Electroscope - wikipedia Electroscope For the 19th century conceptual model of television, see Telectroscope. Gilbert 's versorium. An electroscope is an early scientific instrument that is used to detect the presence and magnitude of electric charge on a body. It was the first electrical measuring instrument. The first electroscope, a pivoted needle called the versorium, was invented by British physician William Gilbert around 1600. The pith - ball electroscope and the gold - leaf electroscope are two classical types of electroscope that are still used in physics education to demonstrate the principles of electrostatics. A type of electroscope is also used in the quartz fiber radiation dosimeter. Electroscopes were used by the Austrian scientist Victor Hess in the discovery of cosmic rays. Electroscopes detect electric charge by the motion of a test object due to the Coulomb electrostatic force. Since the electric potential or voltage of an object with respect to ground equals its charge divided by its capacitance to ground, an electroscope can be regarded as a crude voltmeter. However, the accumulation of enough charge to detect with an electroscope requires hundreds or thousands of volts, so electroscopes are only used with high - voltage sources such as static electricity and electrostatic machines. Electroscopes generally give only a rough, qualitative indication of the magnitude of the charge. An instrument that measures charge quantitatively is called an electrometer. Pith ball electroscope from the 1870s, showing attraction to charged object How it works Contents 1 Pith - ball electroscope 2 Gold - leaf electroscope 3 See also 4 Footnotes 5 Bibliography 6 External links Pith - ball electroscope The pith - ball electroscope, invented by British schoolmaster and physicist John Canton in 1754, consists of a small ball of lightweight nonconductive substance, originally a spongy plant material called pith, although modern electroscopes frequently use plastic balls. The ball is suspended by a silk thread from the hook of an insulated stand. In order to test the presence of a charge on an object, the object is brought near to the uncharged pith ball. If the object is charged, the ball will be attracted to it and move toward it. The attraction occurs because of induced polarization of the atoms inside the pith ball. The pith is a nonconductor, so the electrons in the ball are bound to atoms of the pith and are not free to leave the atoms and move about in the ball, but they can move a little within the atoms. See diagram at right. If, for example, a positively charged object ( B ) is brought near the pith ball ( A ), the negative electrons ( blue minus signs ) in each atom ( yellow ovals ) will be attracted and move slightly toward the side of the atom nearer the object. The positively charged nuclei ( red plus signs ) will be repelled and will move slightly away. Since the negative charges in the pith ball are now nearer the object than the positive charges ( C ), their attraction is greater than the repulsion of the positive charges, resulting in a net attractive force. This separation of charge is microscopic, but since there are so many atoms, the tiny forces add up to a large enough force to move a light pith ball. The pith ball can be charged by touching it to a charged object, so some of the charges on the surface of the charged object move to the surface of the ball. Then the ball can be used to distinguish the polarity of charge on other objects because it will be repelled by objects charged with the same polarity or sign it has, but attracted to charges of the opposite polarity. Often the electroscope will have a pair of suspended pith balls. This allows one to tell at a glance whether the pith balls are charged. If one of the pith balls is touched to a charged object, charging it, the second one will be attracted and touch it, communicating some of the charge to the surface of the second ball. Now both balls have the same polarity charge, so they repel each other. They hang in an inverted ' V ' shape with the balls spread apart. The distance between the balls will give a rough idea of the magnitude of the charge. Gold - leaf electroscope Gold leaf electroscope showing electrostatic induction Play media Using an electroscope to show electrostatic induction. The device has leaves / needle the become charged when introducing a charged rod to it. The leaves bend the leave / needle, and the stronger the static introduced, the more bending occurs. The gold - leaf electroscope was developed in 1787 by British clergyman and physicist Abraham Bennet, as a more sensitive instrument than pith ball or straw blade electroscopes then in use. It consists of a vertical metal rod, usually brass, from the end of which hang two parallel strips of thin flexible gold leaf. A disk or ball terminal is attached to the top of the rod, where the charge to be tested is applied. To protect the gold leaves from drafts of air they are enclosed in a glass bottle, usually open at the bottom and mounted over a conductive base. Often there are grounded metal plates or foil strips in the bottle flanking the gold leaves on either side. These are a safety measure ; if an excessive charge is applied to the delicate gold leaves, they will touch the grounding plates and discharge before tearing. They also capture charge leaking through the air that accumulate on the glass walls, and that increase the sensitivity of the instrument. In the precision instruments the inside of the bottle was occasionally evacuated, to prevent the charge on the terminal from leaking off through the ionization of the air. When the metal terminal is touched with a charged object, the gold leaves spread apart in a ' V '. This is because some of the charge on the object is conducted through the terminal and metal rod to the leaves. Since they receive the same sign charge they repel each other and thus diverge. If the terminal is grounded by touching it with a finger, the charge is transferred through the human body into the earth and the gold leaves close together. The electroscope can also be charged without touching it to a charged object, by electrostatic induction. If a charged object is brought near the electroscope terminal, the leaves also diverge, because the electric field of the object causes charges in the electroscope rod to separate the leaves. Charges of the opposite polarity to the charged object are attracted to the terminal, while charges with the same polarity are repelled to the leaves, causing them to spread. If the electroscope terminal is grounded while the charged object is nearby, by touching it momentarily with a finger, the same polarity charges in the leaves drain away to ground, leaving the electroscope with a net charge of opposite polarity to the object. The leaves close because the charge is all concentrated at the terminal end. When the charged object is moved away, the charge at the terminal spreads into the leaves, causing them to spread apart again. Gold - leaf electroscopes `` Condensing '' electroscope, Rome University physics dept. Electroscope from about 1910 with grounding electrodes inside jar, as described above Kolbe electrometer, precision form of gold - leaf instrument. This has a light pivoted aluminum vane hanging next to a vertical metal plate. When charged the vane is repelled by the plate and hangs at an angle. Homemade electroscope, 1900 See also Electrical measurements Electron Electrometer Radiation Faraday cup electrometer Footnotes Jump up ^ Gilbert, William ; Edward Wright ( 1893 ). On the Lodestone and Magnetic Bodies. John Wiley & Sons. p. 79. a translation by P. Fleury Mottelay of William Gilbert ( 1600 ) Die Magnete, London ^ Jump up to : Derry, Thomas K. ; Williams, Trevor ( 1993 ). A Short History of Technology from Earliest Times to A.D. 1900. Courier Dover. ISBN 0 - 486 - 27472 - 1. p. 609 Jump up ^ Elliott ( 1999 ) Jump up ^ Sherwood, Bruce A. ; Ruth W. Chabay ( 2011 ). Matter and Interactions, 3rd Ed. USA : John Wiley and Sons. pp. 594 -- 596. ISBN 0 - 470 - 50347 - 5. ^ Jump up to : Kaplan MCAT Physics 2010 - 2011. USA : Kaplan Publishing. 