<!-- Title: Eve shift. Fri May 21, 2004 -->
<!-- SciCo: Duncan Carlsmith -->
<!-- DAQAce: Daejung Kong -->
<!-- MonAce: Corrinne Mills -->
<!-- CO: Kyung Kwang Joo -->
<!-- OpManager: J. J.  Schmidt -->
<!-- Notes: -->
<pre>Stack = 82, accRt=0, b0olum=0</pre>


<!-- Date: Fri May 21 16:18:12.473 2004 -->
<table border=0><td><tr><td bgcolor=#e0e030><pre>JJ reports to debug chilled water, need to shut down detector entirely.</pre></td></tr></td></table>
<!-- Author: Carlsmith -->


<!-- Date: Fri May 21 16:35:05.783 2004 -->
Page calorimetry for JJ
<!-- Author: Carlsmith -->


<!-- Date: Fri May 21 16:43:10.455 2004 -->
Many PSM alarms at the beginning of shift, all overvoltages on +5 digital voltage, 4 from 1RR12C_1, one from 1RR18D_1. 
<br><br> 
Many things have tripped off or been turned off as a result of the cooling water problems.  First the silicon (immediately), then the COT, then the MNP, then the CHA, PEM, ...
<!-- Author: corrinne -->


<!-- Date: Fri May 21 19:30:32.562 2004 -->
<table border=0><td><tr><td bgcolor=#e0e030>Online cluster is up. PRD/DEV SUNs are up.</td></tr></td></table>
<!-- Author: Sergei -->


<!-- Date: Fri May 21 19:43:15.860 2004 -->
At 17:15, power was turned off in the collision hall and assembly building and replacement of a circuit breaker initiated. Ultimately two were replaced. The CDF E-log was unavailable. 
A COT LEL baggy alarm appeared breifly and was monitored.  
Power was restored at 18:51. THe detector turn on procedure is underway.
<!-- Author: carlsmith -->


<!-- Date: Fri May 21 19:57:13.184 2004 -->
Phil Schlabach doesn't anticipate any problem with muon systems including the trigger other than a possibility the TDCs got caught in a funny state in which they trigger every crossing. If a test run should reveal this, TDCs will need to be reseated. 
 
TOF responded to a page and did not anticipate any problem. 
<!-- Author: Carlsmith -->
<!-- Comment: Fri May 21 20;28;59 comment by...schlabach -->
if the clock crate didn't get hit, then I don't think the cmu/x/bmu trigger will be effected.
<!-- Comment: Fri May 21 20;49;42 comment by...ps -->
some details are <a href="http://www-cdfonline.fnal.gov/ops/cdf_muon/troubleshooting/ts_guide.html#triggermismatch">here</a>. This covers CMU/X. There are not yet equivalent plots for BMU in TrigMon so if the IMU trigger rates seem through the roof this is likely to be the cause. 
<!-- Date: Fri May 21 19:58:20.217 2004 -->
Tornado warning
<!-- Author: carlsmith -->
<!-- Comment: Fri May 21 20;22;22 comment by...Andrei -->
<pre>is it over?</pre>


<!-- Comment: Fri May 21 20;57;25 comment by...rainer -->

       ILC037-089-220200-
       SEVERE WEATHER STATEMENT
       NATIONAL WEATHER SERVICE CHICAGO IL
       829 PM CDT FRI MAY 21 2004
       ...THE TORNADO WARNING FOR DE KALB COUNTY IN ILLINOIS HAS
       BEEN CANCELLED...
       ...THE SEVERE THUNDERSTORM WARNING FOR DEKALB AND KANE COUNTIES
       CONTINUES TO 900 PM CDT..
       AT 829 PM CDT NATIONAL WEATHER SERVICE DOPPLER RADAR INDICATED THAT
       THE ROTATION WITHIN THE THUNDERSTORM THAT PROMPTED THE TORNADO
       WARNING HAD WEAKENED SIGNIFICANTLY...THUS THE WARNING HAS BEEN
       CANCELED.
       HOWEVER THE THUNDERSTORM IS STILL CAPABLE OF PRODUCING LARGE HAIL
       AND WIND OF UP TO 65 MPH.
       * THE STORM WILL BE NEAR...
       ELBURN AROUND 835 PM CDT.
       NORTH AURORA AROUND 845 PM CDT.
       GENEVA AROUND 850 PM CDT.
       ST. CHARLES AROUND 855 PM CDT.
       ELGIN AROUND 900 PM CDT.
       LAT...LON 4196 8874 4175 8861 4175 8828 4211 8828
       $$
       NNNN
       WHW
<!-- Comment: Fri May 21 20;57;51 comment by...rainer -->
but we have not received the ALL CLEAR yet
<!-- Date: Fri May 21 20:27:20.949 2004 -->
my sirens have stopped in aurora
<!-- Author: schlabach -->
<!-- Comment: Fri May 21 20;51;56 comment by...ps -->
christ, I hope you're all all right, forget the blah, blah




