<!-- Title: Day shift. Sat Oct  1, 2005 -->
<!-- SciCo: Mike Lindgren -->
<!-- DAQAce: John Freeman -->
<!-- MonAce: Sergo Jindariani -->
<!-- CO: Ben Kilminster -->
<!-- OpManager: Bob Wagner -->
<!-- Notes: -->
<pre>Take over with store 4414 ongoing.  Store will be dropped at 
10, then 1 hour for D0 access, followed by shot setup.</pre>


<!-- Date: Sat Oct 1 08:14:17.956 2005 -->
Collider flying wire scan
<!-- Author: Freeman :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204928' target=_top>204928</a>) -->
<!-- Comment: Sat Oct  1 09;13;54 comment by...Freeman -->
Again
<!-- Date: Sat Oct 1 08:16:38.237 2005 -->
HV on west module of CLC is at 80%. I checked the HV directly on the CLC crate downstairs and I saw the problem is on channel 5 of board 3. Set voltage is 1013V, monitored is about 820V. This channel is already known but usually it was ~50V from nominal value. We already planned to change the board as soon as possible. Luminosity measurement is not compromized by this problem, the impact is <1%. Data are definetly good.
<!-- Author: Roberto Rossin (CLC pager) :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204927' target=_top>204927</a>) -->


<!-- Date: Sat Oct 1 08:34:40.613 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-17746&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-17746' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-17747&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-17747' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-17748&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-17748' width='300'></a></center>
<!-- Author: Sergo -->
<!-- Comment: Sat Oct  1 08;47;30 comment by...Sergo -->
Hourly plots 0730-0830
<!-- Date: Sat Oct 1 08:46:39.874 2005 -->
BMU SW 156 tripped. Had to turn it on manually.
<!-- Author: Sergo :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204928' target=_top>204928</a>) -->
<!-- Comment: Sat Oct  1 08;52;02 comment by...Sergo -->
Happened again


<!-- Comment: Sat Oct  1 09;04;21 comment by...Sergo -->
Again, same wedge.
<!-- Comment: Sat Oct  1 09;12;53 comment by...Sergo -->
Talked to BMU expert. She told me to turn this channel off for 1/2 hour. Did so. We'll try turning it back on during quiet time between stores.
<!-- Comment: Tue Oct  4 11;18;25 comment by...Phil Schlabach -->
HV channel SW 156 is attached to the cable for SW 168 at the moment.
<!-- Comment: Wed Oct 19 12;38;29 comment by...Phil -->
I was wrong. This was SW156 that tripped. SE156 and SE168 were the swapped ones. HV channel SW156 was attached to the correct cable.
<!-- Date: Sat Oct 1 08:59:33.387 2005 -->
Noticed that left bar of CPR HV in ifix high voltage summary disappeared for few seconds ~5 and then went back on. Checked last few shift elogs - couldn't find anything related.
<!-- Author: Sergo :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204928' target=_top>204928</a>) -->


<!-- Date: Sat Oct 1 09:08:13.814 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204928">204928</A>
 Terminated at 2005.10.01 09:07:23
<!-- Author: RunControl -->


<!-- Date: Sat Oct 1 09:08:14.108 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204928">204928</A>
TERMINATE: Ended run b/c turning bad BMU channel off
<!-- Author: Yuji x2080 -->
<!-- Comment: Sat Oct  1 09;12;54 comment by...Freeman -->
I'm just pretending to be Yuji here
<!-- Date: Sat Oct 1 09:12:23.630 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204929">204929</A>
 Activated at 2005.10.01 09:11:08
<!-- Author: RunControl -->


<!-- Date: Sat Oct 1 09:12:24.558 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204929">204929</A>
ACTIVATE: RunSet: AAA_CURRENT  Trigger: PHYSICS_3_07 [2,713,594] 
<!-- Author: Freeman x2080 -->






<!-- Date: Sat Oct 1 09:32:00.261 2005 -->
Now it's BMU NE 12 trip ( I guess one that tripped during owl shift). I turned it back on.
<!-- Author: Sergo :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204929' target=_top>204929</a>) -->


<!-- Date: Sat Oct 1 09:38:44.991 2005 -->
I get the following slew of errors at the same time that there's a BMU trip. Hard to figure out what exactly is going on; the good news is that the silicon resonance is not in the "danger" zone of 
300. At any rate, after the inhibit is clear, data taking resumes as normal.   
 
