<!-- Title: Day shift. Thu May 19, 2005 -->
<!-- SciCo: Lee Pondrom -->
<!-- DAQAce: Radu Marginean -->
<!-- MonAce: Pedro Fernanadez -->
<!-- CO: Adrian Buzatu -->
<!-- OpManager: Sung-won Lee -->
<!-- Notes: -->
<pre>Start shift in shot setup.  Final protons loaded. Loading pbars. 
Store #4154.</pre>

<!-- Date: Thu May 19 08:04:40.063 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198227">198227</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_3_02 [1,639,566] 
<!-- Author: Jiyeon x2080 -->


<!-- Date: Thu May 19 08:20:46.359 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198227">198227</A>
TERMINATE: end shot setup.
<!-- Author: Radu x2080 -->



<!-- Date: Thu May 19 08:30:46.339 2005 -->
pbars loaded, scraping started. 9040E9 protons, 1004E9 pbars. Store 4154.
<!-- Author: pondrom -->


<!-- Date: Thu May 19 08:40:11.583 2005 -->
scraping complete, CDFlum 89E30.  
<!-- Author: pondrom -->


<!-- Date: Thu May 19 08:44:22.970 2005 -->
We had HV alarm for TOF that recovered after a few secs.
<!-- Author: Pedro -->


<!-- Date: Thu May 19 08:45:24.235 2005 -->
We had twcie PSM alarm for rack 1RR12C_1 (works for ACNET). Recoveredf after a while.
<!-- Author: Pedro -->

<!-- Date: Thu May 19 08:46:14.483 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198228">198228</A>
TERMINATE: fail config

<!-- Author: Radu x2080 -->



<!-- Date: Thu May 19 08:46:28.043 2005 -->
<TABLE bgcolor=#add8e6> 
<TR><TH>Date</th><th>Time</th><th colspan=3>Quantity</th></tr> 
<tr><td>2005.05.19</td><td>08:41:51</td><td>TEVpr</td><td>8783.43</td><td>E09</td></tr> 
<tr><td>2005.05.19</td><td>08:41:51</td><td>TEVpb</td><td>990.12</td><td>E09</td></tr> 
<tr><td>2005.05.19</td><td>08:42:56</td><td>C:B0ILUM</td><td>87.91</td><td>E30</td></tr> 
<tr><td>2005.05.19</td><td>08:42:56</td><td>C:B0LLUM</td><td>0.00</td><td>E30</td></tr> 
<tr><td>2005.05.19</td><td>08:42:56</td><td>C:B0PLOS</td><td>473.44</td><td>Hz</td></tr> 
<tr><td>2005.05.19</td><td>08:42:56</td><td>C:B0ALOS</td><td>69.28</td><td>Hz</td></tr> 
</TABLE> 
<strong>Luminosity and losses</strong>
<!-- Author: Pedro -->

<!-- Date: Thu May 19 08:51:25.046 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198229">198229</A>
TERMINATE: l3 problems

<!-- Author: Radu x2080 -->
<!-- Comment: Thu May 19 09;41;03 comment by...Jeff Miles -->
After the shot setup run was ended, the Level3 proxy never closed partition 0.  So L3 failed on all susbsequent attempts at transitions.  The L3 display claimed that it was in the partition state even though the Replies and Acknowledgements window indicated L3 trouble, a fact that Radu faithfully reported to me over the phone but whose significance I was unable to interpret without looking at the logs.  The brute-force solution was simply to clean up Level3 (and I cleaned up EVB before that, for completeness).  Had I understood the problem immediately, a more elegant solution would have been to simply "Send Abort" to L3proxy for partition 0 with the EVB/L3 Ace Control Panel.
<!-- Date: Thu May 19 08:53:01.758 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-8567&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-8567' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-8568&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-8568' width='300'></a></center>
<!-- Author: Pedro -->
<!-- Comment: Thu May 19 08;53;36 comment by...Pedro -->
Radiation and losses during shot setup and scraping for store 4154.

