<!-- Title: Eve shift. Sat Dec  9, 2006 -->
<!-- SciCo: Michael Gold -->
<!-- DAQAce: Nagesh Kulkarni -->
<!-- MonAce: Greg Feild -->
<!-- CO: Nick Van Remortel -->
<!-- OpManager: Mark Mattson -->
<!-- Notes: -->
<pre>no beam.  doing calibrations following access on day sift.</pre>

<!-- Date: Sat Dec 9 16:25:40.059 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229855>229855</A>
TERMINATE: BOTHPLUGS_NORMAL done
<!-- Author: Greg -->


<!-- Date: Sat Dec 9 16:27:29.389 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229856>229856</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: Greg -->


<!-- Date: Sat Dec 9 16:29:43.797 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229856>229856</A>
TERMINATE: BOTHPLUGS_MM_C done

<!-- Author: Greg -->


<!-- Date: Sat Dec 9 16:29:45.534 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229856>229856</A>
CALIBRATION: <U><B>pluglaser</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#pluglaser target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Dec  9 16;42;09 comment by...greg -->
<pre>The plug high voltage was not on ...</pre>
<!-- Date: Sat Dec 9 16:34:08.494 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229857>229857</A>
ACTIVATE: RunSet: BOTHPLUGS_MM 
<!-- Author: Greg -->


<!-- Date: Sat Dec 9 16:36:26.808 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229857>229857</A>
TERMINATE: BOTHPLUGS_MM  done

<!-- Author: Greg -->



<!-- Date: Sat Dec 9 16:46:24.011 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229858>229858</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL 
<!-- Author: Greg -->


<!-- Date: Sat Dec 9 16:51:42.480 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229859>229859</A>
ACTIVATE: RunSet: ACE_TOF_QIE_NEW 
<!-- Author: greg -->


<!-- Date: Sat Dec 9 16:51:43.501 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229859>229859</A>
CALIBRATION: <U><B>tofqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#tofqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Dec  9 17;59;10 comment by...Nick vR -->
Calibration looks fine.
<!-- Date: Sat Dec 9 16:52:45.001 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229859>229859</A>
TERMINATE: ACE_TOF_QIE_NEW done
<!-- Author: greg -->


<!-- Date: Sat Dec 9 16:54:41.313 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229858>229858</A>
TERMINATE: BOTHPLUGS_NORMAL done
<!-- Author: Nagesh -->


<!-- Date: Sat Dec 9 16:54:42.460 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229860>229860</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: greg -->


<!-- Date: Sat Dec 9 16:54:44.747 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229861>229861</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: Nagesh -->


<!-- Date: Sat Dec 9 16:56:01.918 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229860>229860</A>
CALIBRATION: <U><B>cemled</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cemled target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Dec  9 18;01;35 comment by...Nick vR -->
calib checked and OK
<!-- Date: Sat Dec 9 16:57:51.146 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229860>229860</A>
TERMINATE: ACE_LED_ALL
<!-- Author: greg -->


<!-- Date: Sat Dec 9 16:57:52.752 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229861>229861</A>
CALIBRATION: <U><B>pluglaser</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#pluglaser target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Dec  9 18;03;18 comment by...Nick vR -->
Calibrations look good
<!-- Date: Sat Dec 9 16:57:53.666 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229861>229861</A>
TERMINATE: BOTH PLUG MMC done
<!-- Author: Nagesh -->



<!-- Date: Sat Dec 9 16:58:02.029 2006 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2006&action=view&page=-15880&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-15880  width='300' align='top'></a></td><td>Bscqie calibrations check (RUn 229854) 
All look fine. Small deviation from reference for pedestal RMS 
BSC 1-east.</td></tr></table>
<!-- Author: Nick van Remortel -->

<!-- Date: Sat Dec 9 16:58:10.204 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229862>229862</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: greg -->


