<!-- Title: Day shift. Thu Dec 11, 2003 -->
<!-- SciCo: Vaia Papadimitriou -->
<!-- DAQAce: Dale Stentz -->
<!-- MonAce: Taka Maruyama -->
<!-- CO: Fodar Prokoshyn -->
<!-- OpManager: Mary Convery+Kaori -->
<!-- Notes: -->
<pre>Taking cosmics. 
Plug will close today. This will take about 3 hours. 
After closed we need to do several calibrations and test runs 
to verify all is fine. 
 
Run Control test and SVT test anticipated.</pre>


<!-- Date: Thu Dec 11 08:43:42.762 2003 -->
COT HV looks somehow going up. 
Guessing this is due to experts' working. 
<!-- Author: Taka -->
<!-- Comment: Thu Dec 11 08;44;53 comment by...Taka -->
<pre>COT is now 5% of full voltage (against nominal "ON") now.
</pre>


<!-- Comment: Thu Dec 11 08;47;23 comment by...Taka -->
40% of full, now. 
<!-- Date: Thu Dec 11 08:46:34.027 2003 -->
Paging CSL. Electrician needs to do some work and their  
system on 3rd floor has to be shut down.
<!-- Author: vaia -->


<!-- Date: Thu Dec 11 08:50:31.992 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176075">176075</A>
 Terminated at 2003.12.11 08:49:35
<!-- Author: RunControl -->


<!-- Date: Thu Dec 11 08:50:32.121 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176075">176075</A>
TERMINATE: ending the run for CSL
<!-- Author: Dale x2080 -->
<!-- Comment: Thu Dec 11 09;12;21 comment by...Dale -->
CSL has been successfully shutdown.
<!-- Date: Thu Dec 11 08:53:18.224 2003 -->
<pre>Added b0etdc00 to the VxMon display. 
</pre>
<!-- Author: W.Badgett -->


<!-- Date: Thu Dec 11 08:55:08.143 2003 -->
<pre>Working on AcnetMonitor again;   ignore ProcMon  
errors for AcnetMonitor and DownTime. 
</pre>
<!-- Author: W.Badgett -->


<!-- Date: Thu Dec 11 09:09:52.934 2003 -->
I have powered down the CSL disks per instructions from Tony so 
the electrician can do some work. The CSL computer is still up 
and this configuration will cause error messages in 
some log (says Tony - I did not get the full details). 
Tony will be here in about 30 minutes to bring system back 
up if electrician is finished.
<!-- Author: JJ -->


<!-- Date: Thu Dec 11 09:11:25.038 2003 -->
<pre>AcnetMonitor and DownTimeEntries have been  
moved to the new server node b0dap85. 
 
ProcMon has been updated. 
</pre>
<!-- Author: W.Badgett -->



<!-- Date: Thu Dec 11 09:19:12.316 2003 -->
<pre>Testing some new RunControl that addresses  
a particular problem with the rtserver. 
 
Have declared "fer devel" current temporarily, and  
rebooting crates. 
</pre>
<!-- Author: W.Badgett -->


<!-- Date: Thu Dec 11 09:21:51.687 2003 -->
PEM, PHA, PSH has global alarm. 
looks like experts' workings.
<!-- Author: Taka -->
<!-- Comment: Thu Dec 11 09;26;21 comment by...Dale and Taka -->
<pre>Lots of movement on the east side [where the alarm happed] - we are not aware of the procedure for when they move the plug but logical guess is just to turn it off.  Currently the plug is off.  We will bring back up once the plug has been offically closed up.</pre>


<!-- Date: Thu Dec 11 09:27:24.910 2003 -->
b0dap30 is out of connection. 
rebooted?
<!-- Author: Taka -->
<!-- Comment: Thu Dec 11 09;32;44 comment by...Taka -->
I can't test for SVT (SVDD beam position readout with latest 
hardware configuration) since I cannot login b0dap30 right now. 


<!-- Date: Thu Dec 11 09:33:44.178 2003 -->
<table border=0><td><tr><td bgcolor=#e0e030><pre> 
Dear ACEs  
 
     We have implemented two checks in TevMon (in coordination with silicon SPL's) 
 
TEVMON work: 
 
     * Avoid fake alarms when luminosity becomes negative (but  
       scraping signal is = 3) 
 
     * Detect possible drop in luminosity by more than 20% in  the    future  
 
 
</pre></td></tr></td></table>
<!-- Author: Mario -->


<!-- Date: Thu Dec 11 09:40:11.152 2003 -->
<pre>Vaia suggested that it might be better to have COT calibration 
after detector side working is over. 
</pre>
<!-- Author: Taka -->


<!-- Date: Thu Dec 11 09:45:15.197 2003 -->
<pre>The crew down in b0 are moving the SE CMX arch back into position (or closer to x =0 anyway)</pre>
<!-- Author: Dale -->
<!-- Comment: Thu Dec 11 09;47;02 comment by...Dale -->
NE arch also moving in.