<!-- Date: Thu May 19 09:12:33.015 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198230">198230</A>
 Activated at 2005.05.19 09:12:29
<!-- Author: RunControl -->


<!-- Date: Thu May 19 09:12:57.150 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198230">198230</A>
ACTIVATE: RunSet: AAA_CURRENT  Trigger: PHYSICS_3_02 [1,639,566] 
<!-- Author: Radu x2080 -->



<!-- Date: Thu May 19 09:17:48.713 2005 -->
T:L1COLI went to zero for a while. Switched SIlicon to STDBY.
<!-- Author: Pedro :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198230' target=_top>198230</a>) -->
<!-- Comment: Thu May 19 09;19;47 comment by...quenchmonkey -->
good catch.
<!-- Comment: Thu May 19 11;30;29 comment by...Marcel -->
<table border=0><td><tr><td bgcolor=#e0e030><h2> GOOD JOB  </h2>
this probably saved us</td></tr></td></table>
<!-- Comment: Thu May 19 11;41;30 comment by...Ray Culbertson -->
TevMon correctly alarmed on this at 09:15:07
<!-- Date: Thu May 19 09:19:29.329 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198230">198230</A>
 Terminated at 2005.05.19 09:19:24
<!-- Author: RunControl -->




<!-- Date: Thu May 19 09:19:55.192 2005 -->
Tevatron quench. beam gone.
<!-- Author: pondrom -->


<!-- Date: Thu May 19 09:20:01.816 2005 -->
<table border=0><td><tr><td bgcolor=#e0e030>Lost the beam.</td></tr></td></table>
<!-- Author: Pedro -->
<!-- Comment: Thu May 19 09;31;20 comment by...Pedro -->
Switched to end of store condition. Global alarms for COT, Muon system, etc.
<!-- Date: Thu May 19 09:21:00.931 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198230">198230</A>
TERMINATE: end, Tevatron quench
<!-- Author: Radu x2080 -->



<!-- Date: Thu May 19 09:22:17.138 2005 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2005&action=view&page=-8569&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2005&action=view&page=-8569  width='300' align='top'></a></td><td>Post quench plots</td></tr></table>
<!-- Author: Pedro -->


<!-- Date: Thu May 19 09:31:03.133 2005 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2005&action=view&page=-8570&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2005&action=view&page=-8571  width='300' align='top'></a></td><td>Quench plots.  The TEL went off a few minutes before the 
quench (good catch above).  </td></tr></table>
<!-- Author: R.J. Tesarek -->




<!-- Date: Thu May 19 10:10:53.537 2005 -->
<pre>EVB upgrade test: 
- tried to start a L2torture run and got L2TO from fcal00. Take it out.</pre>
<!-- Author: markus -->
<!-- Comment: Thu May 19 10;22;18 comment by...convery -->
looks like the smxr in fcal00 got scrambled by the quench...


<!-- Date: Thu May 19 10:42:26.189 2005 -->
Powercycling fcal00 did not help.  Header errors indicate a problem in the SMD crate in the tunnel.
<!-- Author: convery -->

<!-- Date: Thu May 19 10:45:30.782 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198236">198236</A>
ACTIVATE: RunSet: SVT_B0SVT09_TEST 
<!-- Author: pgiova -->


<!-- Date: Thu May 19 10:57:55.123 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198236">198236</A>
TERMINATE: ending
<!-- Author: pgiova -->


<!-- Date: Thu May 19 10:59:12.103 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198238">198238</A>
ACTIVATE: RunSet: SVT_B0SVT09_TEST 
<!-- Author: pgiova -->


<!-- Date: Thu May 19 11:05:53.182 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198238">198238</A>
TERMINATE: ending

<!-- Author: pgiova -->


<!-- Date: Thu May 19 11:07:25.861 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198239">198239</A>
ACTIVATE: RunSet: SVT_B0SVT09_TEST 
<!-- Author: pgiova -->



<!-- Date: Thu May 19 11:08:01.382 2005 -->
<pre>Am using b0cot02 (COT_02) crate to test out the  
so-called "early-early" mode using TDC circular  
header buffers... 
</pre>
<!-- Author: W.Badgett -->

<!-- Date: Thu May 19 11:10:24.399 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198244">198244</A>
ACTIVATE: RunSet: ACE_QIE Calorimeter QIE calibration in order to prepare for CEM XEF calibration.
<!-- Author: radum -->


<!-- Date: Thu May 19 11:11:10.002 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198244">198244</A>
TERMINATE: end CAL QIE calibrations. 
<!-- Author: radum -->
<!-- Comment: Thu May 19 11;12;09 comment by...Radu -->
CCAL 8 and 14 have been exculded.

