<!-- Title: Owl shift. Sat Feb 11, 2006 -->
<!-- SciCo: Stephan Lammel -->
<!-- DAQAce: Brian Mohr -->
<!-- MonAce: Dean Hidas -->
<!-- CO: Diego Cauz -->
<!-- OpManager: Greg Feild -->
<!-- Notes: -->
<pre>shot setup in progress</pre>


<!-- Date: Sat Feb 11 00:06:49.014 2006 -->
no new CO tonight, I'm doing the shift on my own
<!-- Author: diego :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211533' target=_top>211533</a>) -->


<!-- Date: Sat Feb 11 00:07:58.945 2006 -->
ProcMon gives RTServer error: 
 
2006.02.11 00:02:09 Status: CHECKING 
2006.02.11 00:04:09 Status: CHECK FAILED java.lang.InterruptedException 
Sat Feb 11 00:02:15 CST 2006   procAction  rtserver   check   b0server01 
 
I checked again and this time it was OK.  It seems to have cleared itself.
<!-- Author: Brian :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211533' target=_top>211533</a>) -->


<!-- Date: Sat Feb 11 00:14:22.132 2006 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2006&action=view&page=-2237&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-2237  width='300' align='top'></a></td><td>I also forgot POST-QUENCH CHECKOUT. 
This is the BLM summary plots. And there are no pinky cell in IMON.</td></tr></table>
<!-- Author: Young-Jang -->
<!-- Comment: Sat Feb 11 01;48;21 comment by...Ankush -->
According to the time stamp, you have the wrong BLM summary plot.  According to the previous shift elog the quench occured around<a href=
'http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2006&action=view&page=123&frame=2&anchor=23:42:15.233'>
22:19 10 Feb 2006</a>.  The time stamp for this is 00:05:06 11 Feb 06.  To get the correct ione, goto to ACNET page E11 and choose "select date" or an option similar to that.  If you find an entry around the quench then paste that plot in.  If you can&#39t find an entry then the BLMs never latched and no data was recorded => This was a very clean quench for Silicon.</p>
<!-- Comment: Sat Feb 11 02;09;13 comment by...Ankush -->
<p>Ooops....you should goto ACNET page E2 to ge the BLM plots(!!)</p>
<!-- Comment: Sat Feb 11 02;12;12 comment by...Dean -->
Got it.  Directions for how I did it are below along with the (hopefully) correct plot.
<!-- Date: Sat Feb 11 00:41:47.919 2006 -->
final protons loaded
<!-- Author: Stephan Lammel :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211533' target=_top>211533</a>) -->


<!-- Date: Sat Feb 11 00:42:59.681 2006 -->
<TABLE bgcolor=#cdc9a5> 
<TR><TH>Date</th><th>Time</th><th>BLM</th><th colspan=2>Dose</th></tr> 
<tr><td>2006.02.11</td><td>00:39:03</td><td>W Inner BLM</td><td>0.79</td><td>RADS</td></tr> 
<tr><td>2006.02.11</td><td>00:39:03</td><td>W Outer BLM</td><td>0.00</td><td>RADS</td></tr> 
<tr><td>2006.02.11</td><td>00:39:03</td><td>E Inner BLM</td><td>0.66</td><td>RADS</td></tr> 
<tr><td>2006.02.11</td><td>00:39:03</td><td>E Outer BLM</td><td>1.87</td><td>RADS</td></tr> 
</TABLE> 
<strong>Integrated dosage</strong>
<!-- Author: Dean -->


<!-- Date: Sat Feb 11 01:12:30.187 2006 -->
final anti-protons loaded
<!-- Author: Stephan Lammel :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211533' target=_top>211533</a>) -->


<!-- Date: Sat Feb 11 01:25:00.678 2006 -->
Busy timeout in vrb/evb.  Required two auto HRRs to clear.
<!-- Author: Brian :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211533' target=_top>211533</a>) -->


<!-- Date: Sat Feb 11 01:31:33.052 2006 -->
Trigger Manager Error:  Events Pending Timeout.  Auto HRR.
<!-- Author: Brian :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211533' target=_top>211533</a>) -->
<!-- Comment: Sat Feb 11 01;37;44 comment by...Brian -->
Again.
<!-- Date: Sat Feb 11 01:34:15.562 2006 -->
done with scraping, store 4638
<!-- Author: Stephan Lammel :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211533' target=_top>211533</a>) -->


<!-- Date: Sat Feb 11 01:39:20.856 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211533">211533</A>
TERMINATE: End shotsetup.
<!-- Author: sasha x2080 -->



<!-- Date: Sat Feb 11 01:41:39.962 2006 -->
<pre>store 4638: protons:       70.9 E10 
            anti-protons:  19.6 E10 
            luminosity:    150 E30 
            C:LOSTP        550 Hz 
            C:LOSTPB       670 Hz 
            C:B0PAGC	   130 Hz</pre>
<!-- Author: Stephan Lammel :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211533' target=_top>211533</a>) -->


<!-- Date: Sat Feb 11 01:42:41.518 2006 -->
L3 Relay Failure when trying to end the run.  Two crates b0svt09 and b0svt08 in error.  (Note, not necessary to page L3 immediately unless > 2 crates in error.)
<!-- Author: Brian :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211533' target=_top>211533</a>) -->



<!-- Date: Sat Feb 11 01:53:49.582 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2238&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2238' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2239&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2239' width='300'></a></center>
<!-- Author: Dean -->
<!-- Comment: Sat Feb 11 01;54;43 comment by...Dean -->
We got an integrated radiation dose alarm (svx) right after flattop.  Here are the integrated and inst plots for the BLMs

<!-- Comment: Sat Feb 11 02;00;00 comment by...Ulrich (RadCo pager) -->
The west (proton) side BLMs saw between 20 rad/min and 30 rad/min during the squeeze. The alarm limit is 19 rad/min. We had very similar problems in December 2005.


