<!-- Title: Day shift. Thu Oct 30, 2003 -->
<!-- SciCo: Brig Williams -->
<!-- DAQAce: Anant Gajjar -->
<!-- MonAce: &nbsp; -->
<!-- CO: &nbsp; -->
<!-- OpManager: &nbsp; -->
<!-- Notes: -->
<pre>First of all, it took 3 minutes for me to make this shift entry. 
 
Stop run, release L3 because of Electrical work on 3rd floor. 
 
Frank C. and Tom P. debugging various COT readout issues. 
 
There is some air in the portcard cooling (from rupture 
of ISL line yesteday) + a new leak in an SVX cooling line.</pre>


<!-- Date: Thu Oct 30 08:58:55.063 2003 -->
L3 back.
<!-- Author: Brig -->


<!-- Date: Thu Oct 30 09:05:06.399 2003 -->
Steve at Cryo calls. They will be working on CDFS4 (controls 
solenoid) most of the shift. We will probably see some alarms.  
<!-- Author: Brig -->


<!-- Date: Thu Oct 30 09:33:22.397 2003 -->
<pre>CMX experts have switch the HV cables for wedges 18-19 and 20 for CMX NE miniskirts.  This means that HV channel  1 --> 7 and vice versa.  Turning on HV.  
 
Please page if something trips.  (CMX pager = 266-2712)</pre>
<!-- Author: Ingyin Zaw, John Strologas, Dale Stentz -->


<!-- Date: Thu Oct 30 09:46:16.827 2003 -->
<pre>I am trying to restart the calib consumer proxy but had problems logging into b0dap62 from b0dap52. 
 
I did a kticket and the ssh b0dap62 -l cdfcalib 
however it still asked me for a password. The only work around was to login as cdfdaq and then ksu cdfcalib on b0dap62. 
 
Could someone investigate this?</pre>
<!-- Author: Anant -->


<!-- Date: Thu Oct 30 09:47:34.876 2003 -->
IMU steel coming out now. North wall will be moved out by 
about lunchtime.
<!-- Author: Brig -->

<!-- Date: Thu Oct 30 09:51:50.817 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=172553">172553</A>
ACTIVATE: cal qie calibration
<!-- Author: Anant 2080 -->


<!-- Date: Thu Oct 30 09:52:42.143 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=172553">172553</A>
TERMINATE: cal qie
<!-- Author: Anant 2080 -->



<!-- Date: Thu Oct 30 09:59:03.495 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-11818&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-11818' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-11819&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-11819' width='300'></a></center>
<!-- Author: Anant -->
<!-- Comment: Thu Oct 30 09;59;30 comment by...Anan -->
<pre>cal qie bad pedestals and slopes</pre>

<!-- Date: Thu Oct 30 10:01:39.835 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=172554">172554</A>
ACTIVATE: clc qie calibration
<!-- Author: Anant 2080 -->


<!-- Date: Thu Oct 30 10:01:56.514 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=172554">172554</A>
TERMINATE: clc qie
<!-- Author: Anant 2080 -->
<!-- Comment: Thu Oct 30 10;06;30 comment by...Anant -->
<pre>dbana plot looks fine

SloErr/Slope MEAN histogram had a mean of 0.0004</pre>
<!-- Date: Thu Oct 30 10:04:58.002 2003 -->
Roberta D. has replaced the power supply for "the  
crate in the West Alcove". We believe that this is 
FCAL_1W. We need someone to replug in the cards and 
check that everything is OK as soon as possible. Roberto 
will also replace the power supply on the east side soon.  
<!-- Author: Brig -->
<!-- Comment: Thu Oct 30 10;43;02 comment by...Brig -->
Koji Terashi will come in around 12:30 - 1:00 to
check out FCAL_1W.
<!-- Date: Thu Oct 30 10:14:28.557 2003 -->
<pre>I can't do the bsc qie calibration as b0fcal00 is won't partition. It is red in VxWorks and I can't shepherd it or even log into it.</pre>
<!-- Author: Anant -->
<!-- Comment: Thu Oct 30 15;34;20 comment by...Brig -->
ADMEM cards were all pulled out because of the 
replacement of the power supply. They have now
been replaced and checked out.
<!-- Date: Thu Oct 30 10:18:33.435 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=172557">172557</A>
ACTIVATE: tof qie calibration
<!-- Author: Anant 2080 -->


<!-- Date: Thu Oct 30 10:18:44.639 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=172557">172557</A>
TERMINATE: tof qie
<!-- Author: Anant 2080 -->


<!-- Date: Thu Oct 30 10:21:24.745 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=164126">164126</A>
RUNSTATUS: L1T The Live time (and other) accounting is all messed up for this run.
<!-- Author: Carla -->


<!-- Date: Thu Oct 30 10:22:09.003 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=172560">172560</A>
ACTIVATE: cem led calibration
<!-- Author: Anant 2080 -->


