Seismometer - wikipedia Seismometer This article needs additional citations for verification. Please help improve this article by adding citations to reliable sources. Unsourced material may be challenged and removed. ( July 2014 ) Japan Meteorological Agency Optical Electromagnetic Seismometer Part of a series on Earthquakes Types Foreshock Aftershock Blind thrust Doublet Interplate Intraplate Megathrust Remotely triggered Slow Submarine Supershear Tsunami Earthquake swarm Causes Fault movement Volcanism Induced seismicity Characteristics Epicenter Hypocenter Shadow zone Seismic waves P - wave S - wave Measurement Seismometer Seismic magnitude scales Seismic intensity scales Prediction Coordinating Committee for Earthquake Prediction Forecasting Other topics Shear wave splitting Adams -- Williamson equation Flinn -- Engdahl regions Earthquake engineering Seismite Seismology Earth Sciences Portal Related topics A seismometer is an instrument that measures motion of the ground, caused by, for example, an earthquake, a volcanic eruption, or the use of explosives. Records of seismic waves allow seismologists to map the interior of the Earth and to locate and measure the size of events like these. Contents 1 Basic principles 2 Nomenclature 3 History 3.1 Ancient era 3.2 Modern designs 4 Modern instruments 4.1 Teleseismometers 4.2 Strong - motion seismometers 4.3 Other forms 4.4 Interconnected seismometers 5 Recording 6 See also 7 References 8 External links Basic principles A simple seismometer, sensitive to up - down motions of the Earth, is like a weight hanging from a spring, both suspended from a frame that moves along with any motion detected. The relative motion between the weight ( called the mass ) and the frame provides a measurement of the vertical ground motion. A rotating drum is attached to the frame and a pen is attached to the weight, thus recording any ground motion in a seismogram. Any movement of the ground moves the frame. The mass tends not to move because of its inertia, and by measuring the movement between the frame and the mass, the motion of the ground can be determined. Early seismometers used optical levers or mechanical linkages to amplify the small motions involved, recording on soot - covered paper or photographic paper. Modern instruments use electronics. In some systems, the mass is held nearly motionless relative to the frame by an electronic negative feedback loop. The motion of the mass relative to the frame is measured, and the feedback loop applies a magnetic or electrostatic force to keep the mass nearly motionless. The voltage needed to produce this force is the output of the seismometer, which is recorded digitally. In other systems the weight is allowed to move, and its motion produces an electrical charge in a coil attached to the mass which voltage moves through the magnetic field of a magnet attached to the frame. This design is often used in a geophone, which is used in exploration for oil and gas. Seismic observatories usually have instruments measuring three axes : north - south ( y - axis ), east - west ( x-axis ), and vertical ( z - axis ). If only one axis is measured, it is usually the vertical because it is less noisy and gives better records of some seismic waves. The foundation of a seismic station is critical. A professional station is sometimes mounted on bedrock. The best mountings may be in deep boreholes, which avoid thermal effects, ground noise and tilting from weather and tides. Other instruments are often mounted in insulated enclosures on small buried piers of unreinforced concrete. Reinforcing rods and aggregates would distort the pier as the temperature changes. A site is always surveyed for ground noise with a temporary installation before pouring the pier and laying conduit. Originally, European seismographs were placed in a particular area after a destructive earthquake. Today, they are spread to provide appropriate coverage ( in the case of weak - motion seismology ) or concentrated in high - risk regions ( strong - motion seismology ). Nomenclature The word derives from the Greek σεισμός, seismós, a shaking or quake, from the verb σείω, seíō, to shake ; and μέτρον, métron, to measure, and was coined by David Milne - Home in 1841, to describe an instrument designed by Scottish physicist James David Forbes. Seismograph is another Greek term from seismós and γράφω, gráphō, to draw. It is often used to mean seismometer, though it is more applicable to the older instruments in which the measuring and recording of ground motion were combined, than to modern systems, in which these functions are separated. Both types provide a continuous