2009. p. 329. ISBN 1 - 4277 - 9875 - 3. Archived from the original on 2014 - 01 - 31. Jump up ^ Paul E. Tippens, Electric Charge and Electric Force, Powerpoint presentation, p. 27 - 28, 2009, S. Polytechnic State Univ. Archived April 19, 2012, at the Wayback Machine. on DocStoc.com website Jump up ^ Henderson, Tom ( 2011 ). `` Charge and Charge Interactions ''. Static Electricity, Lesson 1. The Physics Classroom. Retrieved 2012 - 01 - 01. Jump up ^ Winn, Will Winn ( 2010 ). Introduction to Understandable Physics Vol. 3 : Electricity, Magnetism and Ligh. USA : Author House. p. 20.4. ISBN 1 - 4520 - 1590 - 2. ^ Jump up to : ( Anon. ) ( 2001 ) Bibliography `` Fleming, J.A. ''. Encyclopædia Britannica. 9 ( 11th ed. ). The Encyclopædia Britannica Co. 1910. pp. 239 -- 240. Elliott, P. ( 1999 ). `` Abraham Bennet F.R.S. ( 1749 -- 1799 ) : a provincial electrician in eighteenth - century England '' ( PDF ). Notes and Records of the Royal Society of London. 53 ( 1 ) : 59 -- 78. doi : 10.1098 / rsnr. 1999.0063. External links `` Pith - ball electroscope ''. Physics demonstration resource. St. Mary 's University. Retrieved 2015 - 05 - 28. `` Computer simulation of electroscopes ''. Molecular Workbench. Concord Consortium. `` Pith Ball and Charged Rod Video ''. St. Mary 's Physics YouTube Channel. St. Mary 's Physics Online. `` Electroscope or برق نما ''. oloom blogfa. nedaye oloom. Archived from the original on 2014 - 10 - 03. Sensors Acoustic, sound, vibration Geophone Hydrophone Microphone Seismometer Automotive, transportation Air -- fuel ratio meter Blind spot monitor Crankshaft position sensor Curb feeler Defect detector Engine coolant temperature sensor Hall effect sensor MAP sensor Mass flow sensor Omniview technology Oxygen sensor Parking sensors Radar gun Speed sensor Speedometer Throttle position sensor Tire - pressure monitoring system Torque sensor Transmission fluid temperature sensor Turbine speed sensor Variable reluctance sensor Vehicle speed sensor Water sensor Wheel speed sensor Chemical Breathalyzer Carbon dioxide sensor Carbon monoxide detector Catalytic bead sensor Chemical field - effect transistor Electrochemical gas sensor Electrolyte -- insulator -- semiconductor sensor Electronic nose Fluorescent chloride sensors Holographic sensor Hydrocarbon dew point analyzer Hydrogen sensor Hydrogen sulfide sensor Infrared point sensor Ion selective electrode Microwave chemistry sensor Nitrogen oxide sensor Nondispersive infrared sensor Olfactometer Optode Oxygen sensor Pellistor pH glass electrode Potentiometric sensor Redox electrode Smoke detector Zinc oxide nanorod sensor Electric, magnetic, radio Current sensor Electroscope Galvanometer Hall effect sensor Hall probe Magnetic anomaly detector Magnetometer MEMS magnetic field sensor Metal detector Planar Hall sensor Radio direction finder Test light Environment, weather, moisture Actinometer Bedwetting alarm Ceilometer Dew warning Electrochemical gas sensor Fish counter Frequency domain sensor Gas detector Hook gauge evaporimeter Humistor Hygrometer Leaf sensor Psychrometer Pyranometer Pyrgeometer Rain gauge Rain sensor SNOTEL Snow gauge Soil moisture sensor Stream gauge Tide gauge Weather radar Flow, fluid velocity Air flow meter Anemometer Flow sensor Gas meter Mass flow sensor Water metering Ionising radiation, subatomic particles Bubble chamber Cloud chamber Geiger -- Müller tube Geiger counter Ionization chamber Neutron detection Particle detector Proportional counter Scintillation counter Semiconductor detector Scintillator Thermoluminescent dosimeter Wire chamber Navigation instruments Airspeed indicator Machmeter Altimeter Attitude indicator Depth gauge Fluxgate compass Gyroscope Inertial navigation system Inertial reference unit Magnetic compass MHD sensor Ring laser gyroscope Turn coordinator Variometer Vibrating structure gyroscope Yaw - rate sensor Position, angle, displacement Accelerometer Angular rate sensor Auxanometer Capacitive displacement sensor Capacitive sensing Gravimeter Inclinometer