<!-- Date: Fri May 21 21:06:50.858 2004 -->
All clear. CLC crate dead. Paged expert.
<!-- Author: carlsmith -->
<!-- Comment: Fri May 21 21;10;40 comment by...schlabach -->
great, see comments above
<!-- Date: Fri May 21 21:07:27.933 2004 -->
<pre>The b0clc01 CLC_01 crate's power supply does not  
appear to be working.   I checked the 20A fuse  
in the back of the crate and it looked continuous. 
The circuit supplying AC is OK, too.   Crate is located  
in 1RR27E 
 
This is a special crate;  I hope we have spare supplies. 
</pre>
<!-- Author: W.Badgett -->
<!-- Comment: Fri May 21 21;08;27 comment by...WB -->
<pre>This is the top crate in the rack 1RR27E</pre>



<!-- Date: Fri May 21 21:18:54.375 2004 -->
<pre>COT LV back on before the tornado. 
COT HV now brought to standby.  So far so good...</pre>
<!-- Author: K Burkett -->


<!-- Date: Fri May 21 21:25:07.451 2004 -->
I assumed Rainer was there; some comfort. 2 sirens here already and all hell seems to be breaking loose in Aurora.
<!-- Author: schlabach -->
<!-- Comment: Fri May 21 21;58;52 comment by...ps -->
the all hell cleared - 
<!-- Date: Fri May 21 21:28:18.019 2004 -->
I restarted icicle. 
<br><br> 
Also, I checked the muon3 pc and it is okay.  I brought the CMP back up from a tripped state -- it had tripped back when much of the HV was being turned off earlier. 
<br><br> 
We continue to get PSM alarms due to overvoltages from the +5V digital in 1RR12C_1.  Usually the value has gone up to +5.77V.  
<!-- Author: corrinne -->


<!-- Date: Fri May 21 21:30:17.431 2004 -->
Cheng-Ju says he has the XFT crates for the next 20 min, since the flash ram got corrupted by the power cycle.  
<!-- Author: corrinne -->
<!-- Comment: Fri May 21 22;12;32 comment by...Cheng-Ju -->
<pre>I am done reprogramming flashrams.  Released all XFT crate.</pre>
<!-- Date: Fri May 21 21:36:21.277 2004 -->
Cooling warnings  from L00 and SVX, both for "Air Separator Level" being at 16.9in when the low warning is at 17 in.  Also, for L00, the "East Pressure at the COT Face" is at 13.0 psia.  I think that that alarm has been around beofre, though.   
<br> 
I see no ignored/silenced alarms, and no other warnings for any of the silicon systems  
<!-- Author: corrinne -->


<!-- Date: Fri May 21 21:39:51.051 2004 -->
<pre>SVT SPY MONITOR error 
 
Bill started "svtspymon" at b0dap85, 
I started "spy_manager" at b0dap85. 
and click "PUCH TO FORCE CHECK". 
Now Fine</pre>
<!-- Author: Daejung -->


<!-- Date: Fri May 21 21:49:56.184 2004 -->
Access in progress to bring up solenoid. 
 
L3 still having trouble. 
 
MCR says we have at least an hour of quiet time. So starting calibrations.
<!-- Author: carlsmith -->


<!-- Date: Fri May 21 21:57:47.720 2004 -->
<table border=0><td><tr><td bgcolor=#e0e030><pre>The Reset Crate (RESET_00, b0res00 and b0res01) processors  
are not booting properly.   Under investigation. 
 
