<!-- Title: Owl shift. Fri Jun  1, 2001 -->
<!-- SciCo: None -->
<!-- DAQAce: Aaron, Yanwen -->
<!-- MonAce: Michael Riveline -->
<!-- CO: &nbsp; -->
<!-- OpManager: &nbsp; -->
<!-- Notes: -->
<pre>Try to run with as much of the detector as possible. 
 
Take calibrations. 
 
Take long cosmic runs using COSMIC_1[3,126] 
</pre>


<!-- Date: Fri Jun 1 01:04:38 2001 -->
Inherited cosmics run from previous shift which includes almost 
all available crates (except IMU01,  and of course no SVX). 
 
Will try to take about 10^6 events and then do calibrations.
<!-- Author: Aaron :: (run <a href='/cgi-bin/runSummaryRun.cgi?116441' target=_top>116441</a>) -->


<!-- Date: Fri Jun 1 01:28:41 2001 -->
<pre>COT high voltage testing completed at 1:22AM.  COT was at 100% for approximately 1.5hours.  High voltage turned off and locked out.</pre>
<!-- Author: Aseet Mukherjee -->


<!-- Date: Fri Jun 1 01:50:46 2001 -->
<pre>	From (East) Plug closing at 15:00, until 24:00, Carl Bromberg and Gene Flanagan (Michigan State University) installed 72 TSU counters between the steel slabs of the East toroid. All counters have been cabled. All counters show readout of PAD (post-PMT Amplifer & Discriminator) current, through the CCU (48 channel Control & Concentrator Unit). Tomorrow morning, signal checks will be made. We hope to complete these checks by the end of access on Friday. The power relay in CCU17 (SE toroid TSU counters) does not always stay on. A spare will not be available until early next week. 
	At some point the TSU counter positions should be surveyed. Due to irregularities in the toroid iron surface, and radius, some counters had to be displaced (&lt; 1") from their nominal mounting points.  </pre>
<!-- Author: Carl Bromberg -->


<!-- Date: Fri Jun 1 02:55:47 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-1012&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-1012  width='300' align='top'></td><td>After ~938k events, we got a done timeout from the hadron tdc crate.  HRR fixed it.  Here is the error message.</a></td></tr></table>
<!-- Author: Aaron -->


<!-- Date: Fri Jun 1 03:10:24 2001 -->
<pre>End Run 116441 with 1002987 events 
COSMIC_1[3,126]</pre>
<!-- Author: Yanwen :: (run <a href='/cgi-bin/runSummaryRun.cgi?116441' target=_top>116441</a>) -->


<!-- Date: Fri Jun 1 03:43:09 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-1013&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-1013  width='300' align='top'></td><td>Run 116442 Calorimeter QIE Calib.  </a></td></tr></table>
<!-- Author: Yanwen -->
<!-- Comment: Fri Jun  1 03;44;12 comment by...Aaron -->
ADMEM in ccal03 failed QIE calibration.


<!-- Date: Fri Jun 1 03:55:30 2001 -->
<table border=0><td><tr><td bgcolor=#e0e030><pre>Run 116443. LED Calib. No error message seen. 
But, after 'End-&gt;IDLE' 
1 crate/s : b0ccal04(4), in error.[RXPT]</pre></td></tr></td></table>
<!-- Author: Yanwen :: (run <a href='/cgi-bin/runSummaryRun.cgi?116443' target=_top>116443</a>) -->


<!-- Date: Fri Jun 1 04:02:39 2001 -->
<pre>We can not do Xenon Calib. because Aaron forgot the passwd.</pre>
<!-- Author: Yanwen -->


<!-- Date: Fri Jun 1 04:11:47 2001 -->
Got message that CLC Qie calibration failed. 
 
