<!-- Title: Day shift. Wed Sep 19, 2001 -->
<!-- SciCo: Morris Binkley -->
<!-- DAQAce: Eric Brubaker -->
<!-- MonAce: MooHyun Ahn -->
<!-- CO: Jonatan Piedra -->
<!-- OpManager: Jay Dittmann -->
<!-- Notes: -->
<pre>Store 706 is to be dropped momentarily.  Today will be a day of Tev Studies.</pre>


<!-- Date: Wed Sep 19 08:51:40 2001 -->
COT Cooling Alarm on PDT-SEP. 
It disappeared a few minuates later.
<!-- Author: MooHyun -->


<!-- Date: Wed Sep 19 08:55:53 2001 -->
MCR dropped store at about 8:50.
<!-- Author: Erik -->


<!-- Date: Wed Sep 19 09:40:04 2001 -->
COT Cooling Alarm again. 
PDT-SEP value dropped to 3.8 (low linit is 5.00).
<!-- Author: MooHyun -->


<!-- Date: Wed Sep 19 09:44:24 2001 -->
<table><tr><td><a href=/elog/notebooks/2001/Wed_Sep_19_2001_Day.094359.ps align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-2297  align='top'></td><td>Hot channel in the PEM Wedge Occupancy plot?</a></td></tr></table>
<!-- Author: Jonatan -->
<!-- Comment: Wed Sep 19 10;21;48 comment by...greg V. -->
<pre>You should also try to paste the pmt plots.
It looks like it might be a hot pmt...
I will post the pmt plot shortly</pre>
<!-- Date: Wed Sep 19 09:48:53 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-2298&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-2298  width='300' align='top'></td><td>Luminosities during store 706</a></td></tr></table>
<!-- Author: MooHyun -->


<!-- Date: Wed Sep 19 09:49:37 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-2299&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-2299  width='300' align='top'></td><td>Losses during store 706</a></td></tr></table>
<!-- Author: MooHyun -->


<!-- Date: Wed Sep 19 10:10:51 2001 -->
Temporarily abandoning L2 work to allow other tests to happen.  The results for this morning's work begin <a href= 
'/cgi/elog/det-system-elog.pl?nb=l2&action=view&page=22&frame=2&anchor=10:05:24'>here</a> in the L2 e-log.  A bug in the tracklist firmware has been fixed.
<!-- Author: Matt -->



<!-- Date: Wed Sep 19 10:29:12 2001 -->
<table><tr><td><a href=/elog/notebooks/2001/Wed_Sep_19_2001_Day.102858.ps align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-2303  align='top'></td><td><PRE>PEM Tower plot which shows the possibly hot pmt East, Eta=13, Phi=15, pmt=0 Wow, it's sure easier to read this plot with the new thick lines! </PRE></a></td></tr></table>
<!-- Author: Greg V. -->


<!-- Date: Wed Sep 19 10:41:56 2001 -->
COT Cooling Alarm. 
PDT-SEP dropped to 0.4 (low limit is 5.00).
<!-- Author: MooHyun -->
<!-- Comment: Wed Sep 19 10;45;06 comment by...MooHyun -->
Cryo siad they know and work on it.


<!-- Date: Wed Sep 19 10:45:54 2001 -->
CALIBRATION/QIEPED/CONSUMER 
 
Attempted to test new consumer compiled with  
version 4.0.0 of the offline, but there was a  
problem getting the consumer to start. Need  
to consult with Bill Badgett.
<!-- Author: Jack Cranshaw -->


<!-- Date: Wed Sep 19 11:24:15 2001 -->
CLC LED gain calirbrations are done now. Run numbers: 
127099 
127100 
127101 
127102 
127103 
We rebooted the CLC crate after calibrations(11.20a.m.)
<!-- Author: Alex Pronko :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=127009' target=_top>127009</a>) -->


<!-- Date: Wed Sep 19 11:27:38 2001 -->
tested (successfully) enable and disabling of LED inhibit for 
daq runs.  
run numbers: 1f088 1f089 (hex of course) 
Was unable to test qie calibrations because of external issues. 
Will test that this afternoon.  
<!-- Author: Steve Vejcik :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=0x1f088' target=_top>0x1f088</a>) -->


<!-- Date: Wed Sep 19 11:34:04 2001 -->
<pre>CMU HV scan continues... 
Anode HV is set at 2.375kV. Will 
leave at this setting till we accumulate 
enough data.</pre>
<!-- Author: hyunsoo -->


<!-- Date: Wed Sep 19 11:48:38 2001 -->
CLC Alarm twicce, but it cleared by itself.
<!-- Author: MooHyun -->


<!-- Date: Wed Sep 19 12:30:27 2001 -->
CSL problem. This is the third tiem we've seen this problem 
since the csl code was "improved" yseterday. We went 
back to the code that existed before the improvement 
and will investigate the problem with the new code.  
<!-- Author: Tony , Hiro -->