<!-- Comment: Sat Feb 11 02;15;17 comment by...Dean -->
D0 also got radiation alarms.  see the MCR <a href="http://www-bd.fnal.gov/cgi-mcr/elog.pl?nb=2006&action=view&page=112&scroll=false">elog</a>
<!-- Comment: Sat Feb 11 02;19;19 comment by...Dean -->
People informed about the radiation alarm: silicon main pager, silicon SPL (ankush), radco, and MCR.
<!-- Date: Sat Feb 11 02:01:39.802 2006 -->
<h3>Wrong Post-Quench BLM plot in e-log</h3> 
 
<p>I was just looking at the  
<a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2006&action=view&page=124&frame=2&anchor=00:14:22.132'> 
BLM plots from the quench</a>.  Unfortnately these are are wrong ones.  The quench occured around 22:19 10 Feb 2006. The BLM plots are from 00:05:06 11 Feb 2006.  You&#39ll find instructions on how to get the correct plot as comments to the original posting. Sorry for the hassle as the previous shift crew should have done this for you.</p> 
 
  
<!-- Author: Ankush -->
<!-- Comment: Sat Feb 11 02;07;04 comment by...Dean -->
I think I have the correct plot which I'll post below along with how I got it.
<!-- Date: Sat Feb 11 02:04:40.424 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211534">211534</A>
 Activated at 2006.02.11 02:04:28
<!-- Author: RunControl -->


<!-- Date: Sat Feb 11 02:04:41.588 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211534">211534</A>
ACTIVATE: RunSet: AAA_NOSILICON  Trigger: TEST_TWO_TRK1.5 [12,768,600] 
<!-- Author: Brian x2080 -->




<!-- Date: Sat Feb 11 02:07:39.520 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2240&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2240' width='300'></a></center>
<!-- Author: Dean -->
<!-- Comment: Sat Feb 11 02;09;52 comment by...Dean -->
BLM summary for silicon post quench ckeckout.
<!-- Comment: Sat Feb 11 02;11;08 comment by...Dean -->
To get an old BLM summary go to E2, select "Display Logged Data", then "List Dates".  From here you can choose a date/time.  Then "Plot FIFO".
<!-- Comment: Sat Feb 11 02;13;56 comment by...Dean -->
Ah wait, you don't have to hit "Plot FIFO", it'll just come up on it's own after you select the date, choose "BLM Summary"
<!-- Comment: Sat Feb 11 02;17;30 comment by...Ankush -->
Thanks Dean :-)

So Silicon did receive some non-zero radiation during the last qunech. But nothing worth crying over....considering how much we just received as part of the integrated dose alarm(!!)
<!-- Date: Sat Feb 11 02:09:01.582 2006 -->
Setup for special COT run as per instructions <a href=http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2006&action=copyentry&page=123&time=16:35:58.667>here</a>.    Now running with prescale set to 2000.
<!-- Author: Brian :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211534' target=_top>211534</a>) -->




<!-- Date: Sat Feb 11 02:11:01.156 2006 -->
<pre>Output of script to lower the thresholds: 
 
