<!-- Title: Owl shift. Thu May 15, 2008 -->
<!-- SciCo: Dave Toback -->
<!-- DAQAce: Will Johnson -->
<!-- MonAce: &nbsp; -->
<!-- CO: Amanda Deisher -->
<!-- OpManager: Steve Hahn -->
<!-- Notes: -->
<pre>DZero is still in access. Shot setup to follow. Maybe 12:15 or 12:30?  
 
Waiting for solenoid to finish ramping... </pre>

<!-- Date: Thu May 15 00:08:11.714 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262313>262313</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL 
<!-- Author: Will -->


<!-- Date: Thu May 15 00:14:48.102 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262314>262314</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: Will -->


<!-- Date: Thu May 15 00:17:40.183 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262314>262314</A>
CALIBRATION: <U><B>pluglaser</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#pluglaser target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Thu May 15 00;20;21 comment by...Amanda -->
Empty canvases.  Rerunning.
<!-- Date: Thu May 15 00:19:44.173 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262315>262315</A>
ACTIVATE: RunSet: BOTHPLUGS_MM 
<!-- Author: Will -->



<!-- Date: Thu May 15 00:23:03.359 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262316>262316</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL Trying again, but this time with laser switched to enable
<!-- Author: Will -->



<!-- Date: Thu May 15 00:23:06.615 2008 -->
Cryo calls and tells us Solenoid is ready to go. They also tell us we can silence the annoying pinging that's happening. We do so... 
<!-- Author: Toback -->


<!-- Date: Thu May 15 00:28:27.815 2008 -->
Trigmon has died twice during calibrations.  I'll restart again, but I won't get too alarmed until we start taking cosmics.
<!-- Author: Amanda :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262316' target=_top>262316</a>) -->

<!-- Date: Thu May 15 00:30:10.901 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262317>262317</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: Will -->


<!-- Date: Thu May 15 00:32:36.907 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262317>262317</A>
CALIBRATION: <U><B>pluglaser</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#pluglaser target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Thu May 15 00;56;58 comment by...Amanda -->
Looks okay.  The plot is a little difference from the reference and is shown below.
<!-- Date: Thu May 15 00:33:48.252 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262318>262318</A>
ACTIVATE: RunSet: BOTHPLUGS_MM 
<!-- Author: Will -->



<!-- Date: Thu May 15 00:35:39.547 2008 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2008&action=view&page=-2386&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-2386  width='300' align='top'></a></td><td>The plug laser calibration still shows more structure in west wedges 17-20 than the reference plot.  I'll email the experts.</td></tr></table>
<!-- Author: Amanda  -->

<!-- Date: Thu May 15 00:40:33.462 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262319>262319</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: Will -->


<!-- Date: Thu May 15 00:40:34.814 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262319>262319</A>
CALIBRATION: <U><B>cemled</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cemled target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Thu May 15 00;50;49 comment by...Amanda -->
Still looks good.
<!-- Date: Thu May 15 00:42:34.931 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262320>262320</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Will -->


<!-- Date: Thu May 15 00:42:36.585 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262320>262320</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Thu May 15 00;49;26 comment by...Amanda -->
Looks good
<!-- Date: Thu May 15 00:45:08.226 2008 -->
MCR call. They'll start now... 
<!-- Author: Toback -->

<!-- Date: Thu May 15 00:47:17.020 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262321>262321</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Will -->




<!-- Date: Thu May 15 00:52:35.072 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262322>262322</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: Will -->


<!-- Date: Thu May 15 00:52:36.506 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262322>262322</A>
CALIBRATION: <U><B>calxef</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calxef target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Thu May 15 00;56;06 comment by...Amanda -->
Matches the plots on the CO known problem page.  Wedges 3E and 5E show up as red in the noise plot (upper-right corner; this is the 2nd of the two calibration plots). 



