# Presentation: 145837

## Traceability and Legal Metrology

## 4 Parts of a Measurement

- Device Under Test
- 		(quartz, rubidium, and cesium)
- Reference
- 	 	(cesium oscillator or transfer standard)
- Method
- 		(measurement system and procedure)
- Uncertainty Analysis
- 		(statistics and data reduction)

## ISO definition of traceability

- 	The property of the result of a measurement or the value of a standard whereby it can be related to stated references, usually national or international standards, through an unbroken chain of comparisons all having stated uncertainties.

## The Traceability Chain

- The traceability chain is a series of comparisons between the device under test to a reference.  The final comparison in the chain is made using the International System (SI) units as a reference.
- Each comparison is a link in the chain.
- The uncertainty of each comparison (link) must be known and documented.

## The Traceability Chain (cont.)

- National metrology institutes (NMIs) like NIST provide the ultimate measurement references for their countries.  The intent of all NMIs is to realize the SI units as closely as possible.
- Although the goal is to establish traceability to the SI, this is often done by comparing to an NMI that in turn compares its references to the SI.

## Time and Frequency SI Units

-  Second (s)
- standard unit for time interval
- intrinsic, defined using cesium 133 atom
- one of 7 base SI units
-   Hertz (Hz)
- standard unit for frequency (s-1)
- events per second
- one of 21  SI units derived from base units

## The Traceability Pyramid

## The Importance of Traceability

- Establishing *traceability* to national and international standards provides evidence that measurements are being made correctly
- Traceable measurements are essential elements of:
-  quality control systems
-  trade agreements
-  allowing compatible products to be                                          	manufactured all over the world

## The Traceability Problem

-  Technical
- usually not a difficult technical problem, especially if requirements are modest
-   Legal
- requires repeatable, reliable procedures
- requires documentation of traceability chain for quality control systems
- may be contractual requirement or a prerequisite for doing business

## ISO Guide 17025  -   5.6.2.1

- 	For calibration laboratories, the program for calibration of equipment shall be designed and operated so as to ensure that calibrations and measurements made by the laboratory are traceable to the SI (Systeme International) units of measurement.

## Laboratory Accreditation

- 	Calibration laboratories that seek accreditation through a program such as NVLAP must comply with ISO 17025 and document the traceability chain for each  measurement.  The traceability chain must link the measurement back to the SI unit, and the documentation must include the uncertainty of each link of the chain.

## Types of Time and Frequency Information

-  Date and Time-of-Day
-  records when an event happened
-  Time Interval
-  duration between two events
-   Frequency
-  rate of a repetitive event

## Everything Relates to the Second	

- By counting seconds, we can determine  date and time-of-day
- By counting events per second, we can measure frequency

## SI Definition of the Second	

-    The duration of 9,192,631,770 periods of the radiation corresponding to the transition between two hyperfine levels of the ground state of the cesium-133 atom.

## Uncertainties of physical realizations of the base SI units

## Coordinated Universal Time (UTC)

- About 50 NMIs do a continuous comparison of their standards (over 200 atomic oscillators are involved).  The average value of these standards is used to create the UTC time scale.  The BIPM in France handles the data and computes the uncertainties.
- BIPM Circular T shows difference between UTC and UTC(k), or the UTC maintained by each NMI.

## BIPM Circular T (www.bipm.fr)

## Low accuracy measurements that require traceability

## High accuracy measurements that require traceability

## Other Areas of Metrology that require Traceable Frequency

## Ways to establish traceability to the SI through NIST

- NIST controlled broadcasts
- WWV/WWVH, WWVB, ACTS, ITS
- NIST monitored broadcasts
- LORAN-C, GPS
- NIST controls/monitors measurements
- Frequency Measurement Service

## Traceability chain for signals controlled by a NMI

## The Distribution of UTC(NIST)

## Traceability chain for signals monitored by a NMI

## GPS Monitoring

- More complex than monitoring ground based signals because:
- Each orbiting satellite carries its own frequency standards
- Satellites can only be received when they fly over monitoring station

## NIST GPS Data Archive

## NIST GPS Data Archive

## GPS Traceability Chain A

## GPS Traceability Chain B

## NIST traceable signals and links

## Measuring 1 x 10-8

## Measuring 1 x 10-9

## Measuring 1 x 10-10

## Measuring 1 x 10-11

## Time and Frequency is well suited for remote calibrations

- Since frequency standards are sensitive to shipment and environmental changes, calibrations are often made at the customer’s site
- Radio reference (often GPS) is used as transfer standard
- Calibrations can be monitored by an NMI using a modem or Internet link

## NIST Frequency Measurement and Analysis Service

- Measurements are made remotely at customer’s site
- NIST supplies everything needed
- Measurement process is automated
- NIST downloads data by modem, performs uncertainty analysis, sends calibration reports
- Provides traceability to NIST with uncertainty of 2 x 10-13 per day

## NIST Frequency Measurement and Analysis Service (cont.)

- NIST Service ID Number 76100S
- 5 oscillators can be calibrated at once
- Calibrations are automated
- Recognized by NVLAP
- Replacement parts sent when necessary using overnight delivery service

## FMAS Customer Map (labs with cesium oscillators in red)

- FMAS Customer Map (labs with cesium oscillators in red)

## NIST FMAS Display

## Calibration Reports

- Monthly calibration reports are sent to each NIST FMAS customer.   Report is compliant with ISO Guide 25/17025 and includes a statement of uncertainty.
- The report defines the calibration period as one day (24 hours).
- The report annotates all situations where no data were recorded or where the device under test was out of tolerance.

## Reporting of Measurement Errors and Outages

- The monthly calibration report annotates the following situations:
- No data were recorded
- GPS reception problems
- GPS broadcast errors
- Measurement System Errors including hardware or software failures
- Device under Test errors
- Device under Test substitutions/changes

## FMAS Specifications 

## NIST Policy on Traceability

- For more information about traceability, visit the NIST Traceability Web Site:
- http://nist.gov/traceability/