<!-- Title: Day shift. Fri Feb 21, 2003 -->
<!-- SciCo: Paul Karchin -->
<!-- DAQAce: Sal Rappoccio -->
<!-- MonAce: Gavril Giurgiu -->
<!-- CO: Yves Kemp -->
<!-- OpManager: Eric James -->
<!-- Notes: -->
<pre>store in progress - taking data 
</pre>


<!-- Date: Fri Feb 21 08:29:19 2003 -->
Sync error in L2 pipeline, event 3120028. HRR is not recovering. Run is halted now.
<!-- Author: Sal :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159175' target=_top>159175</a>) -->
<!-- Comment: Fri Feb 21 08;33;37 comment by...Sal -->
Rebooting b0tsi00 from vxworks.

<!-- Date: Fri Feb 21 08:39:45 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159175">159175</A>
 Terminated at 2003.02.21 08:39:08
<!-- Author: RunControl -->


<!-- Date: Fri Feb 21 08:39:46 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159175">159175</A>
TERMINATE: Stopping run to fix b0tsi00 problems.
<!-- Author: Sal -->



<!-- Date: Fri Feb 21 08:45:37 2003 -->
<pre>L2 Problem: minicom tells us: 
 
Read L2 prescales 
XtrpMon: XTRP buffno 1 != event buffno 0 
SvtMon: SVT buffno 1 != event buffno 2 
IsolistMon: Error on pass 2, cluster 0: got 0, expected 3(eta: 30 phi: 2) 
SvtMon: SVT buffno 1 != event buffno 3 
SvtMon: SVT buffno 0 != event buffno 1 
SvtMon: SVT buffno 0 != event buffno 2 
XtrpMon: XTRP buffno 2 != event buffno 3 
XtrpMon: XTRP buffno 3 != event buffno 0 
XtrpMon: XTRP buffno 3 != event buffno 1 
TrackListMonitor saw 3 sequential errors at Feb 21 08:31:43 2003 
SvtMon: SVT buffno 0 != event buffno 1 
FaultManager: Fatal error, sending L2 timeout (L2A 3120154) 
 
Heather Ray is coming in to look at these problems.</pre>
<!-- Author: Sal :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159175' target=_top>159175</a>) -->
<!-- Comment: Fri Feb 21 08;46;36 comment by...Sal -->
Sorry, I am restarting run to try to recover.

<!-- Comment: Fri Feb 21 08;59;06 comment by...Sal -->
Everything fine so far.
<!-- Comment: Fri Feb 21 09;12;11 comment by...hray -->
RC error logger shows TrackList and SVTList Sync errors.  The minicom confirms this : due to buffer number mismatches.  The TrigMon plots from this run show no indication of errors, however.  We will examine these few events offline from our error stream.  This run should not be marked bad.
<!-- Date: Fri Feb 21 08:49:26 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159176">159176</A>
 Activated at 2003.02.21 08:48:59
<!-- Author: RunControl -->


<!-- Date: Fri Feb 21 08:49:27 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159176">159176</A>
ACTIVATE: Restart run after previous L2 problems. Fingers crossed
<!-- Author: Sal -->
<!-- Comment: Fri Feb 21 08;49;48 comment by...Sal -->
Seems to be fine so far...


<!-- Comment: Fri Feb 21 11;35;51 comment by...Saltzberg -->
Keep in mind that L2 is just reporting the error.  It 
actually checks all the BC's and forces you to HRR if 
somebody upstream is out of sync--and forces coldstart
if HRR does not bring them back in sync.  For a long
time people (including us) had confused this situation with actual L2 decision problems.

<!-- Date: Fri Feb 21 08:57:31 2003 -->
current lumi  
B0 13.95 E 30 
D0 13.65 E 30
<!-- Author: Paul -->



<!-- Date: Fri Feb 21 09:10:12 2003 -->
DTO from b0cot15, Bunchcounters in slot 3 (BC=106) and slot 16 (BC=62) disagree (at least). Auto HRR cleared
<!-- Author: Sal :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159176' target=_top>159176</a>) -->
<!-- Comment: Fri Feb 21 09;25;39 comment by...Sal -->
b0cot19: Bunchcounters in slot 5 (BC=3) and slot 10 (BC=62) disagree (at least)
<!-- Date: Fri Feb 21 09:13:19 2003 -->
Stuck Cellid S/B1/W1/L2/C5-9
<!-- Author: Sal :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159176' target=_top>159176</a>) -->
<!-- Comment: Fri Feb 21 09;36;46 comment by...Sal -->
Stuck Cellid in 5 events in Silicon/S/B1/W1/L2/C5-9
<!-- Date: Fri Feb 21 09:19:59 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1286&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1286' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1287&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1287' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1288&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1288' width='300'></a></center>
<!-- Author: Gavril -->



