<!-- Title: Eve shift. Thu Sep 12, 2002 -->
<!-- SciCo: Giorgio Apollinari -->
<!-- DAQAce: Bruce Knuteson -->
<!-- MonAce: John Zhou -->
<!-- CO: Stan Lai -->
<!-- OpManager: Matt Jones -->
<!-- Notes: -->
<pre> 
Data-taking continues with store 1748, L=1E31, pbar stack at 90mA, stacking rate 6.7mA.  
 
Store will be kept until stack reached 120mA. Then CDF goes into an one hour acess. </pre>

<!-- Date: Thu Sep 12 16:08:27 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151436">151436</A>
 Terminated at 2002.09.12 16:08:20
<!-- processed by 131.225.45.171 -->
<!-- Author: RunControl -->


<!-- Date: Thu Sep 12 16:08:40 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151436">151436</A>
TERMINATE: Ending physics run in order to run an hour of testing.
<!-- processed by 131.225.236.161 -->
<!-- Author: Knuteson -->



<!-- Date: Thu Sep 12 16:10:46 2002 -->
Run stopped to allow Mark to check WCAL06 and other tests.
<!-- Author: Xin Wu -->
<!-- Comment: Thu Sep 12 17;39;00 comment by...M Mattson -->
<br>
Earlier today, the CO noticed a zero occupancy channel in WHA 12 East, eta 10 right.  He pasted the plot into the
<a href=
'/cgi/elog/elog.pl?nb=2002&action=view&page=764&frame=2&anchor=11:07:22'>
day shift e-log</a>.<br>
<br>
I ran the diagnostics for wcal06, slot 16, ch 9.  Everything checks out okay.  I have picked Robin and Rick's brains, and we believe the problem lies upstream of the ADMEMs.<br>
<br>
Nikolay checked the HV, which he said look okay.<br>
<br>
Now it is up to the hardware/gain people to investigate.
<!-- Date: Thu Sep 12 16:16:35 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151441">151441</A>
ACTIVATE: Begin silicon D-mode calibration run.
<!-- processed by 131.225.235.212 -->
<!-- Author: Knuteson -->


<!-- Date: Thu Sep 12 16:21:17 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151441">151441</A>
TERMINATE: Ending silicon D-mode calibration run, with DPS-on.
<!-- processed by 131.225.236.161 -->
<!-- Author: Knuteson -->


<!-- Date: Thu Sep 12 16:24:10 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151443">151443</A>
ACTIVATE: Begin silicon D-mode calibration run, DPS-off.
<!-- processed by 131.225.233.51 -->
<!-- Author: Knuteson -->


<!-- Date: Thu Sep 12 16:26:56 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151444">151444</A>
ACTIVATE: dps testing
<!-- processed by 131.225.236.49 -->
<!-- Author: jdl -->


<!-- Date: Thu Sep 12 16:33:01 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151443">151443</A>
TERMINATE: Ending silicon D-mode calibration run, DPS-off.
<!-- processed by 131.225.236.161 -->
<!-- Author: Knuteson -->


<!-- Date: Thu Sep 12 16:36:46 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151444">151444</A>
TERMINATE: end rol tests

<!-- processed by 131.225.235.150 -->
<!-- Author: jdl -->
<!-- Comment: Thu Sep 12 17;57;56 comment by...Greg -->
<pre>Prescales still bouncing around during dps runs.
We have captured the program's output and have 
sent this along to the expert Ke Li.
</pre>
<!-- Date: Thu Sep 12 16:42:01 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151451">151451</A>
 Activated at 2002.09.12 16:42:14
<!-- processed by 131.225.236.161 -->
<!-- Author: RunControl -->


<!-- Date: Thu Sep 12 16:44:56 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151451">151451</A>
 Terminated at 2002.09.12 16:42:54
<!-- processed by 131.225.233.157 -->
<!-- Author: RunControl -->


<!-- Date: Thu Sep 12 16:44:56 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151449">151449</A>
 Activated at 2002.09.12 16:43:55
<!-- processed by 131.225.233.157 -->
<!-- Author: RunControl -->


<!-- Date: Thu Sep 12 16:45:07 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151449">151449</A>
ACTIVATE: Re-starting physics run.
<!-- processed by 131.225.127.7 -->
<!-- Author: Knuteson -->



