<!-- Title: Day shift. Thu Aug 26, 2004 -->
<!-- SciCo: Jonathan Lewis -->
<!-- DAQAce: Gene Flanagan -->
<!-- MonAce: n/a -->
<!-- CO: n/a -->
<!-- OpManager: Rob Roser -->
<!-- Notes: -->
Another day, another 0.83 euros.


<!-- Date: Thu Aug 26 08:16:38.093 2004 -->
the exchange rate US$ to Euro really sucks
<!-- Author: A poor european Fermilab RA payed in US$ -->

<!-- Date: Thu Aug 26 08:25:56.369 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186667">186667</A>
TERMINATE: End cosmics to run calibrations
<!-- Author: Gene(x2080) -->
<!-- Comment: Thu Aug 26 09;41;40 comment by...jdl -->
Note that this run continued through the night unattended. 
<!-- Date: Thu Aug 26 08:27:09.054 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186668">186668</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 08:28:39.096 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186668">186668</A>
TERMINATE: End Cal Qie
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 08:29:36.779 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186669">186669</A>
ACTIVATE: RunSet: CALIB_CLC_QIE 
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 08:31:10.632 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186669">186669</A>
TERMINATE: End CLC Qie
<!-- Author: Gene(x2080) -->



<!-- Date: Thu Aug 26 08:49:32.650 2004 -->
iFix nodes S1, etc. are being updated today from FNAL domain to FERMI domain.  Expect errors due to the outages throughout the day.
<!-- Author: jdl -->

<!-- Date: Thu Aug 26 08:53:45.536 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186670">186670</A>
ACTIVATE: RunSet: CALIB_QIE_FWD 
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 08:55:12.032 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186670">186670</A>
TERMINATE: End BSC Qie
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 08:57:19.433 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186671">186671</A>
ACTIVATE: RunSet: CALIB_RP_SMQIE 
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 08:57:20.119 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186671">186671</A>
TERMINATE: End Roman Pot SMQIE
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 08:59:30.721 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186672">186672</A>
ACTIVATE: RunSet: ACE_TOF_QIE_NEW 
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 09:00:44.116 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186672">186672</A>
TERMINATE: End SMX QIE
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 09:02:50.941 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186673">186673</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 09:04:59.724 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186673">186673</A>
TERMINATE: End SMX Qie
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 09:05:00.023 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186674">186674</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 09:07:16.941 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186674">186674</A>
TERMINATE: End CEM LED
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 09:09:41.125 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186675">186675</A>
TERMINATE: End Cot Calib- as in last shift b0Cot00 missing.
<!-- Author: Gene(x2080) -->



<!-- Date: Thu Aug 26 09:11:36.736 2004 -->
I see the CMX trip - ignore alarms for now
<!-- Author: Dale -->
<!-- Comment: Thu Aug 26 09;14;57 comment by...Dale Stentz -->
3 trips on SW CMX wedge "16" - We'll keep this wedge off for now.

<!-- Comment: Thu Aug 26 11;33;08 comment by...Ingyin -->
<pre>Turned CMX SW miniskirts (IFix wedges "15" to "17") back on. If they trip, try to recover normally. If normal recovery does not work, please turn them off and make a note in the elog. We will look at them later.</pre>
<!-- Date: Thu Aug 26 09:17:45.726 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186676">186676</A>
ACTIVATE: RunSet: TOF_TAC_CALIB 
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 09:26:38.222 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186676">186676</A>
TERMINATE: End TOF TAC calib
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 09:33:01.114 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186677">186677</A>
 Activated at 2004.08.26 09:32:23
<!-- Author: RunControl -->


<!-- Date: Thu Aug 26 09:33:02.133 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186677">186677</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [5,469,442] 
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 09:37:45.097 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186677">186677</A>
 Terminated at 2004.08.26 09:36:47
<!-- Author: RunControl -->


<!-- Date: Thu Aug 26 09:37:47.695 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186677">186677</A>
TERMINATE: End spike run
<!-- Author: Gene(x2080) -->




<!-- Date: Thu Aug 26 09:42:29.728 2004 -->
<pre> 
As in previous 2 shifts lots of bad channels 
 
 
 
