<!-- Title: Owl shift. Wed Apr 13, 2005 -->
<!-- SciCo: Christoph Paus -->
<!-- DAQAce: Oriol Salto -->
<!-- MonAce: Khaldoun Makhoul -->
<!-- CO: Oleg Poukhov -->
<!-- OpManager: Bob Wagner -->
<!-- Notes: -->
<pre>Status: Access is still going on (Silicon, COT TDC replacements) 
Plan  : Get a shot and do some uninterupted, smooth and cool 
        running</pre>

<!-- Date: Wed Apr 13 00:38:23.183 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=196432">196432</a>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Oriol x2080 -->


<!-- Date: Wed Apr 13 00:39:30.746 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=196432">196432</a>
TERMINATE: Done
<!-- Author: Oriol x2080 -->



<!-- Date: Wed Apr 13 00:46:28.561 2005 -->
COT front-end checks out after Eric James swapped TDC's in 
crate 7.  Temporary problem with crate 6 (no calib signals to 
crate).  Eric swapped twin-ax 1->3 fan-out output to crate 6 
to spare channel, fixed.
<!-- Author: ambrose -->


<!-- Date: Wed Apr 13 00:57:07.828 2005 -->
<pre>Access has finished.</pre>
<!-- Author: Christoph Paus -->


<!-- Date: Wed Apr 13 01:03:01.187 2005 -->
Cycled power for ISU because of non-responsible CCU. Everything is OK now.
<!-- Author: Oleg Poukhov -->


<!-- Date: Wed Apr 13 01:43:40.061 2005 -->
<pre>Jianrong and I replaced the TDC boards on slots 11-21  
of b0cot07 with boards modified for fast clear.  Also,  
I replaced the TDC board in slot21 of b0clc00 because  
of errors observed earlier today originating from this  
slot.  Please pay carefull attention to clc monitoring  
plots at the start of the next store to make sure that  
everything looks ok.</pre>
<!-- Author: Eric James -->

<!-- Date: Wed Apr 13 01:48:26.768 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=196438">196438</a>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [16,623,539] 
<!-- Author: Oriol x2080 -->
<!-- Comment: Wed Apr 13 01;52;26 comment by...Oriol -->
Note that this is not the L2 tag in the whiteboard [561].<br>
It was impossible to run with it: L2 TagSet invalid for Pulsar/Linux. Using the deafult one in the RC config for L2_TORTURE.


<!-- Comment: Wed Apr 13 02;11;47 comment by...Oriol -->
Running fine but getting a lot of this messages in the error display:<br>
<br>
(MLE) b0svt06:Messenger:Apr 13, 2005 2:07:02 AM->FER_E_SVT, GB 6.1 nw=15 (mod7=1) ee=60644d event=296002<br>
(MLE) b0svt06:Messenger:Apr 13, 2005 2:07:12 AM->FER_E_SVT, GB 6.1 nw=15 (mod7=1) ee=60674a event=298722<br>
(MLE) b0svt06:Messenger:Apr 13, 2005 2:07:12 AM->FER_E_SVT, GB 6.1 nw=15 (mod7=1) ee=60648b event=298813<br>
(MLE) b0svt06:Messenger:Apr 13, 2005 2:07:40 AM->FER_E_SVT, GB 6.1 nw=15 (mod7=1) ee=686456 event=306521<br>
(MLE) b0svt06:Messenger:Apr 13, 2005 2:07:52 AM->FER_E_SVT, GB 6.1 nw=15 (mod7=1) ee=60646a event=309805<br>
(MLE) b0svt06:Messenger:Apr 13, 2005 2:07:53 AM->FER_E_SVT, GB 6.1 nw=15 (mod7=1) ee=686738 event=310051<br>
(MLE) b0svt06:Messenger:Apr 13, 2005 2:07:53 AM->FER_E_SVT, GB 6.1 nw=15 (mod7=1) ee=686715 event=310077<br>
(MLE) b0svt06:Messenger:Apr 13, 2005 2:07:55 AM->FER_E_SVT, GB 6.1 nw=15 (mod7=1) ee=68677c event=310490<br>
(MLE) b0svt06:Messenger:Apr 13, 2005 2:07:58 AM->FER_E_SVT, GB 6.1 nw=15 (mod7=1) ee=686447 event=311281<br>
(MLE) b0svt06:Messenger:Apr 13, 2005 2:08:02 AM->FER_E_SVT, GB 6.1 nw=15 (mod7=1) ee=686782 event=312399<br>
(MLE) b0svt06:Messenger:Apr 13, 2005 2:08:03 AM->FER_E_SVT, GB 6.1 nw=15 (mod7=1) ee=786744 event=312784<br>
(MLE) b0svt06:Messenger:Apr 13, 2005 2:08:21 AM->FER_E_SVT, GB 6.1 nw=15 (mod7=1) ee=606741 event=317775<br>
(MLE) b0svt06:Messenger:Apr 13, 2005 2:08:25 AM->FER_E_SVT, GB 6.1 nw=15 (mod7=1) ee=686473 event=318670<br>