 
(MLE) b0dap73.fnal.gov:Thread-24:Oct 1, 2005 9:28:48 AM->FrontEnd Error: FCAL_00 CMX_01 VRB_SVX_02  
(MLW) b0dap73.fnal.gov:Thread-24:Oct 1, 2005 9:28:51 AM->Inhibit flag SET  
(MLE) b0cmx01:Messenger:Oct 1, 2005 9:28:53 AM->Runtime Error 1, Event 93577: See nonsequential event number (old 8, new 15) in L2 message 
CT: 2005.10.01 09:28:52 
28'52" 1 crate/s: b0fcal00(26),  in error. [RXPT] 
(MLW) b0xtrp00:Messenger:Oct 1, 2005 9:29:04 AM->No new event ... 
(MLE) b0svx00:Messenger:Oct 1, 2005 9:29:05 AM->Silicon Resonator: Sl 16 Ch 9 -> e074: Bytes: 00168 
 
CT: 2005.10.01 09:29:04 
29'4" b0svx02:Messenger:Oct 1, 2005 9:29:05 AM->SRC Fatal Error:Sl 6 L2A w/o L1A  
 -->   
 Additional Information:  
 
 Attention !!!. FERML_SRC_FATALITY ERROR !!!  
  
 SRC Fatal Error from b0svx02: Sl 6 L2A w/o L1A  
  
 - HALT the run                    
 - record any relevant information in e-log      
 - RECOVER - RUN      
[MLE] 
CT: 2005.10.01 09:29:05 
29'5" 1 crate/s: b0fcal00(26),  in error. [RXPT] 
(RC)   9:29:08 Halt -> HALTED 
(RC)   9:29:14 Recover -> RECOVERED 
(RC)   9:29:17 Run -> ACTIVE 
(MLW) b0xtrp00:Messenger:Oct 1, 2005 9:29:31 AM->No new event ... 
(MLE) b0svx00:Messenger:Oct 1, 2005 9:29:46 AM->Silicon Resonator: Sl 16 Ch 9 -> e074: Bytes: 00168 
 
(RC)   9:29:49 Halt -> HALTED 
(MLW) b0dap73.fnal.gov:Thread-24:Oct 1, 2005 9:31:10 AM->Inhibit flag CLEAR  
(RC)   9:31:19 Recover -> RECOVERED 
(RC)   9:31:21 Run -> ACTIVE
<!-- Author: Freeman :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204929' target=_top>204929</a>) -->
<!-- Comment: Sat Oct  1 09;39;58 comment by...Freeman -->
Apologies for the formatting - I should have had that entry properly html-tagged!
<!-- Comment: Sat Oct  1 09;45;53 comment by...Freeman -->
This episode is quite similar to when the BMU tripped late in the last shift. 
<!-- Date: Sat Oct 1 09:38:55.743 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-17749&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-17749' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-17750&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-17750' width='300'></a></center>
<!-- Author: Ben -->
<!-- Comment: Sat Oct  1 09;42;04 comment by...Ben -->
<pre>The left plot is run 204928.
The right plot is run 204929.

The difference show how many muons we are losing by turning off 
the bad BMU channel.</pre>


<!-- Date: Sat Oct 1 09:42:44.929 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-17752&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-17752' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-17753&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-17753' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-17754&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-17754' width='300'></a></center>
<!-- Author: Sergo -->
<!-- Comment: Sat Oct  1 09;43;14 comment by...Sergo -->
Hourly plotz 0830-0930


<!-- Date: Sat Oct 1 09:52:08.778 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204929">204929</A>
 Terminated at 2005.10.01 09:51:53
<!-- Author: RunControl -->


<!-- Date: Sat Oct 1 09:52:44.549 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204929">204929</A>
TERMINATE: BMU partially turned off in this run
<!-- Author: Freeman x2080 -->


<!-- Date: Sat Oct 1 10:21:47.438 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204930">204930</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Freeman x2080 -->
<!-- Comment: Sat Oct  1 12;48;02 comment by...Ben -->
CAL QIE looks good.