Try again.
<!-- Author: Aaron :: (run <a href='/cgi-bin/runSummaryRun.cgi?116445' target=_top>116445</a>) -->
<!-- Comment: Fri Jun  1 04;22;31 comment by...Aaron -->
<pre>
*******************   Start of FAILURE messages   ************** VALUE    CAP RANGE**
* ADMEM=16 QIE= 1 Failure 14: Sigma(offset) too big          (   12.8131) ( 0) ( 0) *
* ADMEM=16 QIE= 2 Failure 14: Sigma(offset) too big          (   18.5518) ( 0) ( 0) *
* ADMEM=16 QIE= 3 Failure 14: Sigma(offset) too big          (    1.0287) ( 0) ( 0) *
* ADMEM=16 QIE= 4 Failure 14: Sigma(offset) too big          (    1.5909) ( 0) ( 0) *
* ADMEM=16 QIE= 5 Failure 14: Sigma(offset) too big          (   12.3985) ( 0) ( 0) *
* ADMEM=16 QIE= 6 Failure 14: Sigma(offset) too big          (    1.6580) ( 0) ( 0) *
* ADMEM=16 QIE= 7 Failure 14: Sigma(offset) too big          (    1.9111) ( 0) ( 0) *
* ADMEM=16 QIE= 8 Failure 14: Sigma(offset) too big          (    1.6102) ( 0) ( 0) *
* ADMEM=16 QIE= 9 Failure 14: Sigma(offset) too big          (   14.2262) ( 0) ( 0) *
* ADMEM=16 QIE=11 Failure 14: Sigma(offset) too big          (   11.1872) ( 0) ( 0) *
* ADMEM=16 QIE=12 Failure 14: Sigma(offset) too big          (   20.6083) ( 0) ( 0) *
* ADMEM=16 QIE=13 Failure 14: Sigma(offset) too big          (   15.0092) ( 0) ( 0) *
* ADMEM=17 QIE= 0 Failure  6: Raw pedestal RMS TOO big       (   31.9158) ( 0) ( 0) *
* ADMEM=17 QIE= 1 Failure  6: Raw pedestal RMS TOO big       (   36.3168) ( 0) ( 0) *
* ADMEM=17 QIE= 4 Failure  6: Raw pedestal RMS TOO big       (   26.2297) ( 0) ( 0) *
* ADMEM=17 QIE= 5 Failure  6: Raw pedestal RMS TOO big       (   19.4673) ( 0) ( 0) *
* ADMEM=17 QIE= 8 Failure  6: Raw pedestal RMS TOO big       (   41.7117) ( 0) ( 0) *
* ADMEM=17 QIE= 9 Failure  6: Raw pedestal RMS TOO big       (   37.4779) ( 0) ( 0) *
*******************   End of FAILURE messages   *************************************                                       
</pre>
<!-- Comment: Fri Jun  1 04;23;05 comment by...Aaron -->
Next run, 116446 also had same problem.
<!-- Comment: Fri Jun  1 07;12;50 comment by...R.J. Tesarek -->
These detectors are fairly noisy.  All the above messages are
a by product of that noise.  Tollerances for failures for these
ADMEMs need to be updated to accomodate this.


<!-- Date: Fri Jun 1 04:21:33 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-1014&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-1014  width='300' align='top'></td><td>CLC Calibration failed. (Run 116445,116446)</a></td></tr></table>
<!-- Author: Yanwen  -->




<!-- Date: Fri Jun 1 04:26:05 2001 -->
Key number 87 returned. 
  
<!-- Author: Juan Pablo Fernandez -->


<!-- Date: Fri Jun 1 04:30:33 2001 -->
<pre>Run116447. COT Calibration done: no error message seen. </pre>
<!-- Author: Yanwen :: (run <a href='/cgi-bin/runSummaryRun.cgi?116447' target=_top>116447</a>) -->


<!-- Date: Fri Jun 1 04:54:07 2001 -->
<pre>CMX00 failed to coldstart for calibration.  Tried twice and 
then rebooted it. 
 