<!-- Comment: Thu Dec 11 09;50;50 comment by...Dale -->
East arches are in.


<!-- Date: Thu Dec 11 09:55:39.459 2003 -->
Restarted CSL disks after a ~40 minute downtime for 
electrical work on the third floor. 
CSL is back online now.  
<!-- Author: Tony Vaiciulis -->


<!-- Date: Thu Dec 11 09:56:07.552 2003 -->
HV for the plug is on (PEM, PHA, PSH).  No trips on ramp-up.
<!-- Author: Dale -->


<!-- Date: Thu Dec 11 10:03:06.098 2003 -->
I've taken the XFT crates to do some more road file validation.  I'm working from my office.  Page the XFT if you need the crates back (I have the pager right now).
<!-- Author: K. Lannon -->


<!-- Date: Thu Dec 11 10:09:50.938 2003 -->
<pre>Kevin Lannon needs to work with the xft crates.  He will be in his office if we need to contact him or if we need the crates back.  We plan to do a quick series of calibrations (focus on COT and laser Plug calibrations etc.).</pre>
<!-- Author: Dale -->

<!-- Date: Thu Dec 11 10:11:44.336 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176081">176081</A>
ACTIVATE: Starting Calorimeter QIE calibration
<!-- Author: Dale x2080 -->


<!-- Date: Thu Dec 11 10:11:45.381 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176081">176081</A>
TERMINATE: ending Cal QIE calibration
<!-- Author: Dale x2080 -->
<!-- Comment: Thu Dec 11 10;23;11 comment by...Dale -->
<pre>Error messages during calibration (for experts)

b0pcal01:Messenger:10:10:00 AM-&gt;QIE calibration failed
b0pcal01:Readout_Calib:10:10:00 AM-&gt;ADMEM=19 QIE=5 Fail 10: No fitting          (0.0) (0) (0)</pre>
<!-- Comment: Thu Dec 11 12;13;16 comment by...m mattson and v tiwari -->
<br>This is a known dead channel, which does not have an easy fix.
<!-- Date: Thu Dec 11 10:15:01.376 2003 -->
<pre>Put back the old RunControl/FrontEndReadout code so Ace can take  
calibrations after the plug has moved  
in;  also CSL is back up. 
</pre>
<!-- Author: W,Vadgett -->

<!-- Date: Thu Dec 11 10:16:55.620 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176083">176083</A>
ACTIVATE: Startign COT calibration (HV at standby - no B field)
<!-- Author: Dale x2080 -->


<!-- Date: Thu Dec 11 10:16:56.222 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176083">176083</A>
TERMINATE: ending COT calibration
<!-- Author: Dale x2080 -->




<!-- Date: Thu Dec 11 10:26:37.860 2003 -->
<pre>Plug is closed now and CSL has been powered up by Tony. 
We did a COT calibration. Paged Dave Ambrose to make 
sure he does the necessary tests to figure out if COT is OK 
He will be her ein 30 minutes.</pre>
<!-- Author: vaia -->

<!-- Date: Thu Dec 11 10:27:27.635 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176085">176085</A>
ACTIVATE: test SVT run for SVDD beam position readout
<!-- Author: maruyama -->


<!-- Date: Thu Dec 11 10:27:28.723 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176084">176084</A>
 Activated at 2003.12.11 10:26:35
<!-- Author: RunControl -->


<!-- Date: Thu Dec 11 10:28:42.021 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176084">176084</A>
ACTIVATE: Starting Spike run.  L3 accept is at ~95 Hz.
<!-- Author: Dale x2080 -->


<!-- Date: Thu Dec 11 10:30:13.712 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176085">176085</A>
TERMINATE: end test SVT run (SVDD readout with new hardware configuration)
<!-- Author: maruyama -->


<!-- Date: Thu Dec 11 10:32:35.432 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176084">176084</A>
 Terminated at 2003.12.11 10:32:05
<!-- Author: RunControl -->


<!-- Date: Thu Dec 11 10:33:55.295 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176084">176084</A>
TERMINATE: Ending Spike run
<!-- Author: Dale x2080 -->



<!-- Date: Thu Dec 11 10:38:20.423 2003 -->
<pre>COT Calibration says it's OK: 
 