<!-- Date: Thu May 19 11:15:21.617 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198239">198239</A>
TERMINATE: ending
<!-- Author: pgiova -->


<!-- Date: Thu May 19 11:17:28.864 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198246">198246</A>
ACTIVATE: RunSet: SVT_B0SVT09_TEST 
<!-- Author: pgiova -->


<!-- Date: Thu May 19 11:22:04.965 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198247">198247</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: radum -->



<!-- Date: Thu May 19 11:22:36.389 2005 -->
Estimate for completion of CDF access 1230-100 pm
<!-- Author: pondrom -->

<!-- Date: Thu May 19 11:24:03.098 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198246">198246</A>
TERMINATE: ending
<!-- Author: annovi -->


<!-- Date: Thu May 19 11:24:04.075 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198247">198247</A>
TERMINATE: end CEM XEF calibration, CCAL 8,14 were excluded.
<!-- Author: radum -->


<!-- Date: Thu May 19 11:27:27.381 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198248">198248</A>
ACTIVATE: RunSet: SVT_B0SVT09_TEST 
<!-- Author: annovi -->


<!-- Date: Thu May 19 11:28:10.204 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198248">198248</A>
TERMINATE: ending
<!-- Author: annovi -->


<!-- Date: Thu May 19 11:30:23.718 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198250">198250</A>
ACTIVATE: RunSet: SVT_B0SVT09_TEST 
<!-- Author: annovi -->


<!-- Date: Thu May 19 11:30:24.590 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198250">198250</A>
TERMINATE: ending
<!-- Author: annovi -->



<!-- Date: Thu May 19 11:33:12.780 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198245">198245</A>
TERMINATE: Ran the so-called "early-early" or "extra-early" (circular buffer) done mode in b0cot02 
for over 5 million events, without any errors reported by the front-end 
crate.   Now we have to try it with the HEVB to make sure data are OK.
<!-- Author: badgett -->


<!-- Date: Thu May 19 11:35:26.166 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198252">198252</A>
ACTIVATE: RunSet: SVT_B0SVT09_TEST 
<!-- Author: annovi -->


<!-- Date: Thu May 19 11:35:27.135 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198252">198252</A>
TERMINATE: ending
<!-- Author: annovi -->



<!-- Date: Thu May 19 11:38:28.655 2005 -->
About 3 weeks ago, we switched the online production database  
from the old b0dau35 machine to a new machine bzora1.  We have  
just now taken b0dau35 off the network for the first time.   
We believe no one is accessing it for any purpose, but this  
is a test to be sure.  It will remain offline, but powered,  
for at least a week.  In the slim chance of a problem where this  
is suspected, page DAQ.  The machine sysadmin is Maurine Mihalek.  
<!-- Author: Ray Culbertson -->

<!-- Date: Thu May 19 11:39:36.870 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198253">198253</A>
TERMINATE: aborting

<!-- Author: annovi -->


<!-- Date: Thu May 19 11:40:00.973 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198254">198254</A>
ACTIVATE: RunSet: SVT_B0SVT09_TEST 
<!-- Author: annovi -->


<!-- Date: Thu May 19 11:40:02.958 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198254">198254</A>
TERMINATE: ending
<!-- Author: annovi -->



<!-- Date: Thu May 19 11:41:57.580 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198255">198255</A>
ACTIVATE: RunSet: SVT_B0SVT09_TEST 
<!-- Author: annovi -->