Of course, we had to do it twice and call an expert in order to 
confirm this.  See plots below on CAL QIE table and comments from
Mark Mattson.  
<!-- Date: Sat Oct 1 10:22:06.994 2005 -->
<pre> At the end of the store ( after I pressed End of Store button )got SVX alarm while silicon was in transition to standby. 
 SVX Alarm window in ifix says: 
 PS_CAEN_OFF >=1 power supply channels off, see GUI 
  
 Checked silicon power GUI, svx B0W1,10,11   B1W1,10,11   B2W0,1,10,11  were off.  
 
 I checked instructions on Mon Ace help page, they say I can try turning them back on.  Since the whole detector is in standby I'll try bringing these wedges to standby.    </pre>
<!-- Author: Sergo :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204929' target=_top>204929</a>) -->
<!-- Comment: Sat Oct  1 10;30;02 comment by...Sergo -->
Well, I'll talk to Silicon expert first
<!-- Comment: Sat Oct  1 10;41;03 comment by...Sergo -->
 Talked to the expert. She said bringing them to standby was the right thing to do. Did so.
  Now,since we're not taking data, there are some pinkies in SVX. Expert said they will clear after we start taking data.

<!-- Date: Sat Oct 1 10:24:15.509 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204930">204930</A>
TERMINATE: Finished Cal QIE
<!-- Author: Freeman x2080 -->


<!-- Date: Sat Oct 1 10:29:26.730 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204931">204931</A>
ACTIVATE: RunSet: CALIB_CLC_QIE 
<!-- Author: Freeman x2080 -->
<!-- Comment: Sat Oct  1 12;46;27 comment by...Ben -->
CLC QIE looks good.
<!-- Date: Sat Oct 1 10:29:27.708 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204931">204931</A>
TERMINATE: Done CLC QIE
<!-- Author: Freeman x2080 -->



<!-- Date: Sat Oct 1 10:30:37.781 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204932">204932</A>
ACTIVATE: RunSet: CALIB_QIE_FWD 
<!-- Author: Freeman x2080 -->
<!-- Comment: Sat Oct  1 11;48;46 comment by...Ben -->
BSC QIE looks good.
<!-- Date: Sat Oct 1 10:32:11.641 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204932">204932</A>
TERMINATE: Done BSC calibration
<!-- Author: Freeman x2080 -->


<!-- Date: Sat Oct 1 10:36:42.238 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204933">204933</A>
ACTIVATE: RunSet: CALIB_RP_SMQIE 
<!-- Author: Freeman x2080 -->
<!-- Comment: Sat Oct  1 12;49;00 comment by...Ben -->
<pre>Roman Pot QIE looks good.</pre>
<!-- Date: Sat Oct 1 10:36:43.653 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204933">204933</A>
TERMINATE: Done Roman Pot SMQIE
<!-- Author: Freeman x2080 -->


<!-- Date: Sat Oct 1 10:39:09.409 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204934">204934</A>
ACTIVATE: RunSet: ACE_TOF_QIE_NEW 
<!-- Author: Freeman x2080 -->
<!-- Comment: Sat Oct  1 12;49;34 comment by...Ben -->
<pre>TOF QIE looks good</pre>
<!-- Date: Sat Oct 1 10:41:03.568 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204934">204934</A>
TERMINATE: Done TOF QIE
<!-- Author: Freeman x2080 -->



<!-- Date: Sat Oct 1 10:45:04.274 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204935">204935</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Freeman x2080 -->
<!-- Comment: Sat Oct  1 12;50;07 comment by...Ben -->
<pre>SMX QIE looks good.</pre>
<!-- Date: Sat Oct 1 10:45:38.423 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204935">204935</A>
TERMINATE: Done SMX QIE calibration
<!-- Author: Freeman x2080 -->