<!-- Date: Wed Apr 13 01:55:43.164 2005 -->
<h3> Silicon access report </h3> 
We went in two work on two problems, e023 which had been tripping and f404, which has been off for two years. We were able to get f404 back and we have to see how thing look now. <br> 
We also brought back e224, which had some trips a month ago.<br> 
Details can be found in the Silicon e-log
<!-- Author: Ankush/Jen/Marcel/Rong/Sebastian -->



<!-- Date: Wed Apr 13 03:03:51.431 2005 -->
Sergei mounted data1 properly in b0dap10 and now Spy Alpha is working properly.
<!-- Author: Oriol -->


<!-- Date: Wed Apr 13 03:04:55.980 2005 -->
<table border=0><td><tr><td bgcolor=#e0e030>Have remounted /data1 file system on b0dap10.</td></tr></td></table>
<!-- Author: Sergei -->

<!-- Date: Wed Apr 13 03:06:21.203 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=196438">196438</a>
TERMINATE: Let's be prepared for the shot.
<!-- Author: Oriol x2080 -->


<!-- Date: Wed Apr 13 03:34:14.055 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=196439">196439</a>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_3_02 [1,639,563] running w/ the trig table in the whiteboard
<!-- Author: Oriol x2080 -->



<!-- Date: Wed Apr 13 03:45:51.221 2005 -->
<TABLE bgcolor=#cdc9a5> 
<TR><TH>Date</th><th>Time</th><th>BLM</th><th colspan=2>Dose</th></tr> 
<tr><td>2005.04.13</td><td>03:44:42</td><td>W Inner BLM</td><td>0.38</td><td>RADS</td></tr> 
<tr><td>2005.04.13</td><td>03:44:42</td><td>W Outer BLM</td><td>0.00</td><td>RADS</td></tr> 
<tr><td>2005.04.13</td><td>03:44:42</td><td>E Inner BLM</td><td>0.21</td><td>RADS</td></tr> 
<tr><td>2005.04.13</td><td>03:44:42</td><td>E Outer BLM</td><td>0.57</td><td>RADS</td></tr> 
</table> 
<strong>Integrated dosage</strong>
<!-- Author: khaldoun -->


<!-- Date: Wed Apr 13 03:48:54.600 2005 -->
<pre>Final protons. 
</pre>
<!-- Author: Christoph Paus -->


<!-- Date: Wed Apr 13 04:09:35.619 2005 -->
<pre>Final antiprotons are complete.</pre>
<!-- Author: Christoph Paus -->

<!-- Date: Wed Apr 13 04:12:10.356 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=196439">196439</a>
TERMINATE: preparing for datataking!
<!-- Author: Oriol x2080 -->