Until then you cannot issue "reset_crate" commands. 
 
</pre></td></tr></td></table>
<!-- Author: W.Badgett -->

<!-- Date: Fri May 21 21:58:08.683 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182971">182971</a>
ACTIVATE: BSC QIE start
<!-- Author: Daejung x2080 -->


<!-- Date: Fri May 21 21:58:52.718 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182971">182971</a>
TERMINATE: BSC QIE end
<!-- Author: Daejung x2080 -->




<!-- Date: Fri May 21 21:59:54.993 2004 -->
PSM alarm from 1RR21E_2.  Also, I have masked 1RR12C_1 for the time being since its problem is persistent but does not seem to be serious.
<!-- Author: corrinne -->

<!-- Date: Fri May 21 22:01:05.003 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182972">182972</a>
ACTIVATE: TOF QIE start
<!-- Author: Daejung x2080 -->


<!-- Date: Fri May 21 22:01:06.437 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182972">182972</a>
TERMINATE: TOF QIE end
<!-- Author: Daejung x2080 -->



<!-- Date: Fri May 21 22:03:23.551 2004 -->
2RR22H_1 alarm, +12VD and -5.2VD are both staying at 0 volts, but no one seems to be working on this.  
<!-- Author: corrinne -->
<!-- Comment: Fri May 21 22;12;00 comment by...corrinne -->
It turns out that that's b0tsi02.  Steve Hahn is working on it.  He says that it is somehow a monitoring problem, since we wouldn't be able to take even calibrations if it was really not working.  
<!-- Date: Fri May 21 22:06:31.646 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182973">182973</a>
ACTIVATE: COT start
<!-- Author: Daejung x2080 -->


<!-- Date: Fri May 21 22:06:41.296 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182973">182973</a>
TERMINATE: COT end
<!-- Author: Daejung x2080 -->
<!-- Comment: Fri May 21 22;15;12 comment by...K Burkett -->
<pre>COT calibration looks OK.  More drifted channels than normal, but
considering everything that is not unexpected.</pre>


<!-- Date: Fri May 21 22:15:14.614 2004 -->
XFT crate work over and crates released
<!-- Author: carlsmith -->

<!-- Date: Fri May 21 22:18:10.356 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182976">182976</a>
ACTIVATE: TOF TAC start
<!-- Author: Daejung x2080 -->



<!-- Date: Fri May 21 22:18:16.349 2004 -->
Corrine finds b0ccal05, b0pcal05,b0ccal00 troubles. Rob Roser and Steve Hahn off to check.
<!-- Author: carlsmith -->


<!-- Date: Fri May 21 22:19:07.834 2004 -->
SVX alarm due to crate 6 communication failure.  Hopefully this is due to silicon work, o/wise I will go Hockerize the crate...
<!-- Author: corrinne -->
<!-- Comment: Fri May 21 22;23;00 comment by...corrinne -->
It is them.  They had actually warned me earlier but I forgot.
<!-- Date: Fri May 21 22:20:23.630 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182976">182976</a>
TERMINATE: TOF TAC end
<!-- Author: Daejung x2080 -->



<!-- Date: Fri May 21 22:21:15.809 2004 -->
<pre>during MUON calibration 
b0cmx01, b0musc has error, I tried to shepherd, 
but It didn't finished.</pre>
<!-- Author: Daejung -->



<!-- Date: Fri May 21 22:25:45.637 2004 -->
Rong-Shyang points out that there is a "single fan failure" warning from COSWT which is a COT crate.
<!-- Author: corrinne -->