<!-- Date: Thu Sep 12 16:47:15 2002 -->
<center><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-7471&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-7471' width='300'></a><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-7472&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-7472' width='300'></a><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-7473&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-7473' width='300'></a></center>
<!-- Author: JZ -->


<!-- Date: Thu Sep 12 16:57:14 2002 -->
Starting Physics run again. L=1E31. Stack at 105, with stacking rate at 6 mA/h.
<!-- Author: Giorgio Apollinari -->


<!-- Date: Thu Sep 12 17:35:45 2002 -->
scpu_tracer_event_id error on b0eb11.
<!-- Author: Knuteson :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151449' target=_top>151449</a>) -->



<!-- Date: Thu Sep 12 17:41:41 2002 -->
L2 Trigger Rates from SVT seem slightly low. 
Expert was paged.
<!-- Author: Stan :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151449' target=_top>151449</a>) -->
<!-- Comment: Thu Sep 12 19;23;38 comment by...Jim Kraus -->
One of the Two Track Trigger rates looks a little low (barely outside tolerances).  The XTRP plot in TrigMon looks fine and this is a relatively new trigger, so the most likely explanation is that the tolerances for this trigger are too tight.
<!-- Date: Thu Sep 12 17:42:41 2002 -->
SVT tritter appears on the low side (blue color) in TrigMon. SVT expert paged. He promised he would look into them.
<!-- Author: Giorgio Apollinari -->


<!-- Date: Thu Sep 12 17:52:58 2002 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2002&action=view&page=-7474&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2002&action=view&page=-7474  width='300' align='top'></td><td>CES Wedge Strip Occupancy: One channel looks hot.  Phi wedge 12. Expert being paged</a></td></tr></table>
<!-- Author: Stan -->
<!-- Comment: Thu Sep 12 17;53;59 comment by...Stan -->
Karen Byrum says that this problem is known.  No emergency.


<!-- Date: Thu Sep 12 17:54:03 2002 -->
<table border=0><td><tr><td bgcolor=#e0e030><pre>Problem from last night when pressing "Calib_Load"  
after the QIE calibration has been solved in the  
fer package.   Before taking another calibration,  
be sure to: 
 
          setup fer 
          rc 
 
Note the "setup fer" to pick up changes. 
 
</pre></td></tr></td></table>
<!-- Author: W.Badgett -->
<!-- Comment: Thu Sep 12 18;03;33 comment by...W.Badett -->
<pre>Forgot to point out:  There is a new rundb package 
version (v1_9a) which has a small change involving 
warning the user about L2 decision conflicts.

I.e., "TS mode Auto L2 accept" is incompatible 
with "L2 Processors making decisions" and a fatal 
error message is produced.
</pre>

<!-- Date: Thu Sep 12 18:49:06 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151459">151459</A>
ACTIVATE: This is a test comment, please ignore
<!-- processed by 131.225.232.44 -->
<!-- Author: test -->


<!-- Date: Thu Sep 12 18:57:01 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151460">151460</A>
ACTIVATE: This is another test comment, please continue to ignore
<!-- processed by 131.225.236.49 -->
<!-- Author: anotherTest -->


<!-- Date: Thu Sep 12 19:04:08 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151460">151460</A>
IDLE: ignore me
<!-- processed by 131.225.233.10 -->
<!-- Author: anotherTest -->



<!-- Date: Thu Sep 12 19:29:10 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151449">151449</A>
 Terminated at 2002.09.12 19:29:25
<!-- processed by 131.225.236.229 -->
<!-- Author: RunControl -->



<!-- Date: Thu Sep 12 19:29:32 2002 -->
MCR calls. Shot to be dumped in 15 min.
<!-- Author: Giorgio Apollinari -->

<!-- Date: Thu Sep 12 19:30:04 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151449">151449</A>
TERMINATE: Ending physics run; store will be dumped in roughly 15 min.
<!-- processed by 131.225.236.161 -->
<!-- Author: Knuteson -->



<!-- Date: Thu Sep 12 19:37:09 2002 -->
All HV off.
<!-- Author: Giorgio Apollinari -->


