for its transparency, making them lighter and less fragile than glass. Polypropylene ( PP ) is easy to repeatedly autoclave ; however, autoclaving in excess of about 121 ° C ( 250 ° F ) ( depending on the chemical formulation : typical commercial grade polypropylene melts in excess of 177 ° C ( 351 ° F ) ), can warp or damage polypropylene graduated cylinders, affecting accuracy. IHN traditional graduated cylinder ( A in the image ) is usually narrow and tall so as to increase the accuracy and precision of volume measurement ; it has a plastic or glass bottom and a `` spout '' for easy pouring of the measured liquid. An additional version is wide and low. Mixing cylinders ( B in the picture ) have ground glass joints instead of a spout, so they can be closed with a stopper or connect directly with other elements of a manifold. With this kind of cylinder, the metered liquid does not pour directly, but is often removed using a cannula. A graduated cylinder is meant to be read with the surface of the liquid at eye level, where the center of the meniscus shows the measurement line. Typical capacities of graduated cylinders are from 10 mL to 1000 mL. Common uses Graduated cylinders are often used to measure the volume of a liquid. Graduated cylinders are generally more accurate and precise than laboratory flasks and beakers, but they should not be used to perform volumetric analysis ; volumetric glassware, such as a volumetric flask or volumetric pipette, should be used, as it is even more accurate and precise. Graduated cylinders are sometimes used to measure the volume of a solid indirectly by measuring the displacement of a liquid. Scales and accuracy For accuracy the volume on graduated cylinders is depicted on scales with 3 significant digits : 100mL cylinders have 1ml grading divisions while 10mL cylinders have 0.1 mL grading divisions. Two classes of accuracy exist for graduated cylinders. Class A has double the accuracy of class B. Cylinders can have single or double scales. Single scales allow to read the volume from top to bottom ( filling volume ) while double scale cylinders allow reading for filling and pouring ( reverse scale ). Graduated cylinders are calibrated either `` to contain '' ( indicated liquid volume inside the cylinder ) and marked as `` TC '' or `` to deliver '' ( indicated liquid volume poured out, accounting for liquid traces left in the cylinder ) and marked `` TD ''. Formerly the tolerances for `` to deliver '' and `` to contain '' cylinders are distinct ; however now these are the same. Also, the international symbols `` IN '' and `` EX '' are more likely to be used instead of `` TC '' and `` TD '' respectively. Measurement If the reading is done and the value calculated is set to be 40.0 mL. The precise value would be 40.0 ± 0.1 or 40.1 to 39.9 mL If the reading is done and the value calculated is set to be 36.5 mL. The more precise value equates to 36.5 ± ( displaystyle pm ) 0.5 mL or 36.0 to 37.0 mL. To read the volume accurately, the observation must be at an eye level and read at the bottom of a meniscus of the liquid level. For example, in the picture on the bottom of the page, a 100mL graduated cylinder was used to measure the liquid. The cylinder contains 60mL liquid volume. The main reason as to why the reading of the volume is done via meniscus is due to the nature of the liquid in a closed surrounded space. By nature, liquid in the cylinder would be attracted to the wall around it through molecular forces. This forces the liquid surface to develop either a convex or concave shape, depending on the type of the liquid in the cylinder. Reading the liquid at the bottom part of a concave or the top part of the convex liquid is equivalent to reading the liquid at its meniscus. From the picture, the level of the liquid will be read at the bottom of the meniscus, which is the concave. The most accurate of the reading that could be done here is reduced down to 1 mL due to the given means of measurement on the cylinder. From this, the derived error would be one tenth of the least figure. For instance, if the reading is done and the value calculated is set to be 36.5 mL. The error, give or take 0.1 mL, must be included too. Therefore, the more precise value equates to 36.5 ± ( displaystyle pm ) 0.1 ; 36.4 or 36.6 mL. Therefore, there are 3 significant figures can be read from the given graduated cylinder picture. Another example, if the reading is done and the value calculated is set to be 40.0 mL. The precise value would be 40.0 ± ( displaystyle pm ) 0.1 ; 40.1 or 39.9 mL. Additional images Two graduated cylinders. IHN traditional graduated cylinder ( A in the image ), and mixing cylinders ( B in the picture ) References Jump up ^ ( httpmjd3bebdebcueqvyuqewudwevsuh://www.2spi.com/catalog/plasticware/graduated-cylinders.php `` Graduated Cylinders, Plastic - SPI Supplies '' ). www.2spi.com. Retrieved 2015 - 06 - 14. Jump up ^ http://www.elementalscientific.net/store/scripts/prodView.asp?idproduct=1239 Retrieved 15 March 2016 Jump up ^ Pradyot Patnaik ( 2003 ). `` Specifications for volumetric ware ''. Dean 's Handbook of Analytical Chemistry, 2nd Edition. McGraw - Hill. ISBN 978 - 0071410601. Jump up ^ http://www.astm.org/Standards/E1272.htm Jump up ^ `` Graduated Cylinders Information ''. Jump up ^ `` Graduated Cylinders ''. sizes.com. Retrieved 2016 - 02 - 23. Jump up ^ `` graduated cylinder '' ( PDF ). ohlone.edu. Archived from the original ( PDF ) on 2015 - 06 - 26. Retrieved 2015 - 06 - 25. Jump up ^ `` Volume Measurements with a Graduated Cylinder '' ( PDF ). Archived from the original ( PDF ) on 2016 - 02 - 16. Retrieved 2016 - 02 - 04. Jump up ^ `` Math Skills - Scientific Notation ''. www.chem.tamu.edu. Retrieved 2016 - 02 - 12. Jump up ^ Robinson, Michael ; Robinson, Mike ; Taylor, Mike ( 2002 - 01 - 01 ). Maths for Advanced Chemistry. Nelson Thornes. ISBN 9780748765829. Retrieved 15 March 2016. Wikimedia Commons has media related to Graduated cylinder. Laboratory equipment General Heaters Dryers Alcohol burner Bunsen burner Desiccator Heating mantle Hot plate Lab oven Kiln Meker - Fisher burner Striker Teclu burner Water bath Vacuum dry box Mixers Shakers Chemostat Homogenizer Liquid whistle Magnetic stirrer Mortar and pestle Shaker Sonicator Static mixer Stirring rod Vortex mixer Wash bottle Stands Clamps Holders Beaker clamp Clamp holder Tripod Burette clamp Extension clamp Flask clamp Funnel support Iron ring Pinch clamp Retort stand Screw clamp Test tube holder Test tube rack Wire gauze Lab drying rack Containers Storage Agar plate Cryogenic storage dewar Incubator Laminar flow cabinet Microtiter plate Petri dish Picotiter plate Refrigerator Weighing boat Weighing dish Other items Aspirator Autoclave Balance brush Cork borer Crucible Filter paper File Forceps Centrifuge Microscope Pipeclay triangle Spectrophotometer Splint Stopper Scoopula Spatula Test tube brush Wire brush Inoculation needle Inoculation loop Glassware Apparatus Dean - Stark Soxhlet extractor Kipp 's Bottles Boston round Pycnometer Condensers Cold finger Liebig Dishes Evaporating Petri Watch glass Flasks Büchner Vacuum ( Dewar ) Erlenmeyer Fernbach Fleaker Florence Retort Round - bottom Schlenk Volumetric Funnels Büchner Hirsch Dropping Separatory Measuring devices Burette Conical measure Cuvette Eudiometer Graduated cylinder Ostwald viscometer Pipette Pipette ( dropper ) Tubes Boiling Cragie Ignition Nuclear magnetic resonance ( NMR ) Test Thiele Thistle Other items Beaker Gas syringe Vial Analytical chemistry Compositional AutoAnalyzer CHN analyzer Colorimeter Inductively coupled plasma ( ICP ) device Gas chromatograph ( GC ) Liquid Chromatograph ( LC ) Mass spectrometer ( MS ) pH indicator pH meter Microscopy Scanning electron microscope ( SEM ) Transmission electron microscope ( TEM ) Thermochemistry Calorimeter differential scanning Melting point apparatus Thermometer Thermogravimetric analyzer ( TGA ) Other items Analytical balance Colony counter Spiral plater Nuclear magnetic resonance ( NMR ) instrument Plate reader Electronics Ammeter Current source Function generator Galvanostat Multimeter Network analyzer Oscilloscope Pulse generator Potentiostat Spectrum analyzer Time - domain reflectometer Voltage source Voltmeter Safety Personal protective equipment ( PPE ) Lab coat Face shield Respirator Rubber apron Safety shower Eye and hand Acid - resistant gloves Eyewash station Glove box Medical gloves Nitrile gloves Safety glasses Safety goggles Other items Acid ( solvent ) cabinet Biosafety cabinet Fire blanket Fire extinguisher Fume hood Instruments used in medical laboratories Retrieved from `` https://en.wikipedia.org/w/index.php?title=Graduated_cylinder&oldid=813968943 '' Categories : Laboratory glassware Volumetric instruments Talk Contents About Wikipedia Català Čeština Dansk Deutsch Español Esperanto Euskara فارسی Français 한국어 Hrvatski Bahasa Indonesia Italiano עברית Magyar Македонски Nederlands 日本 語 Norsk Norsk nynorsk Polski Português Română Русский Simple English Slovenčina Српски / srpski Srpskohrvatski / српскохрватски Suomi Svenska Tagalog Українська 中文 Edit links This page was last edited on 6 December 2017, at 05 : 19. Text is available under the Creative Commons Attribution - ShareAlike License ; additional terms may apply. By using this site, you agree to the Terms of Use and Privacy Policy. Wikipedia ® is a registered trademark of the Wikimedia Foundation, Inc., a non-profit organization. About Wikipedia

Question: which piece of laboratory equipment would be used to most accurately measure the volume of a liquid
Answer: graduated cylinder