<!-- Date: Sat Dec 9 16:58:17.721 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229863>229863</A>
ACTIVATE: RunSet: BOTHPLUGS_MM 
<!-- Author: Nagesh -->


<!-- Date: Sat Dec 9 16:58:43.281 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229862>229862</A>
TERMINATE: COTCALIB_ACE done
<!-- Author: greg -->


<!-- Date: Sat Dec 9 17:00:26.734 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229863>229863</A>
TERMINATE: BOTHPLUGS_MM done
<!-- Author: Nagesh -->


<!-- Date: Sat Dec 9 17:01:08.310 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229864>229864</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: greg -->


<!-- Date: Sat Dec 9 17:01:56.579 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229862>229862</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Dec  9 18;06;28 comment by...Nick vR -->
Calibration is fine.
<!-- Date: Sat Dec 9 17:02:37.143 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229864>229864</A>
CALIBRATION: <U><B>calxef</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calxef target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Dec  9 18;13;18 comment by...Nick vR -->
One side loks oK, other side is empt. probably didn't switch on.
I paste the plots below.
<!-- Date: Sat Dec 9 17:03:23.492 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229864>229864</A>
TERMINATE: XEF done
<!-- Author: greg -->



<!-- Date: Sat Dec 9 17:03:23.831 2006 -->
Calqie calibration chacked (RUN 229852). All Fine
<!-- Author: Nick VR :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229852' target=_top>229852</a>) -->

<!-- Date: Sat Dec 9 17:05:29.662 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229866>229866</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Nagesh -->


<!-- Date: Sat Dec 9 17:05:30.052 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229865>229865</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: greg -->


<!-- Date: Sat Dec 9 17:06:31.333 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229865>229865</A>
TERMINATE: CALIB_MUON_ALL1 done
<!-- Author: greg -->


<!-- Date: Sat Dec 9 17:06:32.605 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229866>229866</A>
TERMINATE: SMQIECALIB done
<!-- Author: Nagesh -->


<!-- Date: Sat Dec 9 17:06:33.384 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229865>229865</A>
TERMINATE: CALIB_MUON_ALL1 done
<!-- Author: greg -->


<!-- Date: Sat Dec 9 17:06:34.027 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229866>229866</A>
TERMINATE: SMQIECALIB done
<!-- Author: Nagesh -->


<!-- Date: Sat Dec 9 17:08:42.416 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229867>229867</A>
ACTIVATE: RunSet: TOF_TAC_CALIB 
<!-- Author: greg -->


<!-- Date: Sat Dec 9 17:09:13.324 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229866>229866</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Dec  9 18;17;15 comment by...Nick vR -->
PLEASE UPDATE (PUT IN) A REFERNCE GUYS, i HAVE NO IDEA WHAT TO LOOK FOR HERE.
<!-- Comment: Sat Dec  9 18;20;27 comment by...Nick vR -->
Sorry, Ignore my previous comments. Calibration looks OK.
<!-- Date: Sat Dec 9 17:13:41.462 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229867>229867</A>
TERMINATE: TOF_TAC_CALIB done
<!-- Author: greg -->


<!-- Date: Sat Dec 9 17:13:51.973 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229867>229867</A>
CALIBRATION: <U><B>toftac</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#toftac target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Dec  9 18;22;30 comment by...Nick vR -->
Calibration Looks fine
<!-- Date: Sat Dec 9 17:26:01.607 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229868>229868</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [18,791,633] 
<!-- Author: Nagesh -->
<!-- Comment: Sat Dec  9 17;31;00 comment by...greg -->
<pre>Rates and deatimes are as expected.</pre>

<!-- Comment: Sat Dec  9 17;32;05 comment by...greg -->
<pre>no consumers at the moment -- CO is checking the calibrations</pre>


<!-- Date: Sat Dec 9 17:40:53.375 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-15887&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-15887' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-15886&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-15886' width='300'></a></center>
<!-- Author: Nick vR -->
<!-- Comment: Sat Dec  9 17;43;25 comment by...Nick vR -->
problem in clcqie calibration run 229853: big spike in channel 48.
Checked previous calib run 229792: This was also bad!