<!-- Date: Thu Oct 30 10:23:35.994 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=172560">172560</A>
TERMINATE: cem led
<!-- Author: Anant 2080 -->
<!-- Comment: Thu Oct 30 10;26;50 comment by...Anant -->
<pre>A new bad channel has been found:

West wedge 16, Tower 5, PMT 1 the ADC count is 0.00000e+00</pre>
<!-- Date: Thu Oct 30 10:26:17.867 2003 -->
Cryo has finished upgrading S6 and are almost done with S5. 
Now going on to S3. This controls the racks. So we will 
lose the monitoring but there shouldn't be any trips.
<!-- Author: Brig -->


<!-- Date: Thu Oct 30 10:29:49.339 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=172563">172563</A>
ACTIVATE: muon calibration HV on
<!-- Author: Anant 2080 -->


<!-- Date: Thu Oct 30 10:29:50.548 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=172563">172563</A>
TERMINATE: muon calibration
<!-- Author: Anant 2080 -->


<!-- Date: Thu Oct 30 10:33:42.660 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=172565">172565</A>
ACTIVATE: tof tac calibration
<!-- Author: Anant 2080 -->


<!-- Date: Thu Oct 30 10:33:43.394 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=172565">172565</A>
TERMINATE: tof tac
<!-- Author: Anant 2080 -->




<!-- Date: Thu Oct 30 10:43:57.163 2003 -->
Cryo called. They have finished with S3. They are now going 
to work with S1. We will lose monitoring for gas and 
the main alarm panel in the control room.
<!-- Author: Brig -->


<!-- Date: Thu Oct 30 11:05:03.856 2003 -->
<h3>SRC Errors from last night</h3> 
I investigated the SRC errors from last night, and I concluded that none of them were actually real. Most came up as a consequence of power-cycling the TS crate, and one happened amdist a TDC problem bonanza. Interested parties can  
 
<a href= 
'/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=172&frame=2&anchor=10:37:58'> 
read about it.</a>
<!-- Author: Sal -->

<!-- Date: Thu Oct 30 11:07:13.459 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=172568">172568</A>
ACTIVATE: smx qie calibration
<!-- Author: Anant 2080 -->



<!-- Date: Thu Oct 30 11:07:38.396 2003 -->
Cryo calls. They have finished with S1 and also S2.
<!-- Author: Brig -->

<!-- Date: Thu Oct 30 11:08:58.069 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=172568">172568</A>
TERMINATE: smx qie
<!-- Author: Anant 2080 -->
<!-- Comment: Thu Oct 30 11;14;27 comment by...Anant -->
<pre>plots looked fine</pre>


<!-- Date: Thu Oct 30 11:21:00.084 2003 -->
Some work being done on HV Monitoring program. 
Should last less than 1/2 hour?
<!-- Author: Brig -->
<!-- Comment: Thu Oct 30 14;13;24 comment by...Phil Schlabach -->
Restored ~1330 except for BMU. BMU monitoring is disabled because they've disabled their CAENs while they're working. They will re-enable it when they're done. Apparently, this turns the global alarms box light blue.
<!-- Date: Thu Oct 30 11:21:56.638 2003 -->
<table border=0><td><tr><td bgcolor=#e0e030>The solenoid can be powered whenever we want.</td></tr></td></table>
<!-- Author: Brig -->


<!-- Date: Thu Oct 30 11:27:51.043 2003 -->
<pre>Calibrations I haven't done that are usually done: 
 
BSC QIE: b0fcal00 crate errors 
PLUG LASER: HV controls have been disabled on vnode 1 and I wasn't sure if I can go up and turn them on, or whether they are off for a reason. 
PMT SPIKE RUN:cot test are using hardware evb/L3</pre>
<!-- Author: Anant -->


<!-- Date: Thu Oct 30 11:30:21.141 2003 -->
<table border=0><td><tr><td bgcolor=#e0e030>Please use caution if you choose to put HTML into an entry.  Forgetting to use a close tag for each open tag will certainly cause problems.  
 
If you choose to use plain text, PLEASE USE CR's since this text window automatically wraps the text and gives the illusion of CR's.  (click to clear this message) 
</td></tr></td></table>
<!-- Author: Sergei -->


<!-- Date: Thu Oct 30 11:30:32.921 2003 -->
Maria S. would like to do some charge injection tests on ADMEM boards whenever time is available. She would need L1 and L3. 
  
<!-- Author: Brig -->


<!-- Date: Thu Oct 30 11:30:56.724 2003 -->
<pre>It looks like the l2 alpha (b0l2pr01) was using the same  
ip address as the web server. This would explain the slow 
response to the web server and possibly the problems we were seeing with the consumer state manager and other places... 
 
An Alpha was recently swapped in with a MAC address that  
was assigned to the ip address of the web server. 
 