 
COT Calibration 10:14:51 on 2003-12-11 
aveStart:                        304.93    
total numBad:                    548     - o.k. 
   numBad  cells:                  6       
   numBad cables:                  2       
   numBad slots:                   0       
 
aveDrift:                       0.068     
max number drifted in crate:    93        
total numDrifted:              245  
   number of drifted  cells        1       
   number of drifted cables        1      
   number of drifted  slots        1       
 
 
This calibration is o.k. 
*** Timing has drifted, constants updated (i.e new COMPLETE file written to DB)*** 
 
 
Calibration passed quality checks, but there are Potential Bad Cells, Cables or Slots -  
See Expert Section 
  
Out of the last 1 calibrations, constants have been updated 1 times 
 
 
 
Expert Only: 
file: /cdf/people1/cdfdaq/consumers/log/runCotCtt_0_2003-12-11_10:14:51.log 
 
Drifted Cells: 
crate  6 slot 12 asdq  3:  9 
 
 
Drifted Cables: 
crate 19 slot  9 asdq  7 cable  3: 19 
 
 
Drifted Slots: 
crate  5 slot 16: 93 
 
 
Bad Cells: 
crate  2 slot  5 asdq  6: 12 
crate 11 slot  7 asdq  1: 10 
crate 11 slot 17 asdq  1:  9 
crate 16 slot 16 asdq  1: 10 
crate 17 slot  7 asdq  5:  9 
crate 19 slot  9 asdq  3: 10 
 
 
Bad Cables: 
crate  9 slot 17 asdq  1 cable  0: 24 
crate 10 slot 12 asdq  5 cable  2: 24 
 
 
 
 
 
 
</pre>
<!-- Author: Fiodar the CO :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176483' target=_top>176483</a>) -->

<!-- Date: Thu Dec 11 10:43:49.998 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176086">176086</A>
ACTIVATE: Starting Plug Laser calibration [BOTHPLUGS_NORMAL]
<!-- Author: Dale x2080 -->


<!-- Date: Thu Dec 11 10:43:50.850 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176086">176086</A>
TERMINATE: wrong enable state.
<!-- Author: Dale x2080 -->


<!-- Date: Thu Dec 11 10:44:01.527 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176087">176087</A>
ACTIVATE: test SVT run again
<!-- Author: maruyama -->


<!-- Date: Thu Dec 11 10:45:06.195 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176089">176089</A>
ACTIVATE: Starting Plug Laser [BOTH PLUGS_NORMAL] calibration
<!-- Author: Dale x2080 -->



<!-- Date: Thu Dec 11 10:45:06.271 2003 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2003&action=view&page=-13171&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-13171  width='300' align='top'></a></td><td>L1A Rate is 93.5 for spike run</td></tr></table>
<!-- Author: Fiodor the CO -->

<!-- Date: Thu Dec 11 10:45:28.798 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176087">176087</A>
TERMINATE: end SVT test run.
<!-- Author: maruyama -->


<!-- Date: Thu Dec 11 10:51:44.417 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176089">176089</A>
TERMINATE: ending [BOTHPLUGS_NORMAL] calibration
<!-- Author: Dale x2080 -->



<!-- Date: Thu Dec 11 10:55:04.537 2003 -->
<B>Calibrations are OK</B> <small> except that notorius BSC one that still has no reference plots updated</small>
<!-- Author: Fiodor the CO -->
<!-- Comment: Thu Dec 11 11;35;11 comment by...convery -->
references have been updated
<!-- Date: Thu Dec 11 10:56:29.978 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176091">176091</A>
ACTIVATE: Starting Laser Plug Calibration [BOTHPLUGS_MM_C]
<!-- Author: Dale x2080 -->



<!-- Date: Thu Dec 11 10:57:01.821 2003 -->
<br><b>Diode Dosimeters:</b> 
<br>  Tested connectivity of diode dosimeters.  Everything appears to be working correctly. 
<br>
<!-- Author: R.J. Tesarek -->

<!-- Date: Thu Dec 11 10:57:12.538 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176091">176091</A>
TERMINATE: Ending Laser Plug Calibration [BOTHPLUGS_MM_C]
<!-- Author: Dale x2080 -->


<!-- Date: Thu Dec 11 11:06:54.677 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176095">176095</A>
ACTIVATE: Starting Plug Laser Calibration [BOTHPLUGS_MM]
<!-- Author: Dale x2080 -->



<!-- Date: Thu Dec 11 11:07:36.156 2003 -->
<table><tr><td><a href=/elog/notebooks/2003/Thu_Dec_11_2003_Day.110724.ps align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-13173  align='top'></a></td><td>Calorimeter PMT spike rates over the last 20 months.   
Note the increase in the spike rate after the recent 
accelerator start up.</td></tr></table>
<!-- Author: R.J. Tesarek -->