<!-- Date: Thu May 19 11:44:59.029 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198255">198255</A>
TERMINATE: ending
<!-- Author: annovi -->



<!-- Date: Thu May 19 11:45:50.201 2005 -->
<pre>EVB upgrade: 
only got 10% of the rate tests done. Si wants the system for a cosmic run now. I would like to continue testing if there is an opportunity after 4pm this afternoon.</pre>
<!-- Author: markus -->

<!-- Date: Thu May 19 11:46:15.333 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198256">198256</A>
ACTIVATE: RunSet: SVT_B0SVT09_TEST 
<!-- Author: annovi -->


<!-- Date: Thu May 19 11:47:36.785 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198257">198257</A>
ACTIVATE: RunSet: CALIB_RP_SMQIE 
<!-- Author: convery 2080 -->


<!-- Date: Thu May 19 11:48:59.064 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198256">198256</A>
TERMINATE: ending

<!-- Author: annovi -->


<!-- Date: Thu May 19 11:50:09.902 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198259">198259</A>
ACTIVATE: RunSet: SVT_B0SVT09_TEST 
<!-- Author: annovi -->



<!-- Date: Thu May 19 11:54:59.007 2005 -->
New monitor ace for the afternoon Michael Morello.
<!-- Author: pondrom -->

<!-- Date: Thu May 19 11:56:57.925 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198261">198261</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [16,623,539] 
<!-- Author: markus (x2080) -->



<!-- Date: Thu May 19 11:57:12.709 2005 -->
<pre>I'm finished with Level3 maintenance for now  
 
The nodes fixed and included in the run configuration are: 
 
072, 074, 227, 231, 233, 234, 236, 241, 242, 243 
 
</pre>
<!-- Author: Arkadiy -->

<!-- Date: Thu May 19 12:04:58.023 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198259">198259</A>
TERMINATE: ending
<!-- Author: annovi -->



<!-- Date: Thu May 19 12:05:02.913 2005 -->
<h3>CMX access summary</h3> 
We worked on the CMX SW miniskirt (iFIX wedge 15) in order to help prevent trips on the cathode HV board.  We were able to corona dope most of the chambers in the known weak areas.  CMX holds HV fine with the normal current draws, etc.  HV is under the control of the shift crew.
<!-- Author: Dale Stentz and Phil Schlabach -->
<!-- Comment: Thu May 19 12;36;16 comment by...Phil -->
and we closed the fin after we closed the toroid.
<!-- Date: Thu May 19 12:06:57.930 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198262">198262</A>
ACTIVATE: RunSet: SVT_B0SVT09_TEST 
<!-- Author: annovi -->



<!-- Date: Thu May 19 12:07:48.733 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-8572&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-8572' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-8573&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-8573' width='300'></a></center>
<!-- Author: Adrian -->
<!-- Comment: Thu May 19 12;08;38 comment by...Adrian -->
Calibration results for 198244. There are black and blue lines, which may be a proble.

<!-- Comment: Thu May 19 16;48;51 comment by...Kwangzoo -->
a <a href='/cgi/elog/elog.pl?nb=2005&action=view&page=416&frame=2&anchor=15:23:51.629'>comment</a> in the next calibration result (Run 198283).
<!-- Date: Thu May 19 12:10:55.244 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198262">198262</A>
TERMINATE: ending
<!-- Author: annovi -->


<!-- Date: Thu May 19 12:14:28.988 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198263">198263</A>
 Activated at 2005.05.19 12:14:02
<!-- Author: RunControl -->


<!-- Date: Thu May 19 12:14:29.784 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198263">198263</A>
ACTIVATE: RunSet: COSMICS_SILICON_TAPE  Trigger: COSMICS [15,619,538] 
<!-- Author: radum -->


<!-- Date: Thu May 19 12:16:14.239 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198263">198263</A>
 Terminated at 2005.05.19 12:15:35
<!-- Author: RunControl -->


<!-- Date: Thu May 19 12:17:21.470 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198264">198264</A>
 Activated at 2005.05.19 12:17:15
<!-- Author: RunControl -->