We changed this to a free ipaddress. The ipaddress is assigned 
in the bootptab, where the file lists the MAC address and the  
associated ip address. 
</pre>
<!-- Author: Frank Chlebana -->


<!-- Date: Thu Oct 30 11:33:13.497 2003 -->
<table border=0><td><tr><td bgcolor=#e0e030>L2 Alpha VME board used the same IP address as Online web-server 
it was reason for last days problem with access to Online web-server.</td></tr></td></table>
<!-- Author: Sergei -->


<!-- Date: Thu Oct 30 11:40:39.797 2003 -->
<table border=0><td><tr><td bgcolor=#e0e030>The main server for the DAQ is going down (for 
servicing).</td></tr></td></table>
<!-- Author: Brig -->
<!-- Comment: Thu Oct 30 11;56;23 comment by...Brig -->
Server (b0dau30) back up
<!-- Date: Thu Oct 30 11:54:36.441 2003 -->
Tom Phillips noticed that the emergency lights 
behind the L3 display (2RR14C) do not appear to be 
functional. I called both Keith Schuh and George Wyatt, 
who are listed under Satety Systems, Building Utilities  
and left messages.
<!-- Author: Brig -->



<!-- Date: Thu Oct 30 11:58:27.911 2003 -->
CMX Wedge SW6 tripped. We did NOT reset (per 
Muon expert instructions).
<!-- Author: Brig -->
<!-- Comment: Thu Oct 30 13;12;53 comment by...Anant -->
<pre>I powered the wedge on again and it has been fine.</pre>
<!-- Date: Thu Oct 30 12:01:55.008 2003 -->
<table border=0><td><tr><td bgcolor=#e0e030>b0dau30 , main NFS server, was down for 10 minutes to replace  
bad memory DIM</td></tr></td></table>
<!-- Author: Sergei -->


<!-- Date: Thu Oct 30 13:00:56.845 2003 -->
Koji T. and Michele G. have replugged the boards 
in FCAL_1W and checked them out. Everything OK. So 
Roberto D. can now replace the power supply on the 
East side (if he hasn't done so already).
<!-- Author: Brig -->


<!-- Date: Thu Oct 30 13:08:15.457 2003 -->
Dave Ambrose will drop power to one of the wall racks  
(TDC _ TOF) while he works on one of the COT crates.
<!-- Author: Brig -->



<!-- Date: Thu Oct 30 13:17:13.570 2003 -->
Cryo is starting Solenoid check list. Will start 
ramping magnet soon.
<!-- Author: Brig -->


<!-- Date: Thu Oct 30 14:23:03.597 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=172578">172578</A>
ACTIVATE: XTRP Three TRack Board -- test run for base 50 readout

<!-- Author: cmarino -->



<!-- Date: Thu Oct 30 14:32:35.990 2003 -->
Nikolai is turning down the HV on the NW 
CHA.
<!-- Author: Brig -->

<!-- Date: Thu Oct 30 14:36:50.882 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=172583">172583</A>
ACTIVATE: XTRP Three Track Board base = 50, with database set to 50 
<!-- Author: cmarino -->



<!-- Date: Thu Oct 30 14:44:02.483 2003 -->
<p><b>Silicon cooling...</b></p> 
<p> 
...is all OK now.  The leak in SVX BHW3 spotted this morning was fixed by putting on a new pipe fitting.  The air bubbles trapped in the SVX PC lines were blasted out by opening the supply valves quickly (instead of slowly like I did last night).</p>
<!-- Author: Andy H. -->

<!-- Date: Thu Oct 30 14:53:45.583 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=172587">172587</A>
ACTIVATE: L2 Torture test with 3TT in XTRP;  PreFred B0Offset = 25 (default)
<!-- Author: cmarino -->


<!-- Date: Thu Oct 30 15:08:31.860 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=172589">172589</A>
ACTIVATE: L2 Torture with 3TT in XTRP, PreFred B0Offset = 24 (default - 1)
<!-- Author: cmarino -->


<!-- Date: Thu Oct 30 15:12:12.094 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=172590">172590</A>
ACTIVATE: L2 Torture with 3TT in XTRP, PreFred B0Offset = 24 (default - 1) (Take 2)
<!-- Author: cmarino -->



<!-- Date: Thu Oct 30 15:14:52.024 2003 -->
George Wyatt spent the last few days fixing two broken CAEN reset lines for the silicon.  They have been tested and work fine.  The remaining 14 lines were also tested and also work fine.
<!-- Author: Andy H., Florencia C. -->

<!-- Date: Thu Oct 30 15:23:13.721 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=172591">172591</A>
ACTIVATE: L2 Torture test with 3TT in XTRP, PreFred B0Offset = 24 (take 3)
<!-- Author: cmarino -->


