# Presentation: 433629

## Applications of a Pixellated Detection System to Digital Mammography

**V****aleria ****Rosso**
*  **Dipartimento di Fisica, Universita’ di Pisa and Sezione INFN Pisa, Italy*
* *
* *

*  **valeria.rosso**@pi.infn.it*

## Outline

- The detection system, which the Pisa group is working with, consists of a single photon counting chip [G.Bisogni, et al., SPIE  1998 San Diego], bump bonded to a semiconductor detector: the hybrid approach allows to change either the thickness of the detector or the semiconductor type.
- Most important advantages of such system, with respect to a traditional X-rays film/screen device, are the wider linear dynamic range (104 -105) and the higher performance in terms of MTF and DQE [S.R.Amendolia et al., Nucl Instr Meth,** A461**,(2001) 389-392]. Besides the single photon counting architecture allows the detection of image contrasts lower than 3%, that is relevant for mammographic applications [S.R.Amendolia et al., IEEE Trans Nucl Science, **47**(4), (2000), 1478-1482].
-  The detection system
-       300 m Silicon detector
- 	Photon Counting Chip
- X-ray images of details of an accreditation phantom
- Standard mammographic tube
- RMI 156
- Hybrid images of some details using the dual-energy technique

## Calibration measurements

- The chip is an array of 64x64 asynchronous read-out channels, each one equipped with a low noise charge preamplifier, a latched comparator, a digital shaper and a 15 bits counter.
- An energy threshold, common to the 4096 pixels, can be externally selected by means of an external bias voltage Vth; in addition, the threshold of each pixel can be finely adjusted. The preamplifier circuit has an input test sensitive to electrical pulses injected through a calibration capacity (Ctest = 22 fF).

- To optimize the working point of our detection system, a set of calibration measurements is performed. These tests are carried out by sending voltage pulses through the test input of each channel.
- After the fine threshold tuning among the 4096 channels we obtained a total threshold spread  of tot = 350 e-. To make an absolute calibration of the threshold, the silicon detector has been exposed to monochromatic beams whose energy ranges from 14 to 32 keV.

- Calibration curve

## The system

- VME board

- Mother board

## Projection radiographic image of the RMI 156 phantom

- mammographic tube
-    (Mo -Mo) @ 30 kVp
- Dose = 4 mGy
- 300 m Si detec.
- (scan 6x6)

**6 cm**

- Projection radiographic image of the RMI 156 phantom

**8**** cm**

- To realize a partial image of the RMI 156 mammographic phantom with our detection system ( active area ~ 1 cm2) was necessary to perform a scan: 36 images were acquired one after the other and then the total image was built
- Inside the phantom there are: six different size nylon fibers simulate fibrous structures (from 1 to 6), five groups of simulated micro-calcifications (from 7 to 11) and five different sizes tumor-like masses are included in the wax insert (from 12 to 16)

## 18 detection systems based on 200 m GaAs detectors

- 18 detection systems based on 200 m GaAs detectors

**Integrated Mammographic Imaging project**
** ****Funded by the Ministry for University and Scientific Research of Italy**
** ****Research: ****INFN and University Physics Departments of Pisa, Ferrara, Roma, Napoli**
** ****Industries: ****Laben (electronics), AleniaMarconiSystems (detectors, bump-bonding), CAEN (electronics), Pol.Hi.Tech.(scintillators), Gilardoni (X-rays****)**

- Web site:  http://imamint.df.unipi.it

## Dual energy basis decomposition techniques 

- R.E.Alvarez and A. Makovski  ( Phys. Med. Biol. **5**, 733,1976) and L.A. Lehmann, R.E.Alvarez and A. Makovski ( Med. Phys. **8**, 659,1981) presented a technique that using the (E) dependence, is able to identify unknown materials and cancel the contrast between couples of materials.
- To apply the technique we:
-  need two images at different energies: we worked with two radioactive sources  109Cd   <X-ray> = 22.5 keV as low energy beam EL and 125I < X-ray> = 27.7 keV as high energy beam EH.
- decompose and project the images: using the high and low energy images we calculate the log attenuation and then we  decompose the images on the basis set (we considered as basis materials wax and PMMA); varying the projection angle between 0 and 90, we have obtained 90 hybrid images, and among them the most interesting are those in which one can see the contrast cancellation for different pairs of materials: from this angle one can determine the Z of the radiographed materials
- Phantom
- X-ray sources
- Images

## Phantom

-  Ø 2.5 cm

- 700 m

- The difference in %  between water and polyethylene-terephtalate [H2O-PT)/ H2O] is < 10% in the 20-30 keV range.
- The difference between SiO2 and calcium hydroxyapathite is greater

## X-ray sources

- <125I X-ray> = 27.7 keV

-                                  @ 22.5 keV                  @ 27.7 keV
- Contrast: SiO2-wax    (19.0_+_0.2)%                  (9.9_+_0.3)%
- Contrast: PT-wax       (19.6_+_0.2)%                  (12.1_+_0.3)%

- 109Cd image

- 125I image

- The dose in air for the 125I image was 0.3 mGy; while the one for the 109Cd image was 0.4mGy

## 109Cd image

- 109Cd image

- 125I image

- log. atten.
- decomp. on basis
- project

##  @  22 keV 

-  @  22 keV
- SNRSiO2=36.7+0.3
- SNRPT=39.1+0.1

- @ 27 keV
- SNRSiO2=17.0+0.2
- SNRPT=21.9+0.1.

## C between wax and SiO2 is cancelled

- C between wax and SiO2 is cancelled

- C between wax and polymethyl-terephtalate is cancelled

- 44O

- 52O

- Knowing the materials of the phantom details it’s  possible to evaluate the theoretical projection angles for the cancellation of C between SiO2 and wax: 41O, and for the cancellation of C between PT and wax: 47O

## Conclusions

-  The detection system is useful for its use in digital mammography: we have obtained good results in terms of C and MTF
- The prototype for the ImaMInt project will be ready in June 2003 for validation tests.
- The Dual Energy technique can improve the specificity of the projection mammography, bringing information about the different composition of the radiographed details. It may also enhance the contrast for certain detail removing any materials from the image.
- Using  this detection system in conjunction with standard polychromatic beam we have shown that the dose was reduced; using monochromatic X-rays the dose can be further reduced . We hope to achieve still better results using GaAs detectors.

## Slide 14