<!-- Title: 10:40:01  Mon Aug  4 2008 -->
<!-- Crewchief: SashaS (MPC) -->
<!-- Op1: Satyajit (SPL on call) -->
<!-- Op2: &nbsp; -->
<!-- Op3: &nbsp; -->
<!-- Op4: &nbsp; -->
<!-- Op5: &nbsp; -->
<!-- Op6: &nbsp; -->
<!-- Notes: -->
Keep running...


<!-- Date: Tue Aug 5 03:23:50 2008 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12783&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12783  width='300' align='top'></a></td><td>Store 6338: Init Lumi 268 1/ub/s, Lostp flat at 1.5 kHz, included Silicon</td></tr></table>
<!-- Author: SashaS -->


<!-- Date: Tue Aug 5 21:21:00 2008 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12784&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12784  width='300' align='top'></a></td><td>Store 6341: Init Lumi 272 1/ub/s, Lostp is flat at 2kHz. Included Silicon</td></tr></table>
<!-- Author: SashaS -->


<!-- Date: Wed Aug 6 09:21:47 2008 -->
<h3>MCR power glitch and problems with SVX wedge SB3W11</h3> 
There was a power glitch in MCR at about 5:41 this morning and one of RF station was lost. Silicon was put in STBY. Svx SB3W11 ladders L1, L2 and L3 tripped on Bias over-voltage and SB3W11L4 tripped on Bias over-current. SAR recovered wedge SB3W11 to STBY but it continued to trip on L2 Dvdd s/w protection. While still on the phone with the ace I asked him to hockerize CAEN crate #9. After hockerization SAR recovered all affected wedges to STBY. Meanwhile TevMon cleared and I authorized ace to reset CDFpermit, turn Silicon to ON and resume data taking. When rumped up ladder SB3W11L4 continued to trip on Bias overcurrent. While I was on my way to the lab ace put Silicon to STBY and started NO-SILICON run. When I arrived to the lab, I took SB3W11L4 ladder from HWDB and we resumed data taking with Silicon.
<p>I also noticed that SB3W11L2 Bias current is now much higher 400 ua compared to about 260 ua before. 
<p><b>Strip occupancy plots for SB3W11L2 and L4</b>
<ul>
<li>SB3W11L2 before accident
</ul>
<a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12786&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12786' width='300'></a><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12788&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12788' width='300'></a>

<ul>
<li>SB3W11L2 after accident
</ul>
<a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12790&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12790' width='300'></a><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12789&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12789' width='300'></a>

<ul>
<li>SB3W11L4 before accident
</ul>
<a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12791&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12791' width='300'></a><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12792&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12792' width='300'></a>

<!-- Author: SashaS -->
<!-- Comment: Wed Aug  6 15;09;41 comment by...SashaS -->
I put online ladder SB3W11L4. I have increased Bias current trip limit from 500 to 1000 ua. I powered this ladder and the bias current is 480 ua and it seems stable. Will need to re-adjust Imon limits later on.



<!-- Date: Thu Aug 7 01:47:00 2008 -->
<h3>Imon limits adjustment</h3> 
Changed Bias current limits for SB3W11L2 and SB3W11L4
<!-- Author: SashaS -->


<!-- Date: Thu Aug 7 01:49:53 2008 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12793&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12793  width='300' align='top'></a></td><td>Store 6346: Init Lumi 280 1/ub/s, Lostp flattened at 2 kHz, included Silicon</td></tr></table>
<!-- Author: SashaS -->


<!-- Date: Thu Aug 7 12:23:43 2008 -->
<h2> BERT Test Stand Final Report </h2> 
I did some more test these last days and found more good RXs which right now are ready as spares, you can find them in the Silicon office. Also I got three more good TXs, we can consider and  use them as spares. 
 