record of ground motion ; this record distinguishes them from seismoscopes, which merely indicate that motion has occurred, perhaps with some simple measure of how large it was. The technical discipline concerning such devices is called seismometry, a branch of seismology. The concept of measuring the `` shaking '' of something means that the word `` seismograph '' might be used in a more general sense. For example, a monitoring station that tracks changes in electromagnetic noise affecting amateur radio waves presents an rf seismograph. And Helioseismology studies the `` quakes '' on the Sun. History The first seismometer was made in China during the 2nd Century. The first Western description of the device comes from the French physicist and priest Jean de Hautefeuille in 1703. The modern seismometer was developed in the 19th century. Ancient era Replica of Zhang Heng 's seismoscope Houfeng Didong Yi See also : List of Chinese inventions In AD 132, Zhang Heng of China 's Han dynasty invented the first seismoscope ( by the definition above ), which was called Houfeng Didong Yi ( translated as, `` instrument for measuring the seasonal winds and the movements of the Earth '' ). The description we have, from the History of the Later Han Dynasty, says that it was a large bronze vessel, about 2 meters in diameter ; at eight points around the top were dragon 's heads holding bronze balls. When there was an earthquake, one of the mouths would open and drop its ball into a bronze toad at the base, making a sound and supposedly showing the direction of the earthquake. On at least one occasion, probably at the time of a large earthquake in Gansu in AD 143, the seismoscope indicated an earthquake even though one was not felt. The available text says that inside the vessel was a central column that could move along eight tracks ; this is thought to refer to a pendulum, though it is not known exactly how this was linked to a mechanism that would open only one dragon 's mouth. The first earthquake recorded by this seismoscope was supposedly `` somewhere in the east. '' Days later, a rider from the east reported this earthquake. Modern designs An early special - purpose seismometer consisted of a large, stationary pendulum, with a stylus on the bottom. As the earth started to move, the heavy mass of the pendulum had the inertia to stay still within the frame. The result is that the stylus scratched a pattern corresponding with the Earth 's movement. This type of strong - motion seismometer recorded upon a smoked glass ( glass with carbon soot ). While not sensitive enough to detect distant earthquakes, this instrument could indicate the direction of the pressure waves and thus help find the epicenter of a local quake. Such instruments were useful in the analysis of the 1906 San Francisco earthquake. Further analysis was performed in the 1980s, using these early recordings, enabling a more precise determination of the initial fault break location in Marin county and its subsequent progression, mostly to the south. Milne horizontal pendulum seismometer. One of the Important Cultural Properties of Japan. Exhibit in the National Museum of Nature and Science, Tokyo, Japan. By the 13th century, seismographic devices existed in the Maragheh observatory in Persia. French physicist and priest Jean de Hautefeuille built one in 1703. After 1880, most seismometers were descended from those developed by the team of John Milne, James Alfred Ewing and Thomas Gray, who worked as foreign - government advisors in Japan from 1880 to 1895. These seismometers used damped horizontal pendulums. After World War II, these were adapted into the widely used Press - Ewing seismometer. Later, professional suites of instruments for the worldwide standard seismographic network had one set of instruments tuned to oscillate at fifteen seconds, and the other at ninety seconds, each set measuring in three directions. Amateurs or observatories with limited means tuned their smaller, less sensitive instruments to ten seconds. The basic damped horizontal pendulum seismometer swings like the gate of a fence. A heavy weight is mounted on the point of a long ( from 10 cm to several meters ) triangle, hinged at its vertical edge. As the ground moves, the weight stays unmoving, swinging the `` gate '' on the hinge. The advantage of a horizontal pendulum is that it achieves very low frequencies of oscillation in a compact instrument. The `` gate '' is slightly tilted, so the weight tends to slowly return to a central position. The pendulum is adjusted ( before the damping is installed ) to oscillate once per three seconds, or once per thirty