<!-- Title: Day shift. Mon Jul 22, 2002 -->
<!-- SciCo: Liss -->
<!-- DAQAce: Mario Spezziga -->
<!-- MonAce: Jedong Lee -->
<!-- CO: Mark Mattson -->
<!-- OpManager: Steve (Old timer) Hahn -->
<!-- Notes: -->
<pre>Start shift with 6hr access.</pre>


<!-- Date: Mon Jul 22 08:50:25 2002 -->
BSU HV monitor has been showing strange behaviour. Paged an expert. Answer was that they needed an access to solve this.
<!-- Author: Jedong -->


<!-- Date: Mon Jul 22 09:18:03 2002 -->
<pre>Access may only be 2 hrs (& controled, not supervised).  Waiting for word from 9AM meeting.</pre>
<!-- Author: Liss -->


<!-- Date: Mon Jul 22 09:43:17 2002 -->
<pre>The word is 3hr CONTROLLED. </pre>
<!-- Author: Liss -->


<!-- Date: Mon Jul 22 10:24:52 2002 -->
<pre>Fire alarm causes evacuation & dropped interlocks. 
It turns out that this was a fire dept. test - but no one here (including Dee and Ops) knew about this. 
 
Going to supervised access.</pre>
<!-- Author: Liss -->


<!-- Date: Mon Jul 22 11:12:20 2002 -->
I'm taking cmp00, cmx00, cmx01, imu00, and imu01 to test a muon calibration consumer.
<!-- Author: K.Lannon -->

<!-- Date: Mon Jul 22 11:26:47 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148701">148701</A>
ACTIVATE: SMX calibration
<!-- Author: ace 2080 -->



<!-- Date: Mon Jul 22 11:35:10 2002 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2002&action=view&page=-5068&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2002&action=view&page=-5068  width='300' align='top'></td><td>SMX calibration bad channels</a></td></tr></table>
<!-- Author: Mario -->

<!-- Date: Mon Jul 22 11:39:53 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148702">148702</A>
ACTIVATE: SMX calibration
<!-- Author: ace 2080 -->



<!-- Date: Mon Jul 22 11:44:06 2002 -->
BSU HV monitor works fine.
<!-- Author: Jedong -->


<!-- Date: Mon Jul 22 12:11:46 2002 -->
Miniplug access report:<br> 
swapped one inverting miniplug transition board with a new modified version of the board. Walter is going to modify the board which was taken out for future access.
<!-- Author: K. Terashi, W. Stuermer -->

<!-- Date: Mon Jul 22 12:15:40 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148706">148706</A>
 Activated at 2002.07.22 12:12:52
<!-- Author: RunControl -->


<!-- Date: Mon Jul 22 12:15:40 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148706">148706</A>
ACTIVATE: Spike run
<!-- Author: ace 2080 -->


<!-- Date: Mon Jul 22 12:18:09 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148706">148706</A>
 Terminated at 2002.07.22 12:15:23
<!-- Author: RunControl -->



<!-- Date: Mon Jul 22 12:22:57 2002 -->
Access report of SVX - Silicon 
============================== 
 
We did light output measurements on SB5W8L4 and SB5W7L3 and  
repaired the doim cable of SB5W2L0.  
 
Light output measurements of e584 and e573 are ok.  
  
<!-- Author: lester and cigdem  -->


<!-- Date: Mon Jul 22 12:24:09 2002 -->
<h3>Silicon Access Report #1</h3> 
Moved some Fibs around to try and recover some ladders and get 
rid of low level synchronicity errors.  <a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=117&frame=2&anchor=12:21:23'>Details</a>   
<!-- Author: SCN -->


<!-- Date: Mon Jul 22 12:39:25 2002 -->
<pre>Attempts to test the muon calibration consumer have failed.  It 
seemed that the consumer script wasn't being executed.  I may  
have had the run incorrectly configured.  Hopefully there will 
be more tries after consultation with experts. 
 
I'm done with cmp00, cmx00, cmx01, imu00, and imu01.</pre>
<!-- Author: K.Lannon -->

<!-- Date: Mon Jul 22 12:39:45 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148712">148712</A>
 Activated at 2002.07.22 12:37:30
<!-- Author: RunControl -->


<!-- Date: Mon Jul 22 12:40:06 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148712">148712</A>
ACTIVATE: Spike run
<!-- Author: ace 2080 -->
<!-- Comment: Mon Jul 22 13;18;51 comment by...M Mattson -->
<br>
Spike run was at 170ish Hz for L1, so no obvious loose cables.
<!-- Date: Mon Jul 22 12:40:13 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148712">148712</A>
 Terminated at 2002.07.22 12:38:02
<!-- Author: RunControl -->


