Filename-Path :	C:\IV scans\P297\20220900204646.niv



%SENSOR IV TEST
SENSOR SERIAL NUMBER :	20220900204646
MODULE SERIAL NUMBER :	P297
HYBRID SERIAL NUMBER :	N/A
TEST DATE :	11/26/2003	9:27:38 AM
TEMPERATURE :	20 C
HUMIDITY :	34 %

Volts   	I(Microamps)	Wait time (sec)
0.000000	0.000902
10.000000	0.082901
20.000000	0.113839
30.000000	0.136127
40.000000	0.131694
50.000000	0.157222
60.000000	0.152777
70.000000	0.182002
80.000000	0.180428
90.000000	0.197232
100.000000	0.209119
125.000000	0.229020
150.000000	0.248900
175.000000	0.278550
200.000000	0.301380
225.000000	0.309570
250.000000	0.335520
275.000000	0.343430
300.000000	0.366520
325.000000	0.375890
350.000000	0.400410
375.000000	0.404020
400.000000	0.432020
425.000000	0.445460
450.000000	0.458290
475.000000	0.732010
500.000000	2.766200


Filename-Path :	C:\IV scans\P297\20220900204646.niv



%SENSOR IV TEST
SENSOR SERIAL NUMBER :	20220900204646
MODULE SERIAL NUMBER :	P297
HYBRID SERIAL NUMBER :	N/A
TEST DATE :	11/26/2003	9:44:36 AM
TEMPERATURE :	20 C
HUMIDITY :	34 %

Volts   	I(Microamps)	Wait time (sec)
0.000000	0.003021
10.000000	9.601700
20.000000	12.937800
30.000000	15.153700
40.000000	16.388901
50.000000	17.473801
60.000000	19.098101
70.000000	19.271200
80.000000	20.296000
90.000000	21.566000


Filename-Path :	C:\IV scans\P297\20220900204646.niv



%SENSOR IV TEST
SENSOR SERIAL NUMBER :	20220900204646
MODULE SERIAL NUMBER :	P297
HYBRID SERIAL NUMBER :	N/A
TEST DATE :	11/26/2003	9:46:19 AM
TEMPERATURE :	20 C
HUMIDITY :	34 %

Volts   	I(Microamps)	Wait time (sec)
0.000000	0.002973
10.000000	11.141400
20.000000	13.961700
30.000000	15.563200
40.000000	18.220200
50.000000	19.328600
60.000000	20.535400
70.000000	21.591999
80.000000	21.589001
90.000000	21.603001
100.000000	21.660000
125.000000	21.598000
150.000000	21.653000
175.000000	21.642000
200.000000	21.632000


Filename-Path :	C:\IV scans\P297\20220900204646.niv



%SENSOR IV TEST
SENSOR SERIAL NUMBER :	20220900204646
MODULE SERIAL NUMBER :	P297
HYBRID SERIAL NUMBER :	N/A
TEST DATE :	11/26/2003	11:27:46 AM
TEMPERATURE :	20 C
HUMIDITY :	34 %

Volts   	I(Microamps)	Wait time (sec)
0.000000	-0.000063
10.000000	0.015758
20.000000	0.025250
30.000000	0.028335
40.000000	0.031160
50.000000	0.033201
60.000000	0.036265
70.000000	0.039694
80.000000	0.045813
90.000000	0.053998
100.000000	0.060061
125.000000	0.069222
150.000000	0.077384
175.000000	0.085585
200.000000	0.092121
225.000000	0.100192
250.000000	0.106754
275.000000	0.114901
300.000000	0.122466
325.000000	0.129576
350.000000	0.138349
375.000000	0.147620
400.000000	0.156834
425.000000	0.165841
450.000000	0.175278
475.000000	0.183957
500.000000	0.193251