The one before that (dec 5th): run 229505 did not have this problem

<!-- Comment: Sat Dec  9 18;09;53 comment by...sasha -->
I have checked amplitude distributions for this channel and did not find any problems. This peak might be due to internal problem of QIE calibration: some time during CLC ADMEM QIE calibration CLC transition boards which are connected to CLC ADMEMs generate some noise and spoil QIE calibration. Is is a long known problem of the CLC QIE calibration. So my conclusion is that this channel is 100% operational and should not be a problem for data taking/lumi. measurement.
<!-- Date: Sat Dec 9 17:48:11.256 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229850>229850</A>
TERMINATE: done
<!-- Author: fernand -->


<!-- Date: Sat Dec 9 17:57:23.061 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229869>229869</A>
 Activated at 2006.12.09 17:56:42
<!-- Author: RunControl -->


<!-- Date: Sat Dec 9 17:57:27.109 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229869>229869</A>
ACTIVATE: RunSet: ACE_COSMIC_NOTRACKS  Trigger: COSMICS_NOTRACKS [9,731,639] 
<!-- Author: Nagesh -->


<!-- Date: Sat Dec 9 17:58:21.558 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229869>229869</A>
 Terminated at 2006.12.09 17:57:31
<!-- Author: RunControl -->



<!-- Date: Sat Dec 9 18:00:35.788 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229871>229871</A>
 Activated at 2006.12.09 18:00:28
<!-- Author: RunControl -->


<!-- Date: Sat Dec 9 18:00:36.685 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229871>229871</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [7,792,632] 
<!-- Author: Nagesh -->



<!-- Date: Sat Dec 9 18:01:16.752 2006 -->
mcr calls that shot setup in 10-15 minutes
<!-- Author: gold :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229870' target=_top>229870</a>) -->





<!-- Date: Sat Dec 9 18:07:27.518 2006 -->
During the quite time we investigated a svx ladder (SB0W5L2) that 
was having high readout errors in readout time. See details in the 
<a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=330&frame=2&anchor=12:38:51'> 
silicon e-log </a>. 
We also have taken several noise vs bias scans. 
 
  
<!-- Author: Juan Pablo and Roberto :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER= 229850' target=_top> 229850</a>) -->
<!-- Comment: Sat Dec  9 18;09;34 comment by...Juan Pablo -->
Jose Enrique and Satyajit did most of the job and Ankush
gave us his expertise. Thank you very much!


<!-- Date: Sat Dec 9 18:12:25.509 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229871>229871</A>
 Terminated at 2006.12.09 18:12:15
<!-- Author: RunControl -->




<!-- Date: Sat Dec 9 18:14:30.357 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-15890&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-15890' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-15889&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-15889' width='300'></a></center>
<!-- Author: Nick vR -->
<!-- Comment: Sat Dec  9 18;15;30 comment by...Nick vR -->
Run 229864 CALIBRATION: calxef calibration 
-- Sat Dec 9 18:13:18 comment by...Nick vR --  One side loks oK, other side is empt. probably didn't switch on. 





<!-- Comment: Sat Dec  9 19;42;25 comment by...greg -->
<pre>I neglected to turn on the flashers.</pre>
<!-- Date: Sat Dec 9 18:26:31.257 2006 -->
Ending all monitors, preparing for switch to PHYSICS tcl
<!-- Author: Nick vR -->

<!-- Date: Sat Dec 9 18:28:40.207 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229872>229872</A>
 Activated at 2006.12.09 18:28:25
<!-- Author: RunControl -->


<!-- Date: Sat Dec 9 18:28:41.403 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229872>229872</A>
ACTIVATE: RunSet: ACE_COSMIC_NOTRACKS  Trigger: COSMICS_NOTRACKS [9,731,639] 
<!-- Author: Nagesh -->