<h3> Good TXs </h3> 
I used these TXs during my tests. 
<ul> <li> TX 1200(A) </li> 
     <li> TX 1148(B) </li> 
     <li> TX 1146(C) </li> 
     <li> TX 1394(spare) </li> 
     <li> TX 1350(spare) </li> 
     <li> TX 1122(spare) </li> 
 
</ul> 
 
<h3> Good RXs </h3> 
This is the list of all good RXs(9)that we have right now for use in the detector. 
 
<ul> <li> RX 0562: Tested ok in BERT Test Stand </li> 
     <li> RX ????: Tested ok </li> 
     <li> RX 0411: Tested ok </li> 
     <li> RX 0726: Tested ok </li>  
     <li> RX 0752: Tested ok </li> 
     <li> RX 0682: Tested ok </li> 
     <li> RX 0141: Tested ok (8/06/08)</li> 
     <li> RX 0474: Tested ok (8/06/08)</li> 
     <li> RX 0851: Tested ok (8/06/08)</li> 
</ul> 
The BERT Test stand is running very well, if you like you can still testing the RXs. 
Tkanks to Vince Pavlicek and Hank Connor and thanks to everyone by let me work with you. 
Thanks to Fermilab and CDF
<!-- Author: Pedro -->
<!-- Comment: Sat Aug  9 12;21;56 comment by...Satyajit -->
Thanks to you and SebastianC for all the hard/good work! See you 
back soon.

<!-- Date: Thu Aug 7 20:56:39 2008 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12794&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12794  width='300' align='top'></a></td><td>Store 6348: Init Lumi 280 1/ub/s, Lostp flat at 2 kHz, included Silicon</td></tr></table>
<!-- Author: SashaS -->


<!-- Date: Fri Aug 8 07:44:49 2008 -->
<h3>Hockerizations</h3> 
<!-- Author: SashaS -->
<!-- Comment: Fri Aug  8 07;44;50 comment by...SashaS -->
Crate #16. SAR recovered
<!-- Comment: Fri Aug  8 07;47;01 comment by...SashaS -->
Crate #2, last night, 8/7/2008 2:39am. SAR auto-recovery was disabled, ace recovered manually.


<!-- Comment: Fri Aug  8 19;05;42 comment by...SashaS -->
Crate #5, SAR recovered
<!-- Comment: Sat Aug  9 16;07;03 comment by...SashaS -->
Crate #7. SAR recovered
<!-- Date: Fri Aug 8 16:10:40 2008 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12795&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12795  width='300' align='top'></a></td><td>Store 6351: Init Lumi 292 1/ub/s, Lostp 2kHz, included Silicon</td></tr></table>
<!-- Author: SashaS -->



<!-- Date: Fri Aug 8 19:40:34 2008 -->
<h3>Corrupted Readout from CAEN crates</h3>
<!-- Author: SashaS -->
<!-- Comment: Fri Aug  8 19;41;21 comment by...SashaS -->
ISL B5W4L2 Bias0 current read back 0

<!-- Comment: Sat Aug  9 12;10;58 comment by...SashaS -->
SVX B3W10 all ladders Bias voltages and currents read back 0. Hockerized CAEN crate #9. SAR recovered


<!-- Comment: Sat Aug  9 12;30;06 comment by...SashaS -->
In order to fix ISL B5W4L2 Bias0 current read back problem, I have hockerized CAEN crate #15 couple of times. It did not work. Since this ladder returned good data during previous store, I will leave it as is and will keep eye on it. We may need to replace power supply for this ladder to recover correct current monitoring.

<!-- Comment: Sat Aug  9 13;08;26 comment by...Satyajit -->
This seems another candidate we want to just reseat (in an access) 
and see if the problem goes away. If not, we replace it.