<!-- Date: Thu May 19 12:17:32.552 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198264">198264</A>
ACTIVATE: RunSet: COSMICS_SILICON_TAPE  Trigger: COSMICS [15,619,538] 
<!-- Author: radum -->


<!-- Date: Thu May 19 12:19:49.829 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198265">198265</A>
ACTIVATE: RunSet: SVT_B0SVT09_TEST 
<!-- Author: annovi -->


<!-- Date: Thu May 19 12:20:42.264 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198265">198265</A>
TERMINATE: ending
<!-- Author: annovi -->



<!-- Date: Thu May 19 12:20:52.588 2005 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2005&action=view&page=-8574&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2005&action=view&page=-8574  width='300' align='top'></a></td><td>Calibration for 198 247 CEM XEF. It seems ok, according to the  comment Willis made in the reference (as only one of the 2 graphs is zero). I nevertheless wonder about the dead (white) channels. Are they the 2 crates that have been taken out or are there other dead channels?</td></tr></table>
<!-- Author: Adrian  -->

<!-- Date: Thu May 19 12:22:51.874 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198266">198266</A>
ACTIVATE: RunSet: SVT_B0SVT09_TEST 
<!-- Author: annovi -->


<!-- Date: Thu May 19 12:22:52.668 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198266">198266</A>
TERMINATE: trying again
<!-- Author: annovi -->


<!-- Date: Thu May 19 12:23:58.105 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198267">198267</A>
ACTIVATE: RunSet: SVT_B0SVT09_TEST 
<!-- Author: annovi -->


<!-- Date: Thu May 19 12:24:34.148 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198267">198267</A>
TERMINATE: ending
<!-- Author: annovi -->


<!-- Date: Thu May 19 12:24:35.465 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198264">198264</A>
 Terminated at 2005.05.19 12:24:24
<!-- Author: RunControl -->


<!-- Date: Thu May 19 12:25:00.040 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198264">198264</A>
TERMINATE: htdjhtdjhtdl;kh;klxdxlh;klxh;xl;;xlddhf
<!-- Author: radum -->


<!-- Date: Thu May 19 12:26:44.538 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198269">198269</A>
ACTIVATE: RunSet: SVT_B0SVT09_TEST 
<!-- Author: annovi -->


<!-- Date: Thu May 19 12:28:07.477 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198270">198270</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: radum -->


<!-- Date: Thu May 19 12:28:24.270 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198271">198271</A>
 Activated at 2005.05.19 12:28:20
<!-- Author: RunControl -->


<!-- Date: Thu May 19 12:28:25.027 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198271">198271</A>
ACTIVATE: RunSet: COSMICS_SILICON_TAPE  Trigger: COSMICS [15,619,538] 
<!-- Author: radum -->


<!-- Date: Thu May 19 12:29:32.638 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198269">198269</A>
TERMINATE: ending
<!-- Author: annovi -->


<!-- Date: Thu May 19 12:29:33.981 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198270">198270</A>
TERMINATE: end Xenon calibration.
<!-- Author: radum -->


<!-- Date: Thu May 19 12:34:07.485 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198271">198271</A>
 Terminated at 2005.05.19 12:33:55
<!-- Author: RunControl -->


<!-- Date: Thu May 19 12:34:08.524 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198271">198271</A>
TERMINATE: sssssssssssssssss
<!-- Author: radum -->



<!-- Date: Thu May 19 12:37:51.633 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198272">198272</A>
 Activated at 2005.05.19 12:37:33
<!-- Author: RunControl -->


<!-- Date: Thu May 19 12:37:52.147 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198272">198272</A>
ACTIVATE: RunSet: COSMICS_SILICON_TAPE  Trigger: COSMICS [15,619,538] 
hit it again
<!-- Author: radum -->


<!-- Date: Thu May 19 12:41:50.177 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198272">198272</A>
 Terminated at 2005.05.19 12:41:35
<!-- Author: RunControl -->


<!-- Date: Thu May 19 12:41:51.228 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198272">198272</A>
TERMINATE: still not going