b0dap73:461:~&gt; setcv -f lowerStereoASDQ_script 
    File 1 (lowerStereoASDQ_script) opened  
 SL1-A:  DTH(02) = +0225 mV  
 SL1-B:  DTH(02) = +0225 mV  
 SL1-C:  DTH(02) = +0225 mV  
 SL1-D:  DTH(02) = +0225 mV  
 SL2-A:  DTH(02) = +0205 mV  
 SL2-B:  DTH(02) = +0205 mV  
 SL2-C:  DTH(02) = +0205 mV  
 SL2-D:  DTH(02) = +0205 mV  
 SL3-A:  DTH(08) = +0205 mV  
 SL3-B:  DTH(08) = +0205 mV  
 SL3-C:  DTH(08) = +0205 mV  
 SL3-D:  DTH(08) = +0205 mV  
 SL4-A:  DTH(08) = +0205 mV  
 SL4-B:  DTH(08) = +0205 mV  
 SL4-C:  DTH(08) = +0205 mV  
 SL4-D:  DTH(08) = +0205 mV  
 SL5-A:  DTH(14) = +0205 mV  
 SL5-B:  DTH(14) = +0205 mV  
 SL5-C:  DTH(14) = +0205 mV  
 SL5-D:  DTH(14) = +0205 mV  
 SL6-A:  DTH(14) = +0205 mV  
 SL6-B:  DTH(14) = +0205 mV  
 SL6-C:  DTH(14) = +0205 mV  
 SL6-D:  DTH(14) = +0205 mV  
 SL7-A:  DTH(20) = +0205 mV  
 SL7-B:  DTH(20) = +0205 mV  
 SL7-C:  DTH(20) = +0205 mV  
 SL7-D:  DTH(20) = +0205 mV  
 SL8-A:  DTH(20) = +0205 mV  
 SL8-B:  DTH(20) = +0205 mV  
 SL8-C:  DTH(20) = +0205 mV  
 SL8-D:  DTH(20) = +0205 mV  
 SL1-A:  DTH(02) = +0225 mV  
 SL1-B:  DTH(02) = +0225 mV  
 SL1-C:  DTH(02) = +0225 mV  
 SL1-D:  DTH(02) = +0225 mV  
 SL2-A:  DTH(02) = +0205 mV  
 SL2-B:  DTH(02) = +0205 mV  
 SL2-C:  DTH(02) = +0205 mV  
 SL2-D:  DTH(02) = +0205 mV  
 SL3-A:  DTH(08) = +0205 mV  
 SL3-B:  DTH(08) = +0205 mV  
 SL3-C:  DTH(08) = +0205 mV  
 SL3-D:  DTH(08) = +0205 mV  
 SL4-A:  DTH(08) = +0205 mV  
 SL4-B:  DTH(08) = +0205 mV  
 SL4-C:  DTH(08) = +0205 mV  
 SL4-D:  DTH(08) = +0205 mV  
 SL5-A:  DTH(14) = +0205 mV  
 SL5-B:  DTH(14) = +0205 mV  
 SL5-C:  DTH(14) = +0205 mV  
 SL5-D:  DTH(14) = +0205 mV  
 SL6-A:  DTH(14) = +0205 mV  
 SL6-B:  DTH(14) = +0205 mV  
 SL6-C:  DTH(14) = +0205 mV  
 SL6-D:  DTH(14) = +0205 mV  
 SL7-A:  DTH(20) = +0205 mV  
 SL7-B:  DTH(20) = +0205 mV  
 SL7-C:  DTH(20) = +0205 mV  
 SL7-D:  DTH(20) = +0205 mV  
 SL8-A:  DTH(20) = +0205 mV  
 SL8-B:  DTH(20) = +0205 mV  
 SL8-C:  DTH(20) = +0205 mV  
 SL8-D:  DTH(20) = +0205 mV  
    File 1 (lowerStereoASDQ_script) closed  
</pre>
<!-- Author: Brian :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211534' target=_top>211534</a>) -->






<!-- Date: Sat Feb 11 02:16:46.960 2006 -->
SVX trip: B2 W9 tripped from standby to off.  Reset trips and put back ion standby. (we are running without silicon right now for COT HV tests)
<!-- Author: Dean :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211534' target=_top>211534</a>) -->
<!-- Comment: Sat Feb 11 02;24;03 comment by...Ankush -->
Unfortunatly these trips will persist until we able to swap the power supply.  We did try earlier but not successful.
<!-- Comment: Sat Feb 11 02;27;40 comment by...Dean -->
Oscar called and suggested we do a coldstart since we have all of the silicon crates free at the moment (COT tests).  coldstart went through config with no problems.  The trip was a software protection trip: high DVDD.
<!-- Date: Sat Feb 11 02:19:18.604 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211534">211534</A>
 Terminated at 2006.02.11 02:18:06
<!-- Author: RunControl -->


<!-- Date: Sat Feb 11 02:19:19.823 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211534">211534</A>
TERMINATE: EVB busy timeouts.
<!-- Author: Brian x2080 -->




<!-- Date: Sat Feb 11 02:19:36.621 2006 -->
Several busy timeouts (EVB).  HRR did not recover.  Ending run and will restart.  Note evb proxy looks OK when checked.
<!-- Author: Brian :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211534' target=_top>211534</a>) -->
<!-- Comment: Sat Feb 11 02;21;57 comment by...Brian -->
EVB seems to be happy with new run.
<!-- Date: Sat Feb 11 02:21:26.462 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211535">211535</A>
 Activated at 2006.02.11 02:20:45
<!-- Author: RunControl -->


<!-- Date: Sat Feb 11 02:21:27.590 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211535">211535</A>
ACTIVATE: RunSet: AAA_NOSILICON  Trigger: TEST_TWO_TRK1.5 [12,768,600] 
<!-- Author: Brian x2080 -->
<!-- Comment: Sat Feb 11 02;24;14 comment by...Brian -->
Note, I ran the script to set the thresholds when starting this run.  The results were the same as for the previous run.

<!-- Comment: Sat Feb 11 02;31;58 comment by...Brian -->
But I forgot to set the prescales for this run, they were still at 500.
<!-- Date: Sat Feb 11 02:25:26.895 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211535">211535</A>
 Terminated at 2006.02.11 02:25:05
<!-- Author: RunControl -->


<!-- Date: Sat Feb 11 02:26:16.592 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211535">211535</A>
TERMINATE: We now should have about 100k events between the two runs.
<!-- Author: Brian x2080 -->




<!-- Date: Sat Feb 11 02:32:44.562 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211536">211536</A>
 Activated at 2006.02.11 02:32:36
<!-- Author: RunControl -->


<!-- Date: Sat Feb 11 02:34:04.937 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211536">211536</A>
ACTIVATE: RunSet: AAA_NOSILICON  Trigger: TEST_TWO_TRK1.5 [12,768,600] Nominal COT voltage settings
<!-- Author: Brian x2080 -->
<!-- Comment: Sat Feb 11 02;35;15 comment by...Brian -->
Ran script to lower thresholds and also set the prescale to 2000 before activating this run.


