<!-- Title: Eve shift. Mon May 21, 2001 -->
<!-- SciCo: D. Amidei -->
<!-- DAQAce: F. Palmonari -->
<!-- MonAce: G. Veramendi -->
<!-- CO: A. Roy -->
<!-- OpManager: K. Burkett -->
<!-- Notes: -->
<pre>We will pull together the chaos</pre>


<!-- Date: Mon May 21 16:33:46 2001 -->
<pre>I have repaired several of the "minicom" lines to detector crates.  All were problems with the connectors at the patch panel.  The following connections that were reported broken on 5/11 were made to work essentially by unplugging and replugging cables:  ccal01, ccal03, cot01, cot09.  pcal09 was fixed by replacing the RJ-11 jumper (phone cable).  wcal00 works if I use an RJ45 union and a new patch cable.  However, since that configuration does not have the break in the ground, I put the cables back in the default configuration.  There is clearly a problem with the construction of this system.  I'm trying to come up with a long-term solution.</pre>
<!-- Author: Jonathan Lewis -->


<!-- Date: Mon May 21 17:09:02 2001 -->
Reinhard Eckmann working on Had TDC; 
Wilson and Feild working on FRED bit bug, using CCAL;  
Badgett working on RC memory leaks; 
Tesarek and Veramendi tracking down hot tower; 
Burkett making new plan for 8 day access
<!-- Author: D. Amidei -->


<!-- Date: Mon May 21 17:23:53 2001 -->
<br><b>Hot Tower:</b> 
<br>EM trigger tower eta=23,phi=2 found running hot (1.7MHz). 
DCAS reads 0x020.  Tower masked until the problem can be isolated. 
<br> 
<i>Note:  This tower appeared only after the plug closed:</i>
<!-- Author: R.J. Tesarek -->


<!-- Date: Mon May 21 17:34:51 2001 -->
<pre>Found & fixed a pseudo-memory-leak in the resource manager GUIs: 
resusr.PartitionSelector and resusr.ResourceSelector 
 
Have applied a quick fix, but this causes problems with  
the stand-alone Resource Manager GUI. 
</pre>
<!-- Author: W.Badgett -->


<!-- Date: Mon May 21 17:43:47 2001 -->
<pre>Greg updated the firmware for FRED to make single object versions 
of DIRAC diobject triggers work correctly.  We then took several 
runs with TEST_DIRAC_DIJET trigger table.  This table has a 2 Jet 
trigger (ET>2) and a single jet trigger (ET>2).  In previous runs with this table the single jet trigger only ifred when there were 1, 3 or more than 3 towers with Et>2 but not when there were 
exactly 2 towers above threshold.  With the new firmware, the 
single jet trigger fires when there are 1 or more towers above 
threshold.</pre>
<!-- Author: Greg and Peter :: (run <a href='/cgi-bin/runSummaryRun.cgi?115448' target=_top>115448</a>) -->


<!-- Date: Mon May 21 18:10:43 2001 -->
<pre>Completed memory leak studies.   Returned main Run Control  
to normal operation.</pre>
<!-- Author: W.Badgett -->


<!-- Date: Mon May 21 18:56:27 2001 -->
<pre>run 115455: cosmic run 
 
>> table cosmic[3,120] 
 
observed standard prescaled rates: 
L1_TWO_JET0.5		~430 Hz 
L1_TWO_CMU1.5_DPHI120	~181 Hz 
L1_TWO_JET1.0		~101 Hz 
ZERO_BIAS		~1.7 Hz</pre>
<!-- Author: francesco :: (run <a href='/cgi-bin/runSummaryRun.cgi?115455' target=_top>115455</a>) -->


<!-- Date: Mon May 21 19:07:20 2001 -->
<pre>run 115456: 
 
Rick run (ACE_CALORIMETER_MINIMAL) for Photis to  
test the PMT with lasers ON for WHA and CHA 
 
-> end after 92755 events</pre>
<!-- Author: francesco :: (run <a href='/cgi-bin/runSummaryRun.cgi?115456' target=_top>115456</a>) -->


<!-- Date: Mon May 21 19:31:39 2001 -->
PLUG HV back on
<!-- Author: francesco -->
<!-- Comment: Mon May 21 19;40;51 comment by...Willis -->
<pre>The plug HV was off because of the AC power trip when the
plugs were being closed. I turned the HV on for all plug
detectors.</pre>


<!-- Date: Mon May 21 19:57:32 2001 -->
ToF calibrations: DDG scans with various time steps and ranges 
for TAC calibration.  
<!-- Author: Gervasio Gomez -->


