# Presentation: 694725

## FEMCI Poster  Development of Cryogenic Accelerometers for 4K

_**Principal Investigators**_:  Petar Arsenovic/551, S. Irish/542, Brian Ross/549
_**Proposal Title**_:  Development of Cryogenic Accelerometers for 4K
_**Background**_:
- Acceptable finite element model validation requires accurate measurement of the hardware’s response which is complicated for cryogenic hardware due to poor availability of cryogenic material property data.
- To obtain accurate analytic models, it is desirable to measure the structural response of hardware while it is subjected to both a cryogenic temperature environment (~4K) and a controlled vibration test.
- It is currently difficult to perform these measurements because:
- Few commercially available accelerometers rated for cryogenic temperatures are small enough (~<.5”cube) to be useful;
- Accelerometers  rated for only 77K and need to be tested for 4K;
- Need to develop a method to attach accelerometers to flight hardware without damage to hardware.

- 1

## FEMCI Poster  Development of Cryogenic Accelerometers for 4K

_**Objectives**_:
- This proposal will:
- Provide scientists and engineers a new capability to measure the structural performance of their cryogenic hardware.
- Verify certain types of accels will operate at 4K and structural loading.
- Investigate accel attachment methods to obtain a clean signal without damaging the hardware.
_**Context and Innovation**_:
- Being able to develop this capability will enhance the data measurement and analysis technology at GSFC, which supports the Mission and Science Measurement Technologies section of the GSFC Strategic Implementation Plan.
- This capability will be essential for planned missions, ie. Constellation X, in the development of detectors and low temperature hardware.
- Other proposed missions, such as Missing Baryon Explorer (MBE) and ASTRO-G will have hardware development that requires low temperature measurement.

- 2

## FEMCI Poster Development of Cryogenic Accelerometers for 4K

- _Context and Innovation (cont)_:
- Being able to provide small, accurate 4K accels will enable engineers and scientists to more accurately determine the structural performance of cryogenic instruments.  This in turn will provide:
- More efficient, lightweight designs of components;
- Reduce development time of designs;
- Eliminate additional ambient temperature structural testing.
- _Schedule_:  Purchase instrumentation-2 months, investigate bonding materials-1 month, cryogenic testing and data evaluation-2 months
- _Budget_:  15K total for vibration facility, instrumentation, cryogen for dewar, and materials consultation
- _Manpower Estimate_:
- 	.2 manyear: vibration test engineer	.2 manyear: analytical systems engineer
- 	.1 manyear: dewar engineer, 		.1 manyear for technical advisor,
- 	.1 manyear: ManTech-vibration testing 	.1 manyear: Swales materials expert

- 3

## FEMCI Poster:  Vibratable DewarDevelopment of Cryogenic Accelerometers for 4K

- 4

**XRS Filter Vibratable Dewar: Used to Perform LHe Vibration Testing **
**(units are inches, 10 p.s.i burst disc and dewar mounting plate to shaker are not shown)**

## FEMCI Poster:  Current AccelerometersDevelopment of Cryogenic Accelerometers for 4K

- 5

**Current accelerometers are too large and heavy to measure the responses of small, cryogenic components.**

## FEMCI Poster:  New AccelerometersDevelopment of Cryogenic Accelerometers for 4K

- 6

- Purchased Kistler Type 8730A, small accelerometer.  It is .67” tall and .281” diameter and weighs 1.9 grams.
- This accelerometer is being tested for use at LN2 and LHe temperatures.

## FEMCI Poster:  Kistler Accelerometers Mounted to Cold Plate of Vibratable DewarDevelopment of Cryogenic Accelerometers for 4K

- 7

**Kistler accelerometers are shown here mounted to the cold plate of the vibratable dewar.  One is bonded to the plate using Scotchwell 2216 and the other is bolted to the plate with a Fiberglass G-10 block needed for electrical isolation.**

- Kistler Accels

**Current Accels**

## FEMCI Poster:  Examples of Small Cryogenic ComponentsDevelopment of Cryogenic Accelerometers for 4K

- 8

**FEA and the FEA Transition Box contains a small (~ 1” x 1” square) suspended area that houses the detector.  This detector is kept at .065 K for launch.  It was very difficult to predict the structural resonance of the suspended hardware and this was only measured at ambient temperature since instrumentation did not exist that would allow accurate measurement at cryogenic temperature.**

## FEMCI Poster:  Examples of Small Cryogenic ComponentsDevelopment of Cryogenic Accelerometers for 4K

- 9

**ADR contains a salt pill which is ~ 6” tall and 1.5” diameter.  It is suspended from Kevlar straps.  This component is at  4K for launch.  The picture here shows the ADR being tested at ambient conditions with an accelerometer mounted on the top, outer cylinder and one mounted on the top of the salt pill (only the accel cable is shown). These accelerometers were removed for cryogenic testing since this instrumentation was not acceptable for such cold temperatures.  Structural test data was required but not obtained at cold temperatures.**