<!-- Date: Sat Feb 11 02:38:07.091 2006 -->
SVX trip:  single ladder this time: 
 
Sat Feb 11 02:33:49 CST 2006                        success: Channel2 SVX B1 W10 to STANDBY 
Sat Feb 11 02:33:58 CST 2006                        TRIP: SVX  B1  W10 DVDD2 OverCurrent 
 
I put L2 back to standby.
<!-- Author: Dean :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211536' target=_top>211536</a>) -->
<!-- Comment: Sat Feb 11 02;40;27 comment by...Dean -->
I tried looking it up in the CAEN PS converter:

SB1W9 works fine but SB1W10 gives:

Wedge/CAEN Converter

HMM... I don't know this wedge Go back
<!-- Comment: Sat Feb 11 02;42;16 comment by...Dean -->
Again.  I'll reset this one, go through a coldstart again.  Since you silicon people seem to be reading this can you tell me if this is the same problem as SB2W9 or not?  I'll page next time, but I suppose we'll talk to the silicon main pager in 10 miutes anyway to include silicon.
<!-- Comment: Sat Feb 11 03;01;32 comment by...Dean -->
Contacted Oscar about this.  We first tried turning the whole wedge off and back to standby.  This didn't work due to a handshake error and DVDD0 overcurrent.  Then we turned the PC to on, and then each L0-4 individually.  This worked.  We then set the wedge to standby.  Ran a coldstart and everything came through config fine.
<!-- Date: Sat Feb 11 02:38:38.729 2006 -->
Busy Timeout in evb, followed by several more also pending timeout error.  Will clean up evb before starting the next run.  A few HRRs seem to have solved this, hopefully we can collect enough events to satisfy COT.
<!-- Author: Brian :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211536' target=_top>211536</a>) -->



<!-- Date: Sat Feb 11 02:48:59.658 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211536">211536</A>
 Terminated at 2006.02.11 02:48:50
<!-- Author: RunControl -->


<!-- Date: Sat Feb 11 02:49:19.097 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211536">211536</A>
TERMINATE: Too many busy timeout's will start new run.
<!-- Author: Brian x2080 -->


<!-- Date: Sat Feb 11 02:51:28.671 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211537">211537</A>
 Activated at 2006.02.11 02:51:19
<!-- Author: RunControl -->


<!-- Date: Sat Feb 11 02:53:01.676 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211537">211537</A>
ACTIVATE: RunSet: AAA_NOSILICON  Trigger: TEST_TWO_TRK1.5 [12,768,600] Starting new run to collect ~20k events.  Set thresholds and prescale to 1000 this time.  (Rate is 214 Hz, was only 100 Hz before.)
<!-- Author: Brian x2080 -->
<!-- Comment: Sat Feb 11 02;53;42 comment by...Brian -->
Had to set prescale to 2000.  L3 was almost completely green, but recovered with higher prescale.

<!-- Comment: Sat Feb 11 02;54;53 comment by...Brian -->
Rate is 150 Hz.

<!-- Date: Sat Feb 11 02:55:27.460 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211537">211537</A>
 Terminated at 2006.02.11 02:54:58
<!-- Author: RunControl -->


<!-- Date: Sat Feb 11 02:55:28.518 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211537">211537</A>
TERMINATE: Finished collecting COT trigger test events.
<!-- Author: Brian x2080 -->



<!-- Date: Sat Feb 11 02:55:42.263 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2241&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2241' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2242&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2242' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2243&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2243' width='300'></a></center>
<!-- Author: Dean -->


<!-- Date: Sat Feb 11 02:57:13.363 2006 -->
Cleaning up EVB before starting physics runs.  Hopefully this will clear up the problems we have been having.
<!-- Author: Brian -->
<!-- Comment: Sat Feb 11 03;18;10 comment by...Brian -->
Note I actually did the cleanup twice.  The first time the evb proxy was not running on partition 0.  After the second time everything looked OK.
<!-- Date: Sat Feb 11 02:57:22.635 2006 -->
Losses look good.  Just talked to Oscar about the SVX trips and including silicon.  Ramping silicon now.
<!-- Author: Dean :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211537' target=_top>211537</a>) -->


<!-- Date: Sat Feb 11 03:03:28.577 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211538">211538</A>
 Activated at 2006.02.11 03:03:12
<!-- Author: RunControl -->


<!-- Date: Sat Feb 11 03:03:29.950 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211538">211538</A>
ACTIVATE: RunSet: AAA_CURRENT  Trigger: PHYSICS_4_00 [1,783,616] 
<!-- Author: Brian x2080 -->
<!-- Comment: Sat Feb 11 03;04;03 comment by...Brian -->
<pre>This is now a standard phyics run.  Here is the output of the check of the thresholds:

