~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~
                                                 4 Operator Entries
7 October 1995                                  22 Instrument and Other Entries
                                                26 Total Log Entries
~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~

         ************ OPERATOR ENTRIES - CHRONOLOGICALLY *************

TIME: 04:38:49 FROM: J. Bowling
SUBJECT: Site General
0438 GMT.  10-7-95.  All BF's and CF are in condition GREEN.  All launches went
over 10Km.  No problems to report.
-------------------------
TIME: 12:27:57 FROM: Z. Waddell
SUBJECT: Instrument
12:20 GMT 10-07-95 Performed Condensation Inspection of C.F. EBBR:SIROS
EBBR Net Radiometer Dome - (Heavy Condensation)
SIROS               PSP  - (No Condensation)
                    PIR  - (No Condensation)
-------------------------
TIME: 12:31:10 FROM: Z. Waddell
SUBJECT: Site General
The 05:00 GMT to 13:00 GMT shift went well all sites were in condition
green all night and all launches were in excess of 10 Km.
-------------------------
TIME: 20:50:10 FROM: Kristin Waddell
SUBJECT: Facility
All of the Boundary Facilities including the Central Facility are in cond-ition
green.

The Morris Facility operator informed me of problems with his 12:30 CDST
launch.  All of his raw data was showing and continually changing but the height
would only say 500ft. and the time would only say that the launch  had been
going for 1 minute, when actually it lasted for 2 hours.
-------------------------
         **** INSTRUMENT AND OTHER LOG ENTRIES - CHRONOLOGICALLY ****

****     THE FOLLOWING ENTRY WAS GENERATED ON 4 October 1995                ****
****     BUT WAS RECEIVED TODAY, 7 October 1995                             ****

TIME: 00:33:20 FROM: MWR.B4
SUBJECT: Quality_Control

 Time: Wed Oct  4 00:33:20 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 00:33:40 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 00:34:40 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 01:03:40 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 01:04:00 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 01:06:40 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 01:07:20 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 01:07:40 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 01:10:40 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 01:11:00 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 01:12:00 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 01:13:00 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 01:13:20 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 01:13:40 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 03:09:40 1995
MWRLOS.B4, 23tbsky: Value below Minimum.
MWRLOS.B4, 31tbsky: Value below Minimum.

 Time: Wed Oct  4 03:10:40 1995
MWRLOS.B4, 23tbsky: Value now within expected range.
MWRLOS.B4, 31tbsky: Value now within expected range.

 Time: Wed Oct  4 06:07:20 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 06:07:40 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 11:09:40 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 11:10:20 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 11:12:20 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 11:12:40 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 11:26:00 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 11:26:40 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 11:33:40 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 11:34:00 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 11:45:40 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 11:46:00 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 11:46:40 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 11:47:00 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 11:47:40 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 11:48:00 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 11:49:20 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 11:49:40 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 12:02:20 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 12:02:40 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 12:05:00 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 12:05:20 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 15:43:40 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 15:44:20 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 16:53:00 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 16:53:20 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 16:54:00 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 16:54:20 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 16:56:00 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 16:56:20 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 16:56:40 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 16:57:00 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 16:57:40 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 16:58:20 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 16:58:40 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 16:59:40 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 17:00:20 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 17:00:40 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 17:01:00 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 17:02:00 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 17:02:40 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 17:03:00 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 17:03:40 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 17:04:40 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 17:05:00 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 17:27:20 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 17:27:40 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 17:29:00 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 17:29:20 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 17:33:20 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 17:33:40 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 17:34:20 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 17:35:00 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 17:35:20 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 17:35:40 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 17:36:40 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 17:37:00 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 17:37:20 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 17:37:40 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 17:38:00 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 17:38:20 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 17:48:00 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 17:49:00 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 17:49:40 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 17:50:00 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 17:54:20 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 17:58:20 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 17:58:40 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 18:04:20 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 18:04:40 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 18:05:40 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 18:06:00 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 18:07:20 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 18:07:40 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 18:18:00 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 18:19:00 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 18:19:20 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 18:19:40 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 18:20:20 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 18:21:20 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 18:21:40 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 18:22:00 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 18:22:20 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 18:23:40 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 18:24:00 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 18:27:00 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 18:27:20 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 20:42:40 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 20:43:00 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 20:45:00 1995
MWRLOS.B4, liq: Value now within expected range.

 Time: Wed Oct  4 20:45:20 1995
MWRLOS.B4, liq: Value below Minimum.

 Time: Wed Oct  4 20:46:20 1995
MWRLOS.B4, liq: Value now within expected range.

