50 MAS Configuration for 06-630  26 Sep 2006  
01 01 16 0 0.098817 0.0000  0.445  0.464  0.486 0.100    0.00
02 02 16 0 0.104031 0.0000  0.527  0.552  0.570 0.100    0.00
03 03 16 0 0.060894 0.0000  0.628  0.648  0.679 0.100    0.00
04 04 16 0 0.079790 0.0000  0.680  0.700  0.722 0.100    0.00
05 05 16 0 0.061012 0.0000  0.721  0.742  0.763 0.100    0.00
06 06 16 0 0.062741 0.0000  0.802  0.822  0.844 0.100    0.00
07 07 16 0 0.060142 0.0000  0.844  0.864  0.885 0.100    0.00
08 08 16 0 0.051579 0.0000  0.884  0.904  0.925 0.100    0.00
09 09 16 0 0.028634 0.0000  0.924  0.942  0.965 0.100    0.00
10 10 16 0 0.006853 0.0000  1.585  1.612  1.638 0.100    0.00
11 11 16 0 0.006729 0.0000  1.641  1.668  1.694 0.010    0.00
12 12 16 0 0.006582 0.0000  1.697  1.724  1.747 0.010    0.00
13 13 16 0 0.003304 0.0000  1.750  1.776  1.799 0.010    0.00
14 14 16 0 0.003415 0.0000  1.802  1.828  1.852 0.010    0.55
15 15 16 0 0.003305 0.0000  1.856  1.878  1.902 0.010    0.00
16 16 16 0 0.003183 0.0000  1.906  1.928  1.955 0.100    0.00
17 17 16 0 0.003027 0.0000  1.956  1.982  2.005 0.010    0.00
18 18 16 0 0.002956 0.0000  2.006  2.032  2.055 0.010    0.00
19 19 16 0 0.002755 0.0000  2.056  2.082  2.104 0.010    0.00
20 20 16 0 0.002754 0.0000  2.106  2.132  2.154 0.010    0.00
21 21 16 0 0.005680 0.0000  2.155  2.180  2.202 0.010    0.00
22 22 16 0 0.003044 0.0000  2.203  2.230  2.253 0.010    0.00
23 23 16 0 0.003064 0.0000  2.254  2.278  2.301 0.010    0.00
24 24 16 0 0.002879 0.0000  2.305  2.330  2.352 0.010    0.65
25 25 16 0 0.002724 0.0000  2.354  2.378  2.401 0.010    0.00
26 26 16 1 0.956000 0.0000  3.053  3.125  3.201 0.001    0.00
27 27 16 1 0.960000 0.0000  3.220  3.290  3.360 0.001    0.00
28 28 16 1 0.961000 0.0000  3.364  3.445  3.512 0.001    0.00
29 29 16 1 0.949000 0.0000  3.518  3.595  3.667 0.001    0.00
30 30 16 1 0.945000 0.0000  3.672  3.750  3.806 0.001    0.00
31 31 16 1 0.942000 0.0000  3.821  3.905  3.973 0.001    0.00
32 32 16 1 0.940000 0.0000  3.980  4.060  4.126 0.001    0.00
33 33 16 1 0.939000 0.0000  4.143  4.215  4.296 0.001    0.00
34 34 16 1 0.938000 0.0000  4.288  4.360  4.441 0.001    0.70
35 35 16 1 0.935000 0.0000  4.444  4.520  4.586 0.001    0.00
36 36 16 1 0.934000 0.0000  4.587  4.665  4.737 0.001    0.00
37 37 16 1 0.933000 0.0000  4.744  4.820  4.888 0.001    0.00
38 38 16 1 0.932000 0.0000  4.896  4.975  5.036 0.001    0.00
39 39 16 1 0.931000 0.0000  5.039  5.120  5.188 0.001    0.00
40 40 16 1 0.930000 0.0000  5.200  5.260  5.333 0.001    0.00
41 00 16 1 0.932000 0.0000  5.348  5.386  5.427 0.001    0.00
42 42 16 1 0.966000 0.0000  8.227  8.473  8.689 0.010    0.00
43 43 16 1 0.963000 0.0000  9.416  9.655  9.908 0.010    0.00
44 44 16 1 0.960000 0.0000 10.241 10.494 10.741 0.010    0.74
45 45 16 1 0.957000 0.0000 10.767 11.023 11.268 0.010    0.00
46 46 16 1 0.947000 0.0000 11.797 12.054 12.283 0.010    0.00
47 47 16 1 0.943000 0.0000 12.662 12.873 13.127 0.010    0.00
48 48 16 1 0.948000 0.0000 13.066 13.313 13.529 0.010    0.00
49 49 16 1 0.939000 0.0000 13.592 13.893 14.108 0.010    0.00
50 50 16 1 0.944000 0.0000 14.054 14.263 14.495 0.010    0.00
|  |  |  | |        |       |      |      |     |        |
|  |  |  | |        |       |      |      |     |        =(!)
|  |  |  | |        |       |      |      |     =-Scale Factor
|  |  |  | |        |       |      |      =- Right 50% spectral response
|  |  |  | |        |       |      =-Peak 100% spectral response (microns)
|  |  |  | |        |       =-Left 50% spectral response (microns)
|  |  |  | |        =-Calibration intercept (only used for visible bands)
|  |  |  | =--------Calibration slope (only used for visible bands 1-25)
|  |  |  |          Blackbody emmissivity (only used for IR bands 26-50)
|  |  |  =----------0 => visible band, 1 => thermal infrared band"
|  |  =-------------Number of bits in this channel (8 or 10)
|  =----------------Spectral band assigned to this channel
=------------------------- Channel number 
(!) Sensor weighted solar spectral irradiance (Watts/meter2/micron)
at mean Earth-Sun distance