<!-- Date: Thu Dec 11 11:09:04.004 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176095">176095</A>
TERMINATE: Ending Plug Laser [BOTHPLUGS_MM] calibration.
<!-- Author: Dale x2080 -->



<!-- Date: Thu Dec 11 11:12:27.646 2003 -->
<pre>Taking COT crates for checkout after plug move.</pre>
<!-- Author: ambrose -->


<!-- Date: Thu Dec 11 11:18:10.011 2003 -->
<pre>Dale want: 
 
(1) CES, CPR, CCR HVs are tuned off 
(2) plug is stand-by 
 
for SMX calibration. (and I did)</pre>
<!-- Author: Taka -->

<!-- Date: Thu Dec 11 11:22:24.668 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176096">176096</A>
ACTIVATE: Starting SMX QIE calibration
<!-- Author: Dale x2080 -->


<!-- Date: Thu Dec 11 11:23:44.823 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176096">176096</A>
TERMINATE: ending SM (shower max) QIE calibration
<!-- Author: Dale x2080 -->



<!-- Date: Thu Dec 11 11:25:04.424 2003 -->
<pre>Dake finished the calibration and we did turn on 
plug and shower max stuffs. 
</pre>
<!-- Author: Dale and Taka -->
<!-- Comment: Thu Dec 11 11;25;30 comment by...Taka -->
Dake -> Dale 


<!-- Date: Thu Dec 11 11:27:15.860 2003 -->
East side BLM's are back up and running.
<!-- Author: Eva H., Andy H. -->

<!-- Date: Thu Dec 11 11:28:48.359 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176097">176097</A>
 Activated at 2003.12.11 11:27:49
<!-- Author: RunControl -->



<!-- Date: Thu Dec 11 11:29:42.625 2003 -->
I finished SVT test (beam poition readout test with 
latest b0svt06 configuration) 
I checked this bank readout using offline and currently looks 
this test was succeeded. 
Marco will check the data of this test, and if it is OK, 
I will install latest hardware configuration.
<!-- Author: Taka -->


<!-- Date: Thu Dec 11 11:32:08.376 2003 -->
PSM alram: EndWall CONET_1; looks experts' work.
<!-- Author: Taka -->
<!-- Comment: Thu Dec 11 11;32;30 comment by...Taka -->
gone


<!-- Date: Thu Dec 11 11:33:54.726 2003 -->
Dervin called. COSET1 (COT crate ) is in a funny state. 
He wants to power cycle. 
Dave Ambrose had just left for the collision hall anyway, so I asked Dervin to talk to Dave.
<!-- Author: vaia -->


<!-- Date: Thu Dec 11 11:36:59.322 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176097">176097</A>
 Terminated at 2003.12.11 11:36:23
<!-- Author: RunControl -->



<!-- Date: Thu Dec 11 11:36:59.305 2003 -->
<pre>PROCMON complains about SVTSPYMON. 
I tried to re-start the this process, but somehow disk quota 
does not allow to restart the process. 
 
error in b0dap75 is: 
 
&gt;cdf_svt@b0dap75a/~/svtspy#45&gt; spy_manager 
&gt;Starting spy_manager: 
&gt;/cdf/code-common/cdfonline/svtmon/devel/bin/spy_manager_Linux.exe &gt;&! SPY_MANA\ 
GER_2003-12-11_11:12:25.log & 
&gt;[1] 32365 
&gt;cdf_svt@b0dap75a/~/svtspy#46&gt; SPY_MANAGER_2003-12-11_11:12:25.log: Disk quota \ 
exceeded. 
</pre>
<!-- Author: Taka -->
<!-- Comment: Thu Dec 11 11;40;31 comment by...Taka. -->
So, I disabled the PROCMON process for a while. <br>
<b> Please ignore SVTSPYMON process missing for a while </b> 


<!-- Date: Thu Dec 11 11:51:52.919 2003 -->
<pre>Process systems alarms for HVAC happened. 
But, it forced "silenced" very soon, we are sure that this 
is due to experts' work.</pre>
<!-- Author: Taka -->
<!-- Comment: Thu Dec 11 12;02;05 comment by...Dale -->
<pre>Relative Humdity (AT-RP1-RAH) which is around ~19% [low limit 20%].  Process Systems tech has silenced the alarm</pre>
<!-- Date: Thu Dec 11 12:00:03.992 2003 -->
CMX trip on SW wedges 16-17.  Recovered right away.
<!-- Author: Dale and Taka -->