0: 'PHAPED' 
1: 'CHAPED' 
2: 'CEMPED' 
3: 'PEMPED' 
Run, Wedge, Channel, Cap, Detector, Mean(ref), Mean, RMS(ref), Sigma 
186668   0 W   0  0  PHA  194#pm 8  178  0&lt;rms&lt;5   1.52 
186668   0 W   0  1  PHA  191#pm 8  177  0&lt;rms&lt;5   1.45 
186668   0 W   0  2  PHA  191#pm 8  175  0&lt;rms&lt;5   1.66 
186668   0 W   0  3  PHA  197#pm 8  182  0&lt;rms&lt;5   1.62 
186668   0 W   1  0  PHA  180#pm 8  172  0&lt;rms&lt;5   1.66 
186668   0 W   1  1  PHA  179#pm 8  171  0&lt;rms&lt;5   1.73 
186668   0 W   1  2  PHA  182#pm 8  173  0&lt;rms&lt;5   1.58 
186668   0 W   1  3  PHA  186#pm 8  178  0&lt;rms&lt;5   1.76 
186668   2 W   0  0  PHA  194#pm 8  158  0&lt;rms&lt;5   1.77 
186668   2 W   0  1  PHA  196#pm 8  160  0&lt;rms&lt;5   1.90 
186668   2 W   0  2  PHA  199#pm 8  163  0&lt;rms&lt;5   1.74 
186668   2 W   0  3  PHA  202#pm 8  165  0&lt;rms&lt;5   1.74 
186668   3 W   0  0  PHA  193#pm 8  174  0&lt;rms&lt;5   1.24 
186668   3 W   0  1  PHA  192#pm 8  174  0&lt;rms&lt;5   1.39 
186668   3 W   0  2  PHA  191#pm 8  172  0&lt;rms&lt;5   1.51 
186668   3 W   0  3  PHA  191#pm 8  172  0&lt;rms&lt;5   1.33 
186668   4 W   0  0  PHA  181#pm 8  172  0&lt;rms&lt;5   1.49 
186668   4 W   0  1  PHA  179#pm 8  170  0&lt;rms&lt;5   1.43 
186668   4 W   0  2  PHA  177#pm 8  168  0&lt;rms&lt;5   1.42 
186668   4 W   0  3  PHA  180#pm 8  171  0&lt;rms&lt;5   1.33 
186668   4 W   1  0  PHA  184#pm 8  171  0&lt;rms&lt;5   1.39 
186668   4 W   1  1  PHA  180#pm 8  169  0&lt;rms&lt;5   1.46 
186668   4 W   1  2  PHA  183#pm 8  171  0&lt;rms&lt;5   1.30 
186668   4 W   1  3  PHA  185#pm 8  174  0&lt;rms&lt;5   1.36 
186668   6 W   0  0  PHA  176#pm 8  159  0&lt;rms&lt;5   1.55 
186668   6 W   0  1  PHA  175#pm 8  158  0&lt;rms&lt;5   1.37 
186668   6 W   0  2  PHA  178#pm 8  160  0&lt;rms&lt;5   1.32 
186668   6 W   0  3  PHA  181#pm 8  164  0&lt;rms&lt;5   1.32 
186668   6 W   1  0  PHA  166#pm 8  149  0&lt;rms&lt;5   1.55 
186668   6 W   1  1  PHA  170#pm 8  153  0&lt;rms&lt;5   1.58 
186668   6 W   1  2  PHA  167#pm 8  150  0&lt;rms&lt;5   1.58 
186668   6 W   1  3  PHA  173#pm 8  156  0&lt;rms&lt;5   1.49 
186668   7 W   0  0  PHA  183#pm 8  164  0&lt;rms&lt;5   5.69 
186668   7 W   0  1  PHA  182#pm 8  161  0&lt;rms&lt;5   5.64 
186668   7 W   0  2  PHA  185#pm 8  157  0&lt;rms&lt;5   6.22 
186668   7 W   0  3  PHA  187#pm 8  171  0&lt;rms&lt;5   5.83 
186668   9 W   0  0  PHA  176#pm 8  156  0&lt;rms&lt;5   1.49 
186668   9 W   0  1  PHA  172#pm 8  153  0&lt;rms&lt;5   1.42 
186668   9 W   0  2  PHA  173#pm 8  153  0&lt;rms&lt;5   1.42 
186668   9 W   0  3  PHA  177#pm 8  158  0&lt;rms&lt;5   1.52 
186668  10 W   1  2  PHA  178#pm 8  169  0&lt;rms&lt;5   1.47 
186668  11 W   0  0  PHA  198#pm 8  185  0&lt;rms&lt;5   1.56 
186668  11 W   0  1  PHA  197#pm 8  185  0&lt;rms&lt;5   1.44 
186668  11 W   0  2  PHA  198#pm 8  186  0&lt;rms&lt;5   1.54 
186668  11 W   0  3  PHA  204#pm 8  191  0&lt;rms&lt;5   1.56 
186668  12 W   0  0  PHA  190#pm 8  173  0&lt;rms&lt;5   1.57 
186668  12 W   0  1  PHA  191#pm 8  175  0&lt;rms&lt;5   1.50 
186668  12 W   0  2  PHA  191#pm 8  174  0&lt;rms&lt;5   1.36 
186668  12 W   0  3  PHA  195#pm 8  179  0&lt;rms&lt;5   1.53 
186668  13 W   0  0  PHA  185#pm 8  172  0&lt;rms&lt;5   1.44 
Run, Wedge, Channel, Cap, Detector, Mean(ref), Mean, RMS(ref), Sigma 
186668  13 W   0  1  PHA  183#pm 8  169  0&lt;rms&lt;5   1.47 
186668  13 W   0  2  PHA  185#pm 8  171  0&lt;rms&lt;5   1.36 
186668  13 W   0  3  PHA  189#pm 8  175  0&lt;rms&lt;5   1.56 
186668  14 W   1  0  PHA  180#pm 8  162  0&lt;rms&lt;5   1.66 
186668  14 W   1  1  PHA  178#pm 8  160  0&lt;rms&lt;5   1.70 
186668  14 W   1  2  PHA  178#pm 8  160  0&lt;rms&lt;5   1.61 
186668  14 W   1  3  PHA  183#pm 8  164  0&lt;rms&lt;5   1.68 
186668  15 W   0  0  PHA  202#pm 8  191  0&lt;rms&lt;5   1.55 
186668  15 W   0  1  PHA  202#pm 8  190  0&lt;rms&lt;5   1.55 
186668  15 W   0  2  PHA  198#pm 8  187  0&lt;rms&lt;5   1.63 
186668  15 W   0  3  PHA  202#pm 8  191  0&lt;rms&lt;5   1.64 
186668  16 W   0  0  PHA  158#pm 8  150  0&lt;rms&lt;5   1.49 
186668  16 W   1  0  PHA  165#pm 8  146  0&lt;rms&lt;5   1.61 
186668  16 W   1  1  PHA  167#pm 8  149  0&lt;rms&lt;5   1.71 