<!-- Date: Fri Feb 21 09:44:18 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159176">159176</A>
 Terminated at 2003.02.21 09:43:57
<!-- Author: RunControl -->


<!-- Date: Fri Feb 21 09:44:19 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159176">159176</A>
TERMINATE: Stop the run to do COT test runs
<!-- Author: Sal -->



<!-- Date: Fri Feb 21 09:45:54 2003 -->
DTO from b0pcal08. Stopping run for COT test anyway.
<!-- Author: Sal :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159177' target=_top>159177</a>) -->

<!-- Date: Fri Feb 21 09:47:41 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159177">159177</A>
TERMINATE: EVB is red, cleanup in process.
<!-- Author: Sal -->



<!-- Date: Fri Feb 21 09:57:39 2003 -->
L3 was all red, and EVB wouldn't pass coldstart. Cleaned up EVB and L3 farm. Coming back.
<!-- Author: Sal -->

<!-- Date: Fri Feb 21 10:01:26 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159178">159178</A>
 Activated at 2003.02.21 10:01:12
<!-- Author: RunControl -->


<!-- Date: Fri Feb 21 10:02:03 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159178">159178</A>
ACTIVATE: Begin COT test run with threshold 215 mV
<!-- Author: Sal -->



<!-- Date: Fri Feb 21 10:03:14 2003 -->
DTO from b0wcal04: Bunchcounters in slot 16 (BC=0) and slot 6 (BC=125) disagree (at least)
<!-- Author: Sal :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159178' target=_top>159178</a>) -->


<!-- Date: Fri Feb 21 10:03:40 2003 -->
 b0cot17:Messenger:10:03:14 AM->Event 8446: Bunchcounters in slot 3 (BC=65) and slot 18 (BC=80) disagree (at least
<!-- Author: Sal :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159178' target=_top>159178</a>) -->


<!-- Date: Fri Feb 21 10:06:33 2003 -->
<table border=0><td><tr><td bgcolor=#e0e030>Since sometime in August, 2002, the loss rate B0RAT1 (losses in time with the bunches) was recorded in ACNET a factor 10 lower than the actual rate due to an incorrect switch-setting on the rate meter.  This switch was set correctly after the January, 2003 shutdown.  The rate being recorded now in ACNET should be the real rate.  The rates recorded in ACNET since August should be multiplied by 10 to get the true rate...</td></tr></td></table>
<!-- Author: R. Vidal -->

<!-- Date: Fri Feb 21 10:07:54 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159178">159178</A>
 Terminated at 2003.02.21 10:07:32
<!-- Author: RunControl -->


<!-- Date: Fri Feb 21 10:07:55 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159178">159178</A>
TERMINATE: End run after about 10k events at L3 for COT test run
<!-- Author: Sal -->


<!-- Date: Fri Feb 21 10:12:17 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159179">159179</A>
 Activated at 2003.02.21 10:12:12
<!-- Author: RunControl -->


<!-- Date: Fri Feb 21 10:13:02 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159179">159179</A>
ACTIVATE: Begin COT test run, threshold = 205 mV
<!-- Author: Sal -->



<!-- Date: Fri Feb 21 10:13:02 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1295&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1295' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1296&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1296' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1297&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1297' width='300'></a></center>
<!-- Author: Gavril -->


<!-- Date: Fri Feb 21 10:13:49 2003 -->
PSM alarm. clears in a few seconds.
<!-- Author: Gavril :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159179' target=_top>159179</a>) -->

<!-- Date: Fri Feb 21 10:18:43 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159179">159179</A>
 Terminated at 2003.02.21 10:18:15
<!-- Author: RunControl -->


<!-- Date: Fri Feb 21 10:18:44 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159179">159179</A>
TERMINATE: End COT test run after about 10k L3 events
<!-- Author: Sal -->


<!-- Date: Fri Feb 21 10:22:38 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159180">159180</A>
 Activated at 2003.02.21 10:22:39
<!-- Author: RunControl -->


<!-- Date: Fri Feb 21 10:22:57 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159180">159180</A>
ACTIVATE: Begin COT test run, threshold = 200 mV
<!-- Author: Sal -->



<!-- Date: Fri Feb 21 10:24:05 2003 -->
Pipeline out of synch in 89 silicon readout chips
<!-- Author: Sal :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159180' target=_top>159180</a>) -->

<!-- Date: Fri Feb 21 10:28:12 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159180">159180</A>
 Terminated at 2003.02.21 10:28:09
<!-- Author: RunControl -->