-------------------------
****     THE FOLLOWING ENTRY WAS GENERATED ON 4 October 1995                ****
****     BUT WAS RECEIVED TODAY, 7 October 1995                             ****

TIME: 02:30:10 FROM: SONDE.B1
SUBJECT: Quality_Control

 Time: Wed Oct  4 02:30:10 1995
SONDE.B1, wspd: Value above Maximum.

 Time: Wed Oct  4 02:30:12 1995
SONDE.B1, wspd: Value now within expected range.

-------------------------
****     THE FOLLOWING ENTRY WAS GENERATED ON 4 October 1995                ****
****     BUT WAS RECEIVED TODAY, 7 October 1995                             ****

TIME: 11:59:50 FROM: SONDE.B4
SUBJECT: Quality_Control

 Time: Wed Oct  4 11:59:50 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 11:59:52 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:00:10 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:00:12 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:00:20 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:00:22 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:01:00 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:01:02 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:01:40 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:01:42 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:03:20 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:03:22 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:03:40 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:03:42 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:03:50 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:03:52 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:04:00 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:04:02 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:04:40 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:04:42 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:05:10 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:05:12 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:09:50 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:09:52 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:10:10 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:10:12 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:11:10 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:11:12 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:12:40 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:12:42 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:12:50 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:12:52 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:13:30 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:13:32 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:14:00 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:14:02 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:14:20 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:14:22 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:15:20 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:15:22 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:16:00 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:16:02 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:16:10 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:16:12 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:16:20 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:16:22 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:16:40 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:16:42 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:17:10 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:17:12 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:18:10 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:18:12 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:18:40 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:18:42 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:18:50 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:18:52 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:20:00 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:20:02 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:20:30 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:20:32 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:20:50 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:20:52 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:21:20 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:21:22 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:22:50 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:22:52 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:23:40 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:23:42 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:24:00 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:24:02 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:24:10 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:24:12 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:26:20 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:26:22 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:26:50 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:26:52 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:27:20 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:27:22 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:28:00 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:28:02 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:28:10 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:28:12 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:28:50 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:28:52 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:29:20 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:29:22 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:32:50 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:32:52 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:33:20 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:33:22 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:33:40 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:33:42 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:34:30 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:34:32 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:34:40 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:34:42 1995
SONDE.B4, wspd: Value now within expected range.

 Time: Wed Oct  4 12:34:50 1995
SONDE.B4, wspd: Value above Maximum.

 Time: Wed Oct  4 12:34:52 1995
SONDE.B4, wspd: Value now within expected range.

-------------------------
TIME: 00:46:04 FROM: d3c649@roscoe.pnl.gov (Matt Macduff)
SUBJECT: tinman reboot

I shutdown tinman at about 2200 GMT and had SO
connect the replacement disk.  The machine was up
5 minutes later.

I checked and mounted the disk on /files0.

-Matt
-------------------------
TIME: 01:01:22 FROM: ROOT ACCOUNT <root>
SUBJECT: Site Data System - opslog_mail_monitor.pl - Summary Report:
 7 Messages were deleted from /var/spool/mail/opmail
and stored in /home/opslog/logfiles_opmail/logfiles_opmail_10_07_95

-------------------------
TIME: 01:42:16 FROM: EBBR.E20
SUBJECT: Collection

Could NOT RESOLVE checksum for line 28 (4 attempts).
Line 28 mark as bad and continuing.
Could NOT RESOLVE checksum for line 144 (4 attempts).
Line 144 mark as bad and continuing.
-------------------------
TIME: 03:00:00 FROM: SMOS.E8
SUBJECT: Quality_Control

 Time: Sat Oct  7 03:00:00 1995
30SMOS.E8, snow: Value below Minimum.

-------------------------
TIME: 05:43:50 FROM: EBBR.E12
SUBJECT: Collection

   Inst: EBBR.E12
   Host: dorothy
 Status: Collection successful.

-------------------------
TIME: 06:00:00 FROM: EBBR.E4
SUBJECT: Quality_Control

 Time: Sat Oct  7 06:00:00 1995
30EBBR.E4, hum_top: Value above Maximum.

 Time: Sat Oct  7 06:30:00 1995
30EBBR.E4, hum_bot: Value above Maximum.

-------------------------
TIME: 06:30:00 FROM: EBBR.E7
SUBJECT: Quality_Control

 Time: Sat Oct  7 06:30:00 1995
30EBBR.E7, hum_top: Value above Maximum.