186668  16 W   1  2  PHA  164#pm 8  146  0&lt;rms&lt;5   1.63 
186668  16 W   1  3  PHA  171#pm 8  152  0&lt;rms&lt;5   1.65 
186668  17 W   0  0  PHA  169#pm 8  157  0&lt;rms&lt;5   1.44 
186668  17 W   0  1  PHA  168#pm 8  157  0&lt;rms&lt;5   1.55 
186668  17 W   0  2  PHA  167#pm 8  158  0&lt;rms&lt;5   1.56 
186668  17 W   0  3  PHA  170#pm 8  159  0&lt;rms&lt;5   1.63 
186668  18 W   0  0  PHA  188#pm 8  177  0&lt;rms&lt;5   1.47 
186668  18 W   0  1  PHA  186#pm 8  176  0&lt;rms&lt;5   1.51 
186668  18 W   0  2  PHA  183#pm 8  172  0&lt;rms&lt;5   1.74 
186668  18 W   0  3  PHA  188#pm 8  178  0&lt;rms&lt;5   1.40 
186668  18 W   1  0  PHA  193#pm 8  179  0&lt;rms&lt;5   1.21 
186668  18 W   1  1  PHA  191#pm 8  177  0&lt;rms&lt;5   1.16 
186668  18 W   1  2  PHA  192#pm 8  177  0&lt;rms&lt;5   1.34 
186668  18 W   1  3  PHA  192#pm 8  178  0&lt;rms&lt;5   1.37 
186668  19 W   0  0  PHA  185#pm 8  175  0&lt;rms&lt;5   1.49 
186668  19 W   0  1  PHA  184#pm 8  174  0&lt;rms&lt;5   1.45 
186668  19 W   0  2  PHA  186#pm 8  176  0&lt;rms&lt;5   1.51 
186668  19 W   0  3  PHA  188#pm 8  178  0&lt;rms&lt;5   1.47 
186668  20 W   0  0  PHA  165#pm 8  150  0&lt;rms&lt;5   1.38 
186668  20 W   0  1  PHA  164#pm 8  149  0&lt;rms&lt;5   1.50 
186668  20 W   0  2  PHA  165#pm 8  150  0&lt;rms&lt;5   1.45 
186668  20 W   0  3  PHA  167#pm 8  152  0&lt;rms&lt;5   1.27 
186668  20 W   1  0  PHA  160#pm 8  148  0&lt;rms&lt;5   1.60 
186668  20 W   1  1  PHA  166#pm 8  154  0&lt;rms&lt;5   1.66 
186668  20 W   1  2  PHA  162#pm 8  148  0&lt;rms&lt;5   1.73 
186668  20 W   1  3  PHA  164#pm 8  151  0&lt;rms&lt;5   1.63 
186668  21 W   0  0  PHA  156#pm 8  146  0&lt;rms&lt;5   1.43 
186668  21 W   0  1  PHA  157#pm 8  148  0&lt;rms&lt;5   1.45 
186668  21 W   0  2  PHA  160#pm 8  149  0&lt;rms&lt;5   1.30 
186668  21 W   0  3  PHA  162#pm 8  152  0&lt;rms&lt;5   1.63 
186668  22 W   0  0  PHA  170#pm 8  155  0&lt;rms&lt;5   1.48 
186668  22 W   0  1  PHA  176#pm 8  162  0&lt;rms&lt;5   1.52 
186668  22 W   0  2  PHA  171#pm 8  156  0&lt;rms&lt;5   1.32 
186668  22 W   0  3  PHA  176#pm 8  162  0&lt;rms&lt;5   1.54 
Run, Wedge, Channel, Cap, Detector, Mean(ref), Mean, RMS(ref), Sigma 
186668  23 W   0  0  PHA  178#pm 8  165  0&lt;rms&lt;5   1.49 
186668  23 W   0  1  PHA  178#pm 8  165  0&lt;rms&lt;5   1.50 
186668  23 W   0  2  PHA  177#pm 8  164  0&lt;rms&lt;5   1.56 
186668  23 W   0  3  PHA  178#pm 8  165  0&lt;rms&lt;5   1.64 
186668   0 E   0  0  PHA  191#pm 8  172  0&lt;rms&lt;5   1.55 
186668   0 E   0  1  PHA  187#pm 8  170  0&lt;rms&lt;5   1.83 
186668   0 E   0  2  PHA  188#pm 8  170  0&lt;rms&lt;5   1.95 
186668   0 E   0  3  PHA  193#pm 8  175  0&lt;rms&lt;5   1.59 
186668   1 E   0  0  PHA  180#pm 8  172  0&lt;rms&lt;5   1.75 
186668   1 E   0  1  PHA  185#pm 8  176  0&lt;rms&lt;5   1.61 
186668   1 E   0  3  PHA  185#pm 8  177  0&lt;rms&lt;5   1.63 
186668   1 E   1  0  PHA  188#pm 8  175  0&lt;rms&lt;5   1.56 
186668   1 E   1  1  PHA  186#pm 8  174  0&lt;rms&lt;5   1.59 
186668   1 E   1  2  PHA  188#pm 8  176  0&lt;rms&lt;5   1.55 
186668   1 E   1  3  PHA  194#pm 8  181  0&lt;rms&lt;5   1.69 
186668   2 E   0  0  PHA  178#pm 8  163  0&lt;rms&lt;5   1.56 
186668   2 E   0  1  PHA  178#pm 8  163  0&lt;rms&lt;5   1.59 
186668   2 E   0  2  PHA  178#pm 8  163  0&lt;rms&lt;5   1.49 
186668   2 E   0  3  PHA  183#pm 8  168  0&lt;rms&lt;5   1.70 
186668   3 E   0  3  PHA  197#pm 8  189  0&lt;rms&lt;5   1.55 
186668   3 E   1  0  PHA  151#pm 8  139  0&lt;rms&lt;5   1.49 
186668   3 E   1  1  PHA  153#pm 8  141  0&lt;rms&lt;5   1.48 
186668   3 E   1  2  PHA  156#pm 8  145  0&lt;rms&lt;5   1.46 
186668   3 E   1  3  PHA  160#pm 8  148  0&lt;rms&lt;5   1.55 
186668   4 E   0  0  PHA  183#pm 8  165  0&lt;rms&lt;5   1.48 
186668   4 E   0  1  PHA  186#pm 8  168  0&lt;rms&lt;5   1.36 
186668   4 E   0  2  PHA  184#pm 8  166  0&lt;rms&lt;5   1.39 
186668   4 E   0  3  PHA  193#pm 8  175  0&lt;rms&lt;5   1.35 
186668   5 E   0  0  PHA  192#pm 8  179  0&lt;rms&lt;5   1.26 
186668   5 E   0  1  PHA  192#pm 8  180  0&lt;rms&lt;5   1.39 
186668   5 E   0  2  PHA  190#pm 8  177  0&lt;rms&lt;5   1.39 
186668   5 E   0  3  PHA  196#pm 8  184  0&lt;rms&lt;5   1.46 