<!-- Date: Sat Oct 1 10:49:04.202 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-17755&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-17755' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-17756&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-17756' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-17757&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-17757' width='300'></a></center>
<!-- Author: Sergo -->
<!-- Comment: Sat Oct  1 10;49;47 comment by...Sergo -->
Hourly plots including end of store 0930-1030

<!-- Date: Sat Oct 1 10:53:50.206 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204936">204936</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: Freeman x2080 -->
<!-- Comment: Sat Oct  1 12;52;03 comment by...Ben -->
<pre>CEM LED looks good.</pre>
<!-- Date: Sat Oct 1 10:55:57.820 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204936">204936</A>
TERMINATE: Finished CEM LED calibration
<!-- Author: Freeman x2080 -->


<!-- Date: Sat Oct 1 10:55:58.686 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204937">204937</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Freeman x2080 -->
<!-- Comment: Sat Oct  1 12;51;39 comment by...Ben -->
<pre>COT looks good.</pre>
<!-- Date: Sat Oct 1 11:04:16.537 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204937">204937</A>
TERMINATE: Finished COT calibration
<!-- Author: Freeman x2080 -->


<!-- Date: Sat Oct 1 11:04:17.269 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204938">204938</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: Freeman x2080 -->
<!-- Comment: Sat Oct  1 13;01;49 comment by...Ben -->
<pre>Wedge 15 is newly bad.  There is no entry in the CO known problem 
list, but it is mentioned in a previous shift e-log, so apparently
this is a known problem (but it's on a need-to-know basis). 

</pre>
<!-- Date: Sat Oct 1 11:05:29.331 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204938">204938</A>
TERMINATE: Finished CEM XEF calibration
<!-- Author: Freeman x2080 -->



<!-- Date: Sat Oct 1 11:06:18.268 2005 -->
Silicon expert called, asked me to perform cold start for SVX. Did so.
<!-- Author: Sergo -->

<!-- Date: Sat Oct 1 11:10:30.268 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204939">204939</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Freeman x2080 -->


<!-- Date: Sat Oct 1 11:10:31.578 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204939">204939</A>
TERMINATE: Finished muon calibration
<!-- Author: Freeman x2080 -->


<!-- Date: Sat Oct 1 11:14:33.350 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204940">204940</A>
ACTIVATE: RunSet: TOF_TAC_CALIB 
<!-- Author: Freeman x2080 -->
<!-- Comment: Sat Oct  1 13;10;19 comment by...Ben -->
<pre>TOF TAC looks good.</pre>
<!-- Date: Sat Oct 1 11:25:10.865 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204940">204940</A>
TERMINATE: Finished TOF TAC calibration. RXPT loop error for b0puls01 at completion of calibration.
<!-- Author: Freeman x2080 -->


<!-- Date: Sat Oct 1 11:26:41.502 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204928">204928</A>
si: <tt>In SVXMon SVX Chip Status Map, there are some new failures at<br>
Bulhead, Z-side, Layer 4, wedge 7 in bin 5. These are not on the reference plot.  It looks like there are about 3 chips in a row which<br>
are red.  Looking at the Silicon monitor consumer documentation, it <br>
looks like this is acceptable.   I think this documentation would be<br>
more clear if it showed a plot to monitor compared to a reference plot. And said, the difference between these plots is acceptable.</tt>
<!-- Author: Consumer Operator(Ben Kilminster ) -->


<!-- Date: Sat Oct 1 11:26:42.642 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204928">204928</A>
cal: <tt>WHA PMT Occupancy has a new dead tower at eta = 8, phi = 19.  <br>
<br>
<br>
Also,<br>
The PEM tower occupancy is red in the west tower (-21) for all phi wedges.  This fits the criteria that the number of hot consecutive channels is > 12, and so I am tempted not to mark this as good. The east side is almost lit up in tower 21.  Instead, I will mark this as good, and doublecheck that this increased beam splash is reasonable with the experts.<br>
</tt>
<!-- Author: Consumer Operator(Ben Kilminster ) -->
<!-- Comment: Mon Oct  3 08;25;04 comment by...Willis -->
The WHA PMT is okay in high statistics runs. The PEM and PHA
red rings in the high eta tower region are also known as the "rings of fire", and is from the beam (collisions, etc.) -- that is
normal.
<!-- Date: Sat Oct 1 11:26:43.078 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204928">204928</A>
smx: <tt>For the CCR Pad plots, the number of dead consecutive channels is 10, although they are not in a row.   It looks like in these channels, the occupancy is close to zero anyway.   Just being thorough.</tt>
<!-- Author: Consumer Operator(Ben Kilminster ) -->