<!-- Date: Thu Oct 30 15:28:30.851 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=172592">172592</A>
ACTIVATE: L2 Torture with 3TT in the XTRP, PreFred B0Offset = 26 (default + 1)
<!-- Author: cmarino -->



<!-- Date: Thu Oct 30 15:38:21.333 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=172597">172597</A>
ACTIVATE: L2 Torture test with 3TT in XTRP, PreFred B0Offset = 26, BASE = 2.
<!-- Author: cmarino -->


<!-- Date: Thu Oct 30 15:46:29.821 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=172598">172598</A>
ACTIVATE: L2 Torture test with 3TT in the XTRP, PreFred B0Offset = 25 (default), BASE = 2.
<!-- Author: cmarino -->



<!-- Date: Thu Oct 30 15:48:43.001 2003 -->
<font color=red>Shift Summary: </font><pre>(1) Frank C, Tom P, and Dave A. worked on TDC readout most 
of the shift. Fixed a few problems. A few problems remain 
still (with COT05, COT06, COT11..new) 
(2) Work on XFT crates continues in parallel 
(3) Work on silicon cooling leaks in Collision Hall 
(4) Some upgrades to the Solenoid control were performed. 
   Magnet now back at full field 
(5) Cryo cycled through all of their monitoring computers  
   (S1 - S6) doing some upgrades/checks. 
(6) Sections of L3 on and off for electrical work on 3rd floor. 
(7) Main fileserver, b0dau30 down for servicing 
(8) Some work was done on HV monitoring. Should be working 
   except for BMU (disconnected while working on CAEN supplies). 
(9) Most calibrations were done. Calibrations NOT DONE: 
 
     BSC QIE: b0fcal00 crate errors  
     PLUG LASER: HV controls have been disabled on vnode 1 and I wasn't  
      sure if I can go up and turn them on, or whether they are off  
      for a reason.  
      PMT SPIKE RUN:cot test are using hardware evb/L3 
(10) Problem with very long response of online system to e-log entries  
   (especially Start of shift ) believed to be due to the fact that 
   one of the Level 2 Alpha boards had the same IP address as the web 
   server 
(11) The main server (b0dau30) for the DAQ down ~ 15 min for servicing. 
(12) Silicon Cooling now OK again - Andy Hocker 
(13) L2 Torture test with 3TT in XTRP is ongoing. 
(14) George Wyatt spent the last few days fixing two broken CAEN reset lines  
   for the silicon. They have been tested and work fine. 
(15) Emergency lights behind the L3 display (2RR14C) do not appear to  
    be functional. Called George Wyatt, but it is Craig Olsen who should 
    see that it's fixed.</pre> 
<!-- Author: Brig -->


<!-- Date: Thu Oct 30 15:54:08.367 2003 -->
<pre>We strung a set of signal cables from the bottom ASD crate to  
stacks 30-35 on the bottom CMP.  This was done to bypass a 
 broken signal cable for CMP bottom stack 32.  By doing this we 
 have fixed one dead channel in the CMP.  
 
This is a kludge which will remain in place at least until the 
 next long shut-down period (at the moment we cannot locate any 
 surplus CMP signal cable, so the CMX folks were kind enough to 
 lend us some for the time being).   
 
Since the new signal cable length is probably different from 
 that which was originally in place, we can expect the t_0 to 
 change for stacks 30-35 on CMP bottom.  The t_0 change can be 
 handled by the calibration database.</pre>
<!-- Author: Trevor and Phil -->
<!-- Comment: Thu Oct 30 21;00;48 comment by...ps -->
<ul><li>once more into the subfloor for the gipper (as another shirt bites the dust)...</li> <li>The kludge may remain; but, if we leave the cable as is, we have to dress the cable at both ends so it moves with the miniskirt. That may be the way to go, but I want to look at the end of the old cable I haven't looked at yet before giving up, as well as the way we strung it to see if there is a better way. We ran out of time today. And we need to confirm with the data overnight that we did fix it.</li></ul>
<!-- Date: Thu Oct 30 15:55:48.303 2003 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>15:55:01</TD>                                                       
<TD ALIGN=CENTER>::</TD>                                                
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Thu Oct 30 16:05:56.122 2003 -->
<pre>BMU Access Report 
 
Dan replaced the HV board on BMU NE-65.  This difficult 
operation was apparently successful.  BMU HV is on. 
BMU HV control returned to shiftcrew.  </pre>
<!-- Author: Camille Ginsburg and Dan Cyr -->


<!-- Date: Thu Oct 30 16:09:34.709 2003 -->
<pre>Turned CMX wedge 6 [west] back on. If any part of CMX trips,  
follow normal recovery procedures and make note of time and wedge. If the wedge trips multiple times, turn it off. Page  
CMX pager if needed. </pre>
<!-- Author: John Strologas, Ingyin Zaw, Dale Stentz -->