186668   5 E   1  0  PHA  165#pm 8  149  0&lt;rms&lt;5   1.58 
186668   5 E   1  1  PHA  163#pm 8  146  0&lt;rms&lt;5   1.53 
186668   5 E   1  2  PHA  167#pm 8  149  0&lt;rms&lt;5   1.65 
186668   5 E   1  3  PHA  170#pm 8  153  0&lt;rms&lt;5   1.53 
186668   7 E   0  0  PHA  183#pm 8  163  0&lt;rms&lt;5   1.46 
186668   7 E   0  1  PHA  185#pm 8  165  0&lt;rms&lt;5   1.48 
186668   7 E   0  2  PHA  182#pm 8  162  0&lt;rms&lt;5   1.57 
186668   7 E   0  3  PHA  189#pm 8  169  0&lt;rms&lt;5   1.64 
186668   7 E   1  0  PHA  176#pm 8  161  0&lt;rms&lt;5   1.59 
186668   7 E   1  1  PHA  174#pm 8  161  0&lt;rms&lt;5   1.57 
186668   7 E   1  2  PHA  173#pm 8  159  0&lt;rms&lt;5   1.53 
186668   7 E   1  3  PHA  178#pm 8  164  0&lt;rms&lt;5   1.55 
186668   8 E   0  0  PHA  175#pm 8  140  0&lt;rms&lt;5   1.44 
186668   8 E   0  1  PHA  178#pm 8  142  0&lt;rms&lt;5   1.31 
186668   8 E   0  2  PHA  177#pm 8  141  0&lt;rms&lt;5   1.48 
186668   8 E   0  3  PHA  179#pm 8  143  0&lt;rms&lt;5   1.42 
Run, Wedge, Channel, Cap, Detector, Mean(ref), Mean, RMS(ref), Sigma 
186668   9 E   1  0  PHA  181#pm 8  169  0&lt;rms&lt;5   1.59 
186668   9 E   1  1  PHA  181#pm 8  170  0&lt;rms&lt;5   1.52 
186668   9 E   1  2  PHA  182#pm 8  169  0&lt;rms&lt;5   1.66 
186668   9 E   1  3  PHA  184#pm 8  172  0&lt;rms&lt;5   1.52 
186668  10 E   0  0  PHA  177#pm 8  157  0&lt;rms&lt;5   1.34 
186668  10 E   0  1  PHA  181#pm 8  160  0&lt;rms&lt;5   1.66 
186668  10 E   0  2  PHA  179#pm 8  158  0&lt;rms&lt;5   1.47 
186668  10 E   0  3  PHA  183#pm 8  163  0&lt;rms&lt;5   1.48 
186668  11 E   0  0  PHA  167#pm 8  149  0&lt;rms&lt;5   1.50 
186668  11 E   0  1  PHA  168#pm 8  149  0&lt;rms&lt;5   1.46 
186668  11 E   0  2  PHA  168#pm 8  149  0&lt;rms&lt;5   1.43 
186668  11 E   0  3  PHA  172#pm 8  152  0&lt;rms&lt;5   1.53 
186668  12 E   0  0  PHA  195#pm 8  184  0&lt;rms&lt;5   1.59 
186668  12 E   0  1  PHA  192#pm 8  180  0&lt;rms&lt;5   1.58 
186668  12 E   0  2  PHA  195#pm 8  183  0&lt;rms&lt;5   1.48 
186668  12 E   0  3  PHA  200#pm 8  188  0&lt;rms&lt;5   1.45 
186668  13 E   0  0  PHA  195#pm 8  175  0&lt;rms&lt;5   1.53 
186668  13 E   0  1  PHA  196#pm 8  176  0&lt;rms&lt;5   1.53 
186668  13 E   0  2  PHA  196#pm 8  177  0&lt;rms&lt;5   1.47 
186668  13 E   0  3  PHA  200#pm 8  180  0&lt;rms&lt;5   1.41 
186668  14 E   0  0  PHA  175#pm 8  160  0&lt;rms&lt;5   1.34 
186668  14 E   0  1  PHA  177#pm 8  162  0&lt;rms&lt;5   1.54 
186668  14 E   0  2  PHA  176#pm 8  161  0&lt;rms&lt;5   1.60 
186668  14 E   0  3  PHA  179#pm 8  164  0&lt;rms&lt;5   1.55 
186668  15 E   1  0  PHA  165#pm 8  150  0&lt;rms&lt;5   1.43 
186668  15 E   1  1  PHA  171#pm 8  154  0&lt;rms&lt;5   1.58 
186668  15 E   1  2  PHA  170#pm 8  153  0&lt;rms&lt;5   1.51 
186668  15 E   1  3  PHA  170#pm 8  154  0&lt;rms&lt;5   1.56 
186668  16 E   0  0  PHA  185#pm 8  172  0&lt;rms&lt;5   1.33 
186668  16 E   0  1  PHA  180#pm 8  168  0&lt;rms&lt;5   1.30 
186668  16 E   0  2  PHA  181#pm 8  168  0&lt;rms&lt;5   1.26 
186668  16 E   0  3  PHA  186#pm 8  173  0&lt;rms&lt;5   1.49 
186668  17 E   0  0  PHA  189#pm 8  175  0&lt;rms&lt;5   1.69 
186668  17 E   0  1  PHA  179#pm 8  165  0&lt;rms&lt;5   1.53 
186668  17 E   0  2  PHA  185#pm 8  170  0&lt;rms&lt;5   1.46 
186668  17 E   0  3  PHA  188#pm 8  174  0&lt;rms&lt;5   1.64 
186668  17 E   1  0  PHA  170#pm 8  159  0&lt;rms&lt;5   1.54 
186668  17 E   1  1  PHA  164#pm 8  154  0&lt;rms&lt;5   1.73 
186668  17 E   1  2  PHA  168#pm 8  157  0&lt;rms&lt;5   1.57 
186668  17 E   1  3  PHA  172#pm 8  161  0&lt;rms&lt;5   1.59 
186668  18 E   0  0  PHA  189#pm 8  179  0&lt;rms&lt;5   1.56 
186668  18 E   0  1  PHA  186#pm 8  174  0&lt;rms&lt;5   1.67 
186668  18 E   0  2  PHA  187#pm 8  177  0&lt;rms&lt;5   1.59 
186668  18 E   0  3  PHA  190#pm 8  179  0&lt;rms&lt;5   1.51 
186668  19 E   0  1  PHA  196#pm 8  187  0&lt;rms&lt;5   1.58 
186668  19 E   0  3  PHA  201#pm 8  192  0&lt;rms&lt;5   1.54 
186668  19 E   1  0  PHA  175#pm 8  166  0&lt;rms&lt;5   1.61 
186668  19 E   1  1  PHA  175#pm 8  164  0&lt;rms&lt;5   1.46 
186668  19 E   1  2  PHA  173#pm 8  163  0&lt;rms&lt;5   1.61 
186668  19 E   1  3  PHA  182#pm 8  171  0&lt;rms&lt;5   1.42 
 