<!-- Date: Sat Dec 9 18:28:42.483 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229872>229872</A>
 Terminated at 2006.12.09 18:28:32
<!-- Author: RunControl -->


<!-- Date: Sat Dec 9 18:32:17.601 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229873>229873</A>
 Activated at 2006.12.09 18:31:52
<!-- Author: RunControl -->


<!-- Date: Sat Dec 9 18:32:18.272 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229873>229873</A>
ACTIVATE: RunSet: ACE_COSMIC_NOTRACKS  Trigger: COSMICS_NOTRACKS [9,731,639] 
<!-- Author: Nagesh -->


<!-- Date: Sat Dec 9 19:13:22.157 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229873>229873</A>
 Terminated at 2006.12.09 19:13:02
<!-- Author: RunControl -->


<!-- Date: Sat Dec 9 19:17:41.027 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229874>229874</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_4_01 [2,850,663] 
<!-- Author: Nagesh -->



<!-- Date: Sat Dec 9 19:17:55.815 2006 -->
FERML_TM2_HEVB2B_EVENTS_PENDING_TIMEOUT !!!  
 - AUTO HRR will be issued. 
<!-- Author: Nagesh :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229874' target=_top>229874</a>) -->

<!-- Date: Sat Dec 9 19:24:12.185 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229875>229875</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_4_01 [2,850,663] 
<!-- Author: Nagesh -->



<!-- Date: Sat Dec 9 19:32:21.406 2006 -->
<pre>Running AAA_SHOTSETUP. 
We keep having evb problems and BUSY timeouts. 
 
Check SCPUs says cot00 and cot02 are giving load errors. 
WE are shpherding these crates.</pre>
<!-- Author: Greg and Nagesh :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229875' target=_top>229875</a>) -->


<!-- Date: Sat Dec 9 19:34:03.655 2006 -->
<pre>Final protons are loaded. 
Injecting pbars. 
</pre>
<!-- Author: greg :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229876' target=_top>229876</a>) -->


<!-- Date: Sat Dec 9 19:34:34.238 2006 -->
final protons are in. loading pbars
<!-- Author: gold :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229876' target=_top>229876</a>) -->

<!-- Date: Sat Dec 9 19:35:13.127 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229876>229876</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_4_01 [2,850,663] 
<!-- Author: Nagesh -->




<!-- Date: Sat Dec 9 20:11:28.630 2006 -->
pbars are in... preparing to ramp
<!-- Author: gold :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229876' target=_top>229876</a>) -->


<!-- Date: Sat Dec 9 20:14:45.704 2006 -->
12 pbar bunches lost... 36x24 store 
<!-- Author: gold :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229876' target=_top>229876</a>) -->


<!-- Date: Sat Dec 9 20:33:00.739 2006 -->
mcr calls for end of scraping... 77E30 starting lum
<!-- Author: gold :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229877' target=_top>229877</a>) -->

<!-- Date: Sat Dec 9 20:37:03.653 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229877>229877</A>
 Activated at 2006.12.09 20:36:39
<!-- Author: RunControl -->


<!-- Date: Sat Dec 9 20:37:04.069 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229877>229877</A>
ACTIVATE: RunSet: AAA_CURRENT  Trigger: PHYSICS_4_01 [2,850,663] 
<!-- Author: Nagesh -->



<!-- Date: Sat Dec 9 20:52:54.766 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-15891&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-15891' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-15892&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-15892' width='300'></a></center>
<!-- Author: Nagesh -->


<!-- Date: Sat Dec 9 20:59:40.960 2006 -->
<pre>no icicle heartbeat alarm - we restart icicle</pre>
<!-- Author: greg :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229877' target=_top>229877</a>) -->


