<!-- Title: Day shift. Mon Nov 22, 2004 -->
<!-- SciCo: Giovanni Pauletta -->
<!-- DAQAce: Ryo Tsuchiya -->
<!-- MonAce: Pete McNamara -->
<!-- CO: &nbsp; -->
<!-- OpManager: &nbsp; -->
<!-- Notes: -->
<pre>Starting day shift</pre>

<!-- Date: Mon Nov 22 08:29:58.891 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=190037">190037</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [16,466,486] 
<!-- Author: Ryo & Pete -->



<!-- Date: Mon Nov 22 08:30:18.568 2004 -->
I'm working on CMX HV for a pair of wedges on the SW.  Ignore ramping, etc.
<!-- Author: Dale Stentz -->
<!-- Comment: Mon Nov 22 10;02;49 comment by...Dale -->
Finished this an hour ago - <a href=
'/cgi/elog/elog.pl?nb=2004&action=view&page=810&frame=2&anchor=10:01:10.250'>
See CMX special instructions below</a>.
<!-- Date: Mon Nov 22 08:39:26.800 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=190038">190038</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [16,466,486] 
<!-- Author: Ryo & Pete -->



<!-- Date: Mon Nov 22 08:47:06.865 2004 -->
<pre>B0ccal13 is not responding.  I power-cycled it and noticed 
the error light was on the tracer.   Ping is not  
working to the processor.   </pre>
<!-- Author: Kuhlmann -->
<!-- Comment: Mon Nov 22 08;50;58 comment by...m mattson -->
<br>This is a crate with a newer processor.  I've noticed that it seems to take longer to respond (to RC monitoring) after rebooting (or power cycling).  I suggest issuing a "reset_crate", then get a coffee and check on it later.
<!-- Comment: Mon Nov 22 09;01;48 comment by...Frank Chlebana -->
<pre>For the problems with ccal13:

The boot path had an extra "saved" directory which is no longer
present. When rebooting the mvme5500 it could not find the 
kernal.

I changed the boot path on the mvme5500 so that the kernal is
now found.

The same problem with COT08 - change the boot path and tested
rebooting the processor.

looks like the other crates with the MVME5500 have the correct
path.
 
</pre>
<!-- Date: Mon Nov 22 08:49:59.437 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=190039">190039</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [16,466,486] 
<!-- Author: Ryo & Pete -->




<!-- Date: Mon Nov 22 08:53:33.315 2004 -->
<p>Went downstairs anticipating fun with the PISABOX PC, only to 
find it has not been filling the iFIX DB over the weekend. 
I have now killed iFIX and rebooted the PC; it is coming back up 
now.</p>
<!-- Author: Steve Hahn -->

<!-- Date: Mon Nov 22 08:57:43.944 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=190040">190040</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Ryo & Pete -->




<!-- Date: Mon Nov 22 09:06:41.950 2004 -->
Fixed what we believe was a cabling error on wires 16-31 on CES 10E. 
Calibration run then showed problem in b0ccal13 - SMXR 4. (16E). Looking at this now.
<!-- Author: Jimmy Proudfoot, Steve Kuhlmann -->
<!-- Comment: Mon Nov 22 11;31;49 comment by...Proudfoot -->
16E was fixed with minimal intervention
<!-- Date: Mon Nov 22 09:19:00.880 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=190041">190041</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: TEST_CSL_MB_XING [2,541,490] for L3 test
<!-- Author: Ryo & Pete -->


<!-- Date: Mon Nov 22 09:33:51.204 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=190041">190041</A>
TERMINATE: End L3 test with new CS_Pointer_Space file
<!-- Author: Ryo & Pete -->



<!-- Date: Mon Nov 22 09:39:50.989 2004 -->
Test of the new CS_pointer_space (with increased number of L2 and  L3 bits) : Level3 + CSL ran OK; but YMON crashed. The reason is that Consummer have not been built with the new CS_pointer_space. 
The file has to be tagged for 6.0.0int3
<!-- Author: Gilles, Tony and Veronique :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=190041' target=_top>190041</a>) -->


