# Presentation: 861334

## Fermilab Main Injector Beam Position Monitor UpgradeB. Banerjee, W. Barker, S. Bledsoe, T. Boes, C. Briegel, D. Capista, G. Deuerling, R. Dysert, R. Forster, S. Foulkes, W. Haynes, B. Hendricks, T. Kasza,               R. Kutschke, A. Marchionni, M. Olson, V. Pavlicek, L. Piccoli, P. Prieto, S. Rapisarda,             A. Saewert, J. Van Bogaert, M. Votava, R. Webber, M. Wendt, N. Wilcer, S. WolbersFermi National Accelerator Laboratory, P.O. Box 500, Batavia, IL, USA 60510

**Analog**
-  Analog signal processing occurs in both the Combiner Boxes and the Transition Modules.  The Combiner Boxes accept beam signals from the four stripline electrodes, pass them through a low-pass filter (*f*c ≈ 100 MHz) and combine them to form either a vertical measuring or a horizontal measuring signal pair.  The analog Transition Modules interface the signals from the Combiner Boxes to the Echotek digital receiver, adapting signal levels and further filtering frequency content.
**Digital**
- Eight-channel EchoTek ECDR-GC814-FV2 VME64X digital receiver boards are used for beam signal digitization and frequency conversion. Each GC814 channel is comprised of a 14 bit ADC (AD6645), a four-channel digital down DDC (GC-4016), an FPGA accumulator, and 128K x 32 dual-ported RAM. These boards shift the information content of the beam signal from a RF analog form to a stream of digital, baseband I and Q data pairs.  The position, *P*, and intensity, *I*, are given by:

**Fermilab Main Injector Beam Position Monitor Upgrade**B. Banerjee, W. Barker, S. Bledsoe, T. Boes, C. Briegel, D. Capista, G. Deuerling, R. Dysert, R. Forster, S. Foulkes, W. Haynes, B. Hendricks, T. Kasza,               R. Kutschke, A. Marchionni, M. Olson, V. Pavlicek, L. Piccoli, P. Prieto, S. Rapisarda,             A. Saewert, J. Van Bogaert, M. Votava, R. Webber, M. Wendt, N. Wilcer, S. Wolbers*Fermi National Accelerator Laboratory, P.O. Box 500, Batavia, IL, USA 60510*

**Introduction**
- A project to upgrade the readout electronics of the Fermilab Main Injector (MI) Beam Position Monitors (BPMs) was begun in 2005.  This project is one of many improvements to the Fermilab accelerator complex aimed at providing increased luminosity and reliability for the Run 2 collider physics program.  The design builds upon many of the technologies and techniques used to upgrade the BPM electronics in the Recycler Ring, the Tevatron, as well as some of the transfer lines.
**Operational Requirements**
- The Fermilab Main Injector (MI) is designed to accelerate either protons or antiprotons with an injection energy of 8 GeV and an extraction energy as high as 150 GeV.  At any time, however, only one of the two beam species may be present.    The time structure of the beam may be one of the following:

- This work was supported by Fermi National Accelerator Laboratory, operated by Universities Research Association Inc. under contract No. DE-AC02-76CH03000 with the United States Department of Energy

**Performance**

**Figure 2:** Overview of the Main Injector BPM hardware.

- The results presented below were obtained using a pre-production system installed for 11 BPM locations. Figure 3 illustrates, for one BPM, the closed-orbit capability at 53 MHz for a simple MI cycle.  Figure 4 illustrates the closed-orbit response to a more complex MI cycle; both vertical and horizontal measurements are shown.
- Figure 5 summarizes the closed-orbit resolutions at 53 MHz obtained for the 11 BPMs, using the method described in Figure 3.  Figure 5 illustrates the turn-by-turn capability of the system, using anti-protons injected from the Recycler.

**Figure 6:** Turn-by-turn 2.5 MHz data for injection of anti-protons from the Recycler.

**Figure 3**: Closed-orbit proton data, 53 MHz band.

**Figure 5**: RMS resolution for different BPM locations. Horizontal RMS correlates with the MI dispersion function.

- where the coefficients ai were determined from bench test measurements.

**Figure 4**: Closed orbit proton data, 53 MHz band.  Anti-proton and NuMI neutrino production.

- For further details see the paper WPMP01046 contributed to this conference, and references therein.

**53 MHz  Protons:** From 1 bunch up to a full batch of 84 bunches in successive 53 MHz buckets. Up to 6 batches, each one of 84 bunches, can be loaded in the MI.
**53 MHz Anti-protons:** Four consecutive groups of antiprotons, spaced by 396 ns, of typically 5 (up to 9) 53 MHz bunches, are present in the MI.
**2.5 MHz - Protons or anti-protons:** Four bunches in successive 2.5 MHz RF buckets (396 ns spacing).

- The supported measurement types are:
**Closed-orbit:** A narrowband measurement averaging over all bunches for about 40 turns; resolution bandwidth 1 kHz.
**Turn-by-turn:** A wideband measurement averaging over all bunches in 1/14 of the circumference (1/28 for 2.5 MHz); resolution bandwidth 500 kHz ( 250 kHz for 2.5 MHz ).