seconds. The general - purpose instruments of small stations or amateurs usually oscillate once per ten seconds. A pan of oil is placed under the arm, and a small sheet of metal mounted on the underside of the arm drags in the oil to damp oscillations. The level of oil, position on the arm, and angle and size of sheet is adjusted until the damping is `` critical, '' that is, almost having oscillation. The hinge is very low friction, often torsion wires, so the only friction is the internal friction of the wire. Small seismographs with low proof masses are placed in a vacuum to reduce disturbances from air currents. Zollner described torsionally suspended horizontal pendulums as early as 1869, but developed them for gravimetry rather than seismometry. Early seismometers had an arrangement of levers on jeweled bearings, to scratch smoked glass or paper. Later, mirrors reflected a light beam to a direct - recording plate or roll of photographic paper. Briefly, some designs returned to mechanical movements to save money. In mid-twentieth - century systems, the light was reflected to a pair of differential electronic photosensors called a photomultiplier. The voltage generated in the photomultiplier was used to drive galvanometers which had a small mirror mounted on the axis. The moving reflected light beam would strike the surface of the turning drum, which was covered with photo - sensitive paper. The expense of developing photo sensitive paper caused many seismic observatories to switch to ink or thermal - sensitive paper. Modern instruments CMG - 40T triaxial broadband seismometer Modern instruments use electronic sensors, amplifiers, and recording devices. Most are broadband covering a wide range of frequencies. Some seismometers can measure motions with frequencies from 500 Hz to 0.00118 Hz ( 1 / 500 = 0.002 seconds per cycle, to 1 / 0.00118 = 850 seconds per cycle ). The mechanical suspension for horizontal instruments remains the garden-gate described above. Vertical instruments use some kind of constant - force suspension, such as the LaCoste suspension. The LaCoste suspension uses a zero - length spring to provide a long period ( high sensitivity ). Some modern instruments use a `` triaxial '' design, in which three identical motion sensors are set at the same angle to the vertical but 120 degrees apart on the horizontal. Vertical and horizontal motions can be computed from the outputs of the three sensors. Seismometers unavoidably introduce some distortion into the signals they measure, but professionally designed systems have carefully characterized frequency transforms. Modern sensitivities come in three broad ranges : geophones, 50 to 750 V / m ; local geologic seismographs, about 1,500 V / m ; and teleseismographs, used for world survey, about 20,000 V / m. Instruments come in three main varieties : short period, long period and broadband. The short and long period measure velocity and are very sensitive, however they ' clip ' the signal or go off - scale for ground motion that is strong enough to be felt by people. A 24 - bit analog - to - digital conversion channel is commonplace. Practical devices are linear to roughly one part per million. Delivered seismometers come with two styles of output : analog and digital. Analog seismographs require analog recording equipment, possibly including an analog - to - digital converter. The output of a digital seismograph can be simply input to a computer. It presents the data in a standard digital format ( often `` SE2 '' over Ethernet ). Teleseismometers A low - frequency 3 - direction ocean - bottom seismometer ( cover removed ). Two masses for x - and y - direction can be seen, the third one for z - direction is below. This model is a CMG - 40TOBS, manufactured by Güralp Systems Ltd and is part of the Monterey Accelerated Research System. The modern broadband seismograph can record a very broad range of frequencies. It consists of a small `` proof mass '', confined by electrical forces, driven by sophisticated electronics. As the earth moves, the electronics attempt to hold the mass steady through a feedback circuit. The amount of force necessary to achieve this is then recorded. In most designs the electronics holds a mass motionless relative to the frame. This device is called a `` force balance accelerometer ''. It measures acceleration instead of velocity of ground movement. Basically, the distance between the mass and some part of the frame is measured very precisely, by a linear variable differential transformer. Some instruments use a linear variable differential capacitor. That measurement is then amplified by electronic amplifiers