<!-- Date: Mon Nov 22 09:53:46.926 2004 -->
Booked cmp00,cmx00-01 and inhibits (inh00) crates. 
<!-- Author: Donatella T.(x3986) -->
<!-- Comment: Mon Nov 22 10;36;35 comment by...Donatella T. -->
temporarily released the crates.
<!-- Date: Mon Nov 22 10:01:10.250 2004 -->
After <a href= 
'/cgi/elog/elog.pl?nb=2004&action=view&page=804&frame=2&anchor=14:53:23.632'> 
Friday's eposide</a> and the fact that the end of the shutdown is quickly approching, it seems that we need some special instructions for CMX HV trips in our suspect wedge: 
<p><b> 
If CMX West wedge 15 [aka SW miniskirt 10] trips please page CMX and reset the trips (turn the wedge back on).  We at some point we will be in B0 so paging over the intercom in addition to the pager is a good idea.  We'll try this mode of operation durning the day shift for the next 2 days.  Most likely the aces will be requested to keep reseting the wedge until we're in posisition. 
</b></p> 
If the wedge goes bad well into the evening shift then treat the trip as normal: reset the trip once and if it trips again shortly afterwards turn the wedge off.
<!-- Author: Dale Stentz and Ingyin Zaw -->


<!-- Date: Mon Nov 22 10:04:47.829 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=190042">190042</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [16,466,486] 
<!-- Author: Ryo & Pete -->



<!-- Date: Mon Nov 22 10:33:28.640 2004 -->
the CSP discriminator crate is being turned off.
<!-- Author: Pete and Ryo -->


<!-- Date: Mon Nov 22 10:37:27.095 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=190043">190043</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Ryo & Pete -->


<!-- Date: Mon Nov 22 10:57:38.202 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=190045">190045</A>
ACTIVATE: RunSet: ACE_CPRTEST  Trigger: COSMICS_NOTRACKS [9,467,487] 
<!-- Author: Ryo & Pete -->



<!-- Date: Mon Nov 22 11:04:51.177 2004 -->
Sorry, CMX-ers, just accidentally turned off 
CMX HV in two SW wedges, then turned back on.  
<!-- Author: Camille Ginsburg -->
<!-- Comment: Mon Nov 22 11;26;19 comment by...Dale -->
That's ok.
<!-- Date: Mon Nov 22 11:21:27.724 2004 -->
<pre>BMU Access Report 
 
We replaced the HV distribution board in NW258 
and there is now much less noise on those four 
channels.</pre>
<!-- Author: Dan Cyr and Camille Ginsburg -->




<!-- Date: Mon Nov 22 12:03:24.018 2004 -->
Downstairs working on ccal03 and checking on a ccal05 channel.
<!-- Author: m mattson and k chung -->
<!-- Comment: Mon Nov 22 13;20;40 comment by...m mattson and k chung -->
<br>The five channels that were fried in ccal03 (slot 21, ch 0-4) had their CAFE cards replaced.  We tried checking on two channels that are reading out zero pedestals, but were not able to access the PMTs.
<!-- Date: Mon Nov 22 12:40:08.830 2004 -->
Bob Wagner Says Dale (gas tech) will be doing tests on COT, so he is taking the voltage down. Will let us know when back uo.
<!-- Author: G.P. -->


<!-- Date: Mon Nov 22 12:49:29.558 2004 -->
Taking COT crates for check-out.
<!-- Author: ambrose -->
<!-- Comment: Mon Nov 22 13;17;36 comment by...ambrose -->
SL4 cell 15 had bad TDC flat-cable plug-in, made access to fix.  The rest of the COT checks-out.
<!-- Date: Mon Nov 22 13:04:36.279 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=190048">190048</A>
ACTIVATE: RunSet: ACE_QIE calib and download, without b0muca
<!-- Author: Ryo & Pete -->
<!-- Comment: Mon Nov 22 13;14;25 comment by...m mattson and k chung -->
<br>Downloading the FRAM tables for the calorimeter ADMEM/CAFEs.  We removed the b0muca crate, as it would not go from IDLE to READY (problem with readout list??).  The results in the error log look okay without that crate.

