<!-- Title: Day shift. Sun Sep 30, 2007 -->
<!-- SciCo: Marcel Stanitzki -->
<!-- DAQAce: Glenn Strycker -->
<!-- MonAce: &nbsp; -->
<!-- CO: &nbsp; -->
<!-- OpManager: Steve Hahn -->
<!-- Notes: -->
<pre>Ran cosmics overnight ... 
b0inh00 still not working 
Power supply of b0xft02 toasted 
 
We are waiting for the things to come</pre>


<!-- Date: Sun Sep 30 08:18:13.575 2007 -->
<pre>Short Status Summary: 
------------------------- 
 
- currently running COSMICS 
  - L1A rate: 27.4 Hz (normal) 
  - L2  rate: 27.4 Hz 
  - deadtime: 0% (normal) 
 
- b0inh00 is still down (library malfunction) 
- b0xft02 is still down (power supply problem) 
- cooling rack problem is still present (alarm turned off!) 
 
- HV and Global Alarm status: 
  - ON:   CEM/CHA/WHA, GAM (HV says "NOT") 
  - "ON" but no HV: CPR, CLC, MNP, BSC 
  - STD:  PEM/PHA/PSH 
  - OFF:  everything else 
  - WARN: none yellow or red 
  - all heartbeat are ok (green)! 
  - no "pink" or "grey" heartbeats, including COT and PSM! 
 
- Process System Alarms: 
  - status seems to be consistant with known problems, except the rack 
  - yellow: GAS, COT, SILICON, RACKS (new) 
  - green: CRYO, HVAC, WATER, PSM, SOLENOID 
 
- Racks not in use 
  - b0inh00, b0xft02 
  - all other racks in the L2_TORTURE [19,952,639] mode are presumably working 
  - all other racks in COSMICS mode [15,954,639] are presumably working 
 
- Known issues 
  - in TOFQIE calib, WCAL04 is still out 
  - in CEMLED, CALXEF, and TOFTAC calibs, wedges 9-11West are empty 
 
- Shift plan 
  - stop Cosmic run 
  - run L2_TORTURE with all possible crates (see which ones are really out) 
  - run all calibrations 
  - end shift with a nice, quiet COSMICS run</pre>
<!-- Author: Glenn Strycker (Ace) -->


<!-- Date: Sun Sep 30 08:58:28.266 2007 -->
We're still alive and awake.
<!-- Author: G.S. -->

<!-- Date: Sun Sep 30 09:15:23.271 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250472>250472</A>
 Terminated at 2007.09.30 09:15:09
<!-- Author: RunControl -->


<!-- Date: Sun Sep 30 09:15:54.405 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250472>250472</A>
TERMINATE: Good overnight run.  Let's try L2_TORTURE now, just to keep us all on our toes.
<!-- Author: Glenn Strycker -->


<!-- Date: Sun Sep 30 09:19:24.173 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250473>250473</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [19,952,639] -- Normal partition settings (b0inh00 removed by default) and b0xft02/XFT_FINDER_02 removed.
<!-- Author: Glenn Strycker -->
<!-- Comment: Sun Sep 30 09;19;51 comment by...G.S. -->
One L2TO -- Manual HRR recovered.

<!-- Comment: Sun Sep 30 09;30;01 comment by...G.S. -->
L2 Decision TO -- Auto HRR recovered.

<!-- Comment: Sun Sep 30 10;11;43 comment by...G.S. -->
L2 Decision TO -- Auto HRR recovered.

<!-- Comment: Sun Sep 30 10;28;03 comment by...G.S. -->
Another one.

<!-- Comment: Sun Sep 30 10;36;33 comment by...G.S. -->
Again.  Perhaps L2 experts should know of this?  These TOs are getting frequent.  But there are a few systems down, so let's not bother anyone on a Sunday.