<!-- Date: Fri Feb 21 10:28:29 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159180">159180</A>
TERMINATE: End COT test run after about 10k L3 events
<!-- Author: Sal -->



<!-- Date: Fri Feb 21 10:30:53 2003 -->
Completed COT threshold scan runs, all about 10k L3 events: 
<p> 
Run 159178 --> DTH = 215 mV <br> 
Run 159179 --> DTH = 205 mV <br> 
Run 159180 --> DTH = 200 mV
<!-- Author: ambrose -->


<!-- Date: Fri Feb 21 10:36:59 2003 -->
CSL error: CSL Driver stopped, cleared by the next time PROCMON came to it. This was in between runs.
<!-- Author: Sal -->

<!-- Date: Fri Feb 21 10:37:25 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159181">159181</A>
 Activated at 2003.02.21 10:37:23
<!-- Author: RunControl -->


<!-- Date: Fri Feb 21 10:37:32 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159181">159181</A>
ACTIVATE: Begin TOF trigger special run
<!-- Author: Sal -->



<!-- Date: Fri Feb 21 10:38:22 2003 -->
Pipeline out of synch in 89 silicon readout chips
<!-- Author: Sal :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159181' target=_top>159181</a>) -->

<!-- Date: Fri Feb 21 11:01:50 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159181">159181</A>
 Terminated at 2003.02.21 11:01:47
<!-- Author: RunControl -->


<!-- Date: Fri Feb 21 11:02:21 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159181">159181</A>
TERMINATE: END TOF test run. Lots of bunch counter problems, too.
<!-- Author: Sal -->



<!-- Date: Fri Feb 21 11:03:45 2003 -->
<table border=0><td><tr><td bgcolor=#e0e030><pre>Lots of errors!!! Here is a sampling. Please see error log file for full report. Fatal clock error amongst: 
 
(MLE) b0cot18:Messenger:10:59:34 AM-&gt;Runtime Error 1, Event 52195: See nonsequential event number (old 2, new 9) in L2 message 
(MLE) b0ccal12:Messenger:10:59:34 AM-&gt;Event 52196: Bunchcounters in slot 17 (BC=75) and slot 21 (BC=78) disagree (at least) 
(MLE) b0cot12:Messenger:10:59:34 AM-&gt;Runtime Error 2, Event 52195: Timeout on global TDC Done 
(MLE) b0cot18:Messenger:10:59:34 AM-&gt;Event 52195: Bunchcounters in slot 4 (BC=72) and slot 5 (BC=15) disagree (at least) 
(MLE) b0cot12:Messenger:10:59:34 AM-&gt;Runtime Error 2, Event 52195: TDC in slot 4 not ready for readout 
(MLE) b0ccal12:Messenger:10:59:35 AM-&gt;Event 52196: Bunchcounters in slot 17 (BC=75) and slot 16 (BC=78) disagree (at least) 
(MLE) b0cot18:Messenger:10:59:34 AM-&gt;Event 52195: Bunchcounters in slot 4 (BC=72) and slot 6 (BC=15) disagree (at least) 
(MLE) b0cot12:Messenger:10:59:35 AM-&gt;Runtime Error 2, Event 52195: TDC in slot 5 not ready for readout 
(MLE) b0ccal12:Messenger:10:59:35 AM-&gt;Event 52196: Bunchcounters in slot 17 (BC=75) and slot 18 (BC=78) disagree (at least) 
(MLE) b0cot18:Messenger:10:59:35 AM-&gt;Event 52195: Bunchcounters in slot 4 (BC=72) and slot 9 (BC=15) disagree (at least) 
(MLE) b0cot12:Messenger:10:59:35 AM-&gt;Runtime Error 2, Event 52195: TDC in slot 7 not ready for readout</pre></td></tr></td></table>
<!-- Author: Sal :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159181' target=_top>159181</a>) -->
<!-- Comment: Fri Feb 21 11;05;01 comment by...Sal -->
This fried errlog. Clock status is still red...