<!-- Date: Wed Sep 19 12:47:45 2001 -->
return crosspoint crate stopped replying ... turns out that 
both rc front-end tasks (Messenger and Monitor) are suspended - 
I have never seen this before. Rebooting.  
<!-- Author: Arnd -->


<!-- Date: Wed Sep 19 12:58:48 2001 -->
Nikolay Luzhetskiy put <a href= 
'/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=67&frame=2&anchor=11:26:59'> 
this</a> in the Silicon logbook by accident. It says 
a crate controller for HV monitoring in Rack 1RR06C was 
exchanged.  
<!-- Author: SCN -->


<!-- Date: Wed Sep 19 13:13:09 2001 -->
Got an SVX Integrated Dose Alarm. 
Now currnet svx dose rates are all zeros.
<!-- Author: MooHyun -->
<!-- Comment: Wed Sep 19 14;30;03 comment by...MooHyun -->
<pre>Recardo check it.
Instantaneous Dose Rate was high, but it lasted just a few tens of seconds. So it trigged alarm, but not make it trip.
Now it's OK. </pre>
<!-- Comment: Wed Sep 19 14;41;14 comment by...MooHyun -->
Recardo reset SVRAD's.
<!-- Date: Wed Sep 19 13:19:38 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-2304&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-2304  width='300' align='top'></td><td>SVRAD's for last 1 hour.</a></td></tr></table>
<!-- Author: MooHyun -->


<!-- Date: Wed Sep 19 13:48:17 2001 -->
<pre>Traced the EVB+IMU00 problem to a TDC. Replaced with another TDC and tested with EVB. At this point the muon system is completely instrumented with TDCs! 
</pre>
<!-- Author: dan -->
<!-- Comment: Wed Sep 19 13;51;28 comment by...ronmoore -->
<pre>Woo-hoo!</pre>


<!-- Date: Wed Sep 19 14:12:17 2001 -->
Running L3 test for Jonas. Full system, using cosmics_notracks[3,185]. Silicon at standby. All files going to look area. Get the following error: 
<pre> 
Partition:0 Run number: 127131 (0x1f09b) 
(MLE) b0l3gate2.fnal.gov:main:2:11:07 PM->Error on L3 node b0l3gate2 (partition 0) Wed Sep 19 14:11:06 2001 l3proxy  
  in Config (L3ProxyPartition.cc:1319) 
  @L3_PACKAGE_FAIL - Failed to download TCL package on a gateway. Checksum or size is incorrect 
 
(RC)   14:11:13 Config -> ERROR</pre>
<!-- Author: Erik :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=127131' target=_top>127131</a>) -->
<!-- Comment: Wed Sep 19 14;14;29 comment by...Erik -->
On abort, b0xft02 did not respond. Cannot ping it. Issuing reset_crate.

<!-- Comment: Wed Sep 19 14;15;13 comment by...Erik -->
XFT/XTRP removed from run for now.

<!-- Comment: Wed Sep 19 14;19;42 comment by...Erik -->
Same problem from L3 on the second try. In each case, l3m takes a long time in coldStart--on order minutes. Ending these L3 tests.


<!-- Comment: Wed Sep 19 14;35;29 comment by...Erik -->
Ilya says the L3 problem is with tcl size. RC gave 2868 bytes, but L3 sees 37350 bytes. Since this size check failed, L3 threw an error.
<!-- Date: Wed Sep 19 14:23:09 2001 -->
<pre>Finished work on cot pulser crate, readout of scanning ADC. 
</pre>
<!-- Author: Arnd -->


<!-- Date: Wed Sep 19 14:23:26 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-2305&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-2305  width='300' align='top'></td><td>Losses from 12 to 13:15</a></td></tr></table>
<!-- Author: MooHyun -->


<!-- Date: Wed Sep 19 14:27:19 2001 -->
PHA Alarm on West 21 Wedge. 
It cleared by itself a few minuates later
<!-- Author: MooHyun -->





<!-- Date: Wed Sep 19 14:44:27 2001 -->
Doing tests for evb. Set zero bias prescale down to 6001 to manufacture rate.
<!-- Author: Erik :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=127134' target=_top>127134</a>) -->
<!-- Comment: Wed Sep 19 14;55;32 comment by...Erik -->
We've run with silicon and COT for 90k events at 120 Hz (~22 MB/s = ~200 kB/event). No evb problems thus far. Certainly we have passed the time scale for 12 Hz problems in the silicon VRBs that we experienced last Friday after a change in the EVB code.