<!-- Date: Thu May 15 00:58:03.885 2008 -->
Closing all displays and monitors.  Restarting ConCon.
<!-- Author: Amanda -->
<!-- Comment: Thu May 15 01;04;39 comment by...Amanda -->
Everything is running in cosmics mode.  TrigMon seems to be aware of the run number and is updating plots. 
<!-- Date: Thu May 15 01:00:34.374 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262323>262323</A>
 Activated at 2008.05.15 01:00:07
<!-- Author: RunControl -->


<!-- Date: Thu May 15 01:00:35.363 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262323>262323</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,1019,706] 
<!-- Author: Will -->




<!-- Date: Thu May 15 01:19:13.155 2008 -->
<pre>During shot setup 'L2 Control' has turned red in ProcMon, also while trying to open the  L2 log file, the 'slogin -l ace 192.168.0.10' command from b0l2cntr hangs. 
Paged L2, Laura is taking a look</pre>
<!-- Author: Will :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262323' target=_top>262323</a>) -->
<!-- Comment: Thu May 15 01;27;55 comment by...Laura -->
Could happen that if the decision node is busy, it could be difficult to connect to it, both for ACE and Procmon.
Try this check again later(for example when you end the run).
So far nothing wrong and to be worried about it.
<!-- Comment: Thu May 15 01;30;30 comment by...Pedro -->
The decision node is/was too busy - it might be that compressing a file caused temporarily a problem  - everything looked actually fine.  Just redo  the L2 control check. 
<!-- Comment: Thu May 15 01;33;40 comment by...Pedro -->
The log files show that the process is running correctly and responding to TED, so no worries about the ProcMon alarm.
<!-- Date: Thu May 15 01:19:49.972 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262297>262297</A>
<HTML>Run <A HREF=http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN=262297 target=_runSummary>262297</A> has not yet been checked for goodrun status; please do so.</HTML>
<!-- Author: RunControl -->
<!-- Comment: Thu May 15 01;29;39 comment by...Toback -->
Not sure why this wasn't done before. We note that the SVT was marked bad by the CO, but later notes indicate that it was fine. The GoodRun bits indicated that it's bad but I'm guessing that's just an .AND. of all of the bits throughout running. Since the notes say it's good I marked it good. 
<!-- Date: Thu May 15 01:29:25.490 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262323>262323</A>
 Terminated at 2008.05.15 01:29:15
<!-- Author: RunControl -->




<!-- Date: Thu May 15 01:30:09.642 2008 -->
Switching monitors to physics mode.
<!-- Author: Amanda :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262323' target=_top>262323</a>) -->


<!-- Date: Thu May 15 01:30:18.670 2008 -->
Looks like we're starting to inject final protons
<!-- Author: Toback :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262323' target=_top>262323</a>) -->



<!-- Date: Thu May 15 01:34:37.373 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262324>262324</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_5_01 [5,1037,719] 
<!-- Author: Will -->


<!-- Date: Thu May 15 01:37:25.707 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262325>262325</A>
ACTIVATE: RunSet: COLDSTART  Trigger: COSMICS [15,954,639] 
<!-- Author: Will -->



<!-- Date: Thu May 15 01:38:59.406 2008 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2008&action=view&page=-2388&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-2388  width='300' align='top'></a></td><td>During loading protons, there was a tev beam abort, no pinkies, coldstart fine</td></tr></table>
<!-- Author: Will -->

<!-- Date: Thu May 15 01:40:52.934 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262326>262326</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_5_01 [5,1037,719] 
<!-- Author: Will -->



<!-- Date: Thu May 15 01:53:48.467 2008 -->
After losing the protons a couple of times it now looks like we're done injecting final protons. Now injecting anti-protons. We page silicon
<!-- Author: Toback :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262326' target=_top>262326</a>) -->
<!-- Comment: Thu May 15 01;54;23 comment by...Toback -->
And Ops

<!-- Comment: Thu May 15 01;55;30 comment by...Toback -->
Miguel (Silicon expert) is coming in for the shot... 

