50 MAS Configuration for 03-934    
01 01 16 0 0.098746 0.0000  0.452  0.471  0.493 0.100 2069.11
02 02 16 0 0.107461 0.0000  0.534  0.553  0.576 0.100 1842.30
03 03 16 0 0.062272 0.0000  0.633  0.659  0.685 0.100 1573.95
04 04 16 0 0.079941 0.0000  0.683  0.705  0.727 0.100 1431.72
05 05 16 0 0.062110 0.0000  0.725  0.743  0.769 0.100 1293.89
06 06 16 0 0.065703 0.0000  0.807  0.827  0.851 0.100 1070.61
07 07 16 0 0.063254 0.0000  0.850  0.871  0.891 0.100  986.82
08 08 16 0 0.057510 0.0000  0.890  0.909  0.932 0.100  922.05
09 09 16 0 0.032562 0.0000  0.931  0.953  0.972 0.100  844.03
10 10 16 0 0.010057 0.0000  1.597  1.622  1.649 0.100  240.90
11 11 16 0 0.009100 0.0000  1.650  1.676  1.701 0.010  223.62
12 12 16 0 0.008342 0.0000  1.704  1.728  1.754 0.010  197.02
13 13 16 0 0.003923 0.0000  1.755  1.780  1.805 0.010  174.51
14 14 16 0 0.003826 0.0000  1.809  1.832  1.856 0.010  154.55
15 15 16 0 0.003560 0.0000  1.859  1.882  1.906 0.010  138.42
16 16 16 0 0.003528 0.0000  1.908  1.932  1.957 0.100  131.05
17 17 16 0 0.003149 0.0000  1.959  1.984  2.007 0.010  122.32
18 18 16 0 0.003072 0.0000  2.010  2.034  2.058 0.010  108.41
19 19 16 0 0.002988 0.0000  2.059  2.084  2.107 0.010   96.43
20 20 16 0 0.003029 0.0000  2.109  2.134  2.157 0.010   86.04
21 21 16 0 0.006518 0.0000  2.159  2.184  2.207 0.010   76.55
22 22 16 0 0.003668 0.0000  2.209  2.234  2.257 0.010   72.03
23 23 16 0 0.004055 0.0000  2.260  2.284  2.309 0.010   65.83
24 24 16 0 0.004059 0.0000  2.311  2.336  2.360 0.010   58.68
25 25 16 0 0.004478 0.0000  2.361  2.386  2.410 0.010   57.98
26 26 16 1 0.956000 0.0000  3.082  3.165  3.245 0.001   23.83
27 27 16 1 0.960000 0.0000  3.250  3.325  3.396 0.001   18.69
28 28 16 1 0.961000 0.0000  3.399  3.470  3.549 0.001   15.19
29 29 16 1 0.949000 0.0000  3.549  3.625  3.706 0.001   13.15
30 30 16 1 0.945000 0.0000  3.699  3.765  3.832 0.001   11.48
31 31 16 1 0.942000 0.0000  3.853  3.930  4.010 0.001   10.04
32 32 16 1 0.940000 0.0000  4.009  4.085  4.167 0.001    8.77
33 33 16 1 0.939000 0.0000  4.170  4.242  4.323 0.001    7.48
34 34 16 1 0.938000 0.0000  4.318  4.390  4.469 0.001    6.55
35 35 16 1 0.935000 0.0000  4.468  4.540  4.618 0.001    5.68
36 36 16 1 0.934000 0.0000  4.612  4.690  4.772 0.001    4.91
37 37 16 1 0.933000 0.0000  4.768  4.845  4.922 0.001    4.31
38 38 16 1 0.932000 0.0000  4.916  4.990  5.063 0.001    3.82
39 39 16 1 0.931000 0.0000  5.057  5.125  5.220 0.001    3.40
40 40 16 1 0.930000 0.0000  5.212  5.275  5.353 0.001    3.04
41 41 16 1 0.932000 0.0000  5.361  5.399  5.440 0.001    2.76
42 42 16 1 0.966000 0.0000  8.357  8.585  8.763 0.010    0.45
43 43 16 1 0.963000 0.0000  9.480  9.696  9.950 0.010    0.27
44 44 16 1 0.960000 0.0000 10.260 10.514 10.757 0.010    0.20
45 45 16 1 0.957000 0.0000 10.750 10.983 11.262 0.010    0.16
46 46 16 1 0.947000 0.0000 11.782 12.024 12.276 0.010    0.12
47 47 16 1 0.943000 0.0000 12.659 12.883 13.108 0.010    0.09
48 48 16 1 0.948000 0.0000 13.025 13.283 13.502 0.010    0.08
49 49 16 1 0.939000 0.0000 13.535 13.803 14.075 0.010    0.07
50 50 16 1 0.944000 0.0000 14.020 14.263 14.469 0.010    0.06
|  |  |  | |        |       |      |      |     |        |
|  |  |  | |        |       |      |      |     |        =(!)
|  |  |  | |        |       |      |      |     =-Scale Factor
|  |  |  | |        |       |      |      =- Right 50% spectral response
|  |  |  | |        |       |      =-Peak 100% spectral response (microns)
|  |  |  | |        |       =-Left 50% spectral response (microns)
|  |  |  | |        =-Calibration intercept (only used for visible bands)
|  |  |  | =--------Calibration slope (only used for visible bands 1-25)
|  |  |  |          Blackbody emmissivity (only used for IR bands 26-50)
|  |  |  =----------0 => visible band, 1 => thermal infrared band"
|  |  =-------------Number of bits in this channel (8 or 10)
|  =----------------Spectral band assigned to this channel
=------------------------- Channel number 
(!) Sensor weighted solar spectral irradiance (Watts/meter2/micron)
at mean Earth-Sun distance

