Geiger counter - wikipedia Geiger counter Geiger -- Müller counter A `` two - piece '' bench type Geiger -- Müller counter with end - window detector Other names Geiger counter Uses Particle detector Inventor Hans Geiger Walther Müller Related items Geiger -- Müller tube The Geiger counter is an instrument used for measuring ionizing radiation used widely in applications such as radiation dosimetry, radiological protection, experimental physics and the nuclear industry. It detects ionizing radiation such as alpha particles, beta particles and gamma rays using the ionization effect produced in a Geiger -- Müller tube ; which gives its name to the instrument. In wide and prominent use as a hand - held radiation survey instrument, it is perhaps one of the world 's best - known radiation detection instruments. The original detection principle was discovered in 1908 at the Cavendish laboratory, but it was not until the development of the Geiger - Müller tube in 1928 that the Geiger - Müller counter became a practical instrument. Since then it has been very popular due to its robust sensing element and relatively low cost. However, there are limitations in measuring high radiation rates and the energy of incident radiation. Contents 1 Principle of operation 1.1 Readout 1.2 Limitations 2 Types and applications 2.1 Particle detection 2.2 Gamma and X-ray detection 2.3 Neutron detection 2.4 Gamma measurement -- personnel protection and process control 2.5 Physical design 2.6 Guidance on application use 3 History 4 Gallery 5 See also 6 References 7 External links Principle of operation Schematic of a Geiger counter using an `` end window '' tube for low penetration radiation. A loudspeaker is also used for indication Main article : Geiger -- Müller tube A Geiger counter consists of a Geiger - Müller tube, the sensing element which detects the radiation, and the processing electronics, which displays the result. The Geiger - Müller tube is filled with an inert gas such as helium, neon, or argon at low pressure, to which a high voltage is applied. The tube briefly conducts electrical charge when a particle or photon of incident radiation makes the gas conductive by ionization. The ionization is considerably amplified within the tube by the Townsend discharge effect to produce an easily measured detection pulse, which is fed to the processing and display electronics. This large pulse from the tube makes the G-M counter relatively cheap to manufacture, as the subsequent electronics is greatly simplified. The electronics also generates the high voltage, typically 400 -- 600 volts, that has to be applied to the Geiger - Müller tube to enable its operation. Readout There are two types of radiation readout : counts or radiation dose. The counts display is the simplest and is the number of ionizing events displayed either as a count rate, commonly `` counts per second '', or as a total over a set time period ( an integrated total ). The counts readout is normally used when alpha or beta particles are being detected. More complex to achieve is a display of radiation dose rate, displayed in a unit such as the sievert which is normally used for measuring gamma or X-ray dose rates. A G-M tube can detect the presence of radiation, but not its energy which influences the radiation 's ionising effect. Consequently, instruments measuring dose rate require the use of an energy compensated G-M tube, so that the dose displayed relates to the counts detected. The electronics will apply known factors to make this conversion, which is specific to each instrument and is determined by design and calibration. The readout can be analog or digital, and increasingly, modern instruments are offering serial communications with a host computer or network. There is usually an option to produce audible clicks representing the number of ionization events detected. This is the distinctive sound normally associated with hand held or portable Geiger counters. The purpose of this is to allow the user to concentrate on manipulation of the instrument whilst retaining auditory feedback on the radiation rate. Limitations There are two main limitations of the Geiger counter. Because the output pulse from a Geiger - Müller tube is always the same magnitude regardless of the energy of the incident radiation, the tube can not differentiate between radiation types. A