50 MAS Configuration for 06-615  31 Jul 2006  
01 01 16 0 0.094507 0.0000  0.446  0.466  0.486 0.100 1979.74
02 02 16 0 0.100127 0.0000  0.528  0.550  0.570 0.100 1863.08
03 03 16 0 0.058449 0.0000  0.628  0.648  0.679 0.100 1576.93
04 04 16 0 0.076555 0.0000  0.680  0.700  0.722 0.100 1416.27
05 05 16 0 0.058646 0.0000  0.721  0.742  0.763 0.100 1289.09
06 06 16 0 0.060779 0.0000  0.803  0.824  0.845 0.100 1081.31
07 07 16 0 0.058502 0.0000  0.845  0.864  0.886 0.100  979.36
08 08 16 0 0.050155 0.0000  0.885  0.906  0.926 0.100  891.46
09 09 16 0 0.027655 0.0000  0.925  0.946  0.966 0.100  793.47
10 10 16 0 0.006700 0.0000  1.585  1.612  1.639 0.100  244.45
11 11 16 0 0.006572 0.0000  1.641  1.668  1.694 0.010  227.59
12 12 16 0 0.006428 0.0000  1.697  1.722  1.748 0.010  201.69
13 13 16 0 0.003228 0.0000  1.751  1.776  1.800 0.010  176.93
14 14 16 0 0.003366 0.0000  1.803  1.828  1.852 0.010  157.40
15 15 16 0 0.003267 0.0000  1.856  1.878  1.901 0.010  140.22
16 16 16 0 0.003136 0.0000  1.906  1.928  1.955 0.100  131.49
17 17 16 0 0.002966 0.0000  1.956  1.980  2.004 0.010  125.01
18 18 16 0 0.002896 0.0000  2.006  2.030  2.055 0.010  110.05
19 19 16 0 0.002701 0.0000  2.056  2.080  2.104 0.010   99.11
20 20 16 0 0.002700 0.0000  2.105  2.130  2.153 0.010   86.77
21 21 16 0 0.005571 0.0000  2.154  2.178  2.202 0.010   77.05
22 22 16 0 0.002989 0.0000  2.203  2.228  2.252 0.010   73.72
23 23 16 0 0.003011 0.0000  2.253  2.278  2.301 0.010   68.92
24 24 16 0 0.002833 0.0000  2.304  2.328  2.351 0.010   58.50
25 25 16 0 0.002685 0.0000  2.353  2.378  2.400 0.010   58.34
26 26 16 1 0.956000 0.0000  3.057  3.130  3.211 0.001   21.18
27 27 16 1 0.960000 0.0000  3.226  3.290  3.368 0.001   18.21
28 28 16 1 0.961000 0.0000  3.370  3.445  3.520 0.001   15.65
29 29 16 1 0.949000 0.0000  3.525  3.595  3.675 0.001   13.50
30 30 16 1 0.945000 0.0000  3.679  3.750  3.809 0.001   11.76
31 31 16 1 0.942000 0.0000  3.830  3.905  3.980 0.001   10.27
32 32 16 1 0.940000 0.0000  3.987  4.055  4.133 0.001    9.02
33 33 16 1 0.939000 0.0000  4.143  4.215  4.296 0.001    7.69
34 34 16 1 0.938000 0.0000  4.288  4.360  4.441 0.001    6.71
35 35 16 1 0.935000 0.0000  4.451  4.520  4.593 0.001    5.81
36 36 16 1 0.934000 0.0000  4.594  4.665  4.744 0.001    5.02
37 37 16 1 0.933000 0.0000  4.750  4.820  4.895 0.001    4.39
38 38 16 1 0.932000 0.0000  4.902  4.975  5.043 0.001    3.88
39 39 16 1 0.931000 0.0000  5.046  5.120  5.195 0.001    3.45
40 40 16 1 0.930000 0.0000  5.205  5.270  5.336 0.001    3.08
41 00 16 1 0.932000 0.0000  5.348  5.386  5.427 0.001    2.79
42 42 16 1 0.966000 0.0000  8.254  8.503  8.690 0.010    0.47
43 43 16 1 0.963000 0.0000  9.440  9.655  9.925 0.010    0.28
44 44 16 1 0.960000 0.0000 10.277 10.494 10.756 0.010    0.20
45 45 16 1 0.957000 0.0000 10.786 11.023 11.286 0.010    0.16
46 46 16 1 0.947000 0.0000 11.829 12.044 12.308 0.010    0.12
47 47 16 1 0.943000 0.0000 12.688 12.913 13.153 0.010    0.09
48 48 16 1 0.948000 0.0000 13.109 13.323 13.539 0.010    0.08
49 49 16 1 0.939000 0.0000 13.618 13.893 14.137 0.010    0.07
50 50 16 1 0.944000 0.0000 14.097 14.283 14.520 0.010    0.06
|  |  |  | |        |       |      |      |     |        |
|  |  |  | |        |       |      |      |     |        =(!)
|  |  |  | |        |       |      |      |     =-Scale Factor
|  |  |  | |        |       |      |      =- Right 50% spectral response
|  |  |  | |        |       |      =-Peak 100% spectral response (microns)
|  |  |  | |        |       =-Left 50% spectral response (microns)
|  |  |  | |        =-Calibration intercept (only used for visible bands)
|  |  |  | =--------Calibration slope (only used for visible bands 1-25)
|  |  |  |          Blackbody emmissivity (only used for IR bands 26-50)
|  |  |  =----------0 => visible band, 1 => thermal infrared band"
|  |  =-------------Number of bits in this channel (8 or 10)
|  =----------------Spectral band assigned to this channel
=------------------------- Channel number 
(!) Sensor weighted solar spectral irradiance (Watts/meter2/micron)
at mean Earth-Sun distance

