50 MASTER Configuration for 07-927-00  5 Aug 2007  E.Pacific Ocean, CR
01 01 16 0 0.033316 0.0000  0.439  0.464  0.484 0.100 1950.98
02 02 16 0 0.049526 0.0000  0.479  0.504  0.524 0.100 1913.94
03 03 16 0 0.042727 0.0000  0.520  0.544  0.565 0.100 1863.97
04 04 16 0 0.053063 0.0000  0.562  0.584  0.605 0.100 1809.98
05 05 16 0 0.068772 0.0000  0.632  0.656  0.692 0.100 1548.14
06 06 16 0 0.091845 0.0000  0.691  0.714  0.735 0.100 1379.38
07 07 16 0 0.072417 0.0000  0.730  0.754  0.774 0.100 1264.17
08 08 16 0 0.074714 0.0000  0.780  0.804  0.824 0.100 1131.88
09 09 16 0 0.077922 0.0000  0.847  0.870  0.890 0.100  970.61
10 10 16 0 0.062413 0.0000  0.887  0.912  0.930 0.100  880.89
11 11 16 0 0.068164 0.0000  0.928  0.952  0.972 0.010  787.33
12 12 16 0 0.008301 0.0000  1.575  1.608  1.636 0.010  247.23
13 13 16 0 0.007152 0.0000  1.635  1.664  1.691 0.010  229.46
14 14 16 0 0.006224 0.0000  1.694  1.722  1.744 0.010  203.29
15 15 16 0 0.006250 0.0000  1.746  1.776  1.800 0.010  178.03
16 16 16 0 0.004727 0.0000  1.802  1.828  1.852 0.100  157.05
17 17 16 0 0.004450 0.0000  1.855  1.876  1.901 0.010  140.63
18 18 16 0 0.004400 0.0000  1.905  1.936  1.957 0.010  131.32
19 19 16 0 0.004118 0.0000  1.956  1.982  2.004 0.010  123.00
20 20 16 0 0.002665 0.0000  2.058  2.084  2.106 0.010   98.54
21 21 16 0 0.002726 0.0000  2.140  2.166  2.188 0.010   79.55
22 22 16 0 0.002734 0.0000  2.190  2.216  2.238 0.010   74.28
23 23 16 0 0.002562 0.0000  2.240  2.264  2.287 0.010   70.74
24 24 16 0 0.001700 0.0000  2.297  2.324  2.368 0.010   59.69
25 25 16 0 0.002063 0.0000  2.367  2.392  2.428 0.010   56.10
26 00 16 1 0.960790 0.0000  3.073  3.149  3.224 1.000   20.88
27 27 16 1 0.961090 0.0000  3.207  3.280  3.352 0.005   18.46
28 28 16 1 0.964230 0.0000  3.349  3.440  3.507 0.006   15.88
29 29 16 1 0.958200 0.0000  3.511  3.585  3.662 0.006   13.63
30 30 16 1 0.956340 0.0000  3.664  3.745  3.803 0.006   11.86
31 31 16 1 0.955820 0.0000  3.813  3.900  3.970 0.006   10.34
32 32 16 1 0.955660 0.0000  3.975  4.055  4.124 0.006    9.09
33 33 16 1 0.955600 0.0000  4.144  4.215  4.297 0.005    7.68
34 34 16 1 0.955640 0.0000  4.278  4.350  4.431 0.001    6.78
35 35 16 1 0.955340 0.0000  4.439  4.515  4.578 0.003    5.85
36 36 16 1 0.955190 0.0000  4.585  4.670  4.735 0.006    5.04
37 37 16 1 0.955570 0.0000  4.739  4.820  4.885 0.005    4.42
38 38 16 1 0.956050 0.0000  4.887  4.975  5.030 0.005    3.91
39 39 16 1 0.956150 0.0000  5.033  5.120  5.181 0.002    3.48
40 40 16 1 0.966110 0.0000  5.183  5.255  5.327 0.001    3.10
41 41 16 1 0.977220 0.0000  7.622  7.803  7.935 0.001    0.66
42 42 16 1 0.972850 0.0000  7.944  8.195  8.418 0.010    0.54
43 43 16 1 0.965870 0.0000  8.435  8.655  8.815 0.010    0.43
44 44 16 1 0.964880 0.0000  8.887  9.097  9.282 0.010    0.36
45 45 16 1 0.963590 0.0000  9.506  9.716  9.906 0.010    0.27
46 46 16 1 0.964920 0.0000  9.924 10.127 10.316 0.010    0.23
47 47 16 1 0.966730 0.0000 10.377 10.625 10.984 0.010    0.19
48 48 16 1 0.966670 0.0000 11.031 11.327 11.725 0.010    0.15
49 49 16 1 0.966410 0.0000 11.929 12.175 12.447 0.010    0.11
50 00 16 1 0.968970 0.0000 12.693 12.918 13.170 0.010    0.09
|  |  |  | |        |       |      |      |     |        |
|  |  |  | |        |       |      |      |     |        =(!)
|  |  |  | |        |       |      |      |     =-Scale Factor
|  |  |  | |        |       |      |      =- Right 50% spectral response
|  |  |  | |        |       |      =-Peak 100% spectral response (microns)
|  |  |  | |        |       =-Left 50% spectral response (microns)
|  |  |  | |        =-Calibration intercept (only used for visible bands)
|  |  |  | =--------Calibration slope (only used for visible bands 1-25)
|  |  |  |          Blackbody emmissivity (only used for IR bands 26-50)
|  |  |  =----------0 => visible band, 1 => thermal infrared band"
|  |  =-------------Number of bits in this channel (8 or 10)
|  =----------------Spectral band assigned to this channel
=------------------------- Channel number 
(!) Sensor weighted solar spectral irradiance (Watts/meter2/micron)
at mean Earth-Sun distance