<!-- Comment: Fri Feb 21 11;15;44 comment by...Sal -->
Sorry, a more descriptive entry for restarting errlog:
<ol>
<li> errlog was spinning lots of entries in the window (probably
couldn't keep up with what was going on)</li>
<li> Issued abort from crate menu.</li> 
<li> Went to IDLE state, then was fine.</li>
</ol>
<!-- Date: Fri Feb 21 11:06:20 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1298&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1298' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1299&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1299' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1300&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1300' width='300'></a></center>
<!-- Author: Gavril -->
<!-- Comment: Fri Feb 21 11;16;19 comment by...Gavril -->
note high spike in C:B0PBSM
<!-- Date: Fri Feb 21 11:07:18 2003 -->
<table border=0><td><tr><td bgcolor=#e0e030><pre>Clock errors: 
 
(MLE) b0dap59.fnal.gov:AWT-EventQueue-0:11:03:36 AM-&gt;rc.vxmon.ClockStatus@7e208823 
  Status:           0x1c100 
  Fatal Errors: 
    SEQ_CSR1_ERR_3_SEQ_HOLD 
    PCC_CSR_5_RF_MISSING_ERROR 
    PCC_CSR_6_SYNC_TIMING_ERROR 
    PCC_CSR_7_SYNC_MISSING_ERROR 
  Time: :           1045847012 
  TimeNanoSeconds:  0 
  SeqCSR0:          0xfd27 
  SeqCSR1:          0xc 
  PccDelay:         0xffb0 
  PccCSR:           0xffe</pre></td></tr></td></table>
<!-- Author: Sal :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159182' target=_top>159182</a>) -->
<!-- Comment: Fri Feb 21 11;07;48 comment by...Sal -->
After Steve Hahn cleared clock errors on board, all is fine and dandy again.

<!-- Date: Fri Feb 21 11:08:38 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159182">159182</A>
 Activated at 2003.02.21 11:08:36
<!-- Author: RunControl -->


<!-- Date: Fri Feb 21 11:09:54 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159182">159182</A>
ACTIVATE: Begin TOF trigger run. HIP threshold 16k.
<!-- Author: Sal -->





<!-- Date: Fri Feb 21 11:16:36 2003 -->
Stuck Cellid S/B1/W8/L2/C5-9
<!-- Author: Sal :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159182' target=_top>159182</a>) -->
<!-- Comment: Fri Feb 21 11;20;31 comment by...Sal -->
again

<!-- Comment: Fri Feb 21 11;23;23 comment by...Sal -->
Again

<!-- Date: Fri Feb 21 11:27:28 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159182">159182</A>
 Terminated at 2003.02.21 11:27:20
<!-- Author: RunControl -->


<!-- Date: Fri Feb 21 11:27:36 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159182">159182</A>
TERMINATE: END TOF test run
<!-- Author: Sal -->



<!-- Date: Fri Feb 21 11:43:43 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159183">159183</A>
 Activated at 2003.02.21 11:43:39
<!-- Author: RunControl -->


<!-- Date: Fri Feb 21 11:44:31 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159183">159183</A>
ACTIVATE: Begin SVT test run
<!-- Author: Sal -->



<!-- Date: Fri Feb 21 11:44:39 2003 -->
Pipeline out of synch in 148 silicon readout chip
<!-- Author: Sal :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159183' target=_top>159183</a>) -->

<!-- Date: Fri Feb 21 11:54:09 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159183">159183</A>
 Terminated at 2003.02.21 11:53:39
<!-- Author: RunControl -->


<!-- Date: Fri Feb 21 11:54:10 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159183">159183</A>
TERMINATE: End SVT test run
<!-- Author: Sal -->


<!-- Date: Fri Feb 21 12:02:32 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159184">159184</A>
 Activated at 2003.02.21 12:00:44
<!-- Author: RunControl -->


<!-- Date: Fri Feb 21 12:02:33 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159184">159184</A>
ACTIVATE: Begin SVT test run.
<!-- Author: Sal -->



<!-- Date: Fri Feb 21 12:04:42 2003 -->
PSM alarm, 1RR18D_1. This happens pretty regularly, at least three or four times a shift, as far as this ace team's meager experience goes.
<!-- Author: Sal :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159184' target=_top>159184</a>) -->


<!-- Date: Fri Feb 21 12:05:36 2003 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2003&action=view&page=-1303&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-1303  width='300' align='top'></td><td>g</a></td></tr></table>
<!-- Author: g -->


<!-- Date: Fri Feb 21 12:07:46 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1304&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1304' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1305&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1305' width='300'></a></center>
<!-- Author: Gavril -->

<!-- Date: Fri Feb 21 12:11:56 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159184">159184</A>
 Terminated at 2003.02.21 12:11:46
<!-- Author: RunControl -->


<!-- Date: Fri Feb 21 12:12:11 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159184">159184</A>
TERMINATE: End SVT test run
<!-- Author: Sal -->


<!-- Date: Fri Feb 21 12:18:00 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159185">159185</A>
 Activated at 2003.02.21 12:17:45
<!-- Author: RunControl -->


<!-- Date: Fri Feb 21 12:18:01 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159185">159185</A>
ACTIVATE: Begin SVT test run
<!-- Author: Sal -->