<!-- Date: Mon Nov 22 13:17:36.008 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=190048">190048</A>
TERMINATE: no more unexpected problems
<!-- Author: Ryo & Pete -->




<!-- Date: Mon Nov 22 13:24:55.547 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=190049">190049</A>
 Activated at 2004.11.22 13:24:02
<!-- Author: RunControl -->


<!-- Date: Mon Nov 22 13:24:56.395 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=190049">190049</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [5,469,442] 
<!-- Author: Ryo & Pete -->


<!-- Date: Mon Nov 22 13:27:07.637 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=190049">190049</A>
 Terminated at 2004.11.22 13:26:12
<!-- Author: RunControl -->


<!-- Date: Mon Nov 22 13:27:08.837 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=190049">190049</A>
TERMINATE: SMXR card in slot 4 of ccal06 not responding
<!-- Author: Ryo & Pete -->


<!-- Date: Mon Nov 22 13:33:34.331 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=190050">190050</A>
 Activated at 2004.11.22 13:32:25
<!-- Author: RunControl -->


<!-- Date: Mon Nov 22 13:33:35.029 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=190050">190050</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [5,469,442] 
<!-- Author: Ryo & Pete -->
<!-- Comment: Mon Nov 22 13;55;15 comment by...m mattson and k chung -->
<br>SMXR cards in ccal06 removed for this run
<!-- Date: Mon Nov 22 13:34:02.285 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=190050">190050</A>
 Terminated at 2004.11.22 13:33:46
<!-- Author: RunControl -->


<!-- Date: Mon Nov 22 13:36:54.534 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=190050">190050</A>
TERMINATE: L1 rate 30 kHz
<!-- Author: Ryo & Pete -->
<!-- Comment: Mon Nov 22 13;48;49 comment by...m mattson and k chung -->
<br>The L1 trigger rate should normally be around 100 Hz.  The increase in rate appears to be from trigger tower HAD eta 9 phi 2.  There are two ccal channels that feed into this tower, that appear to have noisy PMTs.  The hardware (PMT) experts have been notified.


<!-- Date: Mon Nov 22 14:00:13.710 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=190053">190053</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Ryo & Pete -->



<!-- Date: Mon Nov 22 14:01:24.871 2004 -->
I just updated the XMon database with fits using runs from 184291 to 185848.  These fits were made using the new database method which uses multiple points per each run.
<!-- Author: Charles Plager -->


<!-- Date: Mon Nov 22 14:13:35.584 2004 -->
We stopped HV monitoring to turn on HV for fwd detectors.  When we turned monitoring back on, CLC slot 0 is alarming "KILL" on all channels.  We have seen this before when there is a bad connection with the cables going into the CAEN.  We didn't touch any cables or CLC HV.
<!-- Author: convery, terashi -->
<!-- Comment: Mon Nov 22 15;04;50 comment by...convery -->
it is no longer in alarm
<!-- Comment: Mon Nov 22 15;12;34 comment by...convery -->
Roberto Rossin says they had disconnected the HV cables downstairs.
<!-- Date: Mon Nov 22 14:20:10.031 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=190057">190057</A>
ACTIVATE: RunSet: ACE_CPRTEST  Trigger: COSMICS_NOTRACKS [9,467,487] 
<!-- Author: Ryo & Pete -->




<!-- Date: Mon Nov 22 15:23:44.957 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=190060">190060</A>
ACTIVATE: RunSet: ACE_CPRTEST  Trigger: TEST_CSL_MB_XING [2,541,490] 
<!-- Author: Ryo & Pete -->



<!-- Date: Mon Nov 22 15:23:44.082 2004 -->
COT back to full voltage.
<!-- Author: Stefano Zucchelli -->


<!-- Date: Mon Nov 22 15:29:15.160 2004 -->
Level 3 test under way for about half an hour.
<!-- Author: Stefano Zucchelli -->


<!-- Date: Mon Nov 22 15:34:01.853 2004 -->
I have just talked with Dan Johnson in the MCR (who in turn was 
talking to the TeV coordinator).  We will <u>not</u> go to 
controlled access tonight, so people can retain their SA keys 
at least till tomorrow.
<!-- Author: Steve Hahn -->


<!-- Date: Mon Nov 22 15:39:55.638 2004 -->
turning CMX HV to stand-by
<!-- Author: Dale and Ingyin -->

<!-- Date: Mon Nov 22 15:55:23.646 2004 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>15:55:03</TD>                                                       
<TD ALIGN=CENTER>::</TD>                                                
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Mon Nov 22 15:59:45.256 2004 -->
<font color=red>Shift Summary: </font><pre>Run Summary: 
Busy shift. 
CPR people worked all day 
Muon people worked on CMX and CMP 
COT HV brought down for gas tests. COT crates checked out. 
Lots of other last  minute checks and fixes. -- see log. 
No time for data or calibrations. 
CLC turned on for CLC calibration  
COT back to full voltage 
Level 3 code test for CSL changes under way for about half an hour</pre> 
<!-- Author: G.Pauletta -->