attached to parts of an electronic negative feedback loop. One of the amplified currents from the negative feedback loop drives a coil very like a loudspeaker, except that the coil is attached to the mass, and the magnet is mounted on the frame. The result is that the mass stays nearly motionless. Most instruments measure directly the ground motion using the distance sensor. The voltage generated in a sense coil on the mass by the magnet directly measures the instantaneous velocity of the ground. The current to the drive coil provides a sensitive, accurate measurement of the force between the mass and frame, thus measuring directly the ground 's acceleration ( using f = ma where f = force, m = mass, a = acceleration ). One of the continuing problems with sensitive vertical seismographs is the buoyancy of their masses. The uneven changes in pressure caused by wind blowing on an open window can easily change the density of the air in a room enough to cause a vertical seismograph to show spurious signals. Therefore, most professional seismographs are sealed in rigid gas - tight enclosures. For example, this is why a common Streckeisen model has a thick glass base that must be glued to its pier without bubbles in the glue. It might seem logical to make the heavy magnet serve as a mass, but that subjects the seismograph to errors when the Earth 's magnetic field moves. This is also why seismograph 's moving parts are constructed from a material that interacts minimally with magnetic fields. A seismograph is also sensitive to changes in temperature so many instruments are constructed from low expansion materials such as nonmagnetic invar. The hinges on a seismograph are usually patented, and by the time the patent has expired, the design has been improved. The most successful public domain designs use thin foil hinges in a clamp. Another issue is that the transfer function of a seismograph must be accurately characterized, so that its frequency response is known. This is often the crucial difference between professional and amateur instruments. Most instruments are characterized on a variable frequency shaking table. Strong - motion seismometers Another type of seismometer is a digital strong - motion seismometer, or accelerograph. The data from such an instrument is essential to understand how an earthquake affects man - made structures, through earthquake engineering. The recordings of such instruments are crucial for the assessment of seismic hazard, through engineering seismology. A strong - motion seismometer measures acceleration. This can be mathematically integrated later to give velocity and position. Strong - motion seismometers are not as sensitive to ground motions as teleseismic instruments but they stay on scale during the strongest seismic shaking. Strong motion sensors are used for intensity meter applications. Other forms A Kinemetrics seismograph, formerly used by the United States Department of the Interior. Accelerographs and geophones are often heavy cylindrical magnets with a spring - mounted coil inside. As case moves, the coil tends to stay stationary, so the magnetic field cuts the wires, inducing current in the output wires. They receive frequencies from several hundred hertz down to 1 Hz. Some have electronic damping, a low - budget way to get some of the performance of the closed - loop wide - band geologic seismographs. Strain - beam accelerometers constructed as integrated circuits are too insensitive for geologic seismographs ( 2002 ), but are widely used in geophones. Some other sensitive designs measure the current generated by the flow of a non-corrosive ionic fluid through an electret sponge or a conductive fluid through a magnetic field. Interconnected seismometers Seismometers spaced in an array can also be used to precisely locate, in three dimensions, the source of an earthquake, using the time it takes for seismic waves to propagate away from the hypocenter, the initiating point of fault rupture ( See also Earthquake location ). Interconnected seismometers are also used, as part of the International Monitoring System to detect underground nuclear test explosions, as well as for Earthquake early warning systems. These seismometers are often used as part of a large scale governmental or scientific project, but some organizations such as the Quake - Catcher Network, can use residential size detectors built into computers to detect earthquakes as well. In reflection seismology, an array of seismometers image sub-surface features. The data are reduced to images using algorithms similar to tomography. The data reduction methods resemble those of computer - aided