<!-- Date: Fri Feb 21 12:23:12 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159185">159185</A>
 Terminated at 2003.02.21 12:23:11
<!-- Author: RunControl -->


<!-- Date: Fri Feb 21 12:23:33 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159185">159185</A>
TERMINATE: End SVT test run
<!-- Author: Sal -->


<!-- Date: Fri Feb 21 12:37:06 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159186">159186</A>
ACTIVATE: Begin Silicon D-mode calib, DPS ON
<!-- Author: Sal -->



<!-- Date: Fri Feb 21 12:38:11 2003 -->
MCR called: store will end at 15:30
<!-- Author: Paul -->

<!-- Date: Fri Feb 21 12:42:22 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159186">159186</A>
TERMINATE: End SIlicon D-mode calib, DPS-ON
<!-- Author: Sal -->


<!-- Date: Fri Feb 21 12:46:16 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159187">159187</A>
ACTIVATE: Begin Silicon D-mode, DPS OFF
<!-- Author: Sal -->


<!-- Date: Fri Feb 21 12:50:27 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159187">159187</A>
TERMINATE: End SIlicon D-mode calib, DPS OFF
<!-- Author: Sal -->



<!-- Date: Fri Feb 21 12:53:57 2003 -->
Hearbeat alarm from Muon Scint PC. The PC is downloading some constants into the DB, and should be fine. Oleg said to ignore it.
<!-- Author: Sal -->


<!-- Date: Fri Feb 21 13:02:22 2003 -->
During 'partition' & 'cold start' transition, many imon boxes 
turned pink.  Jason is here (silicon general pager person) looking at the problem.
<!-- Author: kaori -->
<!-- Comment: Fri Feb 21 14;08;15 comment by...Jason Nielsen -->
A current jump in all of Layers 0,1 seems to have happened during a failed ColdStart partition when crates were hanging.  The Abort transition at 13:05 brought the silicon DVDD currents back to normal.
For more details, see 
<a href=
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=142&frame=2&anchor=13:58:11'>
this entry in the silicon e-log</a>.
<!-- Date: Fri Feb 21 13:03:54 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159188">159188</A>
TERMINATE: Everything hanging on coldstart, abort and restart run
<!-- Author: Sal -->


<!-- Date: Fri Feb 21 13:08:48 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159189">159189</A>
 Activated at 2003.02.21 13:08:37
<!-- Author: RunControl -->


<!-- Date: Fri Feb 21 13:09:03 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159189">159189</A>
ACTIVATE: Begin run, silicon wedges now green
<!-- Author: Sal -->



<!-- Date: Fri Feb 21 13:11:53 2003 -->
<table border=0><td><tr><td bgcolor=#e0e030><pre>I have modified physics run's run-configurations to followning: 
 
Deleted: AAA_SHOTSETUP_ROL, AAA_NOSILICON_ROL, AAA_CURRENT_ROL. 
update:  AAA_SHOTSETUP, AAA_NOSILICON, AAA_CURRENT. 
 
There are no more "non-ROL" partitions.  "ROL" is default. 
Newest physics table PHYSICS_1_04[4,255,357] is to be used. 
</pre></td></tr></td></table>
<!-- Author: kaori -->
<!-- Comment: Fri Feb 21 13;54;01 comment by...kaori -->
Also updated ACE_COSMICS_ALL run_config to run the most
updated cosmic trigger table cosmics[9,257,318].  The newest
default physics table and cosmic table have COT trip trigger
in them.
<!-- Date: Fri Feb 21 13:18:48 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1308&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1308' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1309&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1309' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1310&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1310' width='300'></a></center>
<!-- Author: Gavril -->
<!-- Comment: Fri Feb 21 13;32;21 comment by...Gavril -->
note that in the right hand side plot, T:SBDMS is not plotted. I tried a few times with no success.
<!-- Date: Fri Feb 21 13:19:46 2003 -->
Pipeline out of synch in 78 silicon readout chips
<!-- Author: Sal :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159189' target=_top>159189</a>) -->


<!-- Date: Fri Feb 21 13:30:45 2003 -->
When tried to do the hourly SI_TEST2 plot, ACNET got stuck before ploting T:SBDMS. The the rest of the curves were polted, but not T:SBDMS. 
<!-- Author: Gavril :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159189' target=_top>159189</a>) -->



<!-- Date: Fri Feb 21 13:42:13 2003 -->
 b0cot17:Messenger:1:40:15 PM->Event 196425: Bunchcounters in slot 3 (BC=62) and slot 19 (BC=118) disagree (at least)
<!-- Author: Sal :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159189' target=_top>159189</a>) -->


<!-- Date: Fri Feb 21 13:48:23 2003 -->
Stuck Cellid in 5 events in Silicon/S/B1/W6/L2/C0-
<!-- Author: Sal :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159189' target=_top>159189</a>) -->



<!-- Date: Fri Feb 21 14:07:57 2003 -->
heartbeat alarms from PEM, PHA, PSH and ICI.
<!-- Author: Gavril :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159189' target=_top>159189</a>) -->
<!-- Comment: Fri Feb 21 14;08;49 comment by...Gavril -->
first 3 cleared.