further limitation is the inability to measure high radiation rates due to the `` dead time '' of the tube. This is an insensitive period after each ionization of the gas during which any further incident radiation will not result in a count, and the indicated rate is therefore lower than actual. Typically the dead time will reduce indicated count rates above about 10 to 10 counts per second depending on the characteristic of the tube being used. Whilst some counters have circuitry which can compensate for this, for accurate measurements ion chamber instruments are preferred for high radiation rates. Types and applications G-M counter with pancake type probe Laboratory use of a G-M counter with end window probe to measure beta radiation from a radioactive source The intended detection application of a Geiger counter dictates the tube design used. Consequently, there are a great many designs, but they can be generally categorised as `` end - window '', or windowless `` thin - walled '', or `` thick - walled '', and sometimes hybrids of these types. Particle detection The first historical uses of the Geiger principle were for the detection of alpha and beta particles, and the instrument is still used for this purpose today. For alpha particles and low energy beta particles the `` end - window '' type of G-M tube has to be used as these particles have a limited range even in free air, and are easily stopped by a solid material. Therefore, the tube requires a window which is thin enough to allow as many as possible of these particles through to the fill gas. The window is usually made of mica with a density of about 1.5 - 2.0 mg / cm. Alpha particles have the shortest range, and to detect these the window should ideally be within 10 mm of the radiation source due to alpha particle attenuation in free air. However, the G-M tube produces a pulse output which is the same magnitude for all detected radiation, so a Geiger counter with an end window tube can not distinguish between alpha and beta particles. A skilled operator can use varying distance from radiation source to detector to differentiate between alpha and high energy beta particles, but with the detector in close contact with the radiation source the two types are both detected and are indistinguishable. The `` pancake '' Geiger - Muller detector is a variant of the end window probe, but designed with a larger detection area to make checking quicker. However the pressure of the atmosphere against the low pressure of the fill gas limits the window size due to the limited strength of the window membrane. Some beta particles can also be detected by a thin - walled `` windowless '' G-M tube, which has no end window, but allows high energy beta particles to pass through the tube walls. Although the tube walls have a greater stopping power than a thin end window, they still allow these more energetic particles to reach the fill gas. End - window G-M detectors are still used as a general purpose portable radioactive contamination measurement and detection instrument, owing to their relatively low cost, robustness and their relatively high detection efficiency ; particularly with high energy beta particles. However, for discrimination between alpha and beta particles or provision of particle energy information, scintillation counters or proportional counters should be used. Those instrument types are manufactured with much larger detector areas, which means that checking for surface contamination is quicker than with a G-M instrument. Gamma and X-ray detection Geiger counters are widely used to detect gamma radiation, and for this the windowless tube is used. However, efficiency is generally low due to the poor interaction of gamma rays compared with alpha and beta particles. For instance, a chrome steel G-M tube is only about 1 % efficient over a wide range of energies. The article on the Geiger - Muller tube carries a more detailed account of the techniques used to detect photon radiation. For high energy gamma it largely relies on interaction of the photon radiation with the tube wall material, usually 1 -- 2 mm of chrome steel on a `` thick - walled '' tube, to produce electrons within the wall which can enter and ionize the fill gas. This is necessary as the low pressure gas in the tube has little interaction with high energy gamma photons. However, for low energy