<!-- Title: Owl shift. Sat Jul 14, 2007 -->
<!-- SciCo: Greg Feild -->
<!-- DAQAce: Alexander Paramonov -->
<!-- MonAce: &nbsp; -->
<!-- CO: Laura Sartori -->
<!-- OpManager: Alexander Pronko -->
<!-- Notes: -->
<pre> 
Inherit store 5551, L=63E30. 
End of store time not yet known.</pre>


<!-- Date: Sat Jul 14 00:21:37.871 2007 -->
Nat all the morning calibrations were checked. What is going on?  Why are we doing calibrations and not cheking them? Why are we running after that?
<!-- Author: Alexander P. :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244996' target=_top>244996</a>) -->
<!-- Comment: Sat Jul 14 00;23;34 comment by...Sasha -->
We are checking missing calibrations now. This is 14 hours after it should be done.
<!-- Comment: Sat Jul 14 00;41;08 comment by...S & L -->
We've checked the missing calibrations and all of them are good. This accident should not happen ever again. Each calibration must be checked right away.
<!-- Date: Sat Jul 14 00:22:42.179 2007 -->
<br>L2 decision timeout. 
<br>Fixed by an HRR.
<!-- Author: Sasha :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244996' target=_top>244996</a>) -->
<!-- Comment: Sat Jul 14 01;12;55 comment by...Sasha -->
again
<!-- Comment: Sat Jul 14 01;29;34 comment by...Sasha -->
again
<!-- Comment: Sat Jul 14 01;33;04 comment by...Sasha -->
again
<!-- Comment: Sat Jul 14 01;56;32 comment by...Sasha -->
again
<!-- Comment: Sat Jul 14 01;59;31 comment by...Sasha -->
1 more
<!-- Comment: Sat Jul 14 02;27;11 comment by...Sasha -->
again
<!-- Comment: Sat Jul 14 02;50;34 comment by...Sasha -->
again
<!-- Comment: Sat Jul 14 03;09;47 comment by...Sasha -->
1
<!-- Comment: Sat Jul 14 03;21;30 comment by...Sasha -->
1
<!-- Comment: Sat Jul 14 03;21;39 comment by...Sasha -->
1
<!-- Comment: Sat Jul 14 03;46;22 comment by...Sasha -->
again
<!-- Comment: Sat Jul 14 04;13;16 comment by...Sasha -->
again
<!-- Comment: Sat Jul 14 04;57;18 comment by...Sasha -->
and again
<!-- Comment: Sat Jul 14 05;14;48 comment by...Sasha -->
again
<!-- Comment: Sat Jul 14 05;52;08 comment by...Sasha -->
omg
<!-- Comment: Sat Jul 14 05;55;42 comment by...Sasha -->
again
<!-- Comment: Sat Jul 14 06;08;50 comment by...Sasha -->
more
<!-- Comment: Sat Jul 14 06;20;50 comment by...Sasha -->
and more.
<!-- Comment: Sat Jul 14 06;38;00 comment by...Sasha -->
again
<!-- Comment: Sat Jul 14 06;57;17 comment by...Sasha -->
1 more
<!-- Date: Sat Jul 14 00:28:42.146 2007 -->
Fake ProcMon Error. IMON went red. It turned back to green after double ckeking it.
<!-- Author: Sasha :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244996' target=_top>244996</a>) -->


<!-- Date: Sat Jul 14 00:31:52.669 2007 -->
<pre>Tom called in and said that the UPS and several L1 DPS 
triggers have not been enabled properly. 
 
This is a new trigger table so the problem may be there ... 
 
We restarted ScalerMon to see if that fixes the problem. 
 
</pre>
<!-- Author: gregfeild :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244996' target=_top>244996</a>) -->
<!-- Comment: Sat Jul 14 05;36;40 comment by...greg -->
<pre>looks like things kicked in at some point.
</pre>
<!-- Date: Sat Jul 14 01:36:11.168 2007 -->
<pre>Attention !!!. CER_SVXMON_HALT_RECOVER_RUN_ERROR !!!  
 Stuck Cellid S/B0/W0/L4/C7-13 .  
 AUTO HRR will be issued   
 
Fixed by an auto HRR</pre>
<!-- Author: Sasha :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244996' target=_top>244996</a>) -->




<!-- Date: Sat Jul 14 02:01:58.241 2007 -->
<pre>Parsing the run coordinator's elog -  
 