tomographic medical imaging X-ray machines ( CAT - scans ), or imaging sonars. A worldwide array of seismometers can actually image the interior of the Earth in wave - speed and transmissivity. This type of system uses events such as earthquakes, impact events or nuclear explosions as wave sources. The first efforts at this method used manual data reduction from paper seismograph charts. Modern digital seismograph records are better adapted to direct computer use. With inexpensive seismometer designs and internet access, amateurs and small institutions have even formed a `` public seismograph network ''. Seismographic systems used for petroleum or other mineral exploration historically used an explosive and a wireline of geophones unrolled behind a truck. Now most short - range systems use `` thumpers '' that hit the ground, and some small commercial systems have such good digital signal processing that a few sledgehammer strikes provide enough signal for short - distance refractive surveys. Exotic cross or two - dimensional arrays of geophones are sometimes used to perform three - dimensional reflective imaging of subsurface features. Basic linear refractive geomapping software ( once a black art ) is available off - the - shelf, running on laptop computers, using strings as small as three geophones. Some systems now come in an 18 '' ( 0.5 m ) plastic field case with a computer, display and printer in the cover. Small seismic imaging systems are now sufficiently inexpensive to be used by civil engineers to survey foundation sites, locate bedrock, and find subsurface water. Recording Viewing of a Develocorder film Matsushiro Seismological Observatory Further information : Seismogram Today, the most common recorder is a computer with an analog - to - digital converter, a disk drive and an internet connection ; for amateurs, a PC with a sound card and associated software is adequate. Most systems record continuously, but some record only when a signal is detected, as shown by a short - term increase in the variation of the signal, compared to its long - term average ( which can vary slowly because of changes in seismic noise ), also known as a STA / LTA trigger. Prior to the availability of digital processing of seismic data in the late 1970s, the records were done in a few different forms on different types of media. A `` Helicorder '' drum was a device used to record data into photographic paper or in the form of paper and ink. A `` Develocorder '' was a machine that record data from up to 20 channels into a 16 - mm film. The recorded film can be viewed by a machine. The reading and measuring from these types of media can be done by hand. After the digital processing has been used, the archives of the seismic data were recorded in magnetic tapes. Due to the deterioration of older magnetic tape medias, large number of waveforms from the archives are not recoverable. See also Accelerometer Galitzine, Boris Borisovich Geophone IRIS Consortium Lehmann, Inge Milne, John Oldham, Richard Dixon Pacific Northwest Seismic Network Plate tectonics Quake - Catcher Network Richter magnitude scale Seismogram References Jump up ^ `` Blasting Seismographs ''. The World of Explosives. Retrieved 28 March 2018. Jump up ^ Erhard Wielandt 's ' Seismic Sensors and their Calibration ' Archived 2010 - 09 - 24 at the Wayback Machine. - Current ( 2002 ) reference by a widely consulted expert. ^ Jump up to : Reitherman, Robert ( 2012 ). Earthquakes and Engineers : an International History. Reston, VA : ASCE Press. pp. 122 -- 125. ISBN 9780784410714. Archived from the original on 2012 - 07 - 26. Jump up ^ Ben - Menahem, A. ( 2009 ). Historical Encyclopedia of Natural and Mathematical Sciences, Volume 1. Springer. p. 2657. ISBN 9783540688310. Retrieved 28 August 2012. Jump up ^ Richter, C.F. ( 1958 ). Elementary Seismology. San Francisco : W.H. Freeman. Jump up ^ William H.K. Lee ; Paul Jennings ; Carl Kisslinger ; Hiroo Kanamori ( 27 September 2002 ). International Handbook of Earthquake & Engineering Seismology. Academic Press. pp. 283 --. ISBN 978 - 0 - 08 - 048922 - 3. Retrieved 29 April 2013. Jump up ^ `` The RF Seismograph ''. www.nsarc.ca. Retrieved 28 March 2018. Jump up ^ `` The Singing Sun ''. solar-center.stanford.edu. Retrieved 28 March 2018. ^ Jump up to : Sleeswyk AW, Sivin N ( 1983 ). `` Dragons and toads : the Chinese seismoscope of BC. 132 ''. Chinese Science. 