 Time: Sat Oct  7 08:00:00 1995
30EBBR.E7, hum_bot: Value above Maximum.

-------------------------
TIME: 07:26:41 FROM: EBBR.E13
SUBJECT: Quality_Control

 From: Sat Oct  7 06:05:00 1995
   To: Sat Oct  7 07:15:00 1995
5EBBR.E13, wind_s: Value remained constant.

-------------------------
TIME: 07:50:00 FROM: SMOS.E7
SUBJECT: Quality_Control

 Time: Sat Oct  7 07:50:00 1995
1SMOS.E7, rh: Value above Maximum.

 Time: Sat Oct  7 08:00:00 1995
1SMOS.E7, rh: Value now within expected range.

 Time: Sat Oct  7 08:40:00 1995
1SMOS.E7, rh: Value above Maximum.

 Time: Sat Oct  7 09:17:00 1995
1SMOS.E7, rh: Value now within expected range.

 Time: Sat Oct  7 09:21:00 1995
1SMOS.E7, rh: Value above Maximum.

 Time: Sat Oct  7 09:34:00 1995
1SMOS.E7, rh: Value now within expected range.

 Time: Sat Oct  7 09:35:00 1995
1SMOS.E7, rh: Value above Maximum.

 Time: Sat Oct  7 09:00:00 1995
30SMOS.E7, rh: Value above Maximum.

-------------------------
TIME: 08:26:42 FROM: EBBR.E13
SUBJECT: Quality_Control

 From: Sat Oct  7 07:05:00 1995
   To: Sat Oct  7 08:15:00 1995
5EBBR.E13, wind_s: Value remained constant.

-------------------------
TIME: 09:15:00 FROM: SMOS.E20
SUBJECT: Quality_Control

 Time: Sat Oct  7 09:15:00 1995
1SMOS.E20, rh: Value above Maximum.

 Time: Sat Oct  7 09:17:00 1995
1SMOS.E20, rh: Value now within expected range.

 Time: Sat Oct  7 09:18:00 1995
1SMOS.E20, rh: Value above Maximum.

 Time: Sat Oct  7 09:23:00 1995
1SMOS.E20, rh: Value now within expected range.

-------------------------
TIME: 09:26:44 FROM: EBBR.E13
SUBJECT: Quality_Control

 From: Sat Oct  7 08:05:00 1995
   To: Sat Oct  7 09:15:00 1995
5EBBR.E13, wind_s: Value remained constant.

-------------------------
TIME: 09:30:00 FROM: EBBR.E22
SUBJECT: Quality_Control

 Time: Sat Oct  7 09:30:00 1995
30EBBR.E22, hum_top: Value above Maximum.
30EBBR.E22, hum_bot: Value above Maximum.

 Time: Sat Oct  7 10:00:00 1995
30EBBR.E22, hum_bot: Value now within expected range.

 Time: Sat Oct  7 10:30:00 1995
30EBBR.E22, hum_top: Value now within expected range.

-------------------------
TIME: 09:34:00 FROM: SMOS.E7
SUBJECT: Quality_Control

 Time: Sat Oct  7 09:34:00 1995
1SMOS.E7, rh: Value now within expected range.

 Time: Sat Oct  7 09:35:00 1995
1SMOS.E7, rh: Value above Maximum.

 Time: Sat Oct  7 11:14:00 1995
1SMOS.E7, rh: Value now within expected range.

 Time: Sat Oct  7 11:19:00 1995
1SMOS.E7, rh: Value above Maximum.

 Time: Sat Oct  7 11:27:00 1995
1SMOS.E7, rh: Value now within expected range.

 Time: Sat Oct  7 11:30:00 1995
1SMOS.E7, rh: Value above Maximum.

 Time: Sat Oct  7 12:51:00 1995
1SMOS.E7, rh: Value now within expected range.

 Time: Sat Oct  7 12:55:00 1995
1SMOS.E7, rh: Value above Maximum.

 Time: Sat Oct  7 12:57:00 1995
1SMOS.E7, rh: Value now within expected range.

 Time: Sat Oct  7 13:30:00 1995
30SMOS.E7, rh: Value now within expected range.

-------------------------
TIME: 11:30:00 FROM: EBBR.E22
SUBJECT: Quality_Control

 Time: Sat Oct  7 11:30:00 1995
30EBBR.E22, hum_top: Value above Maximum.
30EBBR.E22, hum_bot: Value above Maximum.

 Time: Sat Oct  7 14:00:00 1995
30EBBR.E22, hum_top: Value now within expected range.
30EBBR.E22, hum_bot: Value now within expected range.