<!-- Comment: Sun Aug 10 10;06;21 comment by...SashaS -->
L00 W1 current in all channels read back 0. Walked ace through hockerization of crate #16 (it was a little challange, since our iFix computer vnode2 is down, we had to open hockerization window on vnode1)
<!-- Comment: Sun Aug 10 10;08;07 comment by...SashaS -->
SVX B4W1L0 Bias voltage read back 13 V compared to 90 V settings. PSmon set off SVX HV trigger inhibit. Hockerization of crate #10 fixed the problem
<!-- Date: Sat Aug 9 16:41:37 2008 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12796&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12796  width='300' align='top'></a></td><td>Store 6353: Init Lumi 280 1/ub/s, Lostp flat at 2.5 kHz, included Silicon</td></tr></table>
<!-- Author: SashaS -->




<!-- Date: Sun Aug 10 22:34:09 2008 -->
<h3>Monitoring Summary</h3> 
 
Checked runs 265701 - 265828 (40de5 - 40e64). There seemed to be some problems mid-week with the plots not updating, but appears to be fixed now. No new issues to report
<!-- Author: Mark -->


<!-- Date: Mon Aug 11 00:34:28 2008 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12797&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12797  width='300' align='top'></a></td><td>Store 6354: Init Lumi 250 1/ub/s, Lostp 1.9 kHz, included Silicon</td></tr></table>
<!-- Author: SashaS -->


<!-- Date: Mon Aug 11 09:24:08 2008 -->
<h2>Summary of the week</h2> 
 
<h3>Monday, August 4, 2008</h3> 
<ul> 
 <li> <b>Store 6334</b> has been lost during thunderstorm in the morning. 
</ul> 
 
<h3>Tuesday, August 5, 2008</h3> 
<ul> 
  <li> <b>Store 6338:</b> Init Lumi 268 1/ub/s; Lostp 1.5 kHz 
  <li> <b>Store 6341:</b> Init Lumi 272 1/ub/s; Lostp 2 kHz 
</ul> 
 
<h3>Wednesday, August 6, 2008</h3> 
<ul> 
  <li> MCR power glitch and SVX B3W11 <a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=417&frame=2&anchor=09:21:47'>problem</a> 
</ul> 
 
<h3>Thursday, August 7, 2008</h3> 
<ul> 
  <li> <b>Store 6346:</b> Init Lumi 280 1/ub/s; Lostp 2 kHz 
  <li> <b>Store 6348:</b> Init Lumi 280 1/ub/s; Lostp 2 kHz 
</ul> 
 
<h3>Friday, August 8, 2008</h3> 
<ul> 
  <li> <b>Store 6351:</b> Init Lumi 292 1/ub/s; Lostp 2 kHz 
</ul> 
 
<h3>Saturday, August 9, 2008</h3> 
<ul> 
  <li> <b>Store 6353:</b> Init Lumi 280 1/ub/s; Lostp 2.5 kHz 
  <li> Pbar kicker problem - no pbar accumulation all day. 
  Started pbar production late in the evening at reduced rate 
  <li> <a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2008&action=view&page=666&frame=2&anchor=16:43:49.775'>iFix problem</a> on <i>VNODE2</i>. Possibly some of the system/iFix files are <a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2008&action=view&page=666&frame=2&anchor=21:08:54.997'>corrupted</a> 
</ul> 
 
<h3>Sunday, August 10, 2008</h3> 
<ul> 
  <li> <b>Store 6354:</b> Init Lumi 250 1/ub/s; Lostp 1.9 kHz 
</ul> 
 
<h3>Monday, August 11, 2008</h3> 
<ul> 
  <li> <b>EOS 6354</b> scheduled at 10am following 2h access to fix Linac problem 
</ul> 
 
<h3>General Comments</h3> 
 
<ul> 
  <li> <a href= 
  'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=417&frame=2&anchor=07:44:49'>Auto-Hockerizations</a> 
  <li><a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=417&frame=2&anchor=19:40:34'>Corrupted read back from PS</a> 
  <li> <a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=417&frame=2&anchor=12:23:43'>BERT Test Stand Final Report</a> 
  <li> <a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=417&frame=2&anchor=22:34:09'>Monitoring Summary</a> 
</ul>
<!-- Author: SashaS -->