<!-- Comment: Fri Feb 21 14;20;42 comment by...Gavril -->
the ICI heartbeat pink icon turned green.
<!-- Date: Fri Feb 21 14:17:48 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1311&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1311' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1312&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1312' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1313&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1313' width='300'></a></center>
<!-- Author: Gavril -->
<!-- Comment: Fri Feb 21 14;18;43 comment by...Gavril -->
note the strange shape of T:SBDMS plot.



<!-- Date: Fri Feb 21 14:48:20 2003 -->
<pre>Could not find reference plot for BeamMon/BeamMonitor2 
I was wondering why there are no entries at all.</pre>
<!-- Author: Yves :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159189' target=_top>159189</a>) -->


<!-- Date: Fri Feb 21 14:48:45 2003 -->
A few wedges in SVX B0 AND B1 turned blue for a second. It happened the second time this shift. 
<!-- Author: Gavril :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159189' target=_top>159189</a>) -->
<!-- Comment: Fri Feb 21 14;53;13 comment by...Gavril -->
again - also B2 wedges. 


<!-- Date: Fri Feb 21 14:55:29 2003 -->
offine CO report: I see that the global good run bit for  
all runs since run 158644 is zero indicating the runs 
are not good. I don't why all runs are apparently  
marked bad.  
<!-- Author: Tony Vaiciulis -->


<!-- Date: Fri Feb 21 15:03:02 2003 -->
<table><tr><td><a href=/elog/notebooks/2003/Fri_Feb_21_2003_Day.150256.ps align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-1315  align='top'></td><td>pink ladder.</a></td></tr></table>
<!-- Author: Gavril -->
<!-- Comment: Fri Feb 21 15;08;00 comment by...Gavril -->
it keeps turning pink. current well bellow the minimum limit(505). paged silicon.
<!-- Comment: Fri Feb 21 15;25;00 comment by...Jason Nielsen -->
The hint here is that wedges 4,5,6 from B0,B1,B2 all show these 
current drops.  All of these ladders are powered from the same CAEN crate (SW Bot Top).  These drops are the read errors from this crate.  (The read errors also cause the wash of blue IMon cells mentioned in the previous entry.)
<!-- Date: Fri Feb 21 15:04:23 2003 -->
cyro tech calls - says gas alarm is silenced - next shift tech will attempt to clear it and will call expert if needed - alarm 
is "false"
<!-- Author: Paul -->
<!-- Comment: Fri Feb 21 15;05;59 comment by...Paul -->
Alarm was for pressure differential in outside enclosure.



<!-- Date: Fri Feb 21 15:16:38 2003 -->
b0svx06:Messenger:3:15:25 PM->Silicon Timeout:!DONE- Slots:  20:f840
<!-- Author: Sal :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159189' target=_top>159189</a>) -->

<!-- Date: Fri Feb 21 15:24:46 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159189">159189</A>
 Terminated at 2003.02.21 15:24:43
<!-- Author: RunControl -->



<!-- Date: Fri Feb 21 15:25:01 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159189">159189</A>
TERMINATE: End run due to store dropping soon
<!-- Author: Sal -->



<!-- Date: Fri Feb 21 15:28:48 2003 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2003&action=view&page=-1316&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-1316  width='300' align='top'></td><td>Some X-sections reach a factor two between measured and expected. Did not call an expert, but one has to keep an eye on it.</a></td></tr></table>
<!-- Author: Yves -->


<!-- Date: Fri Feb 21 15:46:22 2003 -->
We had a problem bringing the SVX HV to standby.  One of the SVX wedges does not go down.  Steve Nahn sets this wedge individually and notes that this problem has occurred before.
<!-- Author: Paul -->


<!-- Date: Fri Feb 21 15:54:46 2003 -->
<TABLE bgcolor=#cdc9a5> 
<TR><TH>Date</th><th>Time</th><th>BLM</th><th colspan=2>Dose</th></tr> 
<tr><td>2003.02.21</td><td>15:52:38</td><td>W Inner BLM</td><td>0.00</td><td>RADS</td></tr> 
<tr><td>2003.02.21</td><td>15:52:38</td><td>W Outer BLM</td><td>0.00</td><td>RADS</td></tr> 
<tr><td>2003.02.21</td><td>15:52:38</td><td>E Inner BLM</td><td>0.00</td><td>RADS</td></tr> 
<tr><td>2003.02.21</td><td>15:52:38</td><td>E Outer BLM</td><td>0.00</td><td>RADS</td></tr> 
</TABLE> 
<strong>Integrated dosage</strong> 
  