<!-- Date: Thu Dec 11 12:20:26.223 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176098">176098</A>
ACTIVATE: Starting ACE_COSMCIS_ALL cosmic run
<p>
We have removed CMP00 and Inhibit00 for Steve V and Bill B. has wcal03.
We have included the silicon crates (fibs and SVX...).  
</p>
<!-- Author: Dale x2080 -->
<!-- Comment: Thu Dec 11 12;45;31 comment by...Taka and Dale -->
<pre>Seems that L3 manager is in the 'idle' state despite the fact that everything else made it to 'active' and we seem to be running very smoothly.  Looking at b0dap34 we see 

0 level3@b0dap34:~ 1&gt; ps -au level3
  PID TTY          TIME CMD
 4279 ?        00:00:00 parent_manager
21951 pts/0    00:00:00 tcsh
22401 pts/0    00:00:00 less
30238 ?        00:00:00 l3manager
30803 pts/2    00:00:00 tcsh
30823 pts/2    00:00:00 ps

So it appears the parent_manager and l3manager are there.  No reason to stop.  Seems to be harmless.

We tried a HRR - now it is active.  Problem solved for now.</pre>
<!-- Date: Thu Dec 11 12:35:51.555 2003 -->
There is only one person in control room.  
<!-- Author: Taka -->
<!-- Comment: Thu Dec 11 12;38;24 comment by...Fiodar -->
Now at least three



<!-- Comment: Thu Dec 11 13;56;13 comment by...Fiodor -->
Now its almost 6 and still growing!
<!-- Date: Thu Dec 11 12:58:19.420 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-13174&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-13174' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-13175&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-13175' width='300'></a></center>
<!-- Author: Dale -->
<!-- Comment: Thu Dec 11 13;12;25 comment by...Dale  -->
<pre>Trigmon warrings.  L2 expert was here during the spike run and the general consences is that the occurance of errors (for example data vs. sim) are acceptable.  She noted that the L1 (= L2) acceptance rate was higher then normal (~50 Hz verse an expected 30 Hz).  As the run progresses we will double check the trigmon slides and make sure that a L2 expert relooks at the data.</pre>
<!-- Comment: Fri Dec 12 14;24;52 comment by...GregFeild -->
<pre>The XTRP crate was not in this run so it will send
no data to L2.

</pre>
<!-- Date: Thu Dec 11 12:59:22.559 2003 -->
Silicon experts often come here, so this may be alredy known. 
One new pink guy is in Imon now. 
B1W1L4 AVDD. 
 
We have 3 other pinkies, but this is same as  
 <a href= 
'/cgi/elog/elog.pl?nb=2003&action=view&page=1032&frame=2&anchor=01:19:14.139'> 
Previous shift. </a>  
 
One L00 pink was gone right now.   
<!-- Author: Taka -->


<!-- Date: Thu Dec 11 13:01:29.376 2003 -->
PSM EndWall COSET_1 was alarmed. (Now back)
<!-- Author: Taka -->
<!-- Comment: Thu Dec 11 13;03;03 comment by...Taka -->
<pre>Alarm for same guy</pre>
<!-- Date: Thu Dec 11 13:02:51.903 2003 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2003&action=view&page=-13176&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-13176  width='300' align='top'></a></td><td>Small number of yellows in the SVXMon L00 and ISL chip status slide.  Silicon expert happened to be come by (see Taka entry above) and looked at it and said it was ok for now.</td></tr></table>
<!-- Author: Dale -->



<!-- Date: Thu Dec 11 13:08:41.473 2003 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2003&action=view&page=-13178&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-13178  width='300' align='top'></a></td><td>New pink  guy's history. But it looks like always pink. 
I may be wrong.</td></tr></table>
<!-- Author: Taka -->


<!-- Date: Thu Dec 11 13:09:52.291 2003 -->
<pre>Icicle was died. 
I restart the process. 
</pre>
<!-- Author: Taka -->


<!-- Date: Thu Dec 11 13:11:48.270 2003 -->
VxWorks monitor shows b0xft01 and b0cot13 is red state. 
xft is due to expert. cot is also??  
<!-- Author: Dale and Taka -->
<!-- Comment: Thu Dec 11 13;12;23 comment by...Dale and Taka -->
<pre>Cleared the problem.</pre>



<!-- Comment: Thu Dec 11 13;14;57 comment by...Dale -->
<pre>Lets claify:  Who has what?