<!-- Date: Sun Sep 30 10:50:52.434 2007 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2007&action=view&page=-6517&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2007&action=view&page=-6517  width='300' align='top'></a></td><td>L2 Done TO seems to come from Muon system ...</td></tr></table>
<!-- Author: Marcel -->
<!-- Comment: Sun Sep 30 13;23;00 comment by...Laura -->
There are 2 types of TO:
1. caused by RECES system 
2. caused by L1/XTRP/MUON system (pulsar in l2 crate)   

The second one is the same we had last night and few days before.
Kevin already told me is not a XTRP problems.
Please, give a look to Level2_pulsar_b_muon canvas plot in trigmon 
<!-- Date: Sun Sep 30 11:11:15.111 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250473>250473</A>
TERMINATE: ...time to do calibrations.
<!-- Author: Glenn Strycker -->


<!-- Date: Sun Sep 30 11:15:50.129 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250474>250474</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Glenn Strycker -->


<!-- Date: Sun Sep 30 11:18:36.533 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250474>250474</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep 30 11;28;00 comment by...Glenn Strycker -->
TECHNICALLY ok, but I think the bad channels are just pushed off the list.  See the last two calibrations -- Carolina didn't see anything bad, but when I last calibrated there were a few bad CEM channels at the end of the list.
<!-- Date: Sun Sep 30 11:18:37.612 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250475>250475</A>
ACTIVATE: RunSet: CALIB_CLC_QIE 
<!-- Author: Glenn Strycker -->


<!-- Date: Sun Sep 30 11:20:12.472 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250475>250475</A>
CALIBRATION: <U><B>clcqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#clcqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep 30 11;28;32 comment by...G.S. -->
OK
<!-- Date: Sun Sep 30 11:21:39.917 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250476>250476</A>
ACTIVATE: RunSet: CALIB_QIE_FWD 
<!-- Author: Glenn Strycker -->


<!-- Date: Sun Sep 30 11:21:40.374 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250476>250476</A>
CALIBRATION: <U><B>bscqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#bscqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep 30 11;29;43 comment by...G.S. -->
ok.  A little change in the colors on the first plot.
<!-- Date: Sun Sep 30 11:28:25.676 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250477>250477</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL 
<!-- Author: Glenn Strycker -->





<!-- Date: Sun Sep 30 11:34:05.344 2007 -->
Exciting news ... some visitor tour on the first floor
<!-- Author: Marcel :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250477' target=_top>250477</a>) -->

<!-- Date: Sun Sep 30 11:45:17.338 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250478>250478</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: Glenn Strycker -->


<!-- Date: Sun Sep 30 11:45:42.920 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250478>250478</A>
CALIBRATION: <U><B>pluglaser</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#pluglaser target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep 30 11;58;41 comment by...Glenn Strycker -->
BAD -- The last plot has a mean of 582.4 nm, which is outside of the 550-580 nm window.  This value is becoming worse compared to previous calibrations.  There is also a hot pixel on the first plot at Wedge 8 Tower 21E that I mentioned in the elog yesterday.
<!-- Date: Sun Sep 30 11:47:45.829 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250479>250479</A>
ACTIVATE: RunSet: BOTHPLUGS_MM 
<!-- Author: Glenn Strycker -->




<!-- Date: Sun Sep 30 11:59:56.671 2007 -->
We must skip the PMT Spike Run calibration because it has the INHIBIT_00 crate included in the partition.  I'll try to remove it, but I don't think that it will work.  No big deal -- this crate only has a software glitch that is to be fixed tomorrow.
<!-- Author: Glenn Strycker -->

<!-- Date: Sun Sep 30 12:02:04.252 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250480>250480</A>
 Activated at 2007.09.30 12:01:51
<!-- Author: RunControl -->


<!-- Date: Sun Sep 30 12:04:37.806 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250480>250480</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [7,957,639] 
<!-- Author: Glenn Strycker -->


<!-- Date: Sun Sep 30 12:08:31.365 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250480>250480</A>
 Terminated at 2007.09.30 12:08:11
<!-- Author: RunControl -->
<!-- Comment: Sun Sep 30 12;10;04 comment by...Glenn Strycker -->
Well, it ran, but that sure wasn't 13 minutes long.  I'm skeptical of the validity of this calibration, especially with the INHIBIT turned off.