Current store to be dropped today  
between noon and 1 PM. </pre>
<!-- Author: greg -->



<!-- Date: Sat Jul 14 02:39:46.291 2007 -->
<br>Done TO. for crate/s: b0pcal05 
<br>Shepherded.
<!-- Author: Sasha :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244996' target=_top>244996</a>) -->



<!-- Date: Sat Jul 14 02:58:38.798 2007 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2007&action=view&page=-5816&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2007&action=view&page=-5816  width='300' align='top'></a></td><td>In CSP Occupancy two dead channel: Part BW number 16, Part NE a dead channel 15 
</td></tr></table>
<!-- Author: CO-Laura -->



<!-- Date: Sat Jul 14 03:19:54.714 2007 -->
<br>Done TO. for crate/s: b0htdc00 
<br>Fixed by an HRR.  
<!-- Author: Sasha :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244996' target=_top>244996</a>) -->




<!-- Date: Sat Jul 14 03:31:21.008 2007 -->
iFIx indicated alarms in CSP and BSU. No trigger inhibits are caused bu the alarms. HV summary looks fine.
<!-- Author: Sasha :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244996' target=_top>244996</a>) -->



<!-- Date: Sat Jul 14 03:48:22.076 2007 -->
<pre>Trigger Inhibit is set from the components: VME Power  		[147] VME Crate Power Supplies from iFix 
 
Crate 1RR21F_1. LV recovered on its own shortly after the failure.</pre>
<!-- Author: Sasha :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244996' target=_top>244996</a>) -->


<!-- Date: Sat Jul 14 03:55:09.638 2007 -->
TOF PM 350 tripped. HV was ramped up to normal.
<!-- Author: Sasha :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244996' target=_top>244996</a>) -->



<!-- Date: Sat Jul 14 04:20:23.261 2007 -->
ProcMon Error. "ENCP to tape"  fails check.
<!-- Author: Sasha :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244996' target=_top>244996</a>) -->


<!-- Date: Sat Jul 14 04:21:11.472 2007 -->
L3 Farm problem. c10 became pink and then gray shortly.
<!-- Author: Sasha :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244996' target=_top>244996</a>) -->
<!-- Comment: Sat Jul 14 08;38;45 comment by...Taka -->
When we began new RUN, It looked fine.
<!-- Date: Sat Jul 14 04:42:18.819 2007 -->
<br>Done Timeout detected in crates: CLC_00 
<br>Fixed by an HRR.
<!-- Author: Sasha :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244996' target=_top>244996</a>) -->
<!-- Comment: Sat Jul 14 05;57;08 comment by...Sasha -->
again. 
<!-- Date: Sat Jul 14 05:02:06.093 2007 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2007&action=view&page=-5817&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2007&action=view&page=-5817  width='300' align='top'></a></td><td>New dead tower in PHA Tower plot (eta=-11, phi=12)</td></tr></table>
<!-- Author: CO-Laura -->




<!-- Date: Sat Jul 14 05:44:03.616 2007 -->
<br>Done TO. for crate/s: b0pcal08 
<br>Shepherded.  
<!-- Author: Sasha :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244996' target=_top>244996</a>) -->







<!-- Date: Sat Jul 14 06:35:24.219 2007 -->
<pre>Trigger Inhibit is set from the components: 
 Flying Wire  	[095] Tevatron Flying Wires 
 SVX HV  	[128] Silicon Tracker High Voltage from iFix 
 ISL HV  	[129] Intermediate Silicon High Voltage from iFix 
 
Crate #2 was auto-Hockerized. HV was recovered after that. </pre>
<!-- Author: Sasha :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244996' target=_top>244996</a>) -->
<!-- Comment: Sat Jul 14 06;35;56 comment by...Sasha -->
<br>Trigger Inhibit is set from the components: SVX01 Trip  		<br>[001] Silicon Tracker Trip 01





<!-- Date: Sat Jul 14 07:02:01.433 2007 -->
<br>Done Timeout detected in crates: COT_06 
<br>Fixed by an auto HRR.
<!-- Author: Sasha :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244996' target=_top>244996</a>) -->


<!-- Date: Sat Jul 14 07:32:52.677 2007 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2007&action=view&page=-5818&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2007&action=view&page=-5818  width='300' align='top'></a></td><td>Is this plot something we should ignore? </td></tr></table>
<!-- Author: Laura and Greg -->
<!-- Comment: Sat Jul 14 07;42;30 comment by...Greg and Laura -->
<pre>all the other "ZZZ" test triggers (without 3DMATCH)
are firing.  Perhaps the "match" code is not yet in ...?</pre>