<!-- Date: Sat Oct 1 11:30:36.744 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204941">204941</A>
 Activated at 2005.10.01 11:28:46
<!-- Author: RunControl -->


<!-- Date: Sat Oct 1 11:30:37.038 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204941">204941</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [5,720,540] 
<!-- Author: Freeman x2080 -->


<!-- Date: Sat Oct 1 11:36:52.956 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204941">204941</A>
 Terminated at 2005.10.01 11:36:44
<!-- Author: RunControl -->


<!-- Date: Sat Oct 1 11:37:52.712 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204941">204941</A>
TERMINATE: Finished PMT spike run. Average L1 rate was 64.9 Hz.
<!-- Author: Freeman x2080 -->



<!-- Date: Sat Oct 1 12:00:31.324 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204942">204942</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Freeman x2080 -->
<!-- Comment: Sat Oct  1 12;53;08 comment by...Ben -->
<pre>Good.
This was the second try.  We get the same result as the first try
with 204930.   See that run number for more info.</pre>
<!-- Date: Sat Oct 1 12:01:09.145 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204942">204942</A>
TERMINATE: Finished 2nd Cal QIE calibration
<!-- Author: Freeman x2080 -->


<!-- Date: Sat Oct 1 12:12:20.477 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204943">204943</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: Freeman x2080 -->



<!-- Date: Sat Oct 1 12:12:20.446 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-17758&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-17758' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-17759&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-17759' width='300'></a></center>
<!-- Author: Ben -->
<!-- Comment: Sat Oct  1 12;42;30 comment by...m mattson (ADMEM) -->
<br>PHA 23E was acting up during yesterday's access.  At the time it had an RMS of nearly 20.  Kwangzoo investigated it, and in the
<a href=
'/cgi/elog/elog.pl?nb=2005&action=view&page=818&frame=2&anchor=08:52:53.522'>
day e-log</a>
noticed that the chanel was not as bad after the access.  We suspect a light leak at the phototube/base end, in which case there is little to be done at the FER end.
<p>
I'll add this to the CO Fix-list entry, and we will continue to monitor its behavior.  No action to be taken, this calibration is okay.
<!-- Comment: Sat Oct  1 12;43;04 comment by...m mattson (ADMEM) -->
<br>This is in response to the pager call.
<!-- Comment: Sat Oct  1 12;45;28 comment by...Ben -->
These plots were from the CAL QIE calibration.


Also,  Mark mentioned that we should not be concerned about black slopes on the second page unless they are zero. 

So this calibration is okay.
<!-- Date: Sat Oct 1 12:14:29.914 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204943">204943</A>
TERMINATE: Finished 2nd Xenon calibration
<!-- Author: Freeman x2080 -->


<!-- Date: Sat Oct 1 12:19:58.067 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204944">204944</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: Freeman x2080 -->


<!-- Date: Sat Oct 1 12:21:01.139 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204944">204944</A>
TERMINATE: Finished 2nd LED run
<!-- Author: Freeman x2080 -->



<!-- Date: Sat Oct 1 12:27:39.134 2005 -->
<pre>Had the e-log lock up for a bit, is now back on due 
to help from Steve Hahn, so thanks Steve. 
 