<!-- Comment: Thu May 15 01;59;06 comment by...Amanda -->
Ops is monitoring from home.


<!-- Date: Thu May 15 02:15:56.774 2008 -->
Anti-protons are loaded and they are squeezing... 
<!-- Author: Toback -->


<!-- Date: Thu May 15 02:29:47.330 2008 -->
We call MCR and they tell us they just finished scraping. 
<!-- Author: Toback :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262327' target=_top>262327</a>) -->
<!-- Comment: Thu May 15 02;42;00 comment by...Toback -->
Store #6141
<!-- Date: Thu May 15 02:31:14.688 2008 -->
CLC is up so we tell MCR
<!-- Author: Toback :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262327' target=_top>262327</a>) -->

<!-- Date: Thu May 15 02:35:51.057 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262327>262327</A>
 Activated at 2008.05.15 02:35:23
<!-- Author: RunControl -->


<!-- Date: Thu May 15 02:35:52.066 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262327>262327</A>
ACTIVATE: RunSet: AAA_CURRENT  Trigger: PHYSICS_5_01 [5,1037,719] 
<!-- Author: Will -->



<!-- Date: Thu May 15 02:36:02.401 2008 -->
HV is up and we're off... 
<!-- Author: Toback :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262327' target=_top>262327</a>) -->
<!-- Comment: Thu May 15 02;41;06 comment by...Toback -->
Store 6141 Initial lum is 235E30... 



<!-- Date: Thu May 15 02:44:29.914 2008 -->
Deadtime is at around 8%. Mostly coming from L2 Readout
<!-- Author: Toback :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262327' target=_top>262327</a>) -->


<!-- Date: Thu May 15 02:45:35.680 2008 -->
<center><a href='/cgi/elog/elog.pl?nb=2008&action=view&page=-2389&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2008&action=view&page=-2389' width='300'></a><a href='/cgi/elog/elog.pl?nb=2008&action=view&page=-2390&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2008&action=view&page=-2390' width='300'></a></center>
<!-- Author: Will -->
<!-- Comment: Thu May 15 03;47;21 comment by...Miguel M / Tino C -->
After scraping losses were stable and low < 9 KHz. Therefore, we let the  Silicon detector to be switched on in coincidence with Step 2. Silicon is currently taking good data.
<!-- Date: Thu May 15 02:51:02.132 2008 -->
L2 decision TO 
l2ctrlb in RC is yellow, L2 log file give error "... Packet Error | XFT02...]... Log file has become inaccessible. 
Paged L2  
Mor details shortly  
<!-- Author: Will :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262327' target=_top>262327</a>) -->

<!-- Date: Thu May 15 02:52:44.976 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262327>262327</A>
 Terminated at 2008.05.15 02:52:35
<!-- Author: RunControl -->
<!-- Comment: Thu May 15 03;00;08 comment by...Laura -->
Need to start a new run because the executable crashed.
<!-- Comment: Thu May 15 03;23;13 comment by...Will -->
Problem fixed
<!-- Date: Thu May 15 02:59:23.583 2008 -->
PEM, PHA and PSH have HV trips and turned off
<!-- Author: Toback :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262328' target=_top>262328</a>) -->
<!-- Comment: Thu May 15 03;02;52 comment by...Toback -->
HV now back up for all three.
<!-- Comment: Thu May 15 03;04;22 comment by...Will -->
Was crate 10 that tripped, pushing yellow button and restarting plug HV monitoring fixed
<!-- Date: Thu May 15 03:01:12.449 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262328>262328</A>
 Activated at 2008.05.15 03:00:48
<!-- Author: RunControl -->


<!-- Date: Thu May 15 03:01:13.511 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262328>262328</A>
ACTIVATE: RunSet: AAA_CURRENT  Trigger: PHYSICS_5_01 [5,1037,719] 
<!-- Author: Will -->