photons there is greater gas interaction and the direct gas ionisation effect increases. With decreasing energy the wall effect gives way to a combination of wall effect and direct ionisation, until direct gas ionisation dominates. Due to the variance in response to different photon energies, windowless tubes employ what is known as `` energy compensation '' which attempts to compensate for these variations over a large energy range. Low energy photon radiation such as low energy X rays or gamma rays interacts better with the fill gas. Consequently, a typical design for low energy photon detection for these is a long tube with a thin wall or with an end window. The tube has a larger gas volume than a steel walled tube to give an increased chance of particle interaction. Neutron detection Geiger tube filled with BF for detection of thermal neutrons A variation of the Geiger tube is used to measure neutrons, where the gas used is boron trifluoride or helium - 3 and a plastic moderator is used to slow the neutrons. This creates an alpha particle inside the detector and thus neutrons can be counted. Gamma measurement -- personnel protection and process control The term `` Geiger counter '' is commonly used to mean a hand - held survey type meter, however the Geiger principle is in wide use in installed `` area gamma '' alarms for personnel protection, and in process measurement and interlock applications. A Geiger tube is still the sensing device, but the processing electronics will have a higher degree of sophistication and reliability than that used in a hand held survey meter. Physical design Pancake G-M tube used for alpha and beta detection ; the delicate mica window is usually protected by a mesh when fitted in an instrument. For hand - held units there are two fundamental physical configurations : the `` integral '' unit with both detector and electronics in the same unit, and the `` two - piece '' design which has a separate detector probe and an electronics module connected by a short cable. In the 1930s a mica window was added to the cylindrical design allowing low - penetration radiation to pass through with ease. The integral unit allows single - handed operation, so the operator can use the other hand for personal security in challenging monitoring positions, but the two piece design allows easier manipulation of the detector, and is commonly used for alpha and beta surface contamination monitoring where careful manipulation of the probe is required or the weight of the electronics module would make operation unwieldy. A number of different sized detectors are available to suit particular situations, such as placing the probe in small apertures or confined spaces. Gamma and X-Ray detectors generally use an `` integral '' design so the Geiger -- Müller tube is conveniently within the electronics enclosure. This can easily be achieved because the casing usually has little attentuation, and is employed in ambient gamma measurements where distance from the source of radiation is not a significant factor. However, to facilitate more localised measurements such as `` surface dose '', the position of the tube in the enclosure is sometimes indicated by targets on the enclosure so an accurate measurement can be made with the tube at the correct orientation and a known distance from the surface. There is a particular type of gamma instrument known as a `` hot spot '' detector which has the detector tube on the end of a long pole or flexible conduit. These are used to measure high radiation gamma locations whilst protecting the operator by means of distance shielding. Particle detection of alpha and beta can used in both integral and two - piece designs. A pancake probe ( for alpha / beta ) is generally used to increase the area of detection in two - piece instruments whilst being relatively light weight. In integral instruments using an end window tube there is a window in the body of the casing to prevent shielding of particles. There are also hybrid instruments which have a separate probe for particle detection and a gamma detection tube within the electronics module. The detectors are switchable by the operator, depending the radiation type that is being measured. Guidance on application use In the United Kingdom the HSE has issued a user guidance note on selecting the best portable instrument type for the radiation