David A. = cot** crates
Kevin L. = xft** crates
Bill B.  = wcal03
Steve V. = Inhbit00 and CMP00</pre>
<!-- Date: Thu Dec 11 13:14:17.694 2003 -->
PSM alarm was gone.
<!-- Author: Taka -->



<!-- Date: Thu Dec 11 13:20:21.383 2003 -->
<pre>Trigger rate is ~50Hz. 
Maria wonder if this rate is a little bit higher than that of normal in cosmic with no track.</pre>
<!-- Author: Dale and Taka -->


<!-- Date: Thu Dec 11 13:20:52.298 2003 -->
Icicle died again 
re-started again.
<!-- Author: Taka -->


<!-- Date: Thu Dec 11 13:25:01.893 2003 -->
BIGBROTHER complains that: 
 
red Thu Dec 11 13:02:23 CST 2003 ERROR: Can't connect to 131.225.236.162 
 
We have same problem from yesterday. This is not crucial?
<!-- Author: Dale and Taka -->
<!-- Comment: Thu Dec 11 13;35;02 comment by...Taka and Dale -->
<pre>for b0dap52 - it says "current acknowledgement by root@localhost".  

In this context: current = status (we think)

Thus it must mean that this is known about.  Certainly this is the case.  b0dap52 is/was the pc for the daq ace for elogs etc.  Thus this is expected.  Perhaps remove dap52 from BB?</pre>

<!-- Date: Thu Dec 11 13:50:30.926 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176098">176098</A>
TERMINATE: Ending cosmic run to do L2 torture
<!-- Author: Dale x2080 -->
<!-- Comment: Thu Dec 11 13;55;08 comment by...Dale -->
XTRP experts are here and need to do some L2 torture related tests.


<!-- Date: Thu Dec 11 13:56:35.173 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176106">176106</A>
ACTIVATE: Starting L2 torture
<p>
Without CMP00, XFT**, COT**, wcal03.  A 'control run' for the XTRP experts.
</p>
<!-- Author: Dale x2080 -->


<!-- Date: Thu Dec 11 13:57:20.217 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176106">176106</A>
TERMINATE: ending L2 torture - above 25k events
<!-- Author: Dale x2080 -->



<!-- Date: Thu Dec 11 13:59:39.623 2003 -->
PSM 1st floor: 1RR18F_1. 
Maybe XTRP swapping?
<!-- Author: Taka -->
<!-- Comment: Thu Dec 11 14;01;53 comment by...Taka -->
<pre>gone very soon.</pre>

<!-- Date: Thu Dec 11 14:07:45.649 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176107">176107</A>
ACTIVATE: Starting L2 torture (after XTRP 2 track trigger board swap)
<!-- Author: Dale x2080 -->



<!-- Date: Thu Dec 11 14:07:49.566 2003 -->
Dee is disabling VESDA because Mark will do shuttering in the pit.
<!-- Author: vaia -->
<!-- Comment: Thu Dec 11 15;04;48 comment by...convery -->
Mark and Vivek are done soldering.  I reenabled VESDA.
<!-- Date: Thu Dec 11 14:09:06.034 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-13181&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-13181' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-13180&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-13180' width='300'></a></center>
<!-- Author: Fiodor the CO -->
<!-- Comment: Thu Dec 11 14;09;53 comment by...Fiodor -->
<table border=0><td><tr><td bgcolor=#e0e030>New Dead Block for WHA. Who did this?</td></tr></td></table>
<!-- Comment: Thu Dec 11 14;20;13 comment by...Fotis -->
WCAL03 crate was not included in this run. 
<!-- Date: Thu Dec 11 14:09:10.596 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176107">176107</A>
TERMINATE: ending L2 torture XTRP experts are done.
<!-- Author: Dale x2080 -->
<!-- Comment: Thu Dec 11 14;18;21 comment by...Dale -->
(XTRP board switched back - see below)
<!-- Date: Thu Dec 11 14:16:43.487 2003 -->
Run 176106 was a "reference" L2 torture run using standard Two Track Trigger board #001. 
Run 176107 was a L2 torture run taken with Two Trigger board #003.  This board has modified termination of backplane signals associated with L1A.  We are testing to see if this improved termination removes the low level readout errors seen with XTRP.  After 176107, we put the XTRP crate back into its original configuration with Two Track trigger board #001.  We will look at the torture data offline to see if there are any readout errors. 
<!-- Author: KPitts/JKraus -->


<!-- Date: Thu Dec 11 14:17:30.950 2003 -->
<pre>RTserver "went away" for a bit.  Came back no problems / lost connections.  We follow the instructions via the note from Frank: 
 
b0dap33:111:~&gt; netstat |grep 5101 | wc 
    219    1314   17739 
b0dap33:112:~&gt; netstat | grep FIN 
tcp        0  76745 b0dap33.fnal.gov:5101   b0dap30.fnal.gov:33371  FIN_WAIT1</pre>
<!-- Author: Dale -->
<!-- Comment: Thu Dec 11 14;18;39 comment by...Guillelmo -->
The limit in the number of licences is 220, so it's a very close value...