NOTE: Factor in column 10 is used to scale radiances in
      calibrated data by the following method:
      value_stored = int( radiance / factor ) for each channel
------------------------------------------------------------------------------
------------------------------------------------------------------------------
Title                    MODIS Airborne Simulator (MAS) Level-1B Data
Sensor_ID_Code           M5 105 MAS50
ExperimentName           CC-VEX
AircraftPlatform         NASA ER-2 (806) 
NavFormatCode            Format (A-2)
LocationCode             Inertial (INU)
AltitudeCode             Inertial (INU)
CalibrationVersion       Version 1.0 Calibration
CalibrationName          CC-VEX_0627-0001
SRF_Dataset              /rarray/calibrat/mas/srf/Jun_06/
SRF_FTP                  ftp://asapdata.arc.nasa.gov/mas/srf/Jun_06/
NavDataSource            ER2 navigation recorder
NavDataPath              /nav/2006/er2/06-615
TbackBand                45, 47, 31, 273.0
GranuleVersion           1.0.0
MetadataVersion          1.0.0
GeoLocationProcess       standard
FlightComment            none
------------------------------------------------------------------------------
------------------------------------------------------------------------------
Conversion from counts to radiance in VIS and NIR channels is
rad = (count - (cbb_avg)) * slope

"where,"
rad         = radiance
cbb_avg     = cold black body counts (running average)
slope       = from column 7 in above table
count       = count value


***********************************
IR Emissivity Correction (for MAS-50)
From Chris Moeller (UWisc)
***********************************
This is done in 2 steps:
1. Get the (TBACK Band) scan head count and convert it to a 
brightness temperature
2. Use the scan head temperature and the cold and warm   
black body counts/temperatures for computing the 
corrected slope and intercept for all IR bands.

(note: set to 1.0 for channels 1-25)
MAS BB Emissivity Coefficients
Hemispherical Reflectance (JPL) 9/03
convolved with MAS SRF (ARC) 5/03
Red spot Paint
Values in form Wavelength, Emissivity 
*************************************
For Bands 26 thru 50:
0.95410, 0.95510, 0.95950, 0.94740, 0.94240,
0.93960, 0.93740, 0.93630, 0.93460, 0.93240,
0.93090, 0.93030, 0.92940, 0.92820, 0.92720,
0.92850, 0.96330, 0.95850, 0.95670, 0.95390,
0.94470, 0.94270, 0.94540, 0.93680, 0.94360,
**************************************************
 ---

Config.asc creation date = 12-Sep-2006