3648 entries    307 bad channels 
</pre>
<!-- Author: Gebe :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186668' target=_top>186668</a>) -->
<!-- Comment: Thu Aug 26 09;49;48 comment by...Gene -->
<pre>Again Slopes all zero ...


run  186668  version  1
Run, Wedge, Side, Channel, Cap, Range, Detector, Slope (ref), Slope
186668  13 E  0  0  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  0  0  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  0  0  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  0  0  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  0  0  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  0  0  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  0  0  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  0  0  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  0  1  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  0  1  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  0  1  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  0  1  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  0  1  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  0  1  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  0  1  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  0  1  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  0  2  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  0  2  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  0  2  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  0  2  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  0  2  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  0  2  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  0  2  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  0  2  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  0  3  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  0  3  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  0  3  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  0  3  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  0  3  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  0  3  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  0  3  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  0  3  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  1  0  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  1  0  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  1  0  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  1  0  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  1  0  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  1  0  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  1  0  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  1  0  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  1  1  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  1  1  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  1  1  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  1  1  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  1  1  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  1  1  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  1  1  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  1  1  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  1  2  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  1  2  1  CHA  1.70000#pm0.12500  0.00000
Run, Wedge, Side, Channel, Cap, Range, Detector, Slope (ref), Slope
186668  13 E  1  2  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  1  2  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  1  2  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  1  2  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  1  2  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  1  2  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  1  3  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  1  3  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  1  3  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  1  3  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  1  3  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  1  3  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  1  3  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  1  3  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  2  0  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  2  0  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  2  0  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  2  0  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  2  0  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  2  0  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  2  0  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  2  0  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  2  1  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  2  1  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  2  1  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  2  1  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  2  1  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  2  1  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  2  1  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  2  1  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  2  2  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  2  2  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  2  2  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  2  2  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  2  2  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  2  2  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  2  2  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  2  2  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  2  3  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  2  3  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  2  3  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  2  3  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  2  3  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  2  3  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  2  3  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  2  3  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  3  0  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  3  0  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  3  0  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  3  0  3  CHA  0.42500#pm0.03125  0.00000
Run, Wedge, Side, Channel, Cap, Range, Detector, Slope (ref), Slope
186668  13 E  3  0  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  3  0  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  3  0  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  3  0  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  3  1  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  3  1  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  3  1  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  3  1  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  3  1  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  3  1  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  3  1  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  3  1  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  3  2  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  3  2  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  3  2  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  3  2  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  3  2  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  3  2  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  3  2  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  3  2  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  3  3  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  3  3  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  3  3  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  3  3  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  3  3  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  3  3  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  3  3  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  3  3  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  4  0  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  4  0  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  4  0  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  4  0  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  4  0  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  4  0  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  4  0  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  4  0  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  4  1  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  4  1  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  4  1  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  4  1  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  4  1  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  4  1  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  4  1  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  4  1  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  4  2  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  4  2  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  4  2  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  4  2  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  4  2  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  4  2  5  CHA  0.10625#pm0.00781  0.00000
Run, Wedge, Side, Channel, Cap, Range, Detector, Slope (ref), Slope
186668  13 E  4  2  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  4  2  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  4  3  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  4  3  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  4  3  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  4  3  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  4  3  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  4  3  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  4  3  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  4  3  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  5  0  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  5  0  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  5  0  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  5  0  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  5  0  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  5  0  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  5  0  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  5  0  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  5  1  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  5  1  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  5  1  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  5  1  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  5  1  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  5  1  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  5  1  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  5  1  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  5  2  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  5  2  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  5  2  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  5  2  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  5  2  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  5  2  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  5  2  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  5  2  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  5  3  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  5  3  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  5  3  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  5  3  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  5  3  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  5  3  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  5  3  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  5  3  7  CHA  0.02773#pm0.00195  0.00000
186668  13 E  6  0  0  CHA  3.40000#pm0.25000  0.00000
186668  13 E  6  0  1  CHA  1.70000#pm0.12500  0.00000
186668  13 E  6  0  2  CHA  0.85000#pm0.06250  0.00000
186668  13 E  6  0  3  CHA  0.42500#pm0.03125  0.00000
186668  13 E  6  0  4  CHA  0.21250#pm0.01562  0.00000
186668  13 E  6  0  5  CHA  0.10625#pm0.00781  0.00000
186668  13 E  6  0  6  CHA  0.05313#pm0.00391  0.00000
186668  13 E  6  0  7  CHA  0.02773#pm0.00195  0.00000