measurement application concerned. This covers all radiation protection instrument technologies and is a useful comparative guide to the use of G-M detectors. The guide does not recommend the G-M detector for mixed alpha and beta contamination detection, and they are only recommended as `` satisfactory '' for beta - only contamination. However, for gamma and low - voltage X-rays they are recommended as the best instrument type. History An early alpha particle counter designed by Rutherford and Geiger. Early Geiger - Muller tube made in 1932 by Hans Geiger for laboratory use In 1908 Hans Geiger, under the supervision of Ernest Rutherford at the Victoria University of Manchester ( now the University of Manchester ), developed an experimental technique for detecting alpha particles that would later be used in the Geiger - Müller tube. This early counter was only capable of detecting alpha particles and was part of a larger experimental apparatus. The fundamental ionization mechanism used was discovered by John Sealy Townsend by his work between 1897 and 1901, and is known as the Townsend discharge, which is the ionization of molecules by ion impact. It was not until 1928 that Geiger and Walther Müller ( a PhD student of Geiger ) developed the sealed Geiger - Müller tube which developed the basic ionization principles previously used experimentally. This was relatively small and rugged, and could not only detect alpha and beta radiation such as prior models but also gamma radiation. Now a practical radiation instrument could be produced relatively cheaply, and so the Geiger - Muller counter was born. As the tube output required little electronic processing, a distinct advantage in the thermionic valve era due to minimal valve count and low power consumption, the instrument achieved great popularity as a portable radiation detector. Modern versions of the Geiger counter use the halogen tube invented in 1947 by Sidney H. Liebson. It superseded the earlier Geiger tube because of its much longer life and lower operating voltage, typically 400 - 600 volts. Gallery Use of a `` hot spot '' detector on a long pole to survey waste casks. G-M pancake detector feeding a microcontroller data - logger sending data to a PC via bluetooth. A radioactive rock was placed on top the G-M causing the graph to rise. G-M counters being used as gamma survey monitors, seeking radioactive satellite debris See also Civil Defense geiger counters - Describes hand - held radiation monitors. Note ; both G-M and ion chamber Counting efficiency Dosimeter - Device used by personnel to measure radiation dose they have received Ionization chamber - The simplest of ionising radiation detectors Gaseous ionization detectors - An overview of the main gaseous detector types Geiger -- Müller tube - For a more detailed description of GM tube operation and types Geiger plateau - The correct operating voltage range for a G-M tube Photon counting Radioactive decay - Process by which unstable atoms emit radiation Safecast - Use of G-M counter technology in citizen science Scintillation counter - A gas-less radiation detector sievert The SI unit of effects of low levels of radiation on the human body Becquerel Unit used for measuring counts of ionising actions References ^ Jump up to : ' ' Geiger Muller Tubes ; issue 1 ' ' published by Centronics Ltd, UK. ^ Jump up to : Glenn F Knoll. Radiation Detection and Measurement, third edition 2000. John Wiley and sons, ISBN 0 - 471 - 07338 - 5 Jump up ^ `` G-M detector function and measuring methods ''. Retrieved 2017 - 03 - 07. ^ Jump up to : Selection, use and maintenance of portable monitoring instruments - Ionising Radiation Protection Series No 7, issue 10 / 01. Pub by United Kingdom Health and Safety Executive. ^ Jump up to : Korff, SNTM ( 2012 ) 20 : 271. doi : 10.1007 / s00048 - 012 - 0080 - y Jump up ^ http://www.hse.gov.uk/pubns/irp7.pdf Jump up ^ E. Rutherford and H. Geiger ( 1908 ) `` An electrical method of counting the number of α particles from radioactive substances, '' Proceedings of the Royal Society ( London ), Series A, vol. 81, no. 546, pages 141 -- 161. Jump up ^ John S. Townsend ( 1901 ) `` The conductivity produced in gases by the motion of negatively charged ions, '' Philosophical Magazine, series 6, 1 ( 2 ) : 198 - 227. Jump up ^ See : H. Geiger and W. Müller ( 1928 ), `` Elektronenzählrohr zur