<!-- Date: Thu Dec 11 14:22:27.007 2003 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2003&action=view&page=-13182&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-13182  width='300' align='top'></a></td><td>New hot channel at CSX IE</td></tr></table>
<!-- Author: Fiodar the CO -->
<!-- Comment: Thu Dec 11 15;36;48 comment by...Dale -->
<p>
As Fiodar and other CS* experts will tell you the new hot channel is at the end/bottom of the NE arch.  This makes it an 'easy target'.  Several people are looking at the damage.  It appears that the side has been hit hard by the ladder (or purhaps the other way).  Thus there is a light leak.  It may have happend <a href=
'/cgi/elog/elog.pl?nb=2003&action=view&page=1033&frame=2&anchor=09:45:15.197'>
when the arches were being moved in</a>. Experts are looking into a solution. </p>
<!-- Comment: Thu Dec 11 15;43;53 comment by...Dale -->
<pre>Experts have already tried to tape the light leak.  We are looking at the effect via run 176112.</pre>
<!-- Date: Thu Dec 11 14:28:57.708 2003 -->
<pre>A couple of PSM alarms - all are due to known people working on crates on the 1st or 2nd floor (thanks to SVT expert Taka for telling us in advance).</pre>
<!-- Author: Dale -->


<!-- Date: Thu Dec 11 14:33:30.886 2003 -->
<b>COT summary -</b> 
 
After the plug went in, the micro-coax for SL5 cells 4-5 might have been pinched or otherwise damaged.  The repeater for these 
cells is behind the NE plug rack, which Dervin pulled out for now.  We won't be able to fix this problem until the next time the plug is pulled.  Expect these cells to be noisy in YMon and calibrations.  Returning COT crates to Ace.
<!-- Author: wagner, binkley, ambrose -->

<!-- Date: Thu Dec 11 14:46:17.169 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176109">176109</A>
ACTIVATE: Starting ACE_COSMICS_NOTRACKS 
<p>
<b>Using new TT: COSMICS_NOTRACKS[6,353,409]</b><br>
previous L3 tag was 403
</p>
<p>
Removed these crates (iff they were included):
xft** (Kevin), svt (Taka), wcal03 (Bill), cmp00 (Steve)
</p>

<!-- Author: Dale x2080 -->
<!-- Comment: Thu Dec 11 14;55;26 comment by...Dale -->
<pre>Vivek and Mark told us to expect that CEM 16 west should have low occupancy as they will be working on that transition card area.</pre>


<!-- Comment: Thu Dec 11 15;33;11 comment by...kaori -->
This run is using a new Trigger table.  L3 is using 5.1.0.  It was
a short run but L3 managed to put through events ok.  Only
odd thing I observed with the first look is SVXMon and YMon
crashed.  (SVXMon and YMon have been running well with
the default cosmic runs).  Don't know why they crashed yet.
Next, we'll put Silicon in DAQ (OK-ed with Silicon person (Jason)), and take another run.
<!-- Comment: Thu Dec 11 16;31;26 comment by...kaori -->
YMon crash is at "ConsumerErrorSender.cc:168"
ConsumerErrorSender::close.  Probably unrelated to the new trigger table.
<!-- Comment: Thu Dec 11 19;55;13 comment by...kaori -->
Later, we have run many runs with 409 tag, and observed no
crashes in YMon and SVXMon.  (silicon included).  
<!-- Date: Thu Dec 11 15:02:09.439 2003 -->
<pre>Who was working in the 1RR27D crate (has east CMX and CLC)?  It seems that some one switched the power off and left it that way.  It was easy enough to figure (this happend before we were able to start 176109).</pre>
<!-- Author: Dale -->

<!-- Date: Thu Dec 11 15:03:24.217 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176109">176109</A>
TERMINATE: Ending run to include silicon
<!-- Author: Dale x2080 -->
<!-- Comment: Thu Dec 11 15;04;41 comment by...Dale -->
L1 and L2 accept rate was 46 Hz with L3 at 93 Hz.
<!-- Date: Thu Dec 11 15:04:22.660 2003 -->
Last week, the zener diode for ccal05, slot 18, ch 10 was snipped out (of the transition board).  We have soldered a new one back in.  We have informed Mary (ops manager) that we are done, and she has re-enabled VESDA.
<!-- Author: m mattson, v tiwari -->




<!-- Date: Thu Dec 11 15:12:21.832 2003 -->
I changed the COT bits from "bad" to "good" for runs 175066 
175078, 175075, 175079. 
175075 still looks bad because of the L2 being set to 0 
I suppose.
<!-- Author: vaia -->


<!-- Date: Thu Dec 11 15:17:01.730 2003 -->
<pre>Paged COT. 
A CSX counter is broken in NE, area where COT was working. 
Mike L., Fiodar, Davit, Hyunsoo, Dave are looking into the problem. 
There was a ladder close to the damaged counter.Perhaps the  
ladder was pushed there because of the closing of the arch?</pre>
<!-- Author: vaia -->
<!-- Comment: Thu Dec 11 17;39;30 comment by...Fiodar -->
Light isolation was damnaged in few places on the counter light guide, causing plastics be exposed. this was done by ladder, that was contacting counter in the place of damege.