NOTE: Factor in column 10 is used to scale radiances in
      calibrated data by the following method:
      value_stored = int( radiance / factor ) for each channel
------------------------------------------------------------------------------
------------------------------------------------------------------------------
Title                    MODIS Airborne Simulator (MAS) Level-1B Data
Sensor_ID_Code           M5 108 MAS50
ExperimentName           DFRC_Sep_06
AircraftPlatform         NASA ER-2 (806) 
NavFormatCode            Format (A-2)
LocationCode             Inertial (INU)
AltitudeCode             Inertial (INU)
CalibrationVersion       Version 1.0 Calibration
CalibrationName          Aug_2006
SRF_Dataset              /rarray/calibrat/mas/srf/Aug_06/
SRF_FTP                  ftp://asapdata.arc.nasa.gov/mas/srf/Aug_06/
NavDataSource            ER2 navigation recorder
NavDataPath              /nav/2006/er2/06-630
TbackBand                45, 47, 31, 273.0
GranuleVersion           1.0.0
MetadataVersion          1.0.0
GeoLocationProcess       Standard
FlightComment            Objective was to have co-incident data collection with MAS & AVIRIS
------------------------------------------------------------------------------
------------------------------------------------------------------------------
Conversion from counts to radiance in VIS and NIR channels is
rad = (count - (cbb_avg)) * slope

"where,"
rad         = radiance
cbb_avg     = cold black body counts (running average)
slope       = from column 7 in above table
count       = count value

***********************************
IR Emissivity Correction (for MAS-50)
From Chris Moeller (UWisc)
***********************************
This is done in 2 steps:
1. Get the (TBACK Band) scan head count and convert it to a 
brightness temperature
2. Use the scan head temperature and the cold and warm   
black body counts/temperatures for computing the 
corrected slope and intercept for all IR bands.

(note: set to 1.0 for channels 1-25)
MAS BB Emissivity Coefficients
Hemispherical Reflectance (JPL) 9/03
convolved with MAS SRF (ARC) 5/03
Red spot Paint
Values in form Wavelength, Emissivity 
*************************************
For Bands 26 thru 50:
0.95410, 0.95510, 0.95950, 0.94740, 0.94240,
0.93960, 0.93740, 0.93630, 0.93460, 0.93240,
0.93090, 0.93030, 0.92940, 0.92820, 0.92720,
0.92850, 0.96330, 0.95850, 0.95670, 0.95390,
0.94470, 0.94270, 0.94540, 0.93680, 0.94360,
**************************************************
 ---

Config.asc creation date = 16-Nov-2006