6 : 1 -- 19. ^ Jump up to : Joseph Needham ( 1985 ). Science and Civilisation in China : Paper and Printing. Cambridge University Press. p. 122. ISBN 978 - 0 - 521 - 08690 - 5. Retrieved 16 April 2013. In the Southern Sung dynasty, gift money for bestowing upon officials by the imperial court was wrapped in paper envelopes ( chih pao ) Jump up ^ Needham, Joseph ( 1959 ). Science and Civilization in China, Volume 3 : Mathematics and the Sciences of the Heavens and the Earth. Cambridge : Cambridge University Press. pp. 626 -- 635. Jump up ^ `` Physics of the Zero - Length Spring of Geoscience ''. physics.mercer.edu. Retrieved 28 March 2018. Jump up ^ A Biography of Lucien LaCoste, inventor of the zero - length spring Archived 2007 - 03 - 20 at the Wayback Machine. Jump up ^ `` Redwood City Public Seismic Network ''. psn.quake.net. Retrieved 28 March 2018. Jump up ^ Hutton, Kate ; Yu, Ellen. `` NEWS FLASH ! ! SCSN Earthquake Catalog Completed ! ! '' ( PDF ). Seismological Laboratory, Caltech. Retrieved 4 July 2014. Jump up ^ Fogleman, Kent A. ; Lahr, John C. ; Stephens, Christopher D. ; Page, Robert A. ( June 1993 ). Earthquake Locations Determined by the Southern Alaska Seismograph Network for October 1971 through May 1989 ( Report ). USGS. External links Wikimedia Commons has media related to Seismometers. The history of early seismometers The Lehman amateur seismograph, from Scientific American - not designed for calibrated measurement. Sean Morrisey 's professional design of an amateur teleseismograph Also see Keith Payea 's version Both accessed 2010 - 9 - 29 Morrissey was a professional seismographic instrument engineer. This superior design uses a zero - length spring to achieve a 60 - second period, active feedback and a uniquely convenient variable reluctance differential transducer, with parts scavenged from a hardware store. The frequency transform is carefully designed, unlike most amateur instruments. Morrisey is deceased, but the site remains up as a public service. SeisMac is a free tool for recent Macintosh laptop computers that implements a real - time three - axis seismograph. The Development Of Very - Broad - Band Seismography : Quanterra And The Iris Collaboration discusses the history of development of the primary technology in global earthquake research. Video of seismograph at Hawaiian Volcano Observatory -- on Flickr -- retrieved on 2009 - 06 - 15. Seismoscope -- Research References 2012 Iris EDU -- How Does A Seismometer Work ? Sensors Acoustic, sound, vibration Geophone Hydrophone Microphone Seismometer Automotive, transportation Air -- fuel ratio meter Blind spot monitor Crankshaft position sensor Curb feeler Defect detector Engine coolant temperature sensor Hall effect sensor MAP sensor Mass flow sensor Omniview technology Oxygen sensor Parking sensors Radar gun Speed sensor Speedometer Throttle position sensor Tire - pressure monitoring system Torque sensor Transmission fluid temperature sensor Turbine speed sensor Variable reluctance sensor Vehicle speed sensor Water sensor Wheel speed sensor Chemical Breathalyzer Carbon dioxide sensor Carbon monoxide detector Catalytic bead sensor Chemical field - effect transistor Electrochemical gas sensor Electrolyte -- insulator -- semiconductor sensor Electronic nose Fluorescent chloride sensors Holographic sensor Hydrocarbon dew point analyzer Hydrogen sensor Hydrogen sulfide sensor Infrared point sensor Ion selective electrode Microwave chemistry sensor Nitrogen oxide sensor Nondispersive infrared sensor Olfactometer Optode Oxygen sensor Pellistor pH glass electrode Potentiometric sensor Redox electrode Smoke detector Zinc oxide nanorod sensor Electric, magnetic, radio Current sensor Electroscope Galvanometer Hall effect sensor Hall probe Magnetic anomaly detector Magnetometer MEMS magnetic field sensor Metal detector Planar Hall sensor Radio direction finder Test light Environment, weather, moisture Actinometer Bedwetting alarm Ceilometer Dew warning Electrochemical gas sensor Fish counter Frequency domain sensor Gas detector Hook gauge evaporimeter Humistor Hygrometer Leaf sensor Psychrometer Pyranometer Pyrgeometer Rain gauge Rain sensor SNOTEL Snow gauge Soil moisture sensor Stream gauge Tide gauge Weather radar Flow, fluid velocity Air flow meter Anemometer Flow sensor Gas meter Mass flow sensor Water metering Ionising radiation, subatomic particles Bubble chamber Cloud chamber Geiger -- Müller tube Geiger counter Ionization chamber Neutron detection Particle detector Proportional counter Scintillation counter Semiconductor detector Scintillator Thermoluminescent dosimeter Wire chamber Navigation instruments Airspeed indicator Machmeter Altimeter Attitude indicator Depth gauge Fluxgate compass Gyroscope Inertial navigation system Inertial reference unit Magnetic compass MHD sensor Ring laser gyroscope Turn coordinator Variometer Vibrating structure gyroscope Yaw - rate