<!-- Date: Fri May 21 22:28:58.423 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182977">182977</a>
ACTIVATE: CAL QIE start
<!-- Author: Daejung x2080 -->
<!-- Comment: Fri May 21 22;39;19 comment by...V Tiwari -->
<table border=0><td><tr><td bgcolor=#e0e030>Dead Wedge in WHA 14E. Probably blown fuse in ADMEM. Would require access to south-east botttom crate wcal06, slot 18. South wall needs to be out and access will take 1 hr. Please take another CAL QIE calib.</td></tr></td></table>
<!-- Comment: Fri May 21 23;30;22 comment by...V Tiwari -->
Got lucky and acually the ADMEM configuration memory got erased. Rewrote it through QIETEST and all is back to normal. However there is another problem. To make run control happy we have to do a calibration download for the entire wall. But there is another channel in wall WHA 6E, ch 10 which shows very high mean. The mean has come down from 750 to 500 in two CAL QIE runs 5 minutes apart. Since WHA HV is drifting at this time I'll wait for 20 minutes to check this channel again. If it is okay then I'll do a calibration download otherwise make an access to investigate.
<!-- Comment: Sat May 22 01;16;28 comment by...V Tiwari -->
HV for this channel was wondering about and shifted by 50V. After waiting long enough it quieted down and we were able to do a calib. load for WHA. See my entry in the next <a href=
'/cgi/elog/elog.pl?nb=2004&action=view&page=350&frame=2&anchor=00:31:21.752'>
shift elog </a>. How does a 50V shift in HV cause pedestal to jump to 750 instead of normal 200 is still a mystery. All is well now.

<!-- Date: Fri May 21 22:30:00.236 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182977">182977</a>
TERMINATE: CAL QIE end
<!-- Author: Daejung x2080 -->


<!-- Date: Fri May 21 22:30:23.781 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182978">182978</a>
ACTIVATE: CLC QIE start
<!-- Author: Daejung x2080 -->


<!-- Date: Fri May 21 22:31:40.088 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182978">182978</a>
TERMINATE: CLC end
<!-- Author: Daejung x2080 -->


<!-- Date: Fri May 21 22:33:38.925 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182979">182979</a>
ACTIVATE: SMX start
<!-- Author: Daejung x2080 -->


<!-- Date: Fri May 21 22:34:22.544 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182979">182979</a>
TERMINATE: SMX end
<!-- Author: Daejung x2080 -->



<!-- Date: Fri May 21 22:34:31.126 2004 -->
b0clc00 is also troubled -- yellow in VXworks with a red U, but the power looks fine in the power supply monitor.  looking into it.
<!-- Author: corrinne -->
<!-- Comment: Fri May 21 22;35;53 comment by...corrinne -->
that's b0clc01, actually

<!-- Date: Fri May 21 22:36:31.660 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182980">182980</a>
ACTIVATE: CEM LED start
<!-- Author: Daejung x2080 -->


<!-- Date: Fri May 21 22:38:11.306 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182980">182980</a>
TERMINATE: CEM LED end
<!-- Author: Daejung x2080 -->




<!-- Date: Fri May 21 22:49:34.465 2004 -->
<table border=0><td><tr><td bgcolor=#e0e030>Cryo is calm. About to start powering solenoid.  
 
Judging from the weather radar channel, we might see a repeat of severe weather out of that state north of us in a a few hours. 
 
</td></tr></td></table>
<!-- Author: carlsmith -->


<!-- Date: Fri May 21 22:49:38.821 2004 -->
PSM alarms also from Arch Power Supply Monitoring.  SMDCE_05 and SMDCE_04 are red in channel 5, 6, 7 with the voltages listed as "-25 V".  Since it looks like we have lost monitoring for most of the rest of the arch power supplies (they are magenta), this is probably also a monitoring issue
<!-- Author: corrinne -->
<!-- Comment: Fri May 21 23;02;18 comment by...corrinne -->
Rob showed me that this can be fixed by unplugging and then replugging the power cable at the top of the rack that the muon3 pc lives in (1RR06F).  Apparently the on/off switch won't fix it, you have to use the plug.  
<!-- Date: Fri May 21 22:55:42.801 2004 -->
<pre>During calibration run b0cmx01 failed to partition.  Loggin on  
we see the following errrs, which seem to say the slot 5 is not 
responding 
 
FISION: S_errno_EIO (0x5): VME BERR received! 
*** TDC in slot 5, ID PROM not found (blank check) *** 
</pre>
<!-- Author: Ray Culbertson -->


<!-- Date: Fri May 21 22:56:09.619 2004 -->
<pre>The reset_crate (RESET_00, b0res00 and b0res01) are now  
back in operations.   The reset_crate command will work,  
as well as the secondary shepherding reset.   Some fool  
deleted a boot script & now it's been reconstructed. 
</pre>
<!-- Author: W.Badgett -->