<!-- Date: Wed Apr 13 04:27:56.740 2005 -->
<pre>Store #4097 - scraping complete 
 
TevPR -  8883e9  
TevPB -   975e9 
</pre>
<!-- Author: Christoph Paus -->


<!-- Date: Wed Apr 13 04:30:33.970 2005 -->
<pre>Initial Luminosity 
 
C:B0ILUM - 81.5e30 
</pre>
<!-- Author: Christoph Paus -->


<!-- Date: Wed Apr 13 04:32:46.469 2005 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2005&action=view&page=-6219&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2005&action=view&page=-6219  width='300' align='top'></a></td><td>CAL QIE Calibration (RUN 196342).</td></tr></table>
<!-- Author: Oleg -->
<!-- Comment: Wed Apr 13 09;30;37 comment by...Kwangzoo -->
WHA 17W 0 channel has disappeared.
<!-- Date: Wed Apr 13 04:34:23.900 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=196440">196440</a>
 Activated at 2005.04.13 04:33:59
<!-- Author: RunControl -->


<!-- Date: Wed Apr 13 04:34:24.546 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=196440">196440</a>
ACTIVATE: RunSet: AAA_CURRENT  Trigger: PHYSICS_3_02 [1,639,563] 
<!-- Author: Oriol x2080 -->



<!-- Date: Wed Apr 13 04:35:55.654 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-6220&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-6220' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-6221&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-6221' width='300'></a></center>
<!-- Author: losses & svrad during scraping --khaldoun -->


<!-- Date: Wed Apr 13 04:37:44.372 2005 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2005&action=view&page=-6222&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2005&action=view&page=-6222  width='300' align='top'></a></td><td>CAL QIE Calibration (RUN 196342).</td></tr></table>
<!-- Author: Oleg -->


<!-- Date: Wed Apr 13 04:49:05.615 2005 -->
L2 Decision Timeout.[RXPT].    HRRed.
<!-- Author: Oriol :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=196440' target=_top>196440</a>) -->


<!-- Date: Wed Apr 13 04:56:22.611 2005 -->
Level 2 Decision Timeout. HRRed.<br> 
But the error message does not appear neither in the error display nor in the error file...<br>
<!-- Author: Oriol :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=196440' target=_top>196440</a>) -->
<!-- Comment: Wed Apr 13 04;58;51 comment by...Oriol -->
Level 2 Decision Timeout. HRRed.<br>
Still not appearing in the logs.


<!-- Date: Wed Apr 13 05:36:51.818 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-6223&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-6223' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-6224&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-6224' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-6225&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-6225' width='300'></a></center>
<!-- Author: 4:30-5:30 hourly --khaldoun -->


<!-- Date: Wed Apr 13 05:38:51.295 2005 -->
1 crate/s: b0pcal07(41),  Done TO. [RXPT]. Shepherded it.<br> 
Recieved no message saying "Shepherd was successfull"... but obviously it was.<br>  
<!-- Author: Oriol :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=196440' target=_top>196440</a>) -->


<!-- Date: Wed Apr 13 05:44:12.675 2005 -->
L2 Decision Timeout.[RXPT]. HRRed.  
<!-- Author: Oriol :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=196440' target=_top>196440</a>) -->


<!-- Date: Wed Apr 13 05:57:13.734 2005 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2005&action=view&page=-6226&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2005&action=view&page=-6226  width='300' align='top'></a></td><td>CMP one of the wedges has no High Voltage</td></tr></table>
<!-- Author: Oleg -->
<!-- Comment: Wed Apr 13 06;05;03 comment by...Khaldoun, Oleg & Oriol -->
<b>DQMon alarm</b> complaining about <b>CMP</b> Bottom.<br>
In IFIX, all the detector appears with the HV on, but the CO histogram shows a clear stack with no signal.<br>
Went downstairs and saw that the power supply connected to bottom A2 showed no current in the amperimeter, even though it was ON.<br>