b0dap73:465:~/setcv&gt; setcv -f readBackAll
    File 1 (readBackAll) opened 
 SL1-A:  DTH(02) = +0225 mV 
 SL1-B:  DTH(02) = +0225 mV 
 SL1-C:  DTH(02) = +0225 mV 
 SL1-D:  DTH(02) = +0225 mV 
 SL2-A:  DTH(02) = +0205 mV 
 SL2-B:  DTH(02) = +0205 mV 
 SL2-C:  DTH(02) = +0205 mV 
 SL2-D:  DTH(02) = +0205 mV 
 SL3-A:  DTH(08) = +0225 mV 
 SL3-B:  DTH(08) = +0225 mV 
 SL3-C:  DTH(08) = +0225 mV 
 SL3-D:  DTH(08) = +0225 mV 
 SL4-A:  DTH(08) = +0205 mV 
 SL4-B:  DTH(08) = +0205 mV 
 SL4-C:  DTH(08) = +0205 mV 
 SL4-D:  DTH(08) = +0205 mV 
 SL5-A:  DTH(14) = +0225 mV 
 SL5-B:  DTH(14) = +0225 mV 
 SL5-C:  DTH(14) = +0225 mV 
 SL5-D:  DTH(14) = +0225 mV 
 SL6-A:  DTH(14) = +0205 mV 
 SL6-B:  DTH(14) = +0205 mV 
 SL6-C:  DTH(14) = +0205 mV 
 SL6-D:  DTH(14) = +0205 mV 
 SL7-A:  DTH(20) = +0225 mV 
 SL7-B:  DTH(20) = +0225 mV 
 SL7-C:  DTH(20) = +0225 mV 
 SL7-D:  DTH(20) = +0225 mV 
 SL8-A:  DTH(20) = +0205 mV 
 SL8-B:  DTH(20) = +0205 mV 
 SL8-C:  DTH(20) = +0205 mV 
 SL8-D:  DTH(20) = +0205 mV 
    File 1 (readBackAll) closed 
</pre>

<!-- Date: Sat Feb 11 03:07:03.585 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211538">211538</A>
 Terminated at 2006.02.11 03:06:40
<!-- Author: RunControl -->


<!-- Date: Sat Feb 11 03:07:46.042 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211538">211538</A>
TERMINATE: Tons of busy timeouts.  Paging L3.
<!-- Author: Brian x2080 -->



<!-- Date: Sat Feb 11 03:09:31.838 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2244&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2245' ></a></center>
<!-- Author: Dean -->
<!-- Comment: Sat Feb 11 03;10;45 comment by...Dean -->
imon pinky.  We're dealing with L3 issues right now so we'll see if this clears once we config it or if it's going to be a while maybe we can release the crates and I can do a config on this side.

<!-- Comment: Sat Feb 11 03;18;09 comment by...Oscar -->
This pinky was there in the evening. Ankush and Sasha recover it by powercycling the ladder (i.e. go to off and then to on again). You may try if you have a chance. I will keep an eye on it and probably check it in the morning.<br>
Please, put in the log the evolution of this current (i.e. post any plot whene there are big jumps in the current).
<!-- Comment: Sat Feb 11 03;32;18 comment by...Dean -->
The current looks to be back up.  Took a bit to get a HRR in due to front end crate problems that the L3 people were helping sort out.  I'll post the plot below.
<!-- Date: Sat Feb 11 03:13:22.660 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211539">211539</A>
 Activated at 2006.02.11 03:12:52
<!-- Author: RunControl -->


<!-- Date: Sat Feb 11 03:13:23.425 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211539">211539</A>
ACTIVATE: RunSet: AAA_CURRENT  Trigger: PHYSICS_4_00 [1,783,616] 
<!-- Author: Brian x2080 -->
<!-- Comment: Sat Feb 11 03;20;46 comment by...Brian -->
I cleaned up EVB one more time while L3 pager was responding.


<!-- Date: Sat Feb 11 03:22:52.946 2006 -->
Khaldoun responded to L3 page.  After checking the log files, he found SCPU 11 has repeated error messages (load?) and thinks the problem is with a front end crate rather than evb.  He will try to determine which crate this is and call back.  Meanwhile, we are halted.
<!-- Author: Brian :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211539' target=_top>211539</a>) -->
<!-- Comment: Sat Feb 11 03;30;59 comment by...Brian -->
Khaldoun suggested we are having problems with the crate svt09.  Right now, after shepherding pcal04, we are running, so will page svt unless we see more evb errors and busy timeouts.
<!-- Comment: Sat Feb 11 03;33;40 comment by...Brian -->
Wishful thinking.  We are still having problems with busy timeouts after ~2 minutes of running.
<!-- Date: Sat Feb 11 03:25:53.107 2006 -->
While halted b0pcal04 gave an error.  Shepherding.
<!-- Author: Brian :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211539' target=_top>211539</a>) -->
<!-- Comment: Sat Feb 11 03;29;41 comment by...Brian -->
After successful shepherd, this crate had a heap corrupt error.  Shepherded again, this time reached clear state.  Returned to running and so far no problems.





<!-- Date: Sat Feb 11 03:33:44.616 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2246&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2247' ></a></center>
<!-- Author: Dean -->
<!-- Comment: Sat Feb 11 03;34;31 comment by...Dean -->
Finaly got to an HRR and it looks to be back within the limits.  Took a bit due to L3/front end crate trouble.


<!-- Comment: Sat Feb 11 03;37;03 comment by...Oscar -->
Thanks!!!
<!-- Date: Sat Feb 11 03:36:11.660 2006 -->
<pre>Regarding the repeated EVB errors:  scpu 11 repeatedly complains about data from slot 10, channel 9, which is svt09 if i am not  mistaken.  See error msg at the bottom of this entry. 
 
In the past, either excluding this crate or shepherding it could resolve this issue.  I am not in a position to decide whether this is doable at the moment, but at least I believe that it is the source of the errors thrown by the EVB. 
 