<!-- Author: Gavril -->
<!-- Comment: Fri Feb 21 17;18;49 comment by...Gavril -->
The above numbers are not correct. They were reset after beam was aborted.
<!-- Date: Fri Feb 21 15:55:10 2003 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159175>
159175</A> 
x26DC7</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_1_04 [4,255,357]</TD>                                       
<TD ALIGN=CENTER>01:58:13</TD>                                          
<TD ALIGN=CENTER>08:39:08</TD>                                          
<TD ALIGN=CENTER>06:17:53</TD>                                          
<TD ALIGN=RIGHT>    210,349,336</TD>                                    
<TD ALIGN=RIGHT>     3,120,059</TD>                                     
<TD ALIGN=RIGHT>     909,815</TD>                                       
<TD ALIGN=RIGHT>  388.335</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159176>
159176</A> 
x26DC8</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_1_04 [4,255,357]</TD>                                       
<TD ALIGN=CENTER>08:48:59</TD>                                          
<TD ALIGN=CENTER>09:43:57</TD>                                          
<TD ALIGN=CENTER>00:51:44</TD>                                          
<TD ALIGN=RIGHT>     30,154,745</TD>                                    
<TD ALIGN=RIGHT>       371,225</TD>                                     
<TD ALIGN=RIGHT>     118,510</TD>                                       
<TD ALIGN=RIGHT>   42.739</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159178>
159178</A> 
x26DCA</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_1_04 [4,255,357]</TD>                                       
<TD ALIGN=CENTER>10:01:12</TD>                                          
<TD ALIGN=CENTER>10:07:32</TD>                                          
<TD ALIGN=CENTER>00:05:22</TD>                                          
<TD ALIGN=RIGHT>      2,602,361</TD>                                    
<TD ALIGN=RIGHT>        37,720</TD>                                     
<TD ALIGN=RIGHT>      11,409</TD>                                       
<TD ALIGN=RIGHT>    4.088</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>0</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159179>
159179</A> 
x26DCB</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_1_04 [4,255,357]</TD>                                       
<TD ALIGN=CENTER>10:12:12</TD>                                          
<TD ALIGN=CENTER>10:18:15</TD>                                          
<TD ALIGN=CENTER>00:05:16</TD>                                          
<TD ALIGN=RIGHT>      2,469,960</TD>                                    
<TD ALIGN=RIGHT>        37,157</TD>                                     
<TD ALIGN=RIGHT>      11,017</TD>                                       
<TD ALIGN=RIGHT>    4.035</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>0</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159180>
159180</A> 
x26DCC</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_1_04 [4,255,357]</TD>                                       
<TD ALIGN=CENTER>10:22:39</TD>                                          
<TD ALIGN=CENTER>10:28:09</TD>                                          
<TD ALIGN=CENTER>00:05:08</TD>                                          
<TD ALIGN=RIGHT>      2,428,786</TD>                                    
<TD ALIGN=RIGHT>        39,312</TD>                                     
<TD ALIGN=RIGHT>      10,662</TD>                                       
<TD ALIGN=RIGHT>    3.707</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>0</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159181>
159181</A> 
x26DCD</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>TEST_TOF_HIP_HACK [1,249,318]</TD>                                  
<TD ALIGN=CENTER>10:37:23</TD>                                          
<TD ALIGN=CENTER>11:01:47</TD>                                          
<TD ALIGN=CENTER>00:20:28</TD>                                          
<TD ALIGN=RIGHT>         53,150</TD>                                    
<TD ALIGN=RIGHT>        52,995</TD>                                     
<TD ALIGN=RIGHT>      52,194</TD>                                       
<TD ALIGN=RIGHT>   15.759</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159182>
159182</A> 
x26DCE</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>TEST_TOF_HIP_HACK [1,249,318]</TD>                                  
<TD ALIGN=CENTER>11:08:36</TD>                                          
<TD ALIGN=CENTER>11:27:20</TD>                                          
<TD ALIGN=CENTER>00:18:01</TD>                                          
<TD ALIGN=RIGHT>         25,197</TD>                                    
<TD ALIGN=RIGHT>        25,185</TD>                                     
<TD ALIGN=RIGHT>      25,185</TD>                                       
<TD ALIGN=RIGHT>   13.570</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159183>
159183</A> 
x26DCF</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>SVT_CALIBRATION [7,221,318]</TD>                                    
<TD ALIGN=CENTER>11:43:39</TD>                                          
<TD ALIGN=CENTER>11:53:39</TD>                                          
<TD ALIGN=CENTER>00:09:41</TD>                                          
<TD ALIGN=RIGHT>      2,357,937</TD>                                    
<TD ALIGN=RIGHT>        20,249</TD>                                     
<TD ALIGN=RIGHT>      20,249</TD>                                       
<TD ALIGN=RIGHT>    7.031</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>0</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159184>
159184</A> 
x26DD0</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>SVT_CALIBRATION [7,221,318]</TD>                                    
<TD ALIGN=CENTER>12:00:44</TD>                                          
<TD ALIGN=CENTER>12:11:46</TD>                                          
<TD ALIGN=CENTER>00:09:54</TD>                                          
<TD ALIGN=RIGHT>      2,385,046</TD>                                    
<TD ALIGN=RIGHT>        20,397</TD>                                     
<TD ALIGN=RIGHT>      20,397</TD>                                       
<TD ALIGN=RIGHT>    6.931</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>0</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159185>
159185</A> 
x26DD1</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>SVT_CALIBRATION [7,221,318]</TD>                                    
<TD ALIGN=CENTER>12:17:45</TD>                                          
<TD ALIGN=CENTER>12:23:11</TD>                                          
<TD ALIGN=CENTER>00:05:19</TD>                                          
<TD ALIGN=RIGHT>      1,270,774</TD>                                    
<TD ALIGN=RIGHT>        10,964</TD>                                     
<TD ALIGN=RIGHT>      10,964</TD>                                       
<TD ALIGN=RIGHT>    3.803</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>0</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159189>
159189</A> 
x26DD5</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_1_04 [4,255,357]</TD>                                       
<TD ALIGN=CENTER>13:08:37</TD>                                          
<TD ALIGN=CENTER>15:24:43</TD>                                          
<TD ALIGN=CENTER>02:14:24</TD>                                          
<TD ALIGN=RIGHT>     66,319,414</TD>                                    
<TD ALIGN=RIGHT>       837,503</TD>                                     
<TD ALIGN=RIGHT>     271,250</TD>                                       
<TD ALIGN=RIGHT>   90.347</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>15:55:02</TD>                                                       
<TD ALIGN=CENTER>10:43:15</TD>                                          
<TD ALIGN=RIGHT>     320,416,706</TD>                                   
<TD ALIGN=RIGHT>      4,572,766</TD>                                    
<TD ALIGN=RIGHT>     1,461,652</TD>                                     
<TD ALIGN=RIGHT>  580.344</TD>                                          
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Fri Feb 21 15:55:17 2003 -->
CLC quiet time led calibration is completed.
<!-- Author: Paul -->