Messung schwächster Aktivitäten '' ( Electron counting tube for the measurement of the weakest radioactivities ), Die Naturwissenschaften ( The Sciences ), vol. 16, no. 31, pages 617 -- 618. Geiger, H. and Müller, W. ( 1928 ) `` Das Elektronenzählrohr '' ( The electron counting tube ), Physikalische Zeitschrift, 29 : 839 - 841. Geiger, H. and Müller, W. ( 1929 ) `` Technische Bemerkungen zum Elektronenzählrohr '' ( Technical notes on the electron counting tube ), Physikalische Zeitschrift, 30 : 489 - 493. Geiger, H. and Müller, W. ( 1929 ) `` Demonstration des Elektronenzählrohrs '' ( Demonstration of the electron counting tube ), Physikalische Zeitschrift, 30 : 523 ff. Jump up ^ Liebson, S.H. ( 1947 ). `` The Discharge Mechanism of Self - Quenching Geiger - Mueller Counters ''. Physical Review. 72 ( 7 ) : 602 -- 608. Bibcode : 1947PhRv... 72... 602L. doi : 10.1103 / PhysRev. 72.602. Jump up ^ History of Portable Radiation Detection Instrumentation from the period 1920 -- 60 External links Media related to Geiger counters at Wikimedia Commons How a Geiger counter works. Sensors Acoustic, sound, vibration Geophone Hydrophone Microphone Seismometer Automotive, transportation Air -- fuel ratio meter Blind spot monitor Crankshaft position sensor Curb feeler Defect detector Engine coolant temperature sensor Hall effect sensor MAP sensor Mass flow sensor Omniview technology Oxygen sensor Parking sensors Radar gun Speed sensor Speedometer Throttle position sensor Tire - pressure monitoring system Torque sensor Transmission fluid temperature sensor Turbine speed sensor Variable reluctance sensor Vehicle speed sensor Water sensor Wheel speed sensor Chemical Breathalyzer Carbon dioxide sensor Carbon monoxide detector Catalytic bead sensor Chemical field - effect transistor Electrochemical gas sensor Electrolyte -- insulator -- semiconductor sensor Electronic nose Fluorescent chloride sensors Holographic sensor Hydrocarbon dew point analyzer Hydrogen sensor Hydrogen sulfide sensor Infrared point sensor Ion selective electrode Microwave chemistry sensor Nitrogen oxide sensor Nondispersive infrared sensor Olfactometer Optode Oxygen sensor Pellistor pH glass electrode Potentiometric sensor Redox electrode Smoke detector Zinc oxide nanorod sensor Electric, magnetic, radio Current sensor Electroscope Galvanometer Hall effect sensor Hall probe Magnetic anomaly detector Magnetometer MEMS magnetic field sensor Metal detector Planar Hall sensor Radio direction finder Test light Environment, weather, moisture Actinometer Bedwetting alarm Ceilometer Dew warning Electrochemical gas sensor Fish counter Frequency domain sensor Gas detector Hook gauge evaporimeter Humistor Hygrometer Leaf sensor Psychrometer Pyranometer Pyrgeometer Rain gauge Rain sensor SNOTEL Snow gauge Soil moisture sensor Stream gauge Tide gauge Weather radar Flow, fluid velocity Air flow meter Anemometer Flow sensor Gas meter Mass flow sensor Water metering Ionising radiation, subatomic particles Bubble chamber Cloud chamber Geiger -- Müller tube Geiger counter Ionization chamber Neutron detection Particle detector Proportional counter Scintillation counter Semiconductor detector Scintillator Thermoluminescent dosimeter Wire chamber Navigation instruments Airspeed indicator Machmeter Altimeter Attitude indicator Depth gauge Fluxgate compass Gyroscope Inertial navigation system Inertial reference unit Magnetic compass MHD sensor Ring laser gyroscope Turn coordinator Variometer Vibrating structure gyroscope Yaw - rate sensor Position, angle, displacement Accelerometer Angular rate sensor Auxanometer Capacitive displacement sensor Capacitive sensing Gravimeter Inclinometer Integrated circuit piezoelectric sensor Laser rangefinder Laser surface velocimeter Lidar Linear encoder Linear variable differential transformer Liquid capacitive inclinometers Odometer Photoelectric sensor Piezoelectric accelerometer Position sensor Rotary encoder Rotary variable differential transformer Selsyn Sudden Motion Sensor Tachometer Tilt sensor Ultrasonic thickness gauge Variable reluctance sensor Velocity receiver Optical, light, imaging Charge - coupled device Contact image sensor Electro - optical sensor Flame detector Infrared Kinetic inductance detector LED as light sensor Light - addressable potentiometric sensor Nichols radiometer Optical fiber Photodetector Photodiode Photoelectric