<!-- Date: Mon Jul 22 12:49:42 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148713">148713</A>
 Activated at 2002.07.22 12:47:18
<!-- Author: RunControl -->
<!-- Comment: Mon Jul 22 12;52;51 comment by...Mario -->
Few crates in this run to test if the daq is ok. It seems fine indeed.
<!-- Date: Mon Jul 22 12:49:42 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148713">148713</A>
ACTIVATE: Cosmics
<!-- Author: ace 2080 -->



<!-- Date: Mon Jul 22 13:13:06 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148713">148713</A>
 Terminated at 2002.07.22 13:10:34
<!-- Author: RunControl -->


<!-- Date: Mon Jul 22 13:13:06 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148713">148713</A>
TERMINATE: Ending to start a calibration
<!-- Author: ace 2080 -->


<!-- Date: Mon Jul 22 13:16:37 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148716">148716</A>
ACTIVATE: QIE calibration
<!-- Author: ace 2080 -->



<!-- Date: Mon Jul 22 13:17:06 2002 -->
Used FCAL crates to check QIEs for the ADMEM with new installed transition board. Did QIE calibration on the west miniplug ADMEMs and loaded calibration constants. Also, tried to take some LED 
calibration runs on both miniplugs with HV on. The HV was turned  off after the runs.
<!-- Author: K. Terashi -->


<!-- Date: Mon Jul 22 13:29:13 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148718">148718</A>
ACTIVATE: Calibration of pcal03
<!-- Author: ace 2080 -->


<!-- Date: Mon Jul 22 13:39:30 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148719">148719</A>
 Activated at 2002.07.22 13:36:50
<!-- Author: RunControl -->


<!-- Date: Mon Jul 22 13:42:37 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148719">148719</A>
 Terminated at 2002.07.22 13:40:10
<!-- Author: RunControl -->


<!-- Date: Mon Jul 22 13:43:54 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148718">148718</A>
TERMINATE: Downloaded constants for pcal03
<!-- Author: ace 2080 -->


<!-- Date: Mon Jul 22 13:43:54 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148720">148720</A>
 Activated at 2002.07.22 13:41:08
<!-- Author: RunControl -->


<!-- Date: Mon Jul 22 13:44:30 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148720">148720</A>
 Terminated at 2002.07.22 13:42:01
<!-- Author: RunControl -->


<!-- Date: Mon Jul 22 13:46:02 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148721">148721</A>
 Activated at 2002.07.22 13:42:51
<!-- Author: RunControl -->


<!-- Date: Mon Jul 22 13:46:03 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148721">148721</A>
 Terminated at 2002.07.22 13:43:55
<!-- Author: RunControl -->


<!-- Date: Mon Jul 22 13:56:16 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148722">148722</A>
ACTIVATE: Plug laser calibration
<!-- Author: ace 2080 -->



<!-- Date: Mon Jul 22 14:04:58 2002 -->
<b>Plug Calorimeter Access Report</b><br> 
The details are in the 
<a href= 
'/cgi/elog/det-system-elog.pl?nb=plug&action=view&page=105&frame=2&anchor=12:05:21'> 
endplug e-log</a>.  Zhibin, Rick, and I went downstairs to check on the erratic pcal03 channel. 
<UL> 
<LI>Before the access, the channel was active, although weakened (low pedestals at 155). 
<LI>Took out ADMEM board, took out CAFE card, reseated both. 
<LI>Checked pedestals again.  Higher, but still low (167). 
<LI>Pulled out transition board.  Checked protection diode, which appeared normal (good value for resistance).  Put board back in. 
<LI>Took pedestals again.  Values back at normal (185). 
</UL> 
The experts believe that the problem is/was due to an intermittant connection.  (Electronics tend to die in a distinctive way, then stay dead, rather than changing states between good/bad.)  The channel (PHA eta 11W, phi 12) is good right now.  If it goes bad, again, we would not be surprised.  In which case, different experts (Howard and Willis, who deal with the cables and connections) will be consulted.<br> 
<br> 
If we were very lucky, reseating the boards was enough.  We didn't actually fix anything.  More likely, we will need to continue watching the channel for odd behavior.  If it goes bad (white channel in YMon PHA Tower plot), please have the CO make a note in the endplug e-log, let the SciCo/Ops Manager know, but do not need to call the ADMEM pager.
<!-- Author: M Mattson, R Tesarek, Z Huang -->
<!-- Comment: Mon Jul 22 14;16;37 comment by...M Mattson, CO -->
<br>
For contrast, this past weekend, the problem channel between 0-8 qietest pedestal, and YMon tower plot had a white box.
<!-- Date: Mon Jul 22 14:12:56 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148726">148726</A>
 Activated at 2002.07.22 14:10:44
<!-- Author: RunControl -->


<!-- Date: Mon Jul 22 14:13:25 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148726">148726</A>
ACTIVATE: Cosmics without CCAL and WCAL crates
<!-- Author: ace 2080 -->




<!-- Date: Mon Jul 22 14:22:56 2002 -->
<br><b>I/O Error </b></br><p> 
I've just tried to search for files in the look area, and the following Error message was returned: 
 
<pre> 
---------------------------------------------------------- 
> findfile 148722 
 
  Searching for files from run 148722 (0x244f2)...... 
 