<!-- Date: Thu Sep 12 19:52:01 2002 -->
<center><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-7475&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-7475' width='300'></a><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-7476&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-7476' width='300'></a><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-7477&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-7477' width='300'></a></center>
<!-- Author: John Zhou -->


<!-- Date: Thu Sep 12 19:52:38 2002 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2002&action=view&page=-7478&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2002&action=view&page=-7478  width='300' align='top'></td><td>x</a></td></tr></table>
<!-- Author: John Zhou -->
<!-- Comment: Thu Sep 12 19;56;29 comment by...JZ -->
<pre>Store 1748 Luminosity Summary. Delivered: 805.8, Captured: 617.9.
Eff: 76.7%. This includes silicon D-mode calibration.</pre>
<!-- Date: Thu Sep 12 19:54:36 2002 -->
<br><b>Unmask Trigger Tower:</b> 
<br>Unmask trigger tower masked last night in preparation  
for tests.  The unmasked tower will take effect on the next 
run. 
<br>
<!-- Author: R.J. Tesarek -->


<!-- Date: Thu Sep 12 19:58:35 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151461">151461</A>
TERMINATE: Ending trigger test run.
<!-- processed by 131.225.127.7 -->
<!-- Author: Knuteson -->
<!-- Comment: Thu Sep 12 20;05;42 comment by...Arnd -->
This is the readout list ultimate validation run.

<!-- Comment: Thu Sep 12 20;20;03 comment by...Greg, Arnd, Peter -->
Initial tests indicate success in chain of sending ROL from L2 to TS to FE to CSL to consumers. We are performing further checks offline.
<!-- Date: Thu Sep 12 20:03:16 2002 -->
Control Access started
<!-- Author: Giorgio Apollinari -->



<!-- Date: Thu Sep 12 20:23:45 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151462">151462</A>
 Activated at 2002.09.12 20:22:50
<!-- processed by 131.225.235.212 -->
<!-- Author: RunControl -->


<!-- Date: Thu Sep 12 20:23:45 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151462">151462</A>
ACTIVATE: Starting spike run.
<!-- processed by 131.225.235.212 -->
<!-- Author: Knuteson -->


<!-- Date: Thu Sep 12 20:23:45 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151462">151462</A>
 Terminated at 2002.09.12 20:23:40
<!-- processed by 131.225.235.212 -->
<!-- Author: RunControl -->


<!-- Date: Thu Sep 12 20:23:45 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151462">151462</A>
TERMINATE: Ending spike test.
<!-- processed by 131.225.235.212 -->
<!-- Author: Knuteson -->


<!-- Date: Thu Sep 12 20:31:33 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151463">151463</A>
ACTIVATE: Starting SMX calib run.
<!-- processed by 131.225.236.161 -->
<!-- Author: Knuteson -->


<!-- Date: Thu Sep 12 20:32:04 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151463">151463</A>
TERMINATE: Ending SMX calib run.
<!-- processed by 131.225.235.212 -->
<!-- Author: Knuteson -->


<!-- Date: Thu Sep 12 20:35:32 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151464">151464</A>
ACTIVATE: Starting QIE calib run.
<!-- processed by 131.225.127.10 -->
<!-- Author: Knuteson -->



<!-- Date: Thu Sep 12 20:44:13 2002 -->
<b>Post Access Roundup</b><br> 
Replaced a CAFE card in ccal13, slot 17, channel 3.  This channel was producing x20 trigger rates for last nights spike run.  After the swap, pedestals look normal (in both look-up and pass-through modes).  Spike run at 117 Hz.  Also did showermax and QIE calibrations for this crate.  
<!-- Author: M Mattson, V Tiwari, and the whole Scooby Gang -->

<!-- Date: Thu Sep 12 20:52:24 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151464">151464</A>
TERMINATE: Ending calor QIE calib run of ccal13 only w/ downloading.
<!-- processed by 131.225.236.161 -->
<!-- Author: Knuteson -->
<!-- Comment: Thu Sep 12 20;56;05 comment by...DAQ ace and Mario -->
Checked and downloaded QIE calibration constants in b0ccal13.