<!-- Date: Wed Sep 19 15:06:06 2001 -->
Updated run control / fer with Elliott's changes to the 
statemanager classes and ack window. Also includes a new status 
display line from Bill which will not yet work correctly.  
<!-- Author: Arnd -->
<!-- Comment: Wed Sep 19 18;11;28 comment by...W.Badgett -->
<pre>That is, the "TS Status" line on the main Run Control 
window does not current have the correct info.</pre>
<!-- Date: Wed Sep 19 15:21:22 2001 -->
Trying now to blow up silicon event size by using a silicon calibration run configuration which has silicon in read all mode. We include also CCAL and COT, not knowing if this will conflict with some other settings in this configuration.
<!-- Author: Erik -->
<!-- Comment: Wed Sep 19 15;28;12 comment by...Erik -->
For whatever reason, b0ccal08 does not respond to ColdStart. I am removing all COT and CCAL crates since they are no longer essential and I'll debug b0ccal08 presently.

<!-- Comment: Wed Sep 19 15;43;27 comment by...Erik -->
Then L3 failed partition. Ilya is looking into it and cleaning up L3.

<!-- Comment: Wed Sep 19 15;45;36 comment by...Erik -->
Rick came and started the script that reboots all calorimeter crates. It could be that the config command sent by this SVX calibration run configuration screwed up the ccal crates. I'll also reboot the COT crates to be sure.


<!-- Date: Wed Sep 19 15:51:28 2001 -->
<pre>  L3 failed partition and was found dead because output 
node u01 was hung. I had to push reset button and  
clean up L3 afterwards.</pre>
<!-- Author: Ilya -->


<!-- Date: Wed Sep 19 16:02:08 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-2306&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-2306  width='300' align='top'></td><td>TeV Status during shift. (ERING, IBEAM, LOSTP, B0Q5)</a></td></tr></table>
<!-- Author: MooHyun -->


<!-- Date: Wed Sep 19 16:02:52 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-2307&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-2307  width='300' align='top'></td><td>Losses during Shift</a></td></tr></table>
<!-- Author: MooHyun -->


<!-- Date: Wed Sep 19 16:04:23 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-2309&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-2309  width='300' align='top'></td><td>SVRAD's during shift. Reset after AVX integrated dose alarm by expert.</a></td></tr></table>
<!-- Author: MooHyun -->


<!-- Date: Wed Sep 19 16:13:14 2001 -->
<pre>   We are finishing tests of the new SCPU code for the 
Event Builder. This new version from Steve Tether is supposed 
to cure all kinds of problems incl. 12Hz problem and  
suspended task problem. 
  We have done tests with big system, high rate, 
    + very large COT events 
    + also with Silicon in read-all mode (650k events) 
No problems observed so far. 
 
  Please record any cases when EVB  reports a suspended task 
or if 12Hz problem occurs. Also, please write into e-log 
when HEVB is cleaned up and why. Page L3 pager if abnormal 
failure rate of EVB is seen. 
</pre>
<!-- Author: Ilya and Steve Tether -->


<!-- Date: Wed Sep 19 16:23:48 2001 -->
<font color=red>Shift Summary: </font><pre>Shift started with the end of store 706.  Tevatron studies lasted all 
day. 
 
 
About 1000 hrs Matt quit L2 work.  A bug in tracklist firmware has 
been fixed. 
 
 
CLC did calibrations. 
 
 
Steve Vikec worked some more on the test of led pulsing to stabilize 
the gain.  He had QIE problems which need to be fixed before he 
finishes.  He will be back tomorrow. 
 
 
Nikolay replaced camac controller on calorimetry HV monitor crate.   
 
 
Dan checked all muon TDC crates.  Traced the EVB+IMU00 problem to a 
TDC.  The TDC was replaced and the problem appears fixed. 
 
 
Nathan did some more L2 torture tests.  
 
 
Between 1200 and 1300 hrs there were some collimator studies in the 
tevatron.  The instantaneous rate was not high enough to cause a trip, 
but each incident lasted many seconds and the total integrated dose 
for all incidents was large enough to cause an alarm.  The silicon was   
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   
Delivered Luminosity   0  
Acquired Luminosity    0  
Efficiency             
   
</td></tr></table></pre>
<!-- Author: Morris Binkley -->
<!-- Comment: Wed Sep 19 17;50;07 comment by...KPitts -->
Part of Morris' summary was truncated...here is the final part:<br>

Between 1200 and 1300 hrs there were some collimator studies in the
tevatron.  The instantaneous rate was not high enough to cause a trip,
but each incident lasted many seconds and the total integrated dose
for all incidents was large enough to cause an alarm.  The silicon was
"off" at the time.  We reported to MCR with a request that they inform
us when doing colimator studies near B0. They also promised to be more
careful.  This may require some followup.<br>


We ran a test of new L3 code from Jonas using the full system with
silicon at standby (this was after the incidents described above).
The new code was supposed to pick up SVX calibrations.  This was not
initially successful.<br>


At about 1430 we started test of new EB code for Steve Tether.  We did
this with SVX at standby and COT at varying low thresholds to generate
lots of data.  The tests appeared successful and the new code will
become the standard code.  Page EB pager if there is a problem.
<br>


Tests finished on COT pulse crate and it is now back in the system.