MCR called, and they are about ready for shot setup.</pre>
<!-- Author: lindgren -->

<!-- Date: Sat Oct 1 12:31:52.997 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204945">204945</A>
 Activated at 2005.10.01 12:31:01
<!-- Author: RunControl -->


<!-- Date: Sat Oct 1 12:31:53.429 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204945">204945</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,716,538] 
<!-- Author: Freeman x2080 -->





<!-- Date: Sat Oct 1 12:43:34.708 2005 -->
<pre>I get the following during the cosmics run, solved by an HRR: 
 
Attention !!!. FERML_TM2_HEVB2B_EVENTS_PENDING_TIMEOUT !!!  
 - HRR should help.  
 - If problem persists, you may have to clean up the Event Builder. </pre>
<!-- Author: Freeman :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204945' target=_top>204945</a>) -->












<!-- Date: Sat Oct 1 13:04:09.873 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-17760&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-17760' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-17761&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-17761' width='300'></a></center>
<!-- Author: Ben -->
<!-- Comment: Sat Oct  1 13;06;13 comment by...Ben -->
<pre>These are the CEM XEF calibration plots.  There are 4 wedges that 
are dead on 15 and 16 (E and W). 

Can this be added to the CO known problem list ?
 

 </pre>


<!-- Comment: Sat Oct  1 14;48;51 comment by...Bob Wagner (ANL) -->
This is a known problem.  It is not 4 wedges but 2.  It's left and right phototubes on 15E and 16E.  Calib card problem.
<!-- Comment: Sat Oct  1 21;10;03 comment by...Former C.O. -->
Bob -- could you please update the reference plots, then?
<!-- Date: Sat Oct 1 13:11:12.866 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204945">204945</A>
 Terminated at 2005.10.01 13:11:02
<!-- Author: RunControl -->


<!-- Date: Sat Oct 1 13:12:12.242 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204945">204945</A>
TERMINATE: Getting L2 decision TOs frequently. Going to L2 torture to see how much worse it gets. 
<!-- Author: Freeman x2080 -->


<!-- Date: Sat Oct 1 13:15:52.703 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204946">204946</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [16,718,539] 
<!-- Author: Freeman x2080 -->


<!-- Date: Sat Oct 1 13:37:54.815 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204946">204946</A>
TERMINATE: 20 mins running, no L2 decision TOs
<!-- Author: Freeman x2080 -->


<!-- Date: Sat Oct 1 13:40:32.983 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204947">204947</A>
 Activated at 2005.10.01 13:40:15
<!-- Author: RunControl -->
<!-- Comment: Sat Oct  1 14;38;31 comment by...Freeman -->
This is another ACE_COSMICS_ALL run
<!-- Comment: Sat Oct  1 14;39;34 comment by...Freeman -->
b0tsi registers a Busy TO. Auto HRR recovers the run.
<!-- Comment: Sat Oct  1 14;49;12 comment by...Freeman -->
b0cmu01 registers two busy TOs in rapid succession. Each time HRR returns us to normal running. 
<!-- Date: Sat Oct 1 13:53:50.851 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204929">204929</A>
cal: <tt>For the YMON PEM and PHA tower occupancy plots, same comment as last run about how the highest eta towers 21 and -21 are lit up.<br>
<br>
The WHA occupancy for many towers is low/zero, but this was a short run<br>
at low instantaneous luminosity, so it's probably not important.</tt>
<!-- Author: Consumer Operator(Ben Kilminster ) -->


<!-- Date: Sat Oct 1 13:53:51.136 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204929">204929</A>
smx: <tt>CCR PAs plots has lots of white space in east wedges 13 to 15. <br>
But very low statistics also.</tt>
<!-- Author: Consumer Operator(Ben Kilminster ) -->


<!-- Date: Sat Oct 1 13:53:52.761 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204929">204929</A>
muons: <tt>We had BMU West channel off for this run. </tt>
<!-- Author: Consumer Operator(Ben Kilminster ) -->