<!-- Author: radum -->


<!-- Date: Thu May 19 12:43:05.328 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198273">198273</A>
TERMINATE: aborting
<!-- Author: annovi -->


<!-- Date: Thu May 19 12:48:15.298 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198274">198274</A>
TERMINATE: aborting

<!-- Author: annovi -->


<!-- Date: Thu May 19 12:50:47.351 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198276">198276</A>
 Activated at 2005.05.19 12:50:17
<!-- Author: RunControl -->


<!-- Date: Thu May 19 12:50:48.843 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198276">198276</A>
ACTIVATE: RunSet: COSMICS_SILICON_TAPE  Trigger: COSMICS [15,619,538] 
<!-- Author: radum -->


<!-- Date: Thu May 19 12:54:26.832 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198276">198276</A>
 Terminated at 2005.05.19 12:54:09
<!-- Author: RunControl -->


<!-- Date: Thu May 19 12:55:02.694 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198275">198275</A>
TERMINATE: ending
<!-- Author: annovi -->


<!-- Date: Thu May 19 12:57:44.766 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198277">198277</A>
 Activated at 2005.05.19 12:57:18
<!-- Author: RunControl -->


<!-- Date: Thu May 19 12:57:45.140 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198277">198277</A>
ACTIVATE: RunSet: COSMICS_SILICON_TAPE  Trigger: COSMICS [15,619,538] 
<!-- Author: radum -->


<!-- Date: Thu May 19 12:57:46.337 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198279">198279</A>
ACTIVATE: RunSet: CALIB_RP_SMQIE 
<!-- Author: radum -->


<!-- Date: Thu May 19 12:59:37.042 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198278">198278</A>
ACTIVATE: RunSet: SVT_B0SVT09_TEST 
<!-- Author: annovi -->


<!-- Date: Thu May 19 12:59:39.371 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198279">198279</A>
TERMINATE: end Roman Pot SMQIE calibrations. Lots of errors in the error log.
<!-- Author: radum -->



<!-- Date: Thu May 19 13:14:37.915 2005 -->
CDF access is over.
<!-- Author: pondrom -->

<!-- Date: Thu May 19 13:16:31.661 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198280">198280</A>
ACTIVATE: RunSet: CALIB_RP_SMQIE 
<!-- Author: radum -->
<!-- Comment: Thu May 19 13;31;50 comment by...Adrian -->
rpa qie callibration for run 198280 is OK.
<!-- Date: Thu May 19 13:20:12.924 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198280">198280</A>
TERMINATE: SMQIE calib
<!-- Author: radum -->



<!-- Date: Thu May 19 13:41:28.793 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198277">198277</A>
 Terminated at 2005.05.19 13:40:46
<!-- Author: RunControl -->


<!-- Date: Thu May 19 13:41:30.760 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198277">198277</A>
TERMINATE: end for calibrations
<!-- Author: radum -->


<!-- Date: Thu May 19 13:42:45.219 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198281">198281</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: radum -->


<!-- Date: Thu May 19 13:45:57.705 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198281">198281</A>
TERMINATE: redo

<!-- Author: radum -->


<!-- Date: Thu May 19 13:45:58.195 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198282">198282</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: radum -->


<!-- Date: Thu May 19 13:47:32.648 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198282">198282</A>
TERMINATE: redo calib
<!-- Author: radum -->


<!-- Date: Thu May 19 13:48:02.388 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198283">198283</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: radum -->



<!-- Date: Thu May 19 13:48:02.221 2005 -->
<table border=0><td><tr><td bgcolor=#e0e030><h2> Silicon Instructions for next store </h2> 
For the next store, Silicon will stay out for the first hour of data taking. We will have a look, if all the problems from this morning habe been fixed, before we give the ok to include Silicon<br> 
Please page Marcel (218-9520)as well at the beginning of pbar load. </td></tr></td></table>
<!-- Author: Marcel &Rong-Shyang -->

<!-- Date: Thu May 19 13:48:48.938 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198283">198283</A>
TERMINATE: end calomrimeter qie
<!-- Author: radum -->