Error message from scpu logs: 
 
 
2006/02/11-03:06:32 (2/11) :Warning at event 37513, vrb slot 10, channel 9, byte count 19120: Third trailer word not equal 0x80000000. 
2006/02/11-03:06:32 (2/11) :LoadError at event 37513, vrb slot 10, channel 9, byte count 19120: Tracer words inconsistant. 
2006/02/11-03:06:32 (2/11) :LoadError at event 37513, vrb slot 10, channel 9, byte count 19120: First trailer word not equal 0xaaaaaaaa. 
2006/02/11-03:06:32 (2/11) :LoadError at event 37513, vrb slot 10, channel 9, byte count 19120: Second trailer word not equal 0x55555555. 
</pre>
<!-- Author: Khaldoun :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211539' target=_top>211539</a>) -->



<!-- Date: Sat Feb 11 03:37:57.656 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2248&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2248' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2249&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2249' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2250&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2250' width='300'></a></center>
<!-- Author: Dean -->


<!-- Date: Sat Feb 11 03:48:41.645 2006 -->
Paged SVT.  Taichi responded -- could not find errors with svt09.  He suggested shepherding (which we should have done already).  So far we are running.
<!-- Author: Brian :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211539' target=_top>211539</a>) -->


<!-- Date: Sat Feb 11 03:57:10.018 2006 -->
Still having busy timeout problems.  Paged SVT again.  Taichi responded, is looking into it and will call back.  Meanwhile, we are halted.
<!-- Author: Brian :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211539' target=_top>211539</a>) -->


<!-- Date: Sat Feb 11 04:11:22.673 2006 -->
Stephan is on phone with Alberto (SVT).  Shepherd svt09 one more time and are currently running.
<!-- Author: Brian :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211539' target=_top>211539</a>) -->
<!-- Comment: Sat Feb 11 04;14;25 comment by...Brian -->
Alberto suggests to page DAQ.  This problem has happened recently and DAQ somehow managed to fix it.  Meanwhile we have more busy timeouts and I have halted.

<!-- Date: Sat Feb 11 04:20:11.821 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211533">211533</A>
RUNSTATUS: <PRE>Marked Bad, explanation: shot-setup exercise
CLC 
L1T 
L2T 
L3T 
CCAL 
PCAL 
COT 
CMU 
CMP 
CMX 
IMU 
CSP 
CSX 
ISU 
SVX 
ISL 
L00 
SVT 
CSMX 
PSMX 
TOF 
MNP 
BSC 
RPS 
RSF 
</PRE>
<!-- Author: cdfscico -->


<!-- Date: Sat Feb 11 04:20:39.380 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211534">211534</A>
RUNSTATUS: <PRE>Marked Bad, explanation: without silicon
SVX 
ISL 
L00 
SVT 
</PRE>
<!-- Author: cdfscico -->


<!-- Date: Sat Feb 11 04:21:57.147 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211536">211536</A>
RUNSTATUS: <PRE>Marked Bad, explanation: without silicon
SVX 
ISL 
L00 
SVT 
</PRE>
<!-- Author: cdfscico -->


<!-- Date: Sat Feb 11 04:23:42.096 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211537">211537</A>
RUNSTATUS: <PRE>Marked Bad, explanation: without silicon
SVX 
ISL 
L00 
SVT 
</PRE>
<!-- Author: cdfscico -->



<!-- Date: Sat Feb 11 04:32:44.330 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2251&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2251' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2252&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2252' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2253&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2253' width='300'></a></center>
<!-- Author: Dean -->


<!-- Date: Sat Feb 11 04:40:33.082 2006 -->
Paged DAQ, Ray responded.  He rebooted svt09 and had me shepherd without reset.  We resumed running, but only fow a few minutes and then have busy timeouts again.  He is looking into the tracer word problems Kahldoun posted.
<!-- Author: Brian :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211539' target=_top>211539</a>) -->


<!-- Date: Sat Feb 11 05:02:48.406 2006 -->
Ray called back and said he could not find any illuminating error messages in the logs.  He suggested to power cycle the crate and then shepherd without reset.  We paged Alberto to make sure this was OK and to see if he had special instructions.  We are now running, hopefully for longer than 3 minutes.
<!-- Author: Brian :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211539' target=_top>211539</a>) -->
<!-- Comment: Sat Feb 11 05;33;04 comment by...Ray Culbertson -->
<pre>The power cycle seemed to fix the problem for now.
In the process I noticed a few things:
- svt09 spews quite a bit (much more than 08) 
which makes it hard to log on and do anything.
- at reboot, there is a vme map of the crate
printed which is very different from the hardware db
listing of what is in the crate.
- we used to get tracer word errors in the error log,
I think this was restricted to speed up scpu readout.
These errors are apparently in the evb log, could an evb
person let me know how to access this log?
Or better, link it to the run's web page with other logs?
- the error Khaldoun posted does not have the trace
word content, that error could dump several words
surrounding the buffer end
</pre>


<!-- Date: Sat Feb 11 05:51:54.319 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2254&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2254' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2255&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2255' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2256&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2256' width='300'></a></center>
<!-- Author: Dean -->


<!-- Date: Sat Feb 11 06:03:45.462 2006 -->
Stephan has paged the CAL SPL experts, because CHA PMT Plots shows 4 new DEAD channel, 2 in the same tower. 
Willis responded that nothing can be done at the moment. 
At 8 AM he will also page CAL ADMEM experts.
<!-- Author: diego :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211539' target=_top>211539</a>) -->
<!-- Comment: Sat Feb 11 06;04;32 comment by...diego -->
'he' is for Stephan.