<!-- Date: Fri May 21 23:02:37.740 2004 -->
Mitch returns page for TDC problem. 
MCR OKs 2 hour access.
<!-- Author: carlsmith -->

<!-- Date: Fri May 21 23:06:42.432 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182983">182983</A>
ACTIVATE: Plug BOTHPLUGS_NORMAL start
<!-- Author: Daejung x2080 -->


<!-- Date: Fri May 21 23:13:03.795 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182983">182983</A>
TERMINATE: BOTH end
<!-- Author: Daejung x2080 -->


<!-- Date: Fri May 21 23:13:04.655 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182984">182984</A>
ACTIVATE: CAl Qie start
<!-- Author: Daejung x2080 -->


<!-- Date: Fri May 21 23:14:52.119 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182984">182984</A>
TERMINATE: Cal Qie end
<!-- Author: Daejung x2080 -->


<!-- Date: Fri May 21 23:17:10.207 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182985">182985</A>
ACTIVATE: Plug Laser BOTHPLUGS_NORMAL Start
<!-- Author: Daejung x2080 -->



<!-- Date: Fri May 21 23:17:10.450 2004 -->
<pre>Power cycled the 2RR22H rack, since the PSM was complaining  
about the top crate's -5V and 12V power supplies;  after  
the power brought back up, the green LEDs at the front of  
the supply were green & PSM no longer complained.  
</pre>
<!-- Author: W.Badgett -->
<!-- Comment: Fri May 21 23;17;37 comment by...WB -->
The funny thing is that I don't even think those 
voltages are even used by the RXPT crate.



<!-- Date: Fri May 21 23:18:54.581 2004 -->
Central calorimeter HV is ok after some tunes of the gamma supplies 
(CEM NW drifted down by 9V, CHA NE drifted up 4V and WHA gamma1 drifted up by +3 V) and indvidual channels. The gammas drifted although didn't trip. Maybe because of increased temperature in the 1st floor. <br> 
For now, the voltage seems to be  stable but in the next ~12 hours some channels will drift as the gamma supplies settle. In case of individual detector alarms (CHA,WHA,CEM) the aces can follow the instruction under the PisaBox PC and reset the channels to nominal voltage. If problems persist call 677-5319  
<!-- Author: Fotis -->

<!-- Date: Fri May 21 23:19:44.007 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182985">182985</A>
TERMINATE: plug end
<!-- Author: Daejung x2080 -->


<!-- Date: Fri May 21 23:20:59.094 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182986">182986</A>
ACTIVATE: cal qie start
<!-- Author: Daejung x2080 -->


<!-- Date: Fri May 21 23:21:50.011 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182986">182986</A>
TERMINATE: cal qie end
<!-- Author: Daejung x2080 -->


<!-- Date: Fri May 21 23:26:15.400 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182988">182988</A>
ACTIVATE: Plug laser BOTHPLUGS_NORMAL start
<!-- Author: Daejung x2080 -->



<!-- Date: Fri May 21 23:31:26.811 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182988">182988</A>
TERMINATE: plug laser BOTHPLUGS_NORMAL end
<!-- Author: Daejung x2080 -->


<!-- Date: Fri May 21 23:33:49.828 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182989">182989</A>
ACTIVATE: Plug laser BOTHPLUGS_MM_C
<!-- Author: Daejung x2080 -->


<!-- Date: Fri May 21 23:35:03.262 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182989">182989</A>
TERMINATE: Plug laser BOTHPLUGS_MM_C
<!-- Author: Daejung x2080 -->


<!-- Date: Fri May 21 23:37:04.138 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182990">182990</A>
ACTIVATE: plug laser BOTHPLUGS_MM start
<!-- Author: Daejung x2080 -->


<!-- Date: Fri May 21 23:38:18.545 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182990">182990</A>
TERMINATE: plug laser BOTHPLUGS_MM end
<!-- Author: Daejung x2080 -->



<!-- Date: Fri May 21 23:41:02.006 2004 -->
<pre>Have some sort of problem with SCRAMnet in b0tsi00  
crate -- the SCRAMnet check in the EvbControl  
display shows that b0tsi00 is "NOT in the RING!" 
 