NOTE: Factor in column 10 is used to scale radiances in
      calibrated data by the following method:
      value_stored = int( radiance / factor ) for each channel
------------------------------------------------------------------------------
------------------------------------------------------------------------------
Title                    MODIS/ASTER Airborne Simulator (MASTER) Level-1B Data
Sensor_ID_Code           MS 124 MASTR
ExperimentName           TC4_2007  
AircraftPlatform         NASA ER-2 (809) 
NavFormatCode            Format (A-2)
LocationCode             Inertial (INU)
AltitudeCode             Inertial (INU)
CalibrationVersion       Version 1.1 Calibration
CalibrationName          TC4_0514-0823
SRF_Dataset              /rarray/calibrat/MASTER/srf/Aug_07/
SRF_FTP                  ftp://asapdata.arc.nasa.gov/MASTER/srf/Aug_07/
NavDataSource            ER2 navigation recorder
NavDataPath              /nav/2007/er2/07-927
TbackBand                45, 47, 31, 273.0
GranuleVersion           2.0.0
MetadataVersion          1.0.0
GeoLocationProcess       Standard
MGeoControlFilePath      none
FlightComment            
------------------------------------------------------------------------------
------------------------------------------------------------------------------
  
Conversion from counts to radiance in VIS and NIR channels is
Note: there was no instrument temperature correction performed!

rad = (count - (cbb_avg)) * slope

where,
rad         = radiance
cbb_avg     = cold black body counts (running average)
slope       = from column 7 in above table
count       = count value


***********************************
IR Emissivity Correction (for MAS-50{adopted for MASTER})
From Chris Moeller (UWisc) and Simon Hook (JPL)
***********************************
This is done in 2 steps:
1. Get the Channel 45 scan head count and convert it to a 
brightness temperature using UWisc subroutine TEMBCK.F
(from Liam Gumley (UWisc))
2. Call subroutine MASCAL.F (from Liam Gumley (UWisc)) which
uses the chan45 scan head temperature 
and the cold and warm black body counts/temperatures
for computing
the corrected slope and intercept for all IR bands.

Emissivities used were from Simon Hook's calculation for
BB paint and embedded in MASCAL.F program (mascal_master.f).
BB effective emissivity estimates for all 50 channels
(note: set to 1.0 for channels 1-25)
For Bands 1 thru 50:
1.0,1.0,1.0,1.0,1.0,
1.0,1.0,1.0,1.0,1.0,
1.0,1.0,1.0,1.0,1.0,
1.0,1.0,1.0,1.0,1.0,
1.0,1.0,1.0,1.0,1.0,
0.96079, 0.96109, 0.96423, 0.95820, 0.95634,
0.95582, 0.95566, 0.95560, 0.95564, 0.95534,
0.95519, 0.95557, 0.95605, 0.95615, 0.96611,
0.97722, 0.97285, 0.96587, 0.96488, 0.96359,
0.96492, 0.96673, 0.96667, 0.96641, 0.96897 
*************************
*************************

Config creation date: 26 Jul 2007