ICSMA-21-012-01 SOOIL Dana Diabecare RS Products
A hard-coded physician PIN in the physician menu of the insulin pump allows attackers with physical access to change insulin therapy settings.CVE-2020-27256 has been assigned to this vulnerability. A CVSS v3 base score of 4.6 has been calculated; the CVSS vector string is (AV:P/AC:L/PR:N/UI:N/S:U/C:N/I:H/A:N). An information disclosure vulnerability in the communication protocol of the insulin pump and its AnyDana-i and AnyDana-A mobile applications allows unauthenticated attackers to extract the pump 's keypad lock PIN via Bluetooth Low Energy.CVE-2020-27258 has been assigned to this vulnerability. A CVSS v3 base score of 6.5 has been calculated; the CVSS vector string is (AV:A/AC:L/PR:N/UI:N/S:U/C:H/I:N/A:N). The communication protocol of the insulin pump and its AnyDana-i and AnyDana-A mobile applications use deterministic keys, which allows unauthenticated, physically proximate attackers to brute-force the keys via Bluetooth Low Energy.CVE-2020-27264 has been assigned to this vulnerability. A CVSS v3 base score of 7.6 has been calculated; the CVSS vector string is (AV:A/AC:L/PR:N/UI:N/S:U/C:H/I:L/A:L). A client-side control vulnerability in the insulin pump and its AnyDana-i and AnyDana-A mobile applications allows physically proximate attackers to bypass user authentication checks via Bluetooth Low Energy.CVE-2020-27266 has been assigned to this vulnerability. A CVSS v3 base score of 6.5 has been calculated; the CVSS vector string is (AV:A/AC:L/PR:N/UI:N/S:U/C:H/I:N/A:N). A client-side control vulnerability in the insulin pump and its AnyDana-i and AnyDana-A mobile applications allows physically proximate attackers to bypass checks for default PINs via Bluetooth Low Energy.CVE-2020-27268 has been assigned to this vulnerability. A CVSS v3 base score of 6.5 has been calculated; the CVSS vector string is (AV:A/AC:L/PR:N/UI:N/S:U/C:H/I:N/A:N). The communication protocol of the insulin pump and its AnyDana-i and AnyDana-A mobile applications lacks replay protection measures, which allows unauthenticated, physically proximate attackers to replay communication sequences via Bluetooth Low Energy.CVE-2020-27269 has been assigned to this vulnerability. A CVSS v3 base score of 5.4 has been calculated; the CVSS vector string is (AV:A/AC:H/PR:N/UI:R/S:U/C:L/I:H/A:N). The communication protocol of the insulin pump and its AnyDana-i and AnyDana-A mobile applications does not use adequate measures to protect encryption keys in transit, which allows unauthenticated, physically proximate attackers to sniff the keys via Bluetooth Low Energy.CVE-2020-27270 has been assigned to this vulnerability. A CVSS v3 base score of 5.7 has been calculated; the CVSS vector string is (AV:A/AC:L/PR:N/UI:R/S:U/C:H/I:N/A:N). The communication protocol of the insulin pump and its AnyDana-i and AnyDana-A mobile applications does not use adequate measures to authenticate the pump before exchanging keys, which allows unauthenticated, physically proximate attackers to eavesdrop the keys and spoof the pump via Bluetooth Low Energy.CVE-2020-27272 has been assigned to this vulnerability. A CVSS v3 base score of 5.7 has been calculated; the CVSS vector string is (AV:A/AC:L/PR:N/UI:R/S:U/C:H/I:N/A:N). The communication protocol of the insulin pump and its AnyDana-i and AnyDana-A mobile applications does not use adequate measures to authenticate the communicating entities before exchanging keys, which allows unauthenticated, physically proximate attackers to eavesdrop the authentication sequence via Bluetooth Low Energy.CVE-2020-27276 has been assigned to this vulnerability. A CVSS v3 base score of 5.7 has been calculated; the CVSS vector string is (AV:A/AC:L/PR:N/UI:R/S:U/C:H/I:N/A:N).
CVSS v3.0 score: 5.7