<!-- Title: Eve shift. Mon Jan 24, 2005 -->
<!-- SciCo: karen byrum -->
<!-- DAQAce: jonathan efron -->
<!-- MonAce: paul  dituro -->
<!-- CO: darin acosta -->
<!-- OpManager: sungwon  -->
<!-- Notes: -->
<pre>Access over - do calibrations - the word is a shot is possible on this shift.   </pre>

<!-- Date: Mon Jan 24 16:09:06.014 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=192857">192857</A>
ACTIVATE: RunSet: ACE_LED_ALL cem led

<!-- Author: efron (x2080) -->


<!-- Date: Mon Jan 24 16:14:41.090 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=192857">192857</A>
TERMINATE: cem led calibration

<!-- Author: efron (x2080) -->


<!-- Date: Mon Jan 24 16:20:02.908 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=192858">192858</A>
ACTIVATE: RunSet: CALIB_CLC_QIE 
<!-- Author: efron (x2080) -->


<!-- Date: Mon Jan 24 16:20:27.428 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=192858">192858</A>
TERMINATE: clc done again at request of co

<!-- Author: efron (x2080) -->


<!-- Date: Mon Jan 24 16:26:13.950 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=192859">192859</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL  Trigger: COSMICS [14,503,442] 
<!-- Author: efron (x2080) -->


<!-- Date: Mon Jan 24 16:33:12.331 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=192859">192859</A>
TERMINATE: first of plug calibration runs
<!-- Author: efron (x2080) -->


<!-- Date: Mon Jan 24 16:34:59.712 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=192860">192860</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C  Trigger: COSMICS [14,503,442] 
<!-- Author: efron (x2080) -->


<!-- Date: Mon Jan 24 16:37:05.920 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=192860">192860</A>
TERMINATE: 2/3 of plug calibrations
<!-- Author: efron (x2080) -->


<!-- Date: Mon Jan 24 16:38:38.769 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=192861">192861</A>
ACTIVATE: RunSet: BOTHPLUGS_MM  Trigger: COSMICS [14,503,442] 
<!-- Author: efron (x2080) -->


<!-- Date: Mon Jan 24 16:42:35.012 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=192861">192861</A>
TERMINATE: end 3/3 of plug calibrations
<!-- Author: efron (x2080) -->


<!-- Date: Mon Jan 24 16:49:14.662 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=192862">192862</A>
 Activated at 2005.01.24 16:48:59
<!-- Author: RunControl -->



<!-- Date: Mon Jan 24 16:49:16.425 2005 -->
<table border=0><td><tr><td bgcolor=#e0e030><pre>For tonight, please contact me on my cell  
phone (posted in the phone list) in case of  
DAQ emergencies, rather than the DAQ pager. 
</pre></td></tr></td></table>
<!-- Author: W.Badgett -->

<!-- Date: Mon Jan 24 16:50:03.284 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=192862">192862</A>
ACTIVATE: RunSet: ACE_COSMIC_NOTRACKS  Trigger: COSMICS_NOTRACKS [9,467,511] 
<!-- Author: efron (x2080) -->



<!-- Date: Mon Jan 24 16:56:37.964 2005 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2005&action=view&page=-1146&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2005&action=view&page=-1146  width='300' align='top'></a></td><td>TevMon has reported some high values for MEAN L1COL1, however a check of the Lumberjack plot shows it to be holding steady.</td></tr></table>
<!-- Author: Paul -->
<!-- Comment: Mon Jan 24 18;05;08 comment by...efron -->
TEVMON belly aches again at 6:03 pm
<!-- Comment: Mon Jan 24 18;12;17 comment by...efron -->
TevMon complained again at 6:09.  Restarted TevMon to see if that cures its ills
<!-- Date: Mon Jan 24 17:12:44.364 2005 -->
restarting HV monitor for CPR etc
<!-- Author: convery -->

<!-- Date: Mon Jan 24 17:24:32.572 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=192862">192862</A>
 Terminated at 2005.01.24 17:24:14
<!-- Author: RunControl -->


<!-- Date: Mon Jan 24 17:24:33.665 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=192862">192862</A>
TERMINATE: Request for clc00
<!-- Author: efron (x2080) -->



<!-- Date: Mon Jan 24 17:31:19.275 2005 -->
E2 cold compressor just tripped.  cold compressor has to get changed out.  No estimate on shot setup.
<!-- Author: byrum -->
<!-- Comment: Mon Jan 24 17;45;33 comment by...byrum -->
Estimate is now 21:00 for TeV turn on.




<!-- Date: Mon Jan 24 18:17:44.365 2005 -->
I have two pinkies in the ISL Wedge B4.  I spoke with the Si pager carrier and he explained that the tolerances for some of these chips are so tight during running that they sometimes fall below when idle.  The histories for these two chips are very stable, so they will be ignored.
<!-- Author: Paul -->
<!-- Comment: Mon Jan 24 19;50;03 comment by...bbrau -->
As always, pinkies are only cause for concern *during data taking with silicon*.  If the silicon is at standby, the chips draw different amounts of current than they do during data taking, and this is normal.  Unfortunately, IMon is not smart enough to not mark channels pink when we aren't taking data.  The instructions about pinkies which are on the counter in front of the IMon monitor explain this.