-------------------------
TIME: 12:30:00 FROM: EBBR.E7
SUBJECT: Quality_Control

 Time: Sat Oct  7 12:30:00 1995
30EBBR.E7, home_30: Value below Minimum.

 Time: Sat Oct  7 13:00:00 1995
30EBBR.E7, home_30: Value now within expected range.

 Time: Sat Oct  7 14:30:00 1995
30EBBR.E7, hum_top: Value now within expected range.
30EBBR.E7, hum_bot: Value now within expected range.

-------------------------
TIME: 14:00:00 FROM: EBBR.E4
SUBJECT: Quality_Control

 Time: Sat Oct  7 14:00:00 1995
30EBBR.E4, hum_top: Value now within expected range.
30EBBR.E4, hum_bot: Value now within expected range.

-------------------------
TIME: 14:30:00 FROM: SMOS.E8
SUBJECT: Quality_Control

 Time: Sat Oct  7 14:30:00 1995
30SMOS.E8, snow: Value now within expected range.

-------------------------
TIME: 17:24:20 FROM: ROOT ACCOUNT <root>
SUBJECT: Boot Message from lion
The system, lion, is coming up at Sat Oct  7 17:23:48 GMT 1995
-------------------------
~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~
~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~
Central Facility Weather Observations on 7 October 1995
~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~
          Winds                              Total
Time     Spd/Dir      Press   Temp    RH     Cloud         Weather
(GMT)   (mps/deg)     (mb)    (C)     (%)    Coverage      Type
________________________________________________________________________________
0000     3.6/330      994.4   17.0    38     Scattered
0000     5.4/130      992.2   20.0    46     Scattered
0100     1.3/350      994.4   14.0    57     Clear
0200     0.0/030      994.7   13.0    61     Clear
0300     0.0/030      995.2   12.0    65     Clear
0400     0.0/030      995.3   11.0    66     Clear
0500     0.0/030      995.6   10.0    69     Clear
0600     0.0/030      995.8    9.0    78     Clear
0700     0.0/030      995.9    8.0    79     Clear
0800     0.0/030      996.3    7.0    80     Clear
0900     0.4/260      996.4    6.0    83     Clear
1000     0.0/260      996.3    6.0    85     Clear
1100     0.0/260      996.7    6.0    84     Clear
1200     0.0/270      997.0    5.0    85     Clear
1300     0.0/260      997.2    6.0    88     Clear
1400     0.4/140      997.7   12.0    71     Clear
1500     3.1/120      998.0   15.0    63     Clear
1600     3.6/160      997.8   17.0    57     Clear
1700     4.5/170      997.3   19.0    48     Clear
1800     4.9/170      996.6   21.0    41     Scattered
1900     3.6/180      995.3   23.0    39     Scattered
2000     4.5/150      994.2   22.0    39     Scattered
2100     4.9/140      993.4   23.0    37     Scattered
2200     4.9/180      992.9   23.0    37     Scattered
2300     3.6/140      992.3   21.0    44     Scattered

Cloud Observations
Whole Sky Cloud Types:
Time   Quadrant          Low                  Mid                High
                 Amt Ht(km/10) Type   Amt Ht(km/10) Type   Amt Ht(km/10) Type
_____________________________________________________________________________
0000      NE      .1    015      1
          SE      .1    015      1
          SW      .1    015      1
0000      NE                           .4    060      4
          SE                           .2    060      4
          SW                           .2    060      4
          NW                           .3    060      4
1800      NE                                                .1    090      2
          SW                                                .1    090      2
          NW                                                .3    090      2
1900      NE                                                .1    090      2
          SE                                                .1    090      2
          SW                                                .2    090      2
          NW                                                .3    090      2
2000      NE                                                .3    090      2
          SE                                                .2    090      2
          SW                                                .2    090      2
          NW                                                .4    090      2
2100      NE                           .3    060      4
          SE                           .2    060      4
          SW                           .3    060      4
          NW                           .3    060      4
2200      NE                           .3    060      4
          SE                           .2    060      4
          SW                           .2    060      4
          NW                           .3    060      4
2300      NE                           .3    060      4
          SE                           .2    060      4
          SW                           .3    060      4
          NW                           .4    060      4

Overhead Cloud Types (30 Degree Arc)
Time             Amt Ht(km/10) Type
________________________________________
0000              .1    060      4
2000              .4    090      2
2100              .1    060      4
2200              .1    060      4
2300              .1    060      4

Time   Comments in Weather Observations
________________________________________________________________________________

~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~~