<!-- Comment: Sun Sep 30 12;12;37 comment by...Glenn Strycker -->
L1A = 71.7 Hz.  Still within the recommended 40-100 Hz, but significantly higher than previous recent calibrations.
<!-- Date: Sun Sep 30 12:11:31.756 2007 -->
We had a silicon cooling audible alarm, but didn't see anything bad on the Global Alarm window or the Silcon monitors.
<!-- Author: Glenn Strycker -->
<!-- Comment: Sun Sep 30 12;15;37 comment by...G.S. -->
Marcel is going to page his old Silicon gang and to give them a heads-up...
<!-- Comment: Sun Sep 30 12;27;48 comment by...Marcel -->
Jennifer (Silicon Cooling) and Marcel couldn't find the source of the Alarm and the alarm only sounded once...<br>
We'll keep an eye on it and see if it happens again 

<!-- Comment: Sun Sep 30 12;42;26 comment by...Ulrich -->
This is probably a glitch of the process systems alarms page that happens once in a while. If nothing turns red, it's usually not a problem.
<!-- Date: Sun Sep 30 12:14:26.107 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250481>250481</A>
ACTIVATE: RunSet: ACE_TOF_QIE_NEW 
<!-- Author: Glenn Strycker -->


<!-- Date: Sun Sep 30 12:15:37.200 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250481>250481</A>
CALIBRATION: <U><B>tofqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#tofqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep 30 12;38;28 comment by...Glenn Strycker -->
Bad.  Usual missing channel
<!-- Date: Sun Sep 30 12:19:47.574 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250482>250482</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Glenn Strycker -->


<!-- Date: Sun Sep 30 12:21:57.617 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250482>250482</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep 30 12;39;02 comment by...Glenn Strycker -->
very bad -- looks like it did yesterday
<!-- Date: Sun Sep 30 12:21:58.877 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250483>250483</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: Glenn Strycker -->


<!-- Date: Sun Sep 30 12:22:52.140 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250483>250483</A>
CALIBRATION: <U><B>cemled</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cemled target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep 30 12;39;32 comment by...Glenn Strycker -->
BAD -- same missing as usual
<!-- Date: Sun Sep 30 12:24:57.849 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250484>250484</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Glenn Strycker -->


<!-- Date: Sun Sep 30 12:24:58.857 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250484>250484</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep 30 12;40;20 comment by...Glenn Strycker -->
Ooh, very bad.  Eiko beat me to checking it, so she's already looking at what went wrong (see elog entry below)
<!-- Date: Sun Sep 30 12:29:57.525 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250485>250485</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: Glenn Strycker -->


<!-- Date: Sun Sep 30 12:31:17.948 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250485>250485</A>
CALIBRATION: <U><B>calxef</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calxef target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep 30 12;40;58 comment by...Glenn Strycker -->
BAD -- same problems as usual.
<!-- Date: Sun Sep 30 12:33:43.877 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250486>250486</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Glenn Strycker -->



<!-- Date: Sun Sep 30 12:34:46.539 2007 -->
Eiko (COT) called about bad channels in the COT calibration. She'll have a look and may call us back later 
<!-- Author: Marcel :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250486' target=_top>250486</a>) -->

<!-- Date: Sun Sep 30 12:36:52.130 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250487>250487</A>
ACTIVATE: RunSet: TOF_TAC_CALIB 
<!-- Author: Glenn Strycker -->









<!-- Date: Sun Sep 30 12:50:41.074 2007 -->
<pre>TOFTAC calib has ended... the "terminate" elog entry window popped up and everything... but now RunControl is hanging during the "end" step.  SEVB is still active and isn't resetting.  Marcel tried clicking "end" manually for the sevb crate, and it didn't seem to respond.  The TOFTAC button on the CalibConsumers window also is remaining green, not cyan, as if the calibration is still processing.  We hit "stop" for TOFTAC in the CalibConsumer window, and that didn't stop it either.  Finally we're trying "abort"... sevb is really stuck!</pre>
<!-- Author: Glenn Strycker -->
<!-- Comment: Sun Sep 30 12;52;51 comment by...G.S. -->
We hit reset on RunControl and re-partitioned to see if sevb would come back to green.  It did not.