Holes were finally taped and now counting rate is close to old good times
<!-- Date: Thu Dec 11 15:22:03.226 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176111">176111</A>
ACTIVATE: COSMIC NOTRACK[6,353,<b>409</b>] with Silicon to check the SVXMON.
<!-- Author: Dale x2080 -->


<!-- Date: Thu Dec 11 15:29:11.485 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176111">176111</A>
TERMINATE: ending to give CLC00 to Alex Pronko
<!-- Author: Dale x2080 -->




<!-- Date: Thu Dec 11 15:36:21.973 2003 -->
ProcMon error for SMARTSOCKETS RESPONSE. 
Looks like many people uses run control.
<!-- Author: Taka -->
<!-- Comment: Thu Dec 11 15;46;17 comment by...Dale and Taka -->
0-6 (7/8!!) paritions.
<!-- Date: Thu Dec 11 15:39:12.379 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176112">176112</A>
ACTIVATE: Starting Cosmic Run 
<p>
With the new TT COSMICS_NOTRACKS[6,353,409] and silicon but without CLC 
(the previous run without CLC00 and CLC01). 
</p>
<!-- Author: Dale x2080 -->





<!-- Date: Thu Dec 11 15:50:29.721 2003 -->
We have red status for b0xtrp00 in VxWorks System Monitor. 
We don't know this is OK or not. 
Are experts still working for this one? 
  
<!-- Author: Taka -->

<!-- Date: Thu Dec 11 15:51:35.114 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176112">176112</A>
TERMINATE: Ending to run to give Bill B. the remaining muon crates (except musc00 = muon counters)
<!-- Author: Dale x2080 -->



<!-- Date: Thu Dec 11 15:51:38.541 2003 -->
<pre>Testing new RunControl code on the muon and calorimeter  
timing crates. 
 
</pre>
<!-- Author: W.Badgett -->

<!-- Date: Thu Dec 11 15:55:34.373 2003 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>15:55:02</TD>                                                       
<TD ALIGN=CENTER>::</TD>                                                
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Thu Dec 11 15:55:58.026 2003 -->
Icicle died. I restarted. 
On this shift, we have 3 failures for Icicle. 
It might be better to tell this one to expert.  
<!-- Author: Taka -->


<!-- Date: Thu Dec 11 15:58:54.119 2003 -->
SVT crates are now available for main run control. 
This was used for the validation of Track Fitter  
which we install firmware for wedge00. 
 
<!-- Author: Taka -->

<!-- Date: Thu Dec 11 16:05:38.552 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176118">176118</A>
ACTIVATE: Starting COSMICS_NOTRACKS[6,353,<b>409</b>]
<p>
With CLC** and WCAL03.  With IMU**, CMU**, CMX**, 
MUCamac00 [Bill B. has these now] removed along with CMP00 
(Steve should have this still)
</p>
<!-- Author: Dale x2080 -->



<!-- Date: Thu Dec 11 16:07:53.908 2003 -->
<font color=red>Shift Summary: </font><pre>Plan: 
------ 
 
Be ready to do all calibrations, spike runs etc. 
after plug closes to make sure we have no damage. 
In th emean time have some tests done with Run Control, SVT, etc. 
 
  
 
 
Plug was closed by 10:15 am 
 
We did COT calibration (some bad cell/cable messages) 
(Dave identified a bad cable) 
Did couple of spike runs 
Plug laser calibration 
QIE calibration 
SMX QIE calibration 
 
Tevmon improvements 
10:40 
We are giving the inhibit crate and one CMP crate 
for trigger inhibit work. 
 
 
 
 
A CSX counter in NE corner is broken. 
Investigating how this happened. 
 
They taped it. The channel was too noisy. 
Now the rate is high but improved significantly. 
 
Cosmics in progress 
 
People here to fix plug sourcing hardware</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   0
Delivered Luminosity   0  
Acquired Luminosity    0  
Efficiency             NA
   
</td></tr></table></pre>
<!-- Author: vaia -->


<!-- Date: Thu Dec 11 16:10:12.422 2003 -->
<pre>inhibit work: 
  undefined channels masked off. 
  partitioned resources improved but needs more work. 
  inhibit crate recognizes which partition it is in. 
Hopefully more work tomorrow.</pre>
<!-- Author: steve -->