<!-- Date: Thu May 15 03:05:06.221 2008 -->
We note that the deadtime is now less than 2%. Was around 8 before L2 crashed... 
<!-- Author: Toback :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262328' target=_top>262328</a>) -->
<!-- Comment: Thu May 15 03;06;45 comment by...Will -->
Dead time back up to about 4.8%
<!-- Date: Thu May 15 03:05:37.268 2008 -->
During "Activate" b0cot14 turned red in RC, shepherded
<!-- Author: Will :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262328' target=_top>262328</a>) -->


<!-- Date: Thu May 15 03:18:15.725 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262327>262327</A>
si: <tt>Two ISL ladders are yellow, but this was a very low statistics run.</tt>
<!-- Author: Consumer Operator(Amanda Deisher) -->


<!-- Date: Thu May 15 03:18:16.012 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262327>262327</A>
general: <tt>For a low statistics run, it looks okay.</tt>
<!-- Author: Consumer Operator(Amanda Deisher) -->




<!-- Date: Thu May 15 03:30:23.252 2008 -->
Done TO for crate b0cot02, auto HRR fixed
<!-- Author: Will :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262328' target=_top>262328</a>) -->


<!-- Date: Thu May 15 03:35:49.262 2008 -->
Done TO for pcal11, after auto + manual HRR recieved a 'Task Readout SUSPENDED on node b0pcal11' meesage and this crate turned red, shepherded succesfully
<!-- Author: Will :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262328' target=_top>262328</a>) -->


<!-- Date: Thu May 15 03:43:06.965 2008 -->
b0pcal08 done TO, was unresponsive, so shepherded
<!-- Author: Will :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262328' target=_top>262328</a>) -->

<!-- Date: Thu May 15 03:43:45.098 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262328>262328</A>
general: <tt>First check looks good</tt>
<!-- Author: Consumer Operator(Amanda Deisher) -->




<!-- Date: Thu May 15 04:26:35.776 2008 -->
Done TO for b0cot16, auto HRR fixed
<!-- Author: Will :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262328' target=_top>262328</a>) -->


<!-- Date: Thu May 15 04:42:42.208 2008 -->
<pre>Node b0pcal08 in Error 
Last data more than 120s ago!  Node is not responsive 
 
Shepherding fixed</pre>
<!-- Author: Will :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262328' target=_top>262328</a>) -->


<!-- Date: Thu May 15 05:20:32.951 2008 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2008&action=view&page=-2391&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-2391  width='300' align='top'></a></td><td>WHA PMT has a hot channel at eta=-11, phi=6.  Will email the experts.</td></tr></table>
<!-- Author: Amanda -->


<!-- Date: Thu May 15 05:44:43.095 2008 -->
b0pcal04 in Error, Task Readout SUSPENDED on node b0pcal04.  SHeperhded succesfully
<!-- Author: Will :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262328' target=_top>262328</a>) -->


<!-- Date: Thu May 15 06:42:53.693 2008 -->
CER_SVXMON_HALT_RECOVER_RUN_ERROR !!!  
Stuck Cellid S/B2/W4/L4/C0-6 .  
Auto + Manual HRR fixed
<!-- Author: Will :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262328' target=_top>262328</a>) -->
<!-- Comment: Thu May 15 06;52;36 comment by...Will -->
again
<!-- Date: Thu May 15 06:47:36.039 2008 -->
Done TO. for crate/s:  
    b0imu01    
Auto HRR fixes
<!-- Author: Will :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262328' target=_top>262328</a>) -->