672 entries     672 bad channels


</pre>
<!-- Comment: Thu Aug 26 14;40;44 comment by...m mattson -->
<br>You can ignore PHA channels 0 and 1.  Those are PIN Diodes, used by the PHA experts, and not used for data taking.  During shutdowns, they tend to act flakey.
<!-- Date: Thu Aug 26 09:49:12.986 2004 -->
CEM06W-9R HV high by 334V.  What's up with that?
<!-- Author: jdl -->



<!-- Date: Thu Aug 26 09:54:56.118 2004 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2004&action=view&page=-12018&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2004&action=view&page=-12019  width='300' align='top'></a></td><td> </td></tr></table>
<!-- Author:   -->
<!-- Comment: Thu Aug 26 09;55;36 comment by...Gene -->
<pre>CLC QIE </pre>


<!-- Date: Thu Aug 26 10:05:36.252 2004 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2004&action=view&page=-12020&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2004&action=view&page=-12020  width='300' align='top'></a></td><td> </td></tr></table>
<!-- Author:   -->
<!-- Comment: Thu Aug 26 10;06;12 comment by...Gene -->
<pre>From Cem Led calib looks like yesterdays calib</pre>


<!-- Date: Thu Aug 26 10:15:01.442 2004 -->
<pre>Calib summary 
 
 
BSC Qie 186670: OK 
 
Roman Pot SMQIE 186671: OK 
 
TOF QIE 186672: Done 
 
COT Not complete due to B0cot00 being out. See previous shift 
 
SMX QIE 186673: Bad . 
 
TOF TAC 186676: OK 
 
Spike run 186677: L1 rate 117 hz</pre>
<!-- Author: Gene -->


<!-- Date: Thu Aug 26 10:55:31.601 2004 -->
HV's for all PisaBoxes are normal now.
<!-- Author: Akram Artikov -->


<!-- Date: Thu Aug 26 11:05:02.478 2004 -->
CEM SW Gamma has drifted again by ~5V.  Many channels are showing up red.  We adjusted it by 5V only <a href=/cgi/elog/elog.pl?nb=2004&action=copyentry&page=634&time=13:30:30.365>  
yesterday</a>.
<!-- Author: jdl -->


<!-- Date: Thu Aug 26 11:11:58.299 2004 -->
 S5 coming back.  Cryo taking out S4 (solenoid) for domain change.
<!-- Author: jdl -->

<!-- Date: Thu Aug 26 11:18:05.064 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186679">186679</A>
IDLE: <PRE> 
 Attention!!!. L3_RELAY_FAILURE Error !!!. 
 Level3 internal communications failure. One or more L3 nodes is likely
 to have died.
   Do not panic! Level3 will try to repair itself. We recommend:
     - do End/Abort+Reset ON ALL DEFINED PARTITIONS   
     - exclude all affected subfarms from your run configuration
       (find out list of bad subfarms from Error Logger: 
          if None, exclude nothing!)
     - start another run and page L3 farm pager at closest day/evening shift
     - if >2 subfarms affected page L3 farm IMMEDIATELY
 </PRE>
<!-- Author: ErrorHandler -->


<!-- Date: Thu Aug 26 11:20:46.907 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186679">186679</A>
Partition: <PRE> 
 Attention!!!. L3_RELAY_FAILURE Error !!!. 
 Level3 internal communications failure. One or more L3 nodes is likely
 to have died.
   Do not panic! Level3 will try to repair itself. We recommend:
     - do End/Abort+Reset ON ALL DEFINED PARTITIONS   
     - exclude all affected subfarms from your run configuration
       (find out list of bad subfarms from Error Logger: 
          if None, exclude nothing!)
     - start another run and page L3 farm pager at closest day/evening shift
     - if >2 subfarms affected page L3 farm IMMEDIATELY
 </PRE>
<!-- Author: ErrorHandler -->



<!-- Date: Thu Aug 26 11:27:14.458 2004 -->
<pre>Jonathan Lewis notified me that CEM 9R06W was almost 300V low.  I 
went to the Pisabox Readout computer on 1st floor and read this 
tube's voltage.  It was around 735V.  This is not really a possible 
read value given the setting of the Gamma supplies.  With the Pisa 
box channel adjusted to minimum, we can only go as low as about 
900V.  I did reset the voltage to 1022V.  It took three tries. 
The first try resulted in a voltage about 20V too high.  Second was 
about 10V too high.  Third try worked.  I don't know if there is 
some problem with the readback on this channel.  The behavior is 
certainly strange.  For the moment, will just monitor this and see 
that it stays stable.  I'll check the LED runs to see that the 
voltage has indeed remained near nominal.</pre>
<!-- Author: Bob Wagner (ANL) -->
<!-- Comment: Thu Aug 26 12;32;42 comment by...jdl -->
6W9R now high by 75V
<!-- Comment: Thu Aug 26 15;32;06 comment by...Willis -->
CEM 6W-9R has a known history of moving up and down by 200v or so.
<!-- Date: Thu Aug 26 11:28:25.275 2004 -->
After talking to Willis, I again adjusted the CEM SW Gamma HV.  It turns out that the voltages were about 5V high.  For some unknown reason, the display reports the difference as REF-READING.  This is totally non-standard and confusing.  In any event, the Gamma was reading 1264V while yesteday it was 1260V.  I moved the dial from 1284 to 1279.  We now read 1259V.  We will wait for about an hour for the Pisabox program to run.  I expect everything to be copasetic after that.   
<p> 
All of this begs a question: why don't we put feedback on the gammas?  We can have a DAC to set the voltage remotely and control the drift.  
<!-- Author: jdl -->
<!-- Comment: Thu Aug 26 12;33;18 comment by...jdl -->
SW now looks good except for 6W9R as noted above.
<!-- Date: Thu Aug 26 11:36:35.654 2004 -->
Eric has set the offending TDC in cot00 (slot 16?) to be offline and will try to swap it later today.  It is a rev F board.
<!-- Author: jdl -->