<!-- Date: Sat Dec 9 21:29:01.932 2006 -->
First CO check for physics run 229877: All looks fine. 
IMU problem seems to be fixed already: YMON BSU West has no gap anymore at phi>70.
<!-- Author: Nick :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229877' target=_top>229877</a>) -->


<!-- Date: Sat Dec 9 21:37:15.774 2006 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2006&action=view&page=-15893&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-15893  width='300' align='top'></a></td><td>cell unmarked</td></tr></table>
<!-- Author: gold -->

<!-- Date: Sat Dec 9 21:47:14.298 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229877>229877</A>
 Terminated at 2006.12.09 21:47:08
<!-- Author: RunControl -->
<!-- Comment: Sat Dec  9 22;29;16 comment by...greg,nagesh -->
<pre>Just before TS crisis there was a PHA IFix alarm.
Went upstairs and Sakamotorized East Plug 10.
Time 21:34</pre>
<!-- Date: Sat Dec 9 21:47:41.064 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229877>229877</A>
TERMINATE: We have high dead time
<!-- Author: Nagesh -->


<!-- Date: Sat Dec 9 21:51:20.884 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229878>229878</A>
 Activated at 2006.12.09 21:51:08
<!-- Author: RunControl -->


<!-- Date: Sat Dec 9 21:52:32.003 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229878>229878</A>
 Terminated at 2006.12.09 21:51:38
<!-- Author: RunControl -->


<!-- Date: Sat Dec 9 21:52:33.177 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229878>229878</A>
TERMINATE: tsi freakin out
<!-- Author: Nagesh -->


<!-- Date: Sat Dec 9 21:57:44.491 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229879>229879</A>
 Activated at 2006.12.09 21:57:08
<!-- Author: RunControl -->


<!-- Date: Sat Dec 9 21:58:26.562 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229879>229879</A>
ACTIVATE: RunSet: AAA_CURRENT  Trigger: PHYSICS_4_01 [2,850,663] 
<!-- Author: Nagesh -->



<!-- Date: Sat Dec 9 22:01:51.970 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-15894&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-15894' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-15895&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-15895' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-15896&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-15896' width='300'></a></center>
<!-- Author: Nagesh -->


<!-- Date: Sat Dec 9 22:07:44.132 2006 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2006&action=view&page=-15897&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-15897  width='300' align='top'></a></td><td>CHAPMTCanvas for Run 229877 shows slightly hotter channels  
than reference. ACES think it should be ok. </td></tr></table>
<!-- Author: Nick -->


<!-- Date: Sat Dec 9 22:08:29.658 2006 -->
<pre>FERRML_HIGH_DEADTIME: 
 
During the last run the deadtime shot up and RC alarmed. 
Looking at the scaler GUI we saw that the percent BUSY  
deadtime had gone to zero while the TSI deadtime was ~5%  
(should be zero). 
 
Opening a terminal to the TS crate we saw VME bus errors. 
We brought the run to START and shepherded the crate. 
 
In the new run all is fine again. 
 
</pre>
<!-- Author: greg -->


<!-- Date: Sat Dec 9 22:15:11.131 2006 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2006&action=view&page=-15898&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-15899  width='300' align='top'></a></td><td>SL2 problem?</td></tr></table>
<!-- Author: John and Greg -->
<!-- Comment: Sat Dec  9 22;22;33 comment by...Greg and John -->
More than two of the bars of SL2 looked like the problematic one above.  Michael (our scico) believes that it is just a display problem.



<!-- Comment: Sun Dec 10 01;36;24 comment by...Aseet Mukherjee -->
Mike was partly right.  The display is as it should be when bars red/pink showing current are higher than the blue bars showing voltage.  Normally we set the scale to avoid this, but did not do so after repairs this afternoon.  I adjusted it now.
<!-- Date: Sat Dec 9 23:26:24.625 2006 -->
Run 229879 checked. So far so good.
<!-- Author: Nick :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229879' target=_top>229879</a>) -->
<!-- Comment: Sat Dec  9 23;52;12 comment by...Greg&#39;s arch-nemesis -->
Except for FRED which is still broken.