<!-- Comment: Sat Jul 14 11;12;19 comment by...anadi -->
The ZZZ triggers with 3Dmatch are the triggers for the XFT Stereo which should not fire since the XFT data go to the parasitic PC only.
<!-- Date: Sat Jul 14 07:52:04.266 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244996>244996</A>
 Terminated at 2007.07.14 07:51:20
<!-- Author: RunControl -->


<!-- Date: Sat Jul 14 07:55:12.991 2007 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>PhysicsTable</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L3Accepts</TH>
<TH>DelivLumi nb<SUP>-1</SUP></TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN=244996 target=_runSummary>244996</A> x3BD04</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_4_02 [8,916,694]</FONT></TD>
<TD ALIGN=CENTER>18:30:42</TD>
<TD ALIGN=CENTER>07:51:20</TD>
<TD ALIGN=CENTER>11:52:18</TD>
<TD ALIGN=RIGHT>   5,238,697</TD>   <TD ALIGN=RIGHT>  2,917.71</TD>
<TD ALIGN=RIGHT>  2,587.10</TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>07:55:00</TD>
<TD ALIGN=CENTER>11:52:18</TD>
<TD ALIGN=RIGHT>     5,238,697</TD> <TD ALIGN=RIGHT>  2,917.71</TD>
<TD ALIGN=RIGHT>  2,587.10</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Sat Jul 14 07:56:19.362 2007 -->
<font color=red>Shift Summary: </font><pre> 
A smooth quiet shift - we inherited run 244996, still going! 
 
A few minor problems 
 
- Tom Wright called in and said the prescaling was 
  not working quite right.  We re-started the  
  ScalerMon and it seems the prescaling started 
  working properly again 
 
- A fair number of L2 timeouts; only a minor  
  nuisance but there was no information as 
  to the cause in the ErrorLog. ==&gt; Time to 
  start adding some checks back into the L2 code ...? 
 
- Spontaneous shutoff from Si CAEN 2. Auto-hockerization 
  followed by Ace switching wedges back on. 
 
 
 
Totals 
Date:	2007.07.14 
Shift:	owl 
Delivered luminosity:	1420.7 nb-1 
Acquired luminosity:	1265.4 nb-1 
Efficiency:	89.1 
</pre> 
<!-- Author: greg  -->

<!-- Date: Sat Jul 14 08:06:07.666 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244997>244997</A>
 Activated at 2007.07.14 08:04:40
<!-- Author: RunControl -->


<!-- Date: Sat Jul 14 08:06:08.741 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244997>244997</A>
ACTIVATE: RunSet: AAA_CURRENT_L2UP  Trigger: PHYSICS_4_02 [8,916,694] 
<!-- Author: Alexander Paramonov -->


<!-- Date: Sat Jul 14 08:06:26.907 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244997>244997</A>
 Terminated at 2007.07.14 08:06:23
<!-- Author: RunControl -->


<!-- Date: Sat Jul 14 08:07:23.937 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244997>244997</A>
TERMINATE: L3 did not come up
<!-- Author: Alexander Paramonov -->
<!-- Comment: Sat Jul 14 08;39;58 comment by...Taka -->
Clean up EVB.
<!-- Date: Sat Jul 14 08:11:08.834 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244998>244998</A>
 Activated at 2007.07.14 08:10:02
<!-- Author: RunControl -->


<!-- Date: Sat Jul 14 08:11:09.703 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244998>244998</A>
ACTIVATE: RunSet: AAA_CURRENT_L2UP  Trigger: PHYSICS_4_02 [8,916,694] 
<!-- Author: Alexander Paramonov -->



<!-- Date: Sat Jul 14 08:11:53.753 2007 -->
DTO for crate b0clc00. AUTO HRR fixed it.
<!-- Author: Taka :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=244998' target=_top>244998</a>) -->
<!-- Comment: Sat Jul 14 08;15;55 comment by...Taka -->
again.

<!-- Comment: Sat Jul 14 08;17;51 comment by...Taka -->
again.

<!-- Comment: Sat Jul 14 08;19;23 comment by...Taka -->
again.

<!-- Comment: Sat Jul 14 08;21;10 comment by...Taka -->
again. shepherd and HRR.