<!-- Date: Thu May 15 07:07:31.761 2008 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2008&action=view&page=-2392&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-2392  width='300' align='top'></a></td><td>SVX pinkies for avdd, dvdd, bias and doim for all of B1W3.  Example plot. </td></tr></table>
<!-- Author: Amanda -->
<!-- Comment: Thu May 15 07;09;45 comment by...Amanda -->
HRR didn't work.  Paging silicon.
<!-- Comment: Thu May 15 07;12;00 comment by...Toback -->
Miguel calls back. He's looking at it now... 
<!-- Comment: Thu May 15 07;12;55 comment by...Amanda -->
That's 13 pinkies total SVX B1W3L(1-4).
<!-- Comment: Thu May 15 07;16;44 comment by...Will -->
Miguel had us shepherd fib00, and return to data taking
<!-- Comment: Thu May 15 07;42;43 comment by...Will -->
Miguel then had us disable SAR, and turn problem wedge off, then reinabled SAR.  Didn't help, then hockerized crate 2, and let SAR recover everything, everything is fine now.
<!-- Comment: Thu May 15 07;50;46 comment by...Miguel M -->
The currents for that wedge had dropped to zero. We tried to recover them in different ways. At the end the hockerization was successful.
<!-- Date: Thu May 15 07:07:57.859 2008 -->
Node b0pcal02 in Error 
Last data more than 120s ago!  Node is not responsive 
 
Shepherding was  succesful
<!-- Author: Will :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262328' target=_top>262328</a>) -->
<!-- Comment: Thu May 15 07;10;43 comment by...Will -->
again, shepherded


<!-- Date: Thu May 15 07:11:35.299 2008 -->
Cryo calls Mark Knapp will update the PC CDFCRYO S5. This covers the alcohol purifier and other computers. Will take about 20 minutes. Then they'll do the others. This should just show up as ??? in the monitoring. 
<!-- Author: Toback :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262328' target=_top>262328</a>) -->







<!-- Date: Thu May 15 07:52:15.452 2008 -->
Cryo calls. THey're done with the updates to S2. All went well. They're doing S3 now. 
<!-- Author: Toback :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=262328' target=_top>262328</a>) -->

<!-- Date: Thu May 15 07:56:27.533 2008 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>PhysicsTable</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L3Accepts</TH>
<TH>DelivLumi nb<SUP>-1</SUP></TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN=262327 target=_runSummary>262327</A> x400B7</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_5_01 [5,1037,719]</FONT></TD>
<TD ALIGN=CENTER>02:35:23</TD>
<TD ALIGN=CENTER>02:52:35</TD>
<TD ALIGN=CENTER>00:09:23</TD>
<TD ALIGN=RIGHT>      90,327</TD>   <TD ALIGN=RIGHT>    229.39</TD>
<TD ALIGN=RIGHT>    126.15</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN=262328 target=_runSummary>262328</A> x400B8</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_5_01 [5,1037,719]</FONT></TD>
<TD ALIGN=CENTER>03:00:48</TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>04:13:33</TD>
<TD ALIGN=RIGHT>   2,229,054</TD>   <TD ALIGN=RIGHT>  2,602.66</TD>
<TD ALIGN=RIGHT>  2,256.72</TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>07:55:01</TD>
<TD ALIGN=CENTER>04:22:56</TD>
<TD ALIGN=RIGHT>     2,319,381</TD> <TD ALIGN=RIGHT>  2,832.05</TD>
<TD ALIGN=RIGHT>  2,382.87</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Thu May 15 07:59:38.126 2008 -->
<font color=red>Shift Summary: </font><pre>Started with DZero in access and waiting for Solenoid to finish ramping.  
 
Shot setup starts at 12:45. Store 6141 finished scraping at 2:30 and came in with an initial lum of 235E30 
 
Things started smoothly, but right before L2 executable crashed we had a deadtime of almost 8%. After the fix it went back to 2%, then back up to 5%.  
 
Crate 10 tripped.  
 
Full wedge of SVX went down with Pinkies.  
 
Cryo is doing PC windows updates. Will finish this morning</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   262327,262328
Delivered Luminosity   2977  
Acquired Luminosity    2396  
Efficiency             80.5
   
</td></tr></table></pre>
<!-- Author: Toback -->