sensor Position, angle, displacement Accelerometer Angular rate sensor Auxanometer Capacitive displacement sensor Capacitive sensing Gravimeter Inclinometer Integrated circuit piezoelectric sensor Laser rangefinder Laser surface velocimeter Lidar Linear encoder Linear variable differential transformer Liquid capacitive inclinometers Odometer Photoelectric sensor Piezoelectric accelerometer Position sensor Rotary encoder Rotary variable differential transformer Selsyn Sudden Motion Sensor Tachometer Tilt sensor Ultrasonic thickness gauge Variable reluctance sensor Velocity receiver Optical, light, imaging Active pixel sensor Angle -- sensitive pixel Back - illuminated sensor Charge - coupled device Contact image sensor Electro - optical sensor Flame detector Infrared Kinetic inductance detector LED as light sensor Light - addressable potentiometric sensor Nichols radiometer Optical fiber Photodetector Photodiode Photoelectric sensor Photoionization detector Photomultiplier Photoresistor Photoswitch Phototransistor Phototube Position sensitive device Scintillometer Shack -- Hartmann wavefront sensor Single - photon avalanche diode Superconducting nanowire single - photon detector Transition edge sensor Tristimulus colorimeter Visible - light photon counter Wavefront sensor Pressure Barograph Barometer Boost gauge Bourdon gauge Hot - filament ionization gauge Ionization gauge McLeod gauge Oscillating U-tube Permanent Downhole Gauge Piezometer Pirani gauge Pressure gauge Pressure sensor Tactile sensor Time pressure gauge Force, density, level Bhangmeter Force gauge Hydrometer Level sensor Load cell Magnetic level gauge Nuclear density gauge Piezoelectric sensor Strain gauge Torque sensor Viscometer Thermal, heat, temperature Bimetallic strip Bolometer Calorimeter Exhaust gas temperature gauge Flame detection Gardon gauge Golay cell Heat flux sensor Infrared thermometer Microbolometer Microwave radiometer Net radiometer Quartz thermometer Resistance thermometer Silicon bandgap temperature sensor Special sensor microwave / imager Thermistor Thermocouple Thermometer Proximity, presence Alarm sensor Doppler radar Motion detector Occupancy sensor Passive infrared sensor Proximity sensor Reed switch Stud finder Touch switch Triangulation sensor Wired glove Sensor technology Active pixel sensor Back - illuminated sensor Biochip Biosensor Capacitance probe Carbon paste electrode Catadioptric sensor Digital sensors Displacement receiver Electromechanical film Electro - optical sensor Fabry -- Pérot interferometer Fisheries acoustics Image sensor Image sensor format Inductive sensor Intelligent sensor Lab - on - a-chip Leaf sensor Machine vision Microelectromechanical systems Photoelasticity Quantum sensor Radar Ground - penetrating radar Synthetic aperture radar Radar tracker Sensor array Sensor fusion Sensor grid Sensor node Soft sensor Sonar Staring array Transducer Ultrasonic sensor Video sensor technology Visual sensor network Wheatstone bridge Wireless sensor network Related List of sensors GND : 4180766 - 2 NDL : 00574864 Retrieved from `` https://en.wikipedia.org/w/index.php?title=Seismometer&oldid=832821092 '' Categories : 1880 introductions Chinese inventions Measuring instruments Seismology measurement Hidden categories : Webarchive template wayback links Articles needing additional references from July 2014 All articles needing additional references All articles with unsourced statements Articles with unsourced statements from November 2017 Articles with unsourced statements from July 2014 Wikipedia articles with GND identifiers Talk Contents About Wikipedia Azərbaycanca বাংলা Български Brezhoneg Català Čeština Dansk Deutsch Eesti Español Euskara فارسی Français Gaeilge Galego 한국어 Հայերեն हिन्दी Hrvatski Ido Bahasa Indonesia Íslenska Italiano עברית ქართული Kreyòl ayisyen Кыргызча Latviešu Lietuvių Limburgs Magyar Македонски മലയാളം მარგალური Bahasa Melayu မြန်မာဘာသာ Nederlands नेपाली 日本 語 Norsk Norsk nynorsk ଓଡ଼ିଆ Polski Português Română Русский Simple English Slovenčina Српски / srpski Srpskohrvatski / српскохрватски Basa Sunda Suomi Svenska Tagalog தமிழ் తెలుగు Türkçe Українська اردو Tiếng Việt 中文 53 more Edit links This page was last edited on 28 March 2018, at 05 : 06. Text is available under the Creative Commons Attribution - ShareAlike License ; additional terms may apply. By using this site, you agree to the Terms of Use and Privacy Policy. Wikipedia ® is a registered trademark of the Wikimedia Foundation, Inc., a non-profit organization. About Wikipedia

Question: who developed the first procedure used to measure the local size of an earthquake
Answer: Zhang Heng of China's Han dynasty