NOTE: Factor in column 10 is used to scale radiances in
      calibrated data by the following method:
      value_stored = int( radiance / factor ) for each channel
------------------------------------------------------------------------------
------------------------------------------------------------------------------
Title                    MODIS Airborne Simulator (MAS) Level-1B Data
Sensor_ID_Code           M5 105 MAS50
ExperimentName           THORpex 2003 
AircraftPlatform         NASA ER-2 (806) 
NavFormatCode            Format (A-1)
LocationCode             Inertial (INU)
AltitudeCode             Inertial (INU)
CalibrationVersion       Version 1.0 Calibration
CalibrationName          THORpex_Jan23-Mar19
SRF_Dataset              /karray/calibrat/mas/srf/Mar_03/
SRF_FTP                  ftp://asapdata.arc.nasa.gov/mas/srf/Mar_03/
NavDataSource            ER2 navigation recorder
NavDataPath              /nav/2003/er2/03-934
TbackBand                45, 47, 31, 273.0
GranuleVersion           1.0.0
MetadataVersion          1.0.0
------------------------------------------------------------------------------
------------------------------------------------------------------------------
Conversion from counts to radiance in VIS and NIR channels is
rad = (count - (cbb_avg)) * slope

"where,"
rad         = radiance
cbb_avg     = cold black body counts (running average)
slope       = from column 7 in above table
count       = count value

***********************************
IR Emissivity Correction (for MAS-50)
From Chris Moeller (UWisc)
***********************************
This is done in 2 steps:
1. Get the (TBACK Band) scan head count and convert it to a 
brightness temperature
2. Use the scan head temperature and the cold and warm   
black body counts/temperatures for computing the 
corrected slope and intercept for all IR bands.

(note: set to 1.0 for channels 1-25)
MAS BB Emissivity Coefficients
Hemispherical Reflectance (JPL) 11/00
convolved with MAS SRF (ARC) 10/00
Red spot Paint
Values in form Wavelength, Emissivity 
*************************************
For Bands 26 thru 50:
0.95600, 0.96000, 0.96100, 0.94900, 0.94500,
0.94200, 0.94000, 0.93900, 0.93800, 0.93500,
0.93400, 0.93300, 0.93200, 0.93100, 0.93000,
0.93200, 0.96600, 0.96300, 0.96000, 0.95700,
0.94700, 0.94300, 0.94800, 0.93900, 0.94400,
**************************************************
 ---

Config.asc creation date = 29-Jan-2003