<!-- Date: Thu Sep 12 20:53:45 2002 -->
Shower Max Calibrations show no signs of anomaly. 
Plots generated with DBAna are placed in the binder.
<!-- Author: Stan -->
<!-- Comment: Thu Sep 12 20;54;06 comment by...Stan -->
Run 151463


<!-- Date: Thu Sep 12 21:03:06 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151465">151465</A>
 Activated at 2002.09.12 21:00:21
<!-- processed by 131.225.236.162 -->
<!-- Author: RunControl -->


<!-- Date: Thu Sep 12 21:03:06 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151465">151465</A>
ACTIVATE: Starting PMT spike test run.
<!-- processed by 131.225.236.162 -->
<!-- Author: Knuteson -->


<!-- Date: Thu Sep 12 21:04:26 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151465">151465</A>
 Terminated at 2002.09.12 21:04:45
<!-- processed by 131.225.234.146 -->
<!-- Author: RunControl -->


<!-- Date: Thu Sep 12 21:06:24 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151465">151465</A>
TERMINATE: Ending PMT spike run.
<!-- processed by 131.225.234.146 -->
<!-- Author: Knuteson -->


<!-- Date: Thu Sep 12 21:13:18 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151466">151466</A>
 Activated at 2002.09.12 21:13:28
<!-- processed by 131.225.105.19 -->
<!-- Author: RunControl -->


<!-- Date: Thu Sep 12 21:17:34 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151466">151466</A>
 Terminated at 2002.09.12 21:17:52
<!-- processed by 131.225.236.161 -->
<!-- Author: RunControl -->


<!-- Date: Thu Sep 12 21:17:34 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151466">151466</A>
 Terminated at 2002.09.12 21:17:52
<!-- processed by 131.225.236.161 -->
<!-- Author: RunControl -->


<!-- Date: Thu Sep 12 21:19:12 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151466">151466</A>
ACTIVATE: One more PMT spike run.
<!-- processed by 131.225.236.161 -->
<!-- Author: Knuteson -->


<!-- Date: Thu Sep 12 21:19:13 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151466">151466</A>
ACTIVATE: One more PMT spike run.
<!-- processed by 131.225.236.161 -->
<!-- Author: Knuteson -->


<!-- Date: Thu Sep 12 21:19:21 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151466">151466</A>
TERMINATE: Ending PMT spike run.
<!-- processed by 131.225.236.161 -->
<!-- Author: Knuteson -->



<!-- Date: Thu Sep 12 21:28:03 2002 -->
PMT spike run results show everything is fine. 
L1 trigger rate is 120 Hz.
<!-- Author: Stan :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151466' target=_top>151466</a>) -->


<!-- Date: Thu Sep 12 21:28:14 2002 -->
Rick confirms that "Spike Run" is successfull. Will move to quiet time calibrations, starting with CAL QIE.
<!-- Author: Giorgio Apollinari -->

<!-- Date: Thu Sep 12 21:30:20 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151467">151467</A>
ACTIVATE: Begin calor QIE calib run.
<!-- processed by 131.225.236.229 -->
<!-- Author: Knuteson -->


<!-- Date: Thu Sep 12 21:31:10 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151467">151467</A>
TERMINATE: End calor QIE calib run.
<!-- processed by 131.225.232.165 -->
<!-- Author: Knuteson -->


<!-- Date: Thu Sep 12 21:32:41 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151468">151468</A>
ACTIVATE: Begin ToF QIE calib run.
<!-- processed by 131.225.236.161 -->
<!-- Author: Knuteson -->


<!-- Date: Thu Sep 12 21:33:15 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151468">151468</A>
TERMINATE: Ending ToF QIE calib run.
<!-- processed by 131.225.232.165 -->
<!-- Author: Knuteson -->


<!-- Date: Thu Sep 12 21:39:36 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151469">151469</A>
ACTIVATE: Begin CLC QIE calib run.
<!-- processed by 131.225.232.165 -->
<!-- Author: Knuteson -->


<!-- Date: Thu Sep 12 21:41:04 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151469">151469</A>
TERMINATE: Ending CLC QIE calib run.
<!-- processed by 131.225.236.161 -->
<!-- Author: Knuteson -->


<!-- Date: Thu Sep 12 21:45:35 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151470">151470</A>
ACTIVATE: Begin BSC QIE calib run.
<!-- processed by 131.225.236.161 -->
<!-- Author: Knuteson -->