Integrated circuit piezoelectric sensor Laser rangefinder Laser surface velocimeter Lidar Linear encoder Linear variable differential transformer Liquid capacitive inclinometers Odometer Photoelectric sensor Piezoelectric accelerometer Position sensor Rotary encoder Rotary variable differential transformer Selsyn Sudden Motion Sensor Tachometer Tilt sensor Ultrasonic thickness gauge Variable reluctance sensor Velocity receiver Optical, light, imaging Charge - coupled device Contact image sensor Electro - optical sensor Flame detector Infrared Kinetic inductance detector LED as light sensor Light - addressable potentiometric sensor Nichols radiometer Optical fiber Photodetector Photodiode Photoelectric sensor Photoionization detector Photomultiplier Photoresistor Photoswitch Phototransistor Phototube Position sensitive device Scintillometer Shack -- Hartmann wavefront sensor Single - photon avalanche diode Superconducting nanowire single - photon detector Transition edge sensor Tristimulus colorimeter Visible - light photon counter Wavefront sensor Pressure Barograph Barometer Boost gauge Bourdon gauge Hot - filament ionization gauge Ionization gauge McLeod gauge Oscillating U-tube Permanent Downhole Gauge Piezometer Pirani gauge Pressure gauge Pressure sensor Tactile sensor Time pressure gauge Force, density, level Bhangmeter Force gauge Hydrometer Level sensor Load cell Magnetic level gauge Nuclear density gauge Piezoelectric sensor Strain gauge Torque sensor Viscometer Thermal, heat, temperature Bimetallic strip Bolometer Calorimeter Exhaust gas temperature gauge Flame detection Gardon gauge Golay cell Heat flux sensor Infrared thermometer Microbolometer Microwave radiometer Net radiometer Quartz thermometer Resistance thermometer Silicon bandgap temperature sensor Special sensor microwave / imager Thermistor Thermocouple Thermometer Proximity, presence Alarm sensor Doppler radar Motion detector Occupancy sensor Passive infrared sensor Proximity sensor Reed switch Stud finder Touch switch Triangulation sensor Wired glove Sensor technology Active pixel sensor Back - illuminated sensor Biochip Biosensor Capacitance probe Carbon paste electrode Catadioptric sensor Digital sensors Displacement receiver Electromechanical film Electro - optical sensor Fabry -- Pérot interferometer Fisheries acoustics Image sensor Image sensor format Inductive sensor Intelligent sensor Lab - on - a-chip Leaf sensor Machine vision Microelectromechanical systems Photoelasticity Quantum sensor Radar Ground - penetrating radar Synthetic aperture radar Radar tracker Sensor array Sensor fusion Sensor grid Sensor node Soft sensor Sonar Staring array Transducer Ultrasonic sensor Video sensor technology Visual sensor network Wheatstone bridge Wireless sensor network Related List of sensors Retrieved from `` https://en.wikipedia.org/w/index.php?title=Electroscope&oldid=813352572 '' Categories : Electrostatics Measuring instruments Historical scientific instruments Hidden categories : Webarchive template wayback links All articles with dead external links Articles with dead external links from December 2016 Articles with permanently dead external links Talk Contents About Wikipedia Беларуская Беларуская ( тарашкевіца ) ‎ Català Čeština Deutsch Eesti Español Euskara فارسی Gaeilge हिन्दी Hrvatski Ido Italiano עברית Қазақша Lietuvių Македонски മലയാളം Bahasa Melayu Nederlands 日本 語 Norsk Norsk nynorsk Oʻzbekcha / ўзбекча Polski Português Русский Slovenščina Српски / srpski Srpskohrvatski / српскохрватски Suomi Svenska தமிழ் Татарча / tatarça Türkçe Українська اردو 中文 Edit links This page was last edited on 3 December 2017, at 06 : 32. Text is available under the Creative Commons Attribution - ShareAlike License ; additional terms may apply. By using this site, you agree to the Terms of Use and Privacy Policy. Wikipedia ® is a registered trademark of the Wikimedia Foundation, Inc., a non-profit organization. About Wikipedia

Question: which property is used ti identify the electric charge on the object
Answer: Coulomb electrostatic force