Then, the calibration run was able to start, but it died with 
the following message (this is from console while logged into cmx00) 
 
Monitor: readout/sent rate      2.1 Hz /      2.1 Hz, cur evt is 16096 bytes 
Runtime Error 0: TDC_readAllFifo: slot=7 No valid header, 0x2804a4d, evt=78 
Runtime Error 1: TDC_readAllFifo: slot=7 No valid header, 0x280514e, evt=78 
Runtime Error 2: TDC_readAllFifo: slot=7 No valid header, 0x8280693f, evt=78 
Runtime Error 3: TDC_readAllFifo: slot=7 No valid header, 0x82006141, evt=78 
Runtime Error 4: TDC_readAllFifo: slot=7 No valid header, 0x3c04267, evt=78 
Runtime Error 5: TDC_readAllFifo: slot=7 No valid header, 0x3c0524e, evt=78 
Runtime Error 6: TDC_readAllFifo: slot=7 No valid header, 0x83c0d940, evt=78 
Runtime Error 7: TDC_readAllFifo: slot=7 No valid header, 0x3404a52, evt=78 
Runtime Error 8: TDC_readAllFifo: slot=7 No valid header, 0x83405945, evt=78 
Runtime Error 9: TDC_readAllFifo: slot=7 No valid header, 0x2c05a4d, evt=78 
Runtime Error 10: TDC_readAllFifo: slot=7 No valid header, 0x2c041e1, evt=78 
Runtime Error 11: TDC_readAllFifo: slot=7 No valid header, 0x82c0793c, evt=78 
Runtime Error 12: TDC_readAllFifo: slot=7 No valid header, 0x2402251, evt=78 
Runtime Error 13: TDC_readAllFifo: slot=7 No valid header, 0x2401955, evt=78 
Runtime Error 14: TDC_readAllFifo: slot=7 No valid header, 0x82406941, evt=78 
Runtime Error 15: TDC_readAllFifo: slot=7 No valid header, 0x1801957, evt=78 
Runtime Error 16: TDC_readAllFifo: slot=7 No valid header, 0x180294f, evt=78 
Runtime Error 17: TDC_readAllFifo: slot=7 No valid header, 0x81805140, evt=78 
Runtime Error 18: TDC_readAllFifo: slot=7 No valid header, 0x8100613e, evt=78 
TRACER_enableDone: Front End Fatal Error (-1): Tracer done not set                                                           
 
</pre>
<!-- Author: Aaron :: (run <a href='/cgi-bin/runSummaryRun.cgi?116450' target=_top>116450</a>) -->
<!-- Comment: Fri Jun  1 04;55;17 comment by...Aaron -->
Giving up.


<!-- Date: Fri Jun 1 05:15:16 2001 -->
<pre>Run116451,ran, noticed something gone to database.  
But RC never arrived DONE from CALIBRATING, declared ERROR instead.  
</pre>
<!-- Author: Yanwen :: (run <a href='/cgi-bin/runSummaryRun.cgi?116451' target=_top>116451</a>) -->


<!-- Date: Fri Jun 1 05:22:23 2001 -->
Realized  a mistake in the previous configuration.  
Corrected 'Generic Calibraion' to "QIE Calibration' , try again 
the SMX calib. Succeeded..
<!-- Author: Yanwen :: (run <a href='/cgi-bin/runSummaryRun.cgi?115452' target=_top>115452</a>) -->


<!-- Date: Fri Jun 1 05:27:18 2001 -->
Got cryo alarms at 05:26 which were instantly silenced by techs.
<!-- Author: Aaron -->


<!-- Date: Fri Jun 1 05:29:32 2001 -->
TOF Calibration, no error message seen.  
<!-- Author: Yanwen :: (run <a href='/cgi-bin/runSummaryRun.cgi?116453' target=_top>116453</a>) -->


<!-- Date: Fri Jun 1 05:42:09 2001 -->
taken ~36k events of ACE_CALORIMETER_MINIMAL.(Rick's run).
<!-- Author: Yanwen :: (run <a href='/cgi-bin/runSummaryRun.cgi?116454' target=_top>116454</a>) -->


<!-- Date: Fri Jun 1 06:04:02 2001 -->
Restarting cosmics run (saved as config file ACE_JUN_1). 
 