Contacting Greg Feild for some advice. 
 
Now taking a test run to see how it looks like. 
 
</pre>
<!-- Author: W.Badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182991' target=_top>182991</a>) -->
<!-- Comment: Fri May 21 23;48;12 comment by...W.Badgett -->
<pre>The SCRAMnet cehck "NOT in the RING!" error has 
spontaneously disappeared.  We do not know why.

</pre>
<!-- Date: Fri May 21 23:42:55.427 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182994">182994</A>
TERMINATE: Cal qie end
<!-- Author: Daejung x2080 -->



<!-- Date: Fri May 21 23:43:38.274 2004 -->
<p>CLC crate is now back up and running.</p> 
 
<p>This is a special 6U Wiener crate with a front control unit and a rear power supply unit.  First, I tried just putting in a new power supply unit; no good.  Then, I also put in a new control unit in front; good.  Then, put the old power supply unit in; no good.  So, put new power supply unit back in; good.  That is, in the end, both the power supply and control units were replaced.  I have put the old units on Roberto's desk for him to look at next week.</p>
<!-- Author: Steve Hahn -->


<!-- Date: Fri May 21 23:44:41.762 2004 -->
Reiner says, 1RR21F_1 had a funny smell but there is no obvious electrical problem.
<!-- Author: corrinne -->
<!-- Comment: Sat May 22 09;27;09 comment by...rainer -->
The power supply of this crate later

<a href=
'http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2004&action=view&page=244&frame=2&anchor=21:44:03.555'>
complained</a> and then 
<a href=
'http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2004&action=view&page=351&frame=2&anchor=08:17:32.938'>
failed.</a>
<!-- Date: Fri May 21 23:47:35.211 2004 -->
Power supply CASEB_2 has gone back to purple (communication error).  Guess I will try Rob's trick again and see if it will come back.
<!-- Author: corrinne -->
<!-- Comment: Fri May 21 23;52;43 comment by...corrinne -->
The problem has persisted.  

<!-- Date: Fri May 21 23:55:46.632 2004 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>23:55:03</TD>                                                       
<TD ALIGN=CENTER>::</TD>                                                
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->


<!-- Date: Fri May 21 23:59:04.545 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182997">182997</A>
 Activated at 2004.05.21 23:58:46
<!-- Author: RunControl -->



<!-- Date: Fri May 21 23:59:04.295 2004 -->
<font color=red>Shift Summary: </font><pre>Shift started out with loss of chilled water. Powered off 
the detector and assembly building. Power on at 18:50 but turn on 
was interrupted by a tornado warning. At 21:00, we started fixing numerous 
problems associated with the power/thermal cycling. A CLC crate ps was replaced. 
Calorimetry was tweaked. There is a two hour access scheduled to work yet again 
on the solenoid. Work is in progress in fixing the Dead Wedge in WHA 14E 
issue via software. Stack is at 92. </pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   
Delivered Luminosity   0  
Acquired Luminosity    0  
Efficiency             100
   
</td></tr></table></pre>
<!-- Author: Carlsmith -->

<!-- Date: Sat May 22 00:01:41.962 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182997">182997</A>
 Terminated at 2004.05.22 00:01:09
<!-- Author: RunControl -->


<!-- Date: Sat May 22 00:01:42.970 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182998">182998</A>
ACTIVATE: Cal QIE calibration
<!-- Author: Shabnaz x2080 -->


<!-- Date: Sat May 22 00:03:18.157 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182998">182998</A>
TERMINATE: Cal QIE Calibration done
<!-- Author: Shabnaz x2080 -->


<!-- Date: Sat May 22 00:05:30.280 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=182999">182999</A>
ACTIVATE: CAL QIE for WHA and a possible calib load.
<!-- Author: Shabnaz x2080 -->


<!-- Date: Sat May 22 00:07:42.302 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183000">183000</A>
 Activated at 2004.05.22 00:06:51
<!-- Author: RunControl -->


<!-- Date: Sat May 22 00:07:43.134 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183000">183000</A>
ACTIVATE: Ack!  work, dang it!
<!-- Author: BB x2080 -->