<!-- Date: Thu Aug 26 12:38:56.337 2004 -->
I rebooted b0fib02 a few times to try to help DB people resolve some issues, hope noone minded, doubt anyone noticed...
<!-- Author: SCN -->

<!-- Date: Thu Aug 26 12:46:48.302 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186680">186680</A>
HALTED: <PRE>   
  Done TO. for crate/s: 
    b0ccal12   
 </PRE>
<!-- Author: ErrorHandler -->


<!-- Date: Thu Aug 26 12:46:49.615 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186680">186680</A>
ACTIVE: <PRE>   
  Done TO. for crate/s: 
    b0ccal12   
 </PRE>
<!-- Author: ErrorHandler -->


<!-- Date: Thu Aug 26 12:46:50.675 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186680">186680</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [16,466,444] 
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 12:47:26.178 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186680">186680</A>
TERMINATE: b0ccal12 done time out errors 
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 12:49:39.251 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186681">186681</A>
ACTIVATE: RunSet: COTCALIB_ACE b0cot00 back in
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 12:50:31.230 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186681">186681</A>
TERMINATE: COT calib with b0cot00 working
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 12:51:33.978 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186682">186682</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 12:51:34.150 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186682">186682</A>
TERMINATE: End COT calib
<!-- Author: Gene(x2080) -->



<!-- Date: Thu Aug 26 12:57:48.275 2004 -->
<pre>COT Calib Run: 186682 
 
 
COT Calibration 12:50:13 on 2004-08-26 
aveStart:                        304.83 
total numBad:                    508     - o.k. 
   numBad  cells:                  3 
   numBad cables:                  2 
   numBad slots:                   0 
 
aveDrift:                       0.090 
max number drifted in crate:   148 
total numDrifted:              428 
   number of drifted  cells        2 
   number of drifted cables        3 
   number of drifted  slots        0 
 
 
This calibration is o.k. 
Timing has not changed, no new constants written 
 
 
Calibration passed quality checks, but there are Potential Bad Cells, Cables or Slots - 
See Expert Section 
 
Out of the last 8 calibrations, constants have been updated 5 times 
 
 
 
Expert Only: 
file: /cdf/people1/cdfdaq/consumers/log/runCotCtt_0_2004-08-26_12:50:13.log 
 
Drifted Cells: 
crate 13 slot  8 asdq  3: 11 
crate 18 slot  6 asdq  6: 10 
 
 
Drifted Cables: 
crate 14 slot  8 asdq  7 cable  3: 24 
crate 18 slot  6 asdq  1 cable  0: 20 
crate 18 slot  6 asdq  5 cable  2: 24 
 
 
Bad Cells: 
crate  2 slot  5 asdq  6: 12 
crate 16 slot 16 asdq  1: 10 
crate 17 slot 19 asdq  5:  9 
 
 
Bad Cables: 
crate  7 slot 10 asdq  7 cable  3: 24 
crate 11 slot 17 asdq  1 cable  0: 21 
 
</pre>
<!-- Author: Gene :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186682' target=_top>186682</a>) -->

<!-- Date: Thu Aug 26 13:04:05.789 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186683">186683</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [16,466,444] 
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 13:05:21.223 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186683">186683</A>
ACTIVE: <PRE>   
  Done TO. for crate/s: 
    b0ccal12   b0ccal13   
 </PRE>
<!-- Author: ErrorHandler -->


<!-- Date: Thu Aug 26 13:05:22.582 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186683">186683</A>
ACTIVE: <PRE>   
 
    L2 trigger rate has dropped to 0 Hz!!! 
     Events are not being logged !!! </PRE>
<!-- Author: ErrorHandler -->


<!-- Date: Thu Aug 26 13:05:23.286 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186683">186683</A>
ACTIVE: <PRE>   
 
    L3 trigger rate has dropped to 0 Hz!!! 
     Events are not being logged !!! </PRE>
<!-- Author: ErrorHandler -->


<!-- Date: Thu Aug 26 13:06:16.694 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186683">186683</A>
ACTIVE: <PRE> 
 Attention!!!. FERML_SHEPHERD_SUCCESS !!! 
 
 Successfully Shepherded b0ccal13 
 an additional AUTO HRR will be issued.  
 </PRE>
<!-- Author: ErrorHandler -->


<!-- Date: Thu Aug 26 13:06:17.825 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186683">186683</A>
HRRinProgress: <PRE> 
 Attention!!!. FERML_SHEPHERD_SUCCESS !!! 
 
 Successfully Shepherded b0ccal12 
 an additional AUTO HRR will be issued.  
 </PRE>
<!-- Author: ErrorHandler -->


<!-- Date: Thu Aug 26 13:06:18.285 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186683">186683</A>
HRR: <PRE>  
  Attention!!!.   RXPT Errors LOOP!!!. 
  