<!-- Date: Thu May 19 13:50:55.603 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198284">198284</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: radum -->


<!-- Date: Thu May 19 13:52:57.605 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198284">198284</A>
TERMINATE: end CEM LED calib
<!-- Author: radum -->



<!-- Date: Thu May 19 14:07:36.267 2005 -->
<table border=0><td><tr><td bgcolor=#e0e030>The production online database has gone down for scheduled  
mantainance.  No database operations, including any DAQ,  
will be possible for the next 1.5 hours.</td></tr></td></table>
<!-- Author: Ray Culbertson -->


<!-- Date: Thu May 19 14:08:43.456 2005 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2005&action=view&page=-8576&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2005&action=view&page=-8576  width='300' align='top'></a></td><td><p>Results of work on xenon flash transition cards during access:</p> 
 
<ul> 
<li>23W - found xenon cable for 23W not well seated. Ran again; thought we had fixed problem, but forgot problem was only PIN diodes, not the PMT readouts. Still a problem; could check fuses next time.</li> 
<li>20E - found one picofuse blown on transition card. Fixed; looks good now.</li> 
<li>01E - found picofuse holder broken on transition card. Thanks to Roberto, holder is fixed. Looks good now.</li> 
 
<p>So, after much climbing and swearing, fixed two more xenon flash problems. Remaining problems: 23W and 16E, the latter being the crate (b0ccal13) with shower max problems.</p></td></tr></table>
<!-- Author: Steve Hahn -->


<!-- Date: Thu May 19 14:38:18.256 2005 -->
Roman pot access: Found a blown 8A fuse for the +5V on the SMD crate in the tunnel.  Replaced it but still got readout errors.  Tried swapping the SMXR but no help.  On the expert advice of Mike Lindgren we went back in the tunnel to see what happened if we unplugged the SMC cable while the crate was powered.  Some combination of unplugging and reseating the SMC cable with the crate on/off/on brought things back to life.  Next time this happens we will know EXACTLY what to do.  (Nice to have you back Mike!)
<!-- Author: convery, terashi -->

<!-- Date: Thu May 19 15:07:28.140 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198285">198285</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [16,623,539] 
<!-- Author: vrusu -->



<!-- Date: Thu May 19 15:21:31.348 2005 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2005&action=view&page=-8577&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2005&action=view&page=-8577  width='300' align='top'></a></td><td>Calibration result for 192284 cemled. Graph looks similar to the reference one, but there are still white dead channels and some brown (hot ones). Is it OK?</td></tr></table>
<!-- Author: Adrian -->

<!-- Date: Thu May 19 15:23:21.631 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198287">198287</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: radum -->


<!-- Date: Thu May 19 15:23:51.083 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198286">198286</A>
ACTIVATE: RunSet: SVT_B0SVT09_TEST 
<!-- Author: pgiova -->



<!-- Date: Thu May 19 15:23:51.629 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-8579&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-8579' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-8578&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-8578' width='300'></a></center>
<!-- Author: Adrian -->
<!-- Comment: Thu May 19 15;24;57 comment by...Adrian -->
Result of the calibration for 198283 calqie seems to have the same problem as the previous one (black and blue colors)

<!-- Comment: Thu May 19 16;43;31 comment by...Kwangzoo -->
<pre><b>22W Ch.1 PHA</b> is a <b>PIN diode</b> and a wide RMS for PIN diode channels can be ignored. <b>2W Ch.10 CHA</b> shows a very wide RMS(even above 200) which could be a signal for a real tragedy. But this channel is corresponding to CCAL00, Slot 21, Ch 10 and <b>it is noisy due to the (PMT) light leak. It is going to behave normally as the post-access effect goes away.</b></pre> 
<!-- Comment: Sun May 22 23;00;46 comment by...Kwangzoo -->
To make it clear; It was my misunderstanding. The light leak PMT should be back to normal as soon as we close up the detector and turn the light off. It turned out that the muon wall light left turned on and it kept the channel noisy. Sorry for the confusion.
<!-- Date: Thu May 19 15:25:26.217 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198287">198287</A>
TERMINATE: abort, restart SMQIE calib.