<!-- Date: Thu Sep 12 21:49:10 2002 -->
Shot setup is advertised for 22:00. Will finish the Quiet Time calibration and move to the remaining calibrations.
<!-- Author: Giorgio Apollinari -->


<!-- Date: Thu Sep 12 21:49:22 2002 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2002&action=view&page=-7479&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2002&action=view&page=-7479  width='300' align='top'></td><td>Note Bad channels in black -> QIE Calorimeter calibration</a></td></tr></table>
<!-- Author: Stan -->
<!-- Comment: Thu Sep 12 21;51;37 comment by...Stan -->
These are for QIE Calorimeter pedestals.
The slopes show no bad channels

<!-- Comment: Fri Sep 13 08;32;42 comment by...Mario -->
This is the channel corresponding to the <a href=
'/cgi/elog/elog.pl?nb=2002&action=view&page=765&frame=2&anchor=20:44:13'>replaced card</a>
<!-- Date: Thu Sep 12 21:54:34 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151470">151470</A>
TERMINATE: Aborting BSC QIE calib run after 5 minutes -- unsure whether this run would finish, and hustling to finish a set of calib runs before near-term shot setup.
<!-- processed by 131.225.236.161 -->
<!-- Author: Knuteson -->



<!-- Date: Thu Sep 12 21:54:43 2002 -->
No odd channels in TOF QIE calibration
<!-- Author: Stan :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151468' target=_top>151468</a>) -->

<!-- Date: Thu Sep 12 21:56:49 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151471">151471</A>
ACTIVATE: Begin calor LED calib run.
<!-- processed by 131.225.236.161 -->
<!-- Author: Knuteson -->


<!-- Date: Thu Sep 12 21:58:10 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151471">151471</A>
TERMINATE: End calor LED calib run.
<!-- processed by 131.225.236.162 -->
<!-- Author: Knuteson -->



<!-- Date: Thu Sep 12 22:01:16 2002 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2002&action=view&page=-7480&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2002&action=view&page=-7480  width='300' align='top'></td><td>No problem with CLC QIE calibration plot.  Run 151469</a></td></tr></table>
<!-- Author: Stan -->

<!-- Date: Thu Sep 12 22:05:06 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151472">151472</A>
ACTIVATE: Begin calor XEF calib run.
<!-- processed by 131.225.242.150 -->
<!-- Author: Knuteson -->


<!-- Date: Thu Sep 12 22:05:27 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151472">151472</A>
TERMINATE: End calor XEF calib run.
<!-- processed by 131.225.236.161 -->
<!-- Author: Knuteson -->



<!-- Date: Thu Sep 12 22:11:42 2002 -->
Another iFix program started up out of blue. The old one is still running. By clicking on the windows of the old iFix, the new one went away.
<!-- Author: JZ -->


<!-- Date: Thu Sep 12 22:17:11 2002 -->
In order to exploit Quiet Time, we run SMX Quiet Time immediately after the the CAL QIE, and then went back to CAL LED and CAL Xenon. Apparently this prevents the LED and Xenon calibration to appear in DBANA. 
<!-- Author: Giorgio Apollinari -->

<!-- Date: Thu Sep 12 22:19:18 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151473">151473</A>
ACTIVATE: Begin AAA_shotsetup run in preparation for beam.
<!-- processed by 131.225.236.229 -->
<!-- Author: Knuteson -->


<!-- Date: Thu Sep 12 22:25:16 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151473">151473</A>
TERMINATE: Ending run due to fatal error in b0imu01.  Trapped in an attempted halt; no alternative but abort.  Error logger messages include VISIONread/write failed in slot 11, state transition not allowed, resetting card in slot 11 failed, etc.
<!-- processed by 131.225.236.162 -->
<!-- Author: Knuteson -->


<!-- Date: Thu Sep 12 22:31:07 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151474">151474</A>
TERMINATE: Ending run -- b0imu01 not allowing us to get through cold start.
<!-- processed by 131.225.236.161 -->
<!-- Author: Knuteson -->
<!-- Comment: Thu Sep 12 22;56;16 comment by...Knuteson -->
Camille advised us to hit the reset button on b0imu01.  Doing so (and restarting the run) solved the problem.  Thanks, Camille!
<!-- Date: Thu Sep 12 22:33:10 2002 -->
Paged IMU and CAL experts for AAA_SHOTSETUP problems with coldstarting one b0imu crate and all the CCAL crate.
<!-- Author: Giorgio Apollinari -->