On first event, got a done timeout from hadron tdc.  
(See previous cosmics run) 
 
HRR fixed it. 
 
Let this run until end of shift.
<!-- Author: Aaron :: (run <a href='/cgi-bin/runSummaryRun.cgi?116455' target=_top>116455</a>) -->


<!-- Date: Fri Jun 1 06:35:48 2001 -->
COTW turned off by techs.
<!-- Author: Aaron :: (run <a href='/cgi-bin/runSummaryRun.cgi?116455' target=_top>116455</a>) -->


<!-- Date: Fri Jun 1 07:05:05 2001 -->
<pre>rc gave a java out of memory error: 
 
 
2001.06.01 07:01:59 java.lang.OutOfMemoryError 
Exception in thread "AWT-Motif" 2001.06.01 07:03:02 java.lang.OutOfMemoryE 
rror 
Exception in thread "AWT-Motif" 2001.06.01 07:03:03 java.lang.OutOfMemoryE 
rror 
2001.06.01 07:03:03 java.lang.OutOfMemoryError 
2001.06.01 07:03:03 Exception occurred during event dispatching: 
2001.06.01 07:03:03 java.lang.OutOfMemoryError 
Exception in thread "AWT-Motif" 2001.06.01 07:03:05 java.lang.OutOfMemoryE 
rror 
Exception in thread "AWT-Motif" Exception in thread "AWT-Motif" Exceptionin thread "AWT-Motif" Exception in thread "AWT-Motif" Exception in thread"AWT-Motif" Exception in thread "AWT-Motif" Exception in thread "AWT-Motif 
" Exception in thread "AWT-Motif" Exception in thread "AWT-Motif" Exceptio 
n in thread "AWT-Motif" Exception in thread "AWT-Motif" Exception in threa</pre>
<!-- Author: Aaron :: (run <a href='/cgi-bin/runSummaryRun.cgi?116455' target=_top>116455</a>) -->



<!-- Date: Fri Jun 1 07:18:35 2001 -->
Restarted rc and cosmucs run.
<!-- Author: Aaron :: (run <a href='/cgi-bin/runSummaryRun.cgi?116456' target=_top>116456</a>) -->
<!-- Comment: Fri Jun  1 07;33;24 comment by...Aaron -->
Ended after 105631 events to give CMX00 to expert.


<!-- Date: Fri Jun 1 07:35:47 2001 -->
Released all CCAL crates to Rick so he can look at strange Qie 
problems.
<!-- Author: Aaron -->


<!-- Date: Fri Jun 1 07:51:59 2001 -->
Restarted cosmics run sans ccal03, which Rick has.
<!-- Author: Aaron :: (run <a href='/cgi-bin/runSummaryRun.cgi?116457' target=_top>116457</a>) -->


<!-- Date: Fri Jun 1 07:52:24 2001 -->
Someone configured the SciCo's Netscape to print to the color printer by default.  I set it back to the usual printer.
<!-- Author: Tom LeCompte -->


<!-- Date: Fri Jun 1 07:59:53 2001 -->
<br><b>b0ccal03, ADMEM problems:</b> 
<br>Looking into the problem observed during the QIE calibration, 
I found that a single channel appears to have a QIE chip malfunctioning.  This will need to be replaced while we still have access.  Details may be found in the  
<a href="/cgi/elog/det-system-elog.pl?nb=wedge&action=copyentry&page=59&time=07:54:59"> wedge e-log.</a> 
<br>
<!-- Author: R.J. Tesarek -->


<!-- Date: Fri Jun 1 08:02:09 2001 -->
<font color=red>Shift Summary: </font><pre>End of shift.  Now lets get some beam!!!</pre> 
<!-- Author: Aaron -->