<!-- Date: Sat Oct 1 14:49:48.482 2005 -->
<TABLE bgcolor=#cdc9a5> 
<TR><TH>Date</th><th>Time</th><th>BLM</th><th colspan=2>Dose</th></tr> 
<tr><td>2005.10.01</td><td>14:45:08</td><td>W Inner BLM</td><td>0.42</td><td>RADS</td></tr> 
<tr><td>2005.10.01</td><td>14:45:08</td><td>W Outer BLM</td><td>0.00</td><td>RADS</td></tr> 
<tr><td>2005.10.01</td><td>14:45:08</td><td>E Inner BLM</td><td>0.14</td><td>RADS</td></tr> 
<tr><td>2005.10.01</td><td>14:45:08</td><td>E Outer BLM</td><td>0.32</td><td>RADS</td></tr> 
</TABLE> 
<strong>Integrated dosage</strong>
<!-- Author: Sergo -->

<!-- Date: Sat Oct 1 14:51:05.550 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204947">204947</A>
 Terminated at 2005.10.01 14:49:59
<!-- Author: RunControl -->


<!-- Date: Sat Oct 1 14:51:06.259 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204947">204947</A>
TERMINATE: Finished run; now loading protons so going to AAA_SHOTSETUP
<!-- Author: Freeman x2080 -->
<!-- Comment: Sat Oct  1 14;58;37 comment by...Freeman -->
Busy TO followed by RXPT errors loop for b0cmu00. HRR recovers run.


<!-- Comment: Sat Oct  1 15;02;22 comment by...Freeman -->
b0xft02 and 04 register busy TOs + an RXPT errors loop. HRR recovers the run. 
<!-- Date: Sat Oct 1 15:00:37.920 2005 -->
b0dap56 was frozen. I couldn't login to b0dap56. So I just rebooted . If this happens again, please page consumer.  
<!-- Author: Junji -->


<!-- Date: Sat Oct 1 15:03:56.876 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204948">204948</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_3_07 [2,713,594] 
<!-- Author: Freeman x2080 -->


<!-- Date: Sat Oct 1 15:04:25.939 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204948">204948</A>
TERMINATE: Too many busy TOs. I'm going to clean up EVB. 
<!-- Author: Freeman x2080 -->


<!-- Date: Sat Oct 1 15:12:11.809 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204949">204949</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_3_07 [2,713,594] 
<!-- Author: Freeman x2080 -->


<!-- Date: Sat Oct 1 15:17:11.734 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204949">204949</A>
TERMINATE: Too many busy TOs again!
<!-- Author: Freeman x2080 -->
<!-- Comment: Sat Oct  1 15;27;39 comment by...Bob Wagner (ANL) -->
We went through this yesterday before the store went.  You might want to page the L2 pager and see if that person has any words of advive.
<!-- Date: Sat Oct 1 15:23:49.923 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204950">204950</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_3_07 [2,713,594] 
<!-- Author: Freeman x2080 -->




<!-- Date: Sat Oct 1 15:38:27.089 2005 -->
Icicle HB alarm. Restarted the icicle.
<!-- Author: Sergo -->
<!-- Comment: Sat Oct  1 15;41;35 comment by...Sergo -->
<pre>Nothing happened, restarted icicle again. I'll give it some time to update.</pre>

<!-- Comment: Sat Oct  1 15;47;10 comment by...Sergo -->
cleared
<!-- Date: Sat Oct 1 15:44:49.819 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204950">204950</A>
TERMINATE: Ended to clean up EVB once again. 
<!-- Author: Freeman x2080 -->




<!-- Date: Sat Oct 1 15:49:32.448 2005 -->
Made it to flat top, then aborted, low beta quench.
<!-- Author: lindgren -->

<!-- Date: Sat Oct 1 15:53:02.238 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204951">204951</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_3_07 [2,713,594] 
<!-- Author: Freeman x2080 -->
<!-- Comment: Sat Oct  1 15;53;30 comment by...Freeman -->
This is after the latest EVB/L3 farm cleanup