<!-- Date: Mon May 21 20:06:28 2001 -->
run 115459:  
 
trig type: cosmic[3,120] 
 
notes: 
- wcal02 & 06 to TOF guys 
- SVT to Marco 
- L2 included 
 
rates are still good (see before)
<!-- Author: francesco :: (run <a href='/cgi-bin/runSummaryRun.cgi?115459' target=_top>115459</a>) -->


<!-- Date: Mon May 21 20:23:50 2001 -->
Trigmon: plot for DIRAC EM Trigger occupancy is empty. 
Page L1 calor.  
<!-- Author: Amidei -->


<!-- Date: Mon May 21 20:36:59 2001 -->
Doing some new tests of TDC spy mode readout, to establish 
(again) possible error conditions. STILL a mystery...  
<!-- Author: Arnd :: (run <a href='/cgi-bin/runSummaryRun.cgi?115460' target=_top>115460</a>) -->


<!-- Date: Mon May 21 20:42:10 2001 -->
<pre>Slots 18+19 in cot13 show as empty, don't come up on crate 
reset. 
</pre>
<!-- Author: Arnd -->
<!-- Comment: Mon May 21 20;45;50 comment by...f -->
<pre>>> expert paged </pre>


<!-- Comment: Mon May 21 20;58;34 comment by...Arnd -->
<pre>I changed the tdc readout mode in the hardware DB for that crate
to 'local DONE'.
</pre>
<!-- Comment: Mon May 21 21;44;56 comment by...francesco -->
<pre>>>>> status of b0cot13:

47 /cdf/people1/dittmann/bin % /cdf/people1/dittmann/bin/cratemap_linux b0cot13
[b0cot13]
 b0cot13  2   IDPROM = 0078 001 TRACER__V2.1A
 b0cot13  3   (empty)
 b0cot13  4   (empty)
 b0cot13  5 S IDPROM = 0708 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13  6 S IDPROM = 0314 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13  7 A IDPROM = 0695 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13  8 A IDPROM = 0732 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13  9 A IDPROM = 0510 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13 10 S IDPROM = 0608 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13 11 S IDPROM = 0679 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13 12 S IDPROM = 0175 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13 13 S IDPROM = 0153 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13 14 S IDPROM = 0151 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13 15 A IDPROM = 0736 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13 16 A IDPROM = 0718 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13 17 A IDPROM = 0500 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13 18 A (empty)
 b0cot13 19 A (empty)
 b0cot13 20   (empty)
 b0cot13 21   (empty)                                                           </pre>
<!-- Comment: Mon May 21 21;51;42 comment by...fra. -->
<pre>b0dap59:26:~>reset_crate b0cot13
Crate b0cot13 reset channel is 61
Crate reset complete


....BUT STILL:

49 /cdf/people1/dittmann/bin % /cdf/people1/dittmann/bin/cratemap_linux b0cot13
[b0cot13]
 b0cot13  2   IDPROM = 0078 001 TRACER__V2.1A
 b0cot13  3   (empty)
 b0cot13  4   (empty)
 b0cot13  5 S IDPROM = 0708 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13  6 S IDPROM = 0314 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13  7 A IDPROM = 0695 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13  8 A IDPROM = 0732 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13  9 A IDPROM = 0510 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13 10 S IDPROM = 0608 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13 11 S IDPROM = 0679 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13 12 S IDPROM = 0175 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13 13 S IDPROM = 0153 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13 14 S IDPROM = 0151 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13 15 A IDPROM = 0736 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13 16 A IDPROM = 0718 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13 17 A IDPROM = 0500 009 MICHIGAN TDC LVDS Rev D    DSP =   V 035
 b0cot13 18 A (empty)
 b0cot13 19 A (empty)
 b0cot13 20   (empty)
 b0cot13 21   (empty)                                                           </pre>
<!-- Date: Mon May 21 20:56:07 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-991&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-991  width='300' align='top'></td><td>Looks like a dead channel in CEM, eta =-5,, Phi = 16, Left PMT.</a></td></tr></table>
<!-- Author: Amit -->



<!-- Date: Mon May 21 21:10:36 2001 -->
<pre>PHA bird.. crate 2 wedge 5</pre>
<!-- Author: greg -->


<!-- Date: Mon May 21 21:14:42 2001 -->
<pre>summary of the run's taken with Arnd: 
 
from 115460 to 115464 
 
NOTES:  in the last run one converter node died after we had 
 	to HRR after b0ccal14 gave this error: 
(MLE) b0ccal14:Messenger:9:03:47 PM->Event 35289: Bunchcounters in slot 17 and slot 18 disagree (at least) 
(MLE) b0ccal14:Messenger:9:03:47 PM->Runtime Error 1, Event 35289: Bunch counter mismatch, mismatch count = 1                                                   </pre>
<!-- Author: francesco :: (run <a href='/cgi-bin/runSummaryRun.cgi?from 115460 to 115464' target=_top>from 115460 to 115464</a>) -->