sensor Photoionization detector Photomultiplier Photoresistor Photoswitch Phototransistor Phototube Position sensitive device Scintillometer Shack -- Hartmann wavefront sensor Single - photon avalanche diode Superconducting nanowire single - photon detector Transition edge sensor Tristimulus colorimeter Visible - light photon counter Wavefront sensor Pressure Barograph Barometer Boost gauge Bourdon gauge Hot - filament ionization gauge Ionization gauge McLeod gauge Oscillating U-tube Permanent Downhole Gauge Piezometer Pirani gauge Pressure gauge Pressure sensor Tactile sensor Time pressure gauge Force, density, level Bhangmeter Force gauge Hydrometer Level sensor Load cell Magnetic level gauge Nuclear density gauge Piezoelectric sensor Strain gauge Torque sensor Viscometer Thermal, heat, temperature Bimetallic strip Bolometer Calorimeter Exhaust gas temperature gauge Flame detection Gardon gauge Golay cell Heat flux sensor Infrared thermometer Microbolometer Microwave radiometer Net radiometer Quartz thermometer Resistance thermometer Silicon bandgap temperature sensor Special sensor microwave / imager Thermistor Thermocouple Thermometer Proximity, presence Alarm sensor Doppler radar Motion detector Occupancy sensor Passive infrared sensor Proximity sensor Reed switch Stud finder Touch switch Triangulation sensor Wired glove Sensor technology Active pixel sensor Back - illuminated sensor Biochip Biosensor Capacitance probe Carbon paste electrode Catadioptric sensor Digital sensors Displacement receiver Electromechanical film Electro - optical sensor Fabry -- Pérot interferometer Fisheries acoustics Image sensor Image sensor format Inductive sensor Intelligent sensor Lab - on - a-chip Leaf sensor Machine vision Microelectromechanical systems Photoelasticity Quantum sensor Radar Ground - penetrating radar Synthetic aperture radar Radar tracker Sensor array Sensor fusion Sensor grid Sensor node Soft sensor Sonar Staring array Transducer Ultrasonic sensor Video sensor technology Visual sensor network Wheatstone bridge Wireless sensor network Related List of sensors Radiation protection Main articles Dosimetry Health physics Ionizing radiation Internal dosimetry Radioactive contamination Radioactive sources Radiobiology Measurement quantities & units Absorbed dose Becquerel Committed dose Effective dose Equivalent dose Gray Sievert Instruments and measurement techniques Airborne radioactive particulate monitoring Counts per minute Dosimeter Geiger counter Ion chamber Scintillation counter Proportional counter Semiconductor detector Survey meter Whole - body counting Protection techniques Lead apron Glovebox Potassium iodide Radon mitigation Organisations HPS ( USA ) IAEA ICRU ICRP IRPA SRP ( UK ) UNSCEAR Regulation IRR 1999 ( UK ) NRC ( USA ) ONR ( UK ) Radiation Protection Convention, 1960 See also : the categories Medical physics, Radiation effects, Radioactivity, Radiobiology, and Radiation protection. Retrieved from `` https://en.wikipedia.org/w/index.php?title=Geiger_counter&oldid=800308789 '' Categories : Particle detectors Laboratory equipment Counting instruments Ionising radiation detectors 1908 introductions English inventions Radiation protection Talk Contents About Wikipedia Azərbaycanca Башҡортса Беларуская Български Català Čeština Dansk Deutsch Eesti Ελληνικά Español Esperanto Euskara فارسی Français 한국어 हिन्दी Hrvatski Bahasa Indonesia Íslenska Italiano עברית ಕನ್ನಡ Қазақша Kreyòl ayisyen Кыргызча Latviešu Lietuvių Magyar Македонски മലയാളം Nederlands 日本 語 Norsk Norsk nynorsk ଓଡ଼ିଆ پنجابی Piemontèis Polski Português Română Русский Scots Simple English Slovenčina Српски / srpski Srpskohrvatski / српскохрватски Suomi Svenska Tagalog தமிழ் ไทย Türkçe Українська Tiếng Việt Winaray 粵語 中文 Edit links This page was last edited on 12 September 2017, at 18 : 39. Text is available under the Creative Commons Attribution - ShareAlike License ; additional terms may apply. By using this site, you agree to the Terms of Use and Privacy Policy. Wikipedia ® is a registered trademark of the Wikimedia Foundation, Inc., a non-profit organization. About Wikipedia

Question: radiation is detected by an instrument called a
Answer: The Geiger counter is an instrument used for measuring ionizing radiation used widely in applications such as radiation dosimetry, radiological protection, experimental physics and the nuclear industry.