<!-- Comment: Sat Dec  9 23;54;22 comment by...a fred fan -->
<pre>Fred totally rules!</pre>
<!-- Comment: Sat Dec  9 23;56;31 comment by...Fred Schneider -->
It wasn't a rock, it was a ROCK LOBSTER!
<!-- Date: Sat Dec 9 23:49:00.183 2006 -->
RXPT error on b0svx02. HRR issued. --fixed.
<!-- Author: Nagesh :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229879' target=_top>229879</a>) -->
<!-- Comment: Sat Dec  9 23;53;36 comment by...greg -->
<pre>6 Si resonaces detected.</pre>


<!-- Comment: Sun Dec 10 00;08;15 comment by...Ulrich -->
I only see one resonance at that time.
<!-- Date: Sat Dec 9 23:56:06.624 2006 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>PhysicsTable</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L3Accepts</TH>
<TH>DelivLumi nb<SUP>-1</SUP></TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN=229877 target=_runSummary>229877</A> x381F5</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_4_01 [2,850,663]</FONT></TD>
<TD ALIGN=CENTER>20:36:39</TD>
<TD ALIGN=CENTER>21:47:08</TD>
<TD ALIGN=CENTER>00:55:16</TD>
<TD ALIGN=RIGHT>     246,344</TD>   <TD ALIGN=RIGHT>    237.88</TD>
<TD ALIGN=RIGHT>    196.07</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN=229879 target=_runSummary>229879</A> x381F7</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_4_01 [2,850,663]</FONT></TD>
<TD ALIGN=CENTER>21:57:08</TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>01:47:08</TD>
<TD ALIGN=RIGHT>     505,067</TD>   <TD ALIGN=RIGHT>    356.60</TD>
<TD ALIGN=RIGHT>    324.88</TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>23:55:01</TD>
<TD ALIGN=CENTER>02:42:24</TD>
<TD ALIGN=RIGHT>       751,411</TD> <TD ALIGN=RIGHT>    594.48</TD>
<TD ALIGN=RIGHT>    520.95</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->




<!-- Date: Sun Dec 10 00:02:24.707 2006 -->
<font color=red>Shift Summary: </font><pre>Notes: 
mcr repair status 16:30: 
"The epoxy repairs have halted for the KPS4A [transformer] stand-off. The original piece was too damaged.  
OPs are purchasing a 4'x8'x1/4" piece of Lexan from a local hardware store." 
 
 
store # 5115 36x24 store  with starting Lum = 77E30  
 
 
Runs: 229877 196.1/nb 
      229878   .6/nb 
      229879    
</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   229877 229878 229897
Delivered Luminosity   655.5  
Acquired Luminosity    531.1  
Efficiency             81.6
   
</td></tr></table></pre>
<!-- Author: gold -->


<!-- Date: Sun Dec 10 00:02:58.186 2006 -->
SVX_02 is in Error. HRR
<!-- Author: Naoki :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=229879' target=_top>229879</a>) -->
<!-- Comment: Sun Dec 10 00;07;23 comment by...Ulrich -->
This was (another) resonance. The L1A rate is very high in this run, so you expect more resonances than you may be used to. Is everything alright with the trigger? Please paste the full message into the e-log.


<!-- Comment: Sun Dec 10 00;12;49 comment by...Naoki -->
Error in SVX02 : HRR again

<!-- Comment: Sun Dec 10 00;22;14 comment by...Naoki -->
Total deadtime is 7.40. TSO/ Total DT is 0.01. If Error come agein I will post full message into the e-log. 

<!-- Comment: Sun Dec 10 00;27;32 comment by...twright -->
Because it's 36x24 the lumi-enables will come on at higher per-bunch luminosity than they generally do, I guess, so we are seeing more of the growth terms than normal.