# Presentation: 211323

## Run 2BSMT Low Voltage Power 

** **
**      ****The present location of the SMT DC power supply units creates a problem whenever a power unit needs to be serviced or replaced. The problem being that the iron must be opened to permit access to the power units.**

## Run 2BSMT Low Voltage Power 

**1)****       ****Relocate the power supplies on first floor of movable counting house so that service and or replacement can be done at any time.**
**2)****       ****Find a commercial power supply to supply SMT Low Voltage DC power.**
**3)****       ****Redesign fuse panel for better reliability.**
**4)****       ****Use remote sensing at present fuse panel locations.**
**5)****       ****Route wiring from power supply units across cable bridge up north end to cathedral areas on east and west sides to new fuse panels.**

## Purposed Relocation

## MCH Location

## Power Supply Requirements 

**Need 2 Power Supplies now for  1% and 10% testing**
**Need 8 Power Supplies for final system**
**Requirements:**
**Long distance remote sensing (150 Ft.)**
**Delayed compensation for better stability**
**208 VAC input power**
**Monitoring and control compatibility**
**Power failure– and system reset- signals**
**Remote interlock input**

## Wiener PL500

**DC output characteristics:**
**Remote capabilities**
**Individually sensed **
**Continuously controlling regulator to load**
**Long distance sensing with delayed compensation**
**Voltage adjustment (10 mv resolution)**
**Noise measurements**
**330μF and 1μF ceramic across load **
**100nF HF- conducting to case**
**Wire length from power supply to load**
**0.5m wire:< 10mVpp (0-20MHz ) 3mVrms(0-2MHz)**
** ****10m wire:< 3mVpp (0-300MHz)**

## Wiener PL500

**PL500-front**

## Wiener PL500

**PL500-back**

## Interface Crate 0 Power

## Interface Crate 1 Power

## Interface Crate Power

**Interface Power supply for use with either crate**
- 					 	PL 500 	Power Supply
- 						Suggestions
-    Output	   Voltage    Current	   Power		Voltage	Current	Max
-   Number   Require    Required  Required	Range	 Range 	Power
- 1	    5.25 V     63.00 A	    330.75 W	  5-10V	   80A	600W
- 2	  15.10 V       4.50 A	      67.95 W	12-30V	 11.5A	280W
- 3	    2.75 V     16.28 A	      44.77 W	    2-7V	    30A	210W
- 4	    2.75 V     59.40 A	    163.35 W	    2-7V	  115A	550W
- 5	    2.75 V     52.80 A	    _145.20 _W	    2-7V	  115A	550W
**	****	****Total    752.02 Watts**

## Adapter Card

**Adapter Card Power Requirements**
**                  ****Current        Current       Current    Current   Current**
**PS unit    Per hybrid   Per hybrid    Per End   Total       Per Oct.**
**volts            ma                amps          amps      amps      amps**
**3.3              72                0.072            31.968       63.94      8.00**
**5.0             133               0.133              9.052     118.10      14.76**

## SE Crate 0 Adapter Cards

## SE Crate 1 Adapter Cards

## SW Crate 0 Adapter Cards

## SW Crate 1 Adapter Cards

## Can Bus Interface

**Provide Can Bus interface**
**For 1% and 10% testing**
**1)****       ****PC **
**2)****       ****Network Interface Card (Can Bus card)**
**3)****       ****Network Interface Card (Ethernet card)**
**4)****       ****Lab View Software**
**5)****       ****Lab View / Epics Software**

## New Distribution Panel

## New Fuse Panel

## Voltage Drops VCC

## Voltage Drops 15 V 

## Voltage Drops VDD to Hybrid

## Voltage Drops VDD

## Power Supply cost

**Price for PL500-F8 power supply (quantity 1...4) : 6,045.00$**
**Two Power Supplies have been ordered and will be here in January**
**Special price for 8 power supplies, each : 5,743.00$ **
**Delivery time: about 12 weeks aro**

## Can bus cost

**CAN-USB, USB interface for CAN-bus, with P-CAN light driver,**
**P-Can view software and  WIN-CRM, windows crate **
**manager software  for CAN-bus based WIENER crate **
**remote control (Windows 95 / 98 version / XP)**
**Price per system: 575.00 $**
** ****One has been ordered and is somewhere on site**
**Delivery time: about 3 .. 4 week a.r.o.**