<!-- Date: Fri Feb 21 15:58:36 2003 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2003&action=view&page=-1317&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-1317  width='300' align='top'></td><td>End of shift luminosity.</a></td></tr></table>
<!-- Author: Gavril -->


<!-- Date: Fri Feb 21 16:01:12 2003 -->
Date: 
               2003.02.21 
 Shift: 
               day 
 Delivered luminosity: 
               333.70 nb-1 
 Acquired luminosity: 
               219.16 nb-1 
 Efficiency: 
               65.68
<!-- Author: Gavril -->


<!-- Date: Fri Feb 21 16:03:18 2003 -->
<font color=red>Shift Summary: </font><pre>Run 159175 - inherited from owl shift - physics data  
	     stopped run because of L2 timeout 
 
Run 159176 - physics data - stopped to do special runs  
	     (thought store would end soon) 
 
Run 159177 - aborted 
 
Runs 159178 - 159180 COT threshold tests - 215 mV, 205 mV, 200 mV 10K events each 
 
Runs 159181 - 159182 - TOF trigger special runs 
 
Runs 159183 - 159185 - SVT test runs 
 
Run 159186 - 159187 - silicon D mode calib 
 
Run 159188 - aborted 
 
Run 159189 - physics data until end of store at about 15:30 
 
clc quiet time calibration done</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   159175 - 159189
Delivered Luminosity   333.7  
Acquired Luminosity    219.2  
Efficiency             66%
   
</td></tr></table></pre>
<!-- Author: Paul -->


<!-- Date: Fri Feb 21 16:10:14 2003 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2003&action=view&page=-1318&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-1318  width='300' align='top'></td><td>End of shift rad plot</a></td></tr></table>
<!-- Author: Gavril -->