<!-- Comment: Wed Apr 13 06;25;51 comment by...Christoph Paus -->
<pre>I decided it was not useful to call the CMP experts because
the problem cannot be solved as discussed in yesterdays meeting.</pre>
<!-- Comment: Wed Apr 13 07;21;46 comment by...Dale -->
It appears that the anode (A2) tripped off [switch was on - but HV off due to trip] sometime between when Xiaojian (see his <a href=
'/cgi/elog/elog.pl?nb=2005&action=view&page=305&frame=2&anchor=12:32:29.531'>
summary</a>) last looked at it and this morning.  (Note "D2" was on and holding).
<!-- Comment: Wed Apr 13 09;17;22 comment by...Xiaojian -->
Now CMP bottom A2 has recovered, let me summarize it from the OWL shift of April 12 to the OWL shift of April 13:

1) It started trip at around 3:30am in April 12. As it did 4 times before during this year, a number of tries of ramping it up has failed. So it was turned off, and the run should be marked as CMP bad. 

2) Phil, Dale and Xiaojian did a number of detail test to diagnose the problem. The problem that we found is: The A2 section is very sensitive to the ramping process. The trip happens at a HV around 3000-3600. We could not estimate the magnitude of the tripping current, since the drawing current was always stable and normal until it suddenly tripped, neither the monitor display nor the amperemeter of the portable HV source can capture the tripping current. We used the portable HV source to power the section to the full HV, ignoring the trips during the ramping. It holds the HV without problem. 

3) Dale checked the section at 7:21am on April 13, finding it has tripped. Probably due to the shot setup process around 4am. So, the section has been on full HV for near 20 hours with the portable HV supply. From the experience of the section, leaving the section on full HV for a number of hours seems to be the only way for the section to recover. 

4) Dale connected the section to the CAEN, and ramped it up to full HV around 7:30am. 

  I should point out that fixing the problem will be very difficult due to the following two reasons:

i) Given the location, we need long access, I estimate 5 hours at least. 

ii) The problem section is not fully broken. It works most of the time, except acts up for a number of hours (5 - 20 hours), then recovers itself. Therefore, one needs the long access happens at the time when the section is broken. 

  The section is running well now. From the history, I estimate it may trip badly again after 7 -10 days. 


<!-- Comment: Wed Apr 13 09;18;55 comment by...Xiaojian -->
<pre>Sorry, I have put my comments above in HTML mode, making it not readable. Please ignore it, and check below.


Now CMP bottom A2 has recovered, let me summarize it from the OWL shift of April 12 to the OWL shift of April 13:

1) It started trip at around 3:30am in April 12. As it did 4 times before during this year, a number of tries of ramping it up has failed. So it was turned off, and the run should be marked as CMP bad. 

2) Phil, Dale and Xiaojian did a number of detail test to diagnose the problem. The problem that we found is: The A2 section is very sensitive to the ramping process. The trip happens at a HV around 3000-3600. We could not estimate the magnitude of the tripping current, since the drawing current was always stable and normal until it suddenly tripped, neither the monitor display nor the amperemeter of the portable HV source can capture the tripping current. We used the portable HV source to power the section to the full HV, ignoring the trips during the ramping. It holds the HV without problem. 

3) Dale checked the section at 7:21am on April 13, finding it has tripped. Probably due to the shot setup process around 4am. So, the section has been on full HV for near 20 hours with the portable HV supply. From the experience of the section, leaving the section on full HV for a number of hours seems to be the only way for the section to recover. 

4) Dale connected the section to the CAEN, and ramped it up to full HV around 7:30am. 

  I should point out that fixing the problem will be very difficult due to the following two reasons:

i) Given the location, we need long access, I estimate 5 hours at least. 

ii) The problem section is not fully broken. It works most of the time, except acts up for a number of hours (5 - 20 hours), then recovers itself. Therefore, one needs the long access happens at the time when the section is broken. 

  The section is running well now. From the history, I estimate it may trip badly again after 7 -10 days. 