<!-- Date: Thu Sep 12 22:35:57 2002 -->
CEM, CHA, WHA along with their respective heartbeat went gray on global alarm at about 10:10. They came back in 15 minutes. In the mean time, Icicle died. I restarted.
<!-- Author: JZ -->


<!-- Date: Thu Sep 12 22:46:54 2002 -->
Received evb_proxy_sm_timeout error during attempted partitioning.  Cleaning up evb at evb's request.
<!-- Author: Knuteson -->

<!-- Date: Thu Sep 12 22:54:36 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151475">151475</A>
ACTIVATE: Beginning AAA_shotsetup run.
<!-- processed by 131.225.236.162 -->
<!-- Author: Knuteson -->



<!-- Date: Thu Sep 12 23:55:15 2002 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151436>
151436</A> 
x24F8C</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_1_02 [3,136,317]</TD>                                       
<TD ALIGN=CENTER>14:34:14</TD>                                          
<TD ALIGN=CENTER>16:08:20</TD>                                          
<TD ALIGN=CENTER>01:32:55</TD>                                          
<TD ALIGN=RIGHT>     23,877,974</TD>                                    
<TD ALIGN=RIGHT>       624,721</TD>                                     
<TD ALIGN=RIGHT>     137,303</TD>                                       
<TD ALIGN=RIGHT>   61.134</TD>                                          
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=151449>
151449</A> 
x24F99</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_1_02 [3,136,317]</TD>                                       
<TD ALIGN=CENTER>16:43:55</TD>                                          
<TD ALIGN=CENTER>19:29:25</TD>                                          
<TD ALIGN=CENTER>02:43:46</TD>                                          
<TD ALIGN=RIGHT>     37,285,498</TD>                                    
<TD ALIGN=RIGHT>       982,902</TD>                                     
<TD ALIGN=RIGHT>     219,222</TD>                                       
<TD ALIGN=RIGHT>   96.267</TD>                                          
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>23:55:01</TD>                                                       
<TD ALIGN=CENTER>04:16:42</TD>                                          
<TD ALIGN=RIGHT>      61,163,472</TD>                                   
<TD ALIGN=RIGHT>      1,607,623</TD>                                    
<TD ALIGN=RIGHT>       356,525</TD>                                     
<TD ALIGN=RIGHT>  157.401</TD>                                          
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- processed by 131.225.236.229 -->
<!-- Author: End of Shift Report -->



<!-- Date: Thu Sep 12 23:57:51 2002 -->
<font color=red>Shift Summary: </font><pre>16:00  Data-taking continues with store 1748, L=1E31. Pbar stack at 90 mA,  stacking rate a 6 mA/h. Store will continue until stack goes to 120 mA. Then CDF will have a one hour access. 
 
16:10  Zero occupancy in WHA 12 E, eta 10 right. Hardware/gain people to investigate. 
 
16:16  Silicon D-mode calibration 
  
16:36  Completed dps runs studies. Greg reports still problems with prescales "bouncing around". 
 
16:45  Restarting physics run. 
 
19:29  Store to be dumped in 15 min. Lower HV. 
 
20:03  Started controlled access. 
 
20:44  Report from Access: "Replaced CAFE card inccal 13, slot 17, channel 3. Things OK after swap.". 
 
20:53  SMX Calibration performed. OK. 
 
21:28  Spike runs OK. Continue with Calibrations. All Calibrations check OK on DBANA. 
 
22:19  Shot setup started. Preparing system for AAA_SHOTSETUP run. 
 
22:31  Problems with coldstarting b0imu01. Camille paged, she suggested to reset the crate. It works. 
 
22:54  Running AAA_SHOTSETUP, waiting for proton injection. </pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   151436-151449
Delivered Luminosity   127.2  
Acquired Luminosity    109.6  
Efficiency             86.2%
   
</td></tr></table></pre>
<!-- Author: Giorgio Apollinari -->