<!-- Date: Sat Oct 1 15:55:33.145 2005 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>Physics able</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L1Accepts</TH>
<TH>L2Accepts</TH>
<TH>L3Accepts</TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204928 target=_blank>204928</A> x32080</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_3_07 [2,713,594]</FONT></TD>
<TD ALIGN=CENTER>06:55:33</TD>
<TD ALIGN=CENTER>09:07:23</TD>
<TD ALIGN=CENTER>02:01:31</TD>
<TD ALIGN=RIGHT>     96,132,279</TD>
<TD ALIGN=RIGHT>       963,462</TD>
<TD ALIGN=RIGHT>     360,659</TD>
<TD ALIGN=RIGHT>  128.369</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204929 target=_blank>204929</A> x32081</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_3_07 [2,713,594]</FONT></TD>
<TD ALIGN=CENTER>09:11:08</TD>
<TD ALIGN=CENTER>09:51:53</TD>
<TD ALIGN=CENTER>00:37:45</TD>
<TD ALIGN=RIGHT>     23,325,191</TD>
<TD ALIGN=RIGHT>       254,390</TD>
<TD ALIGN=RIGHT>      96,538</TD>
<TD ALIGN=RIGHT>   36.831</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>15:55:01</TD>
<TD ALIGN=CENTER>02:39:17</TD>
<TD ALIGN=RIGHT>     119,457,470</TD>
<TD ALIGN=RIGHT>      1,217,852</TD>
<TD ALIGN=RIGHT>       457,197</TD>
<TD ALIGN=RIGHT>  165.199</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Sat Oct 1 15:59:28.501 2005 -->
<font color=red>Shift Summary: </font><pre>Started shift with at tail end of store, which was dropped at 10 am.  D0 went in on access for an hour, and we got ready for shot setup by doing calibrations.   
 
Lot's of busy time outs running cosmics, and B0dap56 froze 
and had to be restarted.   
 
Put in protons and pbars, and then had a low beta quench at step 20 in the sqeeze. 
 
Begin shift or beam at 08:00:08 
End shift or beam at 09:52:11 
Delivered luminosity: 113.19 
Acquired luminosity: 96.83 
Totals 
Date:	2005.10.01 
Shift:	day 
Delivered luminosity:	113.2 nb-1 
Acquired luminosity:	96.8 nb-1 
Efficiency:	85.5</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   204928, 204929
Delivered Luminosity   113.2 nb-1  
Acquired Luminosity    96.8 nb-1  
Efficiency             85.5
   
</td></tr></table></pre>
<!-- Author: Lindgren -->

<!-- Date: Sat Oct 1 16:01:36.775 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204951">204951</A>
TERMINATE: Beam Quench during scraping.
<!-- Author: Freeman x2080 -->



<!-- Date: Sat Oct 1 16:03:06.870 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-17762&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-17762' width='300'></a></center>
<!-- Author: pinghan -->
<!-- Comment: Sat Oct  1 16;05;47 comment by...pinghan -->
<pre>CDF Ace Silicon Instructions for POST-QUENCH CHECKOUT:

There is no PINK cell and here is BLMs SUMMARY.</pre>
<!-- Comment: Sat Oct  1 16;21;03 comment by...Sebastian -->
this is not the right time stamp, should be 15:48:54...
<!-- Comment: Sat Oct  1 16;48;41 comment by...Teresa -->
I'll post the right BLM summary, it will be in the evening shift report.
<!-- Date: Sat Oct 1 16:03:41.524 2005 -->
With the silicon coldstart run configuration brought run control through to the ready state. No problems.
<!-- Author: Teresa -->


<!-- Date: Sat Oct 1 16:08:35.301 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204952">204952</A>
 Activated at 2005.10.01 16:07:34
<!-- Author: RunControl -->


<!-- Date: Sat Oct 1 16:08:36.884 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204952">204952</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,716,538] 
<!-- Author: Freeman x2080 -->



<!-- Date: Sat Oct 1 16:13:17.176 2005 -->
When I ended the shot setup run I got the following error message:<br> 
Attention!!! RXPT Errors Loop<br> 
Please check the following crates:<br> 
all teh svt crates, tsi03, puls00, l2cl00, l2cl01, l2cl03, l2cl04, l2cl05, l2pu01, l2pu00, htdc00, etdc00. <br> 
I was able to do the silicon coldstart successfully, and I was able to start an cosmic run just fine. So I'm not sure what else I should do to "check" these crates...advice?
<!-- Author: Teresa :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=204952' target=_top>204952</a>) -->