  Cannot access /cdf/data08: I/O error 
----------------------------------------------------------- 
</pre>
<!-- Author: Phil Y. -->


<!-- Date: Mon Jul 22 14:40:45 2002 -->
<center><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-5072&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-5072' width='300'></a><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-5073&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-5073' width='300'></a><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-5074&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-5074' width='300'></a><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-5075&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-5075' width='300'></a></center>
<!-- Author: Phil Y./Howard Budd -->
<!-- Comment: Mon Jul 22 14;46;39 comment by...Phil/Howard -->
The above plug-laser plots show PHA west 
wedge12 tower3, which has had
a problem. The 1st plot shows a good earlier run 146406
on June 13. The next 3 plots show the channel when it is
bad. The 2nd plot shows the full range. The channel has
3 peaks. The highest peak looks like it is the regular laser signal. The 3rd
plots shows the lower peak, and  the 4th plot shows the middle peak.
The odd feature of the lower and middle peaks is laser values
are quantized in particular bins, with no events in adjacent bins.



<!-- Date: Mon Jul 22 14:47:36 2002 -->
<pre>Called Rick Colombo about data08 problems.  They are working on it. 
 
There is a disk controller problem on sgi2.  They are trying to find a fix.  At minimum they will reboot sgi2, but if they don't have an alternate disk chassis, a reboot will only be a temporary fix.</pre>
<!-- Author: Liss -->


<!-- Date: Mon Jul 22 14:53:53 2002 -->
<pre>We took severl timing scans for the central laser calibration 
study. 
 
Now triggering from the reference pulse - hope this is more 
stable that triggering form the laser. 
 
We had to change the calib interval from 33 to 20 
and changed the pipeline length to 1b in the pulser crate. 
 
We used a run type of central hadron laser and a non-null 
trigger table. In this config we were able to see events in the  
event display (previously it crashed) 
 
I suspect that when we gave it a trigger table the event display 
was able to find calibrations associated with it and did not 
crash... 
</pre>
<!-- Author: Frank Chlebana :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148743' target=_top>148743</a>) -->


<!-- Date: Mon Jul 22 15:14:58 2002 -->
<pre>SMX access report: 
 
Swapped 2 SMD boards with single channel problems. 
One on West Phi  = 6 (SMX5) 
One on West Phi = 5 (SMX6) 
 
Also checked Fan connectors which are used by PSM program 
and rseated BNB module to East PHi=9 and 11. These wedges had 
shown fans as being off intermittently in the past week.    
Don't think i fixed any connections, 
since fans were reading 100% prior to my intervention and they 
still read 100%.  I unmasked these crates in the PSM program. 
 
Unfortunately, West PHi=6 and 7 and the 2 top SW VME crates  
are now not readable by the PSM (purple states).  Roberta 
was notified and this fix will have to wait till the next  
access.  I masked these channels off.  </pre>
<!-- Author: k. byrum -->


<!-- Date: Mon Jul 22 15:21:40 2002 -->
<br> 
Earlier, for the laser calibration, Frank asked for the EventDisplay to be killed and launched on partition 1.  Recently, silicon expert asked for SVXMon to use partition 2.<br> 
<br> 
Obviously, before we get beam, these consumers should be reset to partition 0.
<!-- Author: M Mattson -->

<!-- Date: Mon Jul 22 15:37:39 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148726">148726</A>
 Terminated at 2002.07.22 15:35:13
<!-- Author: RunControl -->
<!-- Comment: Mon Jul 22 15;44;44 comment by...K.Lannon -->
I've taken cmp00, cmx00, cmx01, imu00, and imu01 to have another go at testing the muon calibration consumer.
<!-- Date: Mon Jul 22 15:37:39 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148726">148726</A>
TERMINATE: End to give muon crates away
<!-- Author: ace 2080 -->



<!-- Date: Mon Jul 22 15:57:46 2002 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>15:55:02</TD>                                                       
<TD ALIGN=CENTER>::</TD>                                                
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Mon Jul 22 16:03:20 2002 -->
<font color=red>Shift Summary: </font><pre>A fun filled day. 
1) Controlled access 
2) Fire alarm generated by Firemen testing system which we didn't know 
3) Controlled access becomes Supervised. 
4) Plug work, Silicon work, Shower max work 
5) Stacking in progress.  Stack = 74mA .  </pre> 
<!-- Author: Liss -->