</pre>
<!-- Date: Wed Apr 13 06:10:28.923 2005 -->
Level 2 Decision Timeout. HRRed.<br> 
The error does not appear in any log.<br>  
<!-- Author: Oriol :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=196440' target=_top>196440</a>) -->
<!-- Comment: Wed Apr 13 06;35;11 comment by...Oriol -->
Again: Level 2 Decision Timeout. HRRed.<br>
Again: it appears nowhere in the logs.


<!-- Comment: Wed Apr 13 06;51;37 comment by...Oriol -->
Again: Level 2 Decision Timeout. HRRed.<br>
Again: it appears nowhere in the logs.
<!-- Comment: Wed Apr 13 07;40;42 comment by...Donatella T. -->
looking into this.

<!-- Date: Wed Apr 13 06:39:07.584 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-6227&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-6227' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-6228&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-6228' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-6229&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-6229' width='300'></a></center>
<!-- Author: 5:30-6:35 hourly --khaldoun -->




<!-- Date: Wed Apr 13 07:22:09.507 2005 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2005&action=view&page=-6230&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2005&action=view&page=-6230  width='300' align='top'></a></td><td>CCR Wedge Pad Occupancy - suspicious wedges 36-40. Is it OK?</td></tr></table>
<!-- Author: Oleg -->
<!-- Comment: Wed Apr 13 07;26;25 comment by...Larrt -->
That is normal!  


<!-- Date: Wed Apr 13 07:28:11.075 2005 -->
<a href= 
'/cgi/elog/elog.pl?nb=2005&action=view&page=307&frame=2&anchor=05:57:13.734'>Cont. from above,</a> Dale decided to turn on the anode.  It worked after Dale got the current scale right.  Dale hooked up to the CAEN/iFIX to see if it would hold.  It does at ~.1-.2 microA (on A2).<br><br> 
More details to come.
<!-- Author: Khaldoun, Oriol, and Dale -->
<!-- Comment: Wed Apr 13 07;34;55 comment by...Dale -->
I've reenabled inhibits for CMP - If bottom section 2 trips - turn it off and continue running.  Expert/SPL will likely ask for a new run if CMP continues to hold.  I'm going back to my office.
<!-- Date: Wed Apr 13 07:31:28.743 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-6231&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-6231' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-6232&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-6232' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-6233&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-6233' width='300'></a></center>
<!-- Author: 6:35-7:30 hourly --khaldoun -->



<!-- Date: Wed Apr 13 07:55:49.732 2005 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=196440>
196440</A> 
x2FF58</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_3_02 [1,639,563]</TD>                                       
<TD ALIGN=CENTER>04:33:59</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>03:07:27</TD>                                          
<TD ALIGN=RIGHT>    228,654,615</TD>                                    
<TD ALIGN=RIGHT>     3,248,302</TD>                                     
<TD ALIGN=RIGHT>     834,758</TD>                                       
<TD ALIGN=RIGHT>  725.029</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>07:55:04</TD>                                                       
<TD ALIGN=CENTER>03:07:27</TD>                                          
<TD ALIGN=RIGHT>     228,654,615</TD>                                   
<TD ALIGN=RIGHT>      3,248,302</TD>                                    
<TD ALIGN=RIGHT>       834,758</TD>                                     
<TD ALIGN=RIGHT>  725.029</TD>                                          
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Wed Apr 13 08:00:02.228 2005 -->
Virgil Barnes takes over from Christoph as SciCo at 7:30
<!-- Author: virgil barnes :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=196440' target=_top>196440</a>) -->


<!-- Date: Wed Apr 13 08:19:15.595 2005 -->
<font color=red>Shift Summary: </font><pre> - Access finished succesfully at 0:57  
   - TDCs were successfully replaced  
   - Silicon work was a success as well (2 ladders recovered)  
  
 - shot came in pretty smoothly (around 4)  initial lumi = 82E30  
 - smooth running (CMP was off until around 7:30)  
</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   196440
Delivered Luminosity   662  
Acquired Luminosity    592  
Efficiency             89.4
   
</td></tr></table></pre>
<!-- Author: virgil barnes -->