<!-- Comment: Sun Sep 30 12;54;48 comment by...Glenn Strycker -->
Marcel tried hitting "stop" for Partition_0 and then restarted it, and repartitioned.  SEVB turned green!  I guess we're ok for now, although the TOFTAC calib might be crap.  We'll try taking it again for kicks.


<!-- Date: Sun Sep 30 12:56:55.008 2007 -->
Eiko came by to work on the COT stuff -- she asked us to not put any COT crates in the partition while she's working on them.
<!-- Author: Glenn Strycker -->

<!-- Date: Sun Sep 30 12:58:00.354 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250488>250488</A>
ACTIVATE: RunSet: TOF_TAC_CALIB 
<!-- Author: Glenn Strycker -->


<!-- Date: Sun Sep 30 13:03:32.193 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250489>250489</A>
ACTIVATE: RunSet: COT_ACE Check calibration again after finding b0cot09 slot4-6 look fine in ASDQ calibration.
<!-- Author: eikoyu -->


<!-- Date: Sun Sep 30 13:03:33.015 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250490>250490</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: eikoyu -->


<!-- Date: Sun Sep 30 13:03:34.189 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250488>250488</A>
CALIBRATION: <U><B>toftac</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#toftac target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep 30 13;08;33 comment by...Glenn Strycker -->
BAD -- same bad channels as usual.
<!-- Date: Sun Sep 30 13:05:44.985 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250309>250309</A>
CALIBRATION: <U><B>cot</B></U> calibration FAILED, please investigate
<!-- Author: CDFCALIB -->



<!-- Date: Sun Sep 30 13:12:46.090 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250491>250491</A>
 Activated at 2007.09.30 13:11:58
<!-- Author: RunControl -->


<!-- Date: Sun Sep 30 13:12:54.329 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250491>250491</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,954,639] -- We removed all COT crates for this run as well as INHIBIT_00 and XFT_FINDER_02
<!-- Author: Glenn Strycker -->


<!-- Date: Sun Sep 30 13:13:52.479 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250492>250492</A>
ACTIVATE: RunSet: COTCALIB_ACE after rebooting b0cot09 and b0cot08

<!-- Author: eikoyu -->


<!-- Date: Sun Sep 30 13:15:52.525 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250492>250492</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->


<!-- Date: Sun Sep 30 13:16:19.949 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250493>250493</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: eikoyu -->




<!-- Date: Sun Sep 30 13:25:09.407 2007 -->
<pre>COT calibration failure investigation: 
 
  The result from ACE's calibration around noon indicated that  
there are too many bad channels due to new bad channels in  
b0cot09 slot4-6. We had seen these 3 slots with 13-16 ns offset  
in drift time on August 27th, but the offset went away after  
reseating TDC in slot 4.  
 
  30 mins after the calibration at noon, I came to Fermilab and  
performed an ASDQ calibration. Unfortunately, these three slots  
already self-cured themselves and their drift time looked fine.  
I did a COT calibration and found no problem either. In order to  
make sure, I "reset_crate" b0cot09 and b0cot08 (12 bad channels  
also went away in the second calibration). However, this seemed  
to have caused a disconnection between RTserver and these two  
crates. The complaint was "client not unique". Ray told me last  
Thursday that this required a restart of RTserver. We restarted  
it and rebooted b0cot08 and b0cot09 (instead of reset).  
 