<!-- Date: Sat Feb 11 06:11:45.713 2006 -->
<center><a href='/elog/notebooks/2006/Sat_Feb_11_2006_Owl.061118.ps'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2258' ></a><a href='/elog/notebooks/2006/Sat_Feb_11_2006_Owl.061129.ps'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2260' ></a></center>
<!-- Author: Brian -->
<!-- Comment: Sat Feb 11 06;13;20 comment by...Brian -->
DQMon gives error for CHA -- has found the new dead channels in phi wedge 10.


<!-- Date: Sat Feb 11 06:17:27.045 2006 -->
SVX B2 W9 trip.  reset trips and set to on.  This was software protection trip High DVDD.  L1 looks like it went to 294.0 where the limit is 150.0.  Also, AVDD0 Overcurrent.
<!-- Author: Dean :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211539' target=_top>211539</a>) -->


<!-- Date: Sat Feb 11 06:37:07.407 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2261&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2262' ></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2263&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2264' ></a></center>
<!-- Author: Dean -->
<!-- Comment: Sat Feb 11 06;40;40 comment by...Dean -->
Two pinkies.  Both went away.  one on right was only above the limit for a short time.  Left plot looks to be about 10 below the current limit.


<!-- Date: Sat Feb 11 06:45:52.624 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2265&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2265' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2266&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2266' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2267&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2267' width='300'></a></center>
<!-- Author: Dean -->

<!-- Date: Sat Feb 11 06:51:53.283 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211539">211539</A>
cal: <tt> - CHA PMT Plots shows 4 new DEAD channel, 2 in the same tower<br>
 - PHA Tower Plots show > 12 HOT channels, but this is probably due to high luminosity<br>
 - L2 table Beam still read, but < than 0.05, so it is OK, see known problems list</tt>
<!-- Author: Consumer Operator( Diego Cauz) -->


<!-- Date: Sat Feb 11 06:51:54.681 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211539">211539</A>
trigger: <tt>TrigMon XFTTrigger Monitor2 again the known problem. Error rate is 0.4% < 0.5%</tt>
<!-- Author: Consumer Operator( Diego Cauz) -->


<!-- Date: Sat Feb 11 06:51:55.622 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211539">211539</A>
monitors: <tt>XMon: <br>
 trigger L2_PS100_L1_CEM8_PT8_v2 is out of limits most of the time<br>
 trigger EXPRESS_Z_v7 is out for high lum</tt>
<!-- Author: Consumer Operator( Diego Cauz) -->
<!-- Comment: Sat Feb 11 06;56;59 comment by...diego -->
you can see the XMon plots 
 <a href= 'http://www-cdfonline.fnal.gov/internal/mon/cdfdaq/XMon/TriggersOfInterest/images_1139661816/00.html'> here </a>
<!-- Date: Sat Feb 11 06:54:08.433 2006 -->
Done timeout for b0cot09.  Also TDC not ready global readout error for this crate.  HRR recovered.
<!-- Author: Brian :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211539' target=_top>211539</a>) -->



<!-- Date: Sat Feb 11 07:10:19.900 2006 -->
Halted run so Calorimeter Admem expert (Mark) could log in to the crates and try to resolve the issue with CHA phi wedge 10.  He had me shepherd ccal11 and ccal09 when he was finished.  We have resumed running.
<!-- Author: Brian :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211539' target=_top>211539</a>) -->
<!-- Comment: Sat Feb 11 07;21;29 comment by...m mattson (ADMEM) -->
<br>Responded to the Cal ADMEM page.  As the CO noted, there are four dead channels for CHA phi 10E eta 0-2.  Running the QIETEST programs, the symptoms are consistent with a blown fuse on the CAFE cards for b0ccal11, slot 19, channels 1-4.  (Ch 0 and 5 are okay.)  I do not see any problems for b0ccal09, which handles CHA phi 5E, eta 4 (the other dead channel, next to the Chimney).  I will check with the Cal SPLs if this is urgent enough to ask for an access, or if we can limp along until the extended shutdown.<p>
Short version: 4 out of 5 dead channels can be accounted for by electronics failure.  5th is unknown (possibly PMT/base related).

<!-- Comment: Sat Feb 11 07;30;51 comment by...m mattson (ADMEM) -->
<p>My mistake on CHA 5E, the dead zone was always 4 etas wide.  For some reason, I was thinking 3.<p>
I was checking the consumer slides.  First thing I noticed was that YMon was not running during any of the Cosmics.  I thought YMon and TrigMon were supposed to be running between stores.<p>
Second thing I noticed was that the previous run (211463, Friday from 8am - 2pm), the same 4 channels showed low (but not zero) occupancy.  So most likely the CAFE cards blew a fuse sometime during that run.


<!-- Date: Sat Feb 11 07:34:42.457 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2268&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2268' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2269&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2269' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2270&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2270' width='300'></a></center>
<!-- Author: Dean -->


<!-- Date: Sat Feb 11 07:35:15.615 2006 -->
RPS alarm from ifix.  cleared even before I could get the info window open.
<!-- Author: Dean :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211539' target=_top>211539</a>) -->

<!-- Date: Sat Feb 11 07:42:07.467 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211539">211539</A>
cal: <tt> - CHA problem still there, 4 dead channels. Need intervention.<br>
 - PHA hot channels still there, but OK</tt>
<!-- Author: Consumer Operator( Diego Cauz) -->


<!-- Date: Sat Feb 11 07:42:08.304 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211539">211539</A>
trigger: <tt>XFT FINDER still complaining, slide 1.5.1.2</tt>
<!-- Author: Consumer Operator( Diego Cauz) -->