  >> Please Check the crate/s in the following list: << 
  
   b0ccal12   b0ccal13    
    
  >>        and take the appropriate actions.        <<

  >>>>>>>>>>>>>>>>>>>>>>>>>>>>>>>>>>>>>>>
  >>  NO MORE AUTOMATIC HRRs WILL BE ISSUED.                >>
  >>  ACES ......  YOU HAVE CONTROL of RC NOW.                    >>
  >>  AUTOMATIC HRRs WILL RESUME AFTER 3 MIN.              >>
  >>  OF SMOOTH RUNNING                                                     >>
  >>  (ie. EVENTS INCREMENTING and NO TIMEOUTS)            >>
  >>>>>>>>>>>>>>>>>>>>>>>>>>>>>>>>>>>>>>>
  </PRE>
<!-- Author: ErrorHandler -->


<!-- Date: Thu Aug 26 13:06:20.910 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186683">186683</A>
HRR: <PRE>   
  Done TO. for crate/s: 
    b0ccal12   b0ccal13   
 </PRE>
<!-- Author: ErrorHandler -->


<!-- Date: Thu Aug 26 13:08:26.643 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186683">186683</A>
TERMINATE: done TO erros b0ccal12 & 13
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 13:14:13.832 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186684">186684</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [16,466,444] no b0ccal12,13
<!-- Author: Gene(x2080) -->



<!-- Date: Thu Aug 26 13:32:22.711 2004 -->
Sergei is rebooting the nameserver.
<!-- Author: jdl -->

<!-- Date: Thu Aug 26 13:35:55.949 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186684">186684</A>
ACTIVE: <PRE>   
  Done TO. for crate/s: 
    b0ccal11   
 </PRE>
<!-- Author: ErrorHandler -->


<!-- Date: Thu Aug 26 13:36:38.730 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186684">186684</A>
TERMINATE: done TO errors on b0ccal11
<!-- Author: Gene(x2080) -->



<!-- Date: Thu Aug 26 13:38:55.489 2004 -->
<table border=0><td><tr><td bgcolor=#e0e030>Online DNS (fnsrvcdf2) was rebooted by David Tang (Data-comm group) 
request. It's up now.</td></tr></td></table>
<!-- Author: Sergei -->

<!-- Date: Thu Aug 26 13:42:55.040 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186685">186685</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [16,466,444] 
<!-- Author: Gene(x2080) -->


<!-- Date: Thu Aug 26 13:46:40.373 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186685">186685</A>
ACTIVE: <PRE>   
  Done TO. for crate/s: 
    b0ccal11   
 </PRE>
<!-- Author: ErrorHandler -->


<!-- Date: Thu Aug 26 13:51:38.717 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186685">186685</A>
ACTIVE: <PRE>   
  Done TO. for crate/s: 
    b0ccal11   
 </PRE>
<!-- Author: ErrorHandler -->


<!-- Date: Thu Aug 26 13:56:03.578 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186685">186685</A>
HALTED: <PRE>   
  Done TO. for crate/s: 
    b0ccal11   
 </PRE>
<!-- Author: ErrorHandler -->


<!-- Date: Thu Aug 26 14:11:25.385 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186685">186685</A>
HALTED: <PRE>   
  Done TO. for crate/s: 
    b0ccal11   
 </PRE>
<!-- Author: ErrorHandler -->



<!-- Date: Thu Aug 26 14:12:57.850 2004 -->
iFix computer S2 update done.  S1 started.
<!-- Author: jdl -->


<!-- Date: Thu Aug 26 14:47:58.827 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186685">186685</A>
HALTED: <PRE>   
  Done TO. for crate/s: 
    b0ccal11   
 </PRE>
<!-- Author: ErrorHandler -->


<!-- Date: Thu Aug 26 14:58:15.855 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186685">186685</A>
ACTIVE: <PRE>   
  Done TO. for crate/s: 
    b0ccal11   
 </PRE>
<!-- Author: ErrorHandler -->


<!-- Date: Thu Aug 26 15:07:35.723 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186685">186685</A>
HALTED: <PRE>   
  Done TO. for crate/s: 
    b0ccal11   
 </PRE>
<!-- Author: ErrorHandler -->


<!-- Date: Thu Aug 26 15:23:02.781 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186685">186685</A>
HALTED: <PRE>   
  Done TO. for crate/s: 
    b0ccal11   
 </PRE>
<!-- Author: ErrorHandler -->


<!-- Date: Thu Aug 26 15:24:04.702 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186685">186685</A>
HALTED: <PRE>   
  Done TO. for crate/s: 
    b0ccal11   
 </PRE>
<!-- Author: ErrorHandler -->



<!-- Date: Thu Aug 26 15:30:12.164 2004 -->
Cryo reports that iFix computers are back up.  Alarms are real.
<!-- Author: jdl -->



<!-- Date: Thu Aug 26 15:42:36.488 2004 -->
Plug HV Caen crate 12 tripped off.  Willis power cycled it and it came back.
<!-- Author: jdl -->

<!-- Date: Thu Aug 26 15:56:40.723 2004 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>15:55:01</TD>                                                       
<TD ALIGN=CENTER>::</TD>                                                
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->


<!-- Date: Thu Aug 26 15:59:21.965 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186685">186685</A>
ACTIVE: <PRE>   
  Done TO. for crate/s: 
    b0ccal11   
 </PRE>
<!-- Author: ErrorHandler -->


<!-- Date: Thu Aug 26 16:00:02.782 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=186685">186685</A>
ACTIVE: <PRE>   
  Done TO. for crate/s: 
    b0ccal11   
 </PRE>
<!-- Author: ErrorHandler -->



<!-- Date: Thu Aug 26 16:00:02.931 2004 -->
<font color=red>Shift Summary: </font>Another quiet day.  Arch work continues.  Roberto chased down some PSM problems.  IFix computers now on FERMI domain.  There remains one TDC in cot00 that is offline.  There are a few central PMTs with wrong voltages.  These can await Steve Hahn's return.  A few PSM problems remain. 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   110m high hurdle
Delivered Luminosity   no  
Acquired Luminosity    no  
Efficiency             no
   
</td></tr></table></pre>
<!-- Author: jdl -->


<!-- Date: Thu Aug 26 16:01:59.088 2004 -->
<font color=red>Shift Summary: </font><pre>x</pre> 
<!-- Author: jl -->