<!-- Author: radum -->


<!-- Date: Thu May 19 15:26:28.826 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198289">198289</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: radum -->


<!-- Date: Thu May 19 15:27:07.139 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198286">198286</A>
TERMINATE: Ending
<!-- Author: pgiova -->


<!-- Date: Thu May 19 15:29:49.481 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198289">198289</A>
TERMINATE: end SMXQIE calibration
<!-- Author: radum -->
<!-- Comment: Thu May 19 15;52;56 comment by...Adrian -->
Calibration for SMX QIE 198 289 seems OK.
<!-- Date: Thu May 19 15:33:59.264 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198290">198290</A>
ACTIVATE: RunSet: SVT_B0SVT09_TEST 
<!-- Author: pgiova -->


<!-- Date: Thu May 19 15:34:37.321 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198291">198291</A>
ACTIVATE: RunSet: CALIB_CLC_QIE 
<!-- Author: radum -->


<!-- Date: Thu May 19 15:38:24.513 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198291">198291</A>
TERMINATE: end CLC QIE calibrations.
<!-- Author: radum -->


<!-- Date: Thu May 19 15:43:12.423 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198290">198290</A>
TERMINATE: Ending
<!-- Author: pgiova -->


<!-- Date: Thu May 19 15:44:01.251 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198292">198292</A>
ACTIVATE: RunSet: SVT_B0SVT09_TEST 
<!-- Author: pgiova -->


<!-- Date: Thu May 19 15:44:02.236 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198293">198293</A>
ACTIVATE: RunSet: ACE_TOF_QIE_NEW 
<!-- Author: radum -->


<!-- Date: Thu May 19 15:45:58.216 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198293">198293</A>
TERMINATE: end TOF QIE calibration run.
<!-- Author: radum -->



<!-- Date: Thu May 19 15:53:13.115 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198294">198294</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: radum -->


<!-- Date: Thu May 19 15:53:49.245 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198294">198294</A>
TERMINATE: end COT calibration run.
<!-- Author: radum -->


<!-- Date: Thu May 19 15:56:50.289 2005 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198230>
198230</A> 
x30656</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_3_02 [1,639,566]</TD>                                       
<TD ALIGN=CENTER>09:12:29</TD>                                          
<TD ALIGN=CENTER>09:19:24</TD>                                          
<TD ALIGN=CENTER>00:04:02</TD>                                          
<TD ALIGN=RIGHT>      5,608,561</TD>                                    
<TD ALIGN=RIGHT>        73,458</TD>                                     
<TD ALIGN=RIGHT>      18,348</TD>                                       
<TD ALIGN=RIGHT>   19.339</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>15:55:01</TD>                                                       
<TD ALIGN=CENTER>00:04:02</TD>                                          
<TD ALIGN=RIGHT>       5,608,561</TD>                                   
<TD ALIGN=RIGHT>         73,458</TD>                                    
<TD ALIGN=RIGHT>        18,348</TD>                                     
<TD ALIGN=RIGHT>   19.339</TD>                                          
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->


<!-- Date: Thu May 19 15:59:40.327 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198295">198295</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: radum -->


<!-- Date: Thu May 19 15:59:41.041 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198295">198295</A>
TERMINATE: end muon calibrations run.
<!-- Author: radum -->


<!-- Date: Thu May 19 16:02:50.333 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=198296">198296</A>
ACTIVATE: RunSet: TOF_TAC_CALIB 
<!-- Author: Xiaojian x2080 -->



<!-- Date: Thu May 19 16:03:01.771 2005 -->
<font color=red>Shift Summary: </font><pre>Store lost about 9 am, and we went into access.  We were finished about 1 pm.  Stack is now at 76E10, Stash at 14E10. Thunderstorms are on the way. </pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   none
Delivered Luminosity   0  
Acquired Luminosity    0  
Efficiency             0
   
</td></tr></table></pre>
<!-- Author: Pondrom -->