<!-- Date: Sat Feb 11 07:42:09.420 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211539">211539</A>
monitors: <tt>ObjectMon: all slides OK, except slide Met Monitor 'Met>35' and 'Met>45' with low statistics</tt>
<!-- Author: Consumer Operator( Diego Cauz) -->
<!-- Comment: Sat Feb 11 07;44;07 comment by...diego -->
check XMon plots  <a href= 'http://www-cdfonline.fnal.gov/internal/mon/cdfdaq/XMon/TriggersOfInterest/images_1139664947/00.html'>here</a>


<!-- Date: Sat Feb 11 07:49:34.159 2006 -->
<font color=red>Shift Summary: </font><pre>store 4638 initial lumi 1.5 E32 
XFT/COT high voltage test completed (Aseet and Bob came in) 
trouble with SVT-9 at begin of run, until crate was power cycled 
new CHA dead channels 
</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   211533, 211534, 211536, 211537, 211538, 211539
Delivered Luminosity   2178.5 nb-1  
Acquired Luminosity    1021.1 nb-1  
Efficiency             46.9
   
</td></tr></table></pre>
<!-- Author: Stephan Lammel -->

<!-- Date: Sat Feb 11 07:55:49.692 2006 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>Physics able</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L1Accepts</TH>
<TH>L2Accepts</TH>
<TH>L3Accepts</TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211533 target=_blank>211533</A> x33A4D</TD>
<TD>DAQTEST</TD>
<TD><FONT SIZE=-1>PHYSICS_4_00 [1,783,616]</FONT></TD>
<TD ALIGN=CENTER>21:52:49</TD>
<TD ALIGN=CENTER>01:38:02</TD>
<TD ALIGN=CENTER>03:41:51</TD>
<TD ALIGN=RIGHT>      2,418,771</TD>
<TD ALIGN=RIGHT>       210,879</TD>
<TD ALIGN=RIGHT>      87,410</TD>
<TD ALIGN=RIGHT>   24.449</TD>
<TD ALIGN=CENTER>0</TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211534 target=_blank>211534</A> x33A4E</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>TEST_TWO_TRK1.5 [12,768,600]</FONT></TD>
<TD ALIGN=CENTER>02:04:28</TD>
<TD ALIGN=CENTER>02:18:06</TD>
<TD ALIGN=CENTER>00:07:56</TD>
<TD ALIGN=RIGHT>         60,974</TD>
<TD ALIGN=RIGHT>        60,950</TD>
<TD ALIGN=RIGHT>      60,844</TD>
<TD ALIGN=RIGHT>   64.641</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211535 target=_blank>211535</A> x33A4F</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>TEST_TWO_TRK1.5 [12,768,600]</FONT></TD>
<TD ALIGN=CENTER>02:20:45</TD>
<TD ALIGN=CENTER>02:25:05</TD>
<TD ALIGN=CENTER>00:01:20</TD>
<TD ALIGN=RIGHT>         52,842</TD>
<TD ALIGN=RIGHT>        52,842</TD>
<TD ALIGN=RIGHT>      52,838</TD>
<TD ALIGN=RIGHT>    9.939</TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>0</TD></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211536 target=_blank>211536</A> x33A50</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>TEST_TWO_TRK1.5 [12,768,600]</FONT></TD>
<TD ALIGN=CENTER>02:32:36</TD>
<TD ALIGN=CENTER>02:48:50</TD>
<TD ALIGN=CENTER>00:11:31</TD>
<TD ALIGN=RIGHT>         78,225</TD>
<TD ALIGN=RIGHT>        78,196</TD>
<TD ALIGN=RIGHT>      78,028</TD>
<TD ALIGN=RIGHT>   85.762</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211537 target=_blank>211537</A> x33A51</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>TEST_TWO_TRK1.5 [12,768,600]</FONT></TD>
<TD ALIGN=CENTER>02:51:19</TD>
<TD ALIGN=CENTER>02:54:58</TD>
<TD ALIGN=CENTER>00:03:14</TD>
<TD ALIGN=RIGHT>         28,016</TD>
<TD ALIGN=RIGHT>        28,012</TD>
<TD ALIGN=RIGHT>      27,984</TD>
<TD ALIGN=RIGHT>   22.609</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211538 target=_blank>211538</A> x33A52</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_4_00 [1,783,616]</FONT></TD>
<TD ALIGN=CENTER>03:03:12</TD>
<TD ALIGN=CENTER>03:06:40</TD>
<TD ALIGN=CENTER>00:01:58</TD>
<TD ALIGN=RIGHT>      1,491,961</TD>
<TD ALIGN=RIGHT>        37,532</TD>
<TD ALIGN=RIGHT>       9,505</TD>
<TD ALIGN=RIGHT>   13.998</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211539 target=_blank>211539</A> x33A53</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_4_00 [1,783,616]</FONT></TD>
<TD ALIGN=CENTER>03:12:52</TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>02:46:22</TD>
<TD ALIGN=RIGHT>    273,560,560</TD>
<TD ALIGN=RIGHT>     4,965,041</TD>
<TD ALIGN=RIGHT>   1,006,561</TD>
<TD ALIGN=RIGHT>  851.162</TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>07:55:01</TD>
<TD ALIGN=CENTER>06:54:13</TD>
<TD ALIGN=RIGHT>     277,691,349</TD>
<TD ALIGN=RIGHT>      5,433,452</TD>
<TD ALIGN=RIGHT>     1,323,170</TD>
<TD ALIGN=RIGHT> 1072.560</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->