Now COT calibration result is fine too. 
 </pre>
<!-- Author: Shin-Shan Yu (Eiko) -->
<!-- Comment: Sun Sep 30 13;26;04 comment by...Shin-Shan Yu (Eiko) -->
<pre>Will report this to JDL and TDC experts and see if they have spare TDCs to replace the ones in b0cot09 slot4-6.</pre>

<!-- Date: Sun Sep 30 13:29:44.113 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250491>250491</A>
 Terminated at 2007.09.30 13:29:01
<!-- Author: RunControl -->


<!-- Date: Sun Sep 30 13:29:45.974 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250491>250491</A>
TERMINATE: run went well, but Eiko is now done with the COT crates, so I'll put them back in the partition and take more COSMICS...
<!-- Author: Glenn Strycker -->


<!-- Date: Sun Sep 30 13:33:54.370 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250494>250494</A>
 Activated at 2007.09.30 13:33:14
<!-- Author: RunControl -->


<!-- Date: Sun Sep 30 13:33:55.235 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250494>250494</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,954,639] -- Normal mode, only INHIBIT_00 and XFT_FINDER_02 are removed (COT is back in).
<!-- Author: Glenn Strycker -->



<!-- Date: Sun Sep 30 15:02:17.286 2007 -->
<pre> 
- very quiet 
- one Silicon alarm from IFIX, paged experts but couldn't find anything 
- experts should check the calibrations, if they are new problems 
- Eiko looked at COT calibration and "fixed" some problems 
- running cosmics ... 
</pre>
<!-- Author: Marcel :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250494' target=_top>250494</a>) -->


<!-- Date: Sun Sep 30 15:18:12.738 2007 -->
<P> 
Fixed the problem with the b0inh00 INHIBITS_00 crate &minus;  
in the last rebuild of the "inhibits" product the libinhibits.a  
VxWorks library somehow did not get rebuilt.    I rebuilt it  
by hand and the crate is OK now. 
<P>
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250494' target=_top>250494</a>) -->

<!-- Date: Sun Sep 30 15:46:14.504 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250494>250494</A>
 Terminated at 2007.09.30 15:45:04
<!-- Author: RunControl -->


<!-- Date: Sun Sep 30 15:46:15.682 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250494>250494</A>
TERMINATE: ended to start calibrations. Will include inh00 from now on.
<!-- Author: Carolina Deluca -->


<!-- Date: Sun Sep 30 15:50:56.232 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250495>250495</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep 30 16;06;08 comment by...carolina -->
Lots of bad PHA 0 and 1 channels which following reference recommendations have to be ignored. At the end of the list CEM has mean a little bit deviated from the reference. Mark it as GOOD.
<!-- Date: Sun Sep 30 15:50:57.458 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250495>250495</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Carolina Deluca -->


<!-- Date: Sun Sep 30 15:52:57.011 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250496>250496</A>
ACTIVATE: RunSet: CALIB_CLC_QIE 
<!-- Author: Carolina Deluca -->


<!-- Date: Sun Sep 30 15:54:57.138 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250496>250496</A>
CALIBRATION: <U><B>clcqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#clcqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep 30 16;06;36 comment by...carolina -->
OK
<!-- Date: Sun Sep 30 15:55:13.373 2007 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>PhysicsTable</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L3Accepts</TH>
<TH>DelivLumi nb<SUP>-1</SUP></TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>15:55:00</TD>
<TD ALIGN=CENTER>::</TD>
<TD ALIGN=RIGHT></TD>               <TD ALIGN=RIGHT></TD>
<TD ALIGN=RIGHT></TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->


<!-- Date: Sun Sep 30 15:59:30.483 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250497>250497</A>
ACTIVATE: RunSet: CALIB_QIE_FWD 
<!-- Author: Carolina Deluca -->


<!-- Date: Sun Sep 30 15:59:31.775 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250497>250497</A>
CALIBRATION: <U><B>bscqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#bscqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep 30 16;08;38 comment by...carolina -->
GOOD
<!-- Date: Sun Sep 30 16:00:51.030 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=250498>250498</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL 
<!-- Author: Carolina Deluca -->



<!-- Date: Sun Sep 30 16:04:47.699 2007 -->
<font color=red>Shift Summary: </font><pre>End of shift</pre> 
<!-- Author: Robert Harris -->