<!-- Date: Mon May 21 21:41:59 2001 -->
CMX HV in a strange state. A. Gallas responds to page and is investigating...
<!-- Author: Amidei -->



<!-- Date: Mon May 21 21:50:16 2001 -->
<pre>run 115466: This is a "Monica Run" to check L2 clustering
 
trig. table: INTEGRATION_SUMET_2[2,120] 
 
ended after 113846 events 
 
SUM_ET_5[1] average rate prescaled 172 Hz</pre>
<!-- Author: francesco :: (run <a href='/cgi-bin/runSummaryRun.cgi?115466' target=_top>115466</a>) -->



<!-- Date: Mon May 21 21:57:27 2001 -->
<pre>CMX HV was in Standby for half of the West ring, cause unknown. A. Gallas resets, all OK except known problem in 16,17 E.</pre>
<!-- Author: Amidei -->


<!-- Date: Mon May 21 22:04:21 2001 -->
<pre>run 115468: Another Monica Run
 
as 115466 (see before) taken 21740 events</pre>
<!-- Author: francesco :: (run <a href='/cgi-bin/runSummaryRun.cgi?115468' target=_top>115468</a>) -->


<!-- Date: Mon May 21 22:26:21 2001 -->
<pre>run 115469: cosmic run  
 
&gt;&gt; table cosmics[3,125]  
 
measured average prescaled rates:  
 
L1_TWO_JET0.5           ~460 Hz  
L1_TWO_CMU1.5_DPHI120   ~219 Hz  
L1_TWO_JET1.0           ~103 Hz  
L1_MB_XING_PS1M         ~1.7 Hz 
 
end after 113776 events</pre>
<!-- Author: francesco :: (run <a href='/cgi-bin/runSummaryRun.cgi?115469' target=_top>115469</a>) -->
<!-- Comment: Mon May 21 22;40;10 comment by...francesco -->
<pre>NB: no b0cot13 in this run (just because i've forgot
    to include it)</pre>


<!-- Date: Mon May 21 22:45:11 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-992&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-992  width='300' align='top'></td><td>Looks LikeRinging Problem on CMP Part South , Stack 20, Layer 1, and Part North , stack 25 layer 0.</a></td></tr></table>
<!-- Author: Amit -->


<!-- Date: Mon May 21 23:35:38 2001 -->
<pre>run 115471: cosmic run   
 
&gt;&gt; table cosmics[3,123]   
 
measured average prescaled rates:   
 
L1_TWO_JET0.5           ~450 Hz   
L1_TWO_CMU1.5_DPHI120   ~200 Hz   
L1_TWO_JET1.0           ~100 Hz   
L1_MB_XING_PS1M         ~1.7 Hz  
 
end after events ~750 K 
 
NB: b0cot13 is back in this run  
&gt;&gt;&gt; BUT SLOTS 18 AND 19 HAVE BEEN REMOVED 
 
&gt;&gt; event 24538: b0cot18 runtime error: data truncated 
 
&gt;&gt; L3 error: 
(MLE) b0l3gate2.fnal.gov:main:10:43:49 PM-&gt;Error on L3 node b0l3c01 (partition 0) Mon May 21 22:43:47 2001 l3_node 
  in chain_loop (l3_chain.c:599) 
  @L3_INPUT_FAILURE:  chain 1 code Success (4205 good, 1 bad)                    
 
 
&gt;&gt; ... and more ! check the log file</pre>
<!-- Author: francesco :: (run <a href='/cgi-bin/runSummaryRun.cgi?115471' target=_top>115471</a>) -->


<!-- Date: Mon May 21 23:58:43 2001 -->
<font color=red>Shift Summary: </font><pre>shift summary for Mon eve 
  
  
* JDL repairs several minicom lines 
* EM eta = 23 phi = 2 hot, masked 
* Badgett fixes some memory leaks in RM GUIs 
* update Fred firmware to fix binomial bug 
* plug HV turned on, CMX HV on and OK 
* Cosmic Run 115459: 
  TRIGMON anomalies: 
    a dead channel in CEM, eta =-5,, Phi = 16, Left 
    DIRAC EM occupancy = 0?? (Experts notified) 
* a few "Monica" runs to continue debugging DCAS 
* Arnd works on TDC spy 
* Problem with slots 18,19 in B0cot13 
* End with another long cosmic run 115471 
  using cosmics_1(3,123) as instructed                                                                                  </pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   from 115455 to 115471
Delivered Luminosity   \|||/  pb-1
Acquired Luminosity     O O  pb-1
Efficiency              ---
   
</td></tr></table></pre>
<!-- Author: Dante Amidei -->