<!-- Date: Mon Jan 24 18:21:39.921 2005 -->
Restarting TevMon did not stop its constant complaints.  I turned off the LICOLI warning in TevMon.  Will turn it back on during shot setup 
<!-- Author: efron -->


<!-- Date: Mon Jan 24 18:53:07.087 2005 -->
Upon approval by Ops. Managers, Karen and Rob, we increased gap trigger thresholds for the BSC-2, 3 and 4 counters in the BSC ADMEMs (slot 16 and 17 in CLC00) such that the gap trigger decision is made only on BSC-1. The threshold of BSC-1 is not changed. We would like to keep this gap threshold until we understand better the new scintillation counters (installed during the shutdown) and the new photobues (replaced in last week's access). This change will bring the gap trigger rates somewhat higher, up to around the pre-shutdown level, since the gap trigger decision was essentially made only from BSC-1 in the pre-shutdown.
<!-- Author: K. Terashi -->

<!-- Date: Mon Jan 24 19:00:28.600 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=192863">192863</A>
 Activated at 2005.01.24 18:59:42
<!-- Author: RunControl -->


<!-- Date: Mon Jan 24 19:19:40.098 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=192863">192863</A>
ACTIVATE: RunSet: ACE_COSMIC_NOTRACKS  Trigger: COSMICS_NOTRACKS [9,467,511] 
<!-- Author: efron (x2080) -->



<!-- Date: Mon Jan 24 19:48:29.985 2005 -->
Run coordinator elog updates estimate for replacing cold compressor and cooling it down now to around 22:30-23:00.   Then they would start where they left off which would be to resume the long dry squeeze and follow that by combination shot.
<!-- Author: byrum -->
<!-- Comment: Mon Jan 24 20;03;51 comment by...byrum -->
MCR just calls - Make that estimate midnight when we should have a replaced cold compressor.


<!-- Comment: Mon Jan 24 21;27;11 comment by...byrum -->
Clarification - 22:00 to have a replaced cold compressor, an hour to test it out and then an hour or so of dry squeeze.  Midnight is be the most optimistic estimate for shot setup.
<!-- Date: Mon Jan 24 20:26:15.342 2005 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2005&action=view&page=-1147&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2005&action=view&page=-1147  width='300' align='top'></a></td><td><p>Time to set TRACER done for some of the slower crates 
in the system 
 
<p>This is for Run 192424 L_initial = 109E30 
 
<p>Looks like some COT crates have very long times. This 
is probably due to the noise that was recently seen. 
 
<p>CLC crate has lond times. This will be reduced when 
we shorten the time window for the TDC in the CLC  
crate.</td></tr></table>
<!-- Author: Frank Chlebana -->


<!-- Date: Mon Jan 24 21:19:32.318 2005 -->
TevMon ICOLI alarm back on, things seem stable
<!-- Author: efron -->



<!-- Date: Mon Jan 24 23:00:48.287 2005 -->
Tev Turnon beginning
<!-- Author: byrum -->

<!-- Date: Mon Jan 24 23:55:26.633 2005 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=192793>
192793</A> 
x2F119</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_3_00 [2,566,510]</TD>                                       
<TD ALIGN=CENTER>01:56:55</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>00:42:17</TD>                                          
<TD ALIGN=RIGHT>     50,131,802</TD>                                    
<TD ALIGN=RIGHT>       663,287</TD>                                     
<TD ALIGN=RIGHT>     164,390</TD>                                       
<TD ALIGN=RIGHT>  129.352</TD>                                          
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>23:55:03</TD>                                                       
<TD ALIGN=CENTER>00:42:17</TD>                                          
<TD ALIGN=RIGHT>      50,131,802</TD>                                   
<TD ALIGN=RIGHT>        663,287</TD>                                    
<TD ALIGN=RIGHT>       164,390</TD>                                     
<TD ALIGN=RIGHT>  129.352</TD>                                          
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Mon Jan 24 23:57:05.326 2005 -->
<font color=red>Shift Summary: </font><pre>Start shift ending an access. Tev plan is to do a series of 3 ramps, followed by long dry squeeze and then into shot setup.  We make it through 2 ramps. 
16:09 - Start calibrations. 
17:31 - E2 cold compressor tripped and had to get changed out.  
Other: Have some problems TEVMON reporting high Silicon values with L1COL1 even though silicon is off and no beam.  We disable warning and later in the shift turn it back on. Terashi increased gap trigger thresholds for BSC -2,3,and 4 counters in BSC ADMEN boards (new phototubes are more efficient). 
19:48 - MCR says estimate for next shot setup is now midnight so CO goes home 
and aces flip a coin - Monitoring ace leaves.  both leave phone numbers in case 
something changes. 
23:00 - compressor swapped out and cold - tev turn on begins. 
End shift with Tev starting a series of 3 ramps.</pre> 
<!-- Author: byrum -->