*** File to contain comments about FPP chambers ***
***
6/19/97
Pulser runs for chambers 3 and 4
Chamber 3 pulsed wires 1,2,3,4,5,6,7,8 for 500 events 
Data in directory /adaq2/data5/adaq/data which will be moved to MSS
The files are e89003_ped_h_1511.dat thru  e89003_ped_h_1518.dat
Chamber 4 pulsed wires 2,3,4,5,6,7,8,1 for 500 events
Data in directory /adaq2/data5/adaq/data which will be moved to MSS
The files are e89003_ped_h_1521.dat thru  e89003_ped_h_1528.dat
To get pulses to all of chamber 4 needed to up the pulser amplitude
from 1.5V for chamber 3 pulser runs to 2.0V.
***
6/19/97 No Beam at 20:00 MKJ
CH1 HV  Ch2 HV   Ch3a HV  CH3b HV  Ch4 HV 
------  ------   ------   ------  -------
1875     1875    1900     1875     1875
Ch2 has channel 7.1.10 tripped tried to reset
but tripped again. Left 7.1.10 off
***
6/20/97 at 6.00pm KW
Checked the HV on the chambers, Found Ch2 Channel 7.1.10 has tripped.Tried to reset but tripped at 1850 .Reset to 1840 V.
Channel #  Chamber#   Current (MicroA) High voltage (V)
--------   --------   ---------------  -----------------
7.1.0       1          0.02             1875 5
7.1.1       1          0.02              "
7.1.2       1          0.00              "
7.1.3       1          0.01              "
7.1.4       1          0.00              "
7.1.5       1          0.00              "
7.1.8       2          0.05             1838.9
7.1.9       2          0.06              "
7.1.10      2          0.05              "
7.1.11      2          0.07              "
7.1.12      2          0.09              "
7.1.13      2          0.07              "
8.1.1       3          1.83             1900.0
8.1.2       3          2.55              "
8.1.3       3          3.80              "
8.1.5       3          0.18              "
8.1.8       4          0.01             1875.2
8.1.9       4          0.04              "
8.1.10      4          0.06              "
8.1.11      4          0.01              "
8.1.12      4          0.03              "
8.1.13      4          0.01              "
8.1.16      3          8.3              1875
8.1.17      3          13.7             1875
***
***
6/21/97 at 12:00pm MKJ
8.1.8 was tripped , reset 8.0.1
7.1.10 should be moved to third master HV
Channel #  Chamber#   Current (MicroA) High voltage (V)
--------   --------   ---------------  -----------------
7.1.0       1          0.18             1875 5
7.1.1       1          0.72              "
7.1.2       1          0.18              "
7.1.3       1          0.33              "
7.1.4       1          0.65              "
7.1.5       1          0.19              "
7.1.8       2          0.43             1838.9
7.1.9       2          0.48              "
7.1.10      2          0.26              "
7.1.11      2          0.36              "
7.1.12      2          0.64              "
7.1.13      2          0.22              "
8.1.1       3          9.65             1900.0
8.1.2       3          17.8              "
8.1.3       3          11.1              "
8.1.5       3          0.96              "
8.1.8       4          0.96             1875.2
8.1.9       4          0.96              "
8.1.10      4          0.45              "
8.1.11      4          0.38              "
8.1.12      4          0.24              "
8.1.13      4          0.24              "
8.1.16      3          8.3              1875
8.1.17      3          13.7             1875
***
6/23/97  9:20am MKJ
The threshold voltages were at zero.
Set them to 3.0 for front and 7.0 rear.
****
6/23/97 3.00pm KW
Checked the the HV and the currents, every thing looks fine.But No Beam!!
******
6/25/97 8:30am MKJ
CHeck HV. It is off to FPP. It was done to reduce event size.
********
6/29/97 8:30am MKJ
Turn on HV to FPP since data rate and deadtime low.
*******
6/29/97 12:20am MKJ
Check HV. 7.1.10 has tripped again. Other channels ok.
Could not get 7.1.10 to hold HV at 1840V so left it off.
*****
7/12/97 11:00am MKJ
Check HV. CHambers 1 and 2 have been holding steady at 1875V since
fixing problem with Module 8 on V side of Ch 2. CH2 is drawing around
.2-.3 uA on all channels. CH1 is drawing around .01-.04 uA.
CH3 channels 8.1.1, 8.1.2, 8.1.3, 8.1.5 drawing .35,.30,.90,.20 uA at 1900V.
CH3 channels 8.1.16,8.1.17 drawing 20., 28. uA at 1900V, lower to
1875 and drawing 5.5,10.5 . On 7/9/97 the all CH3 channels drawing less than
1 uA. Channels 8.1.16 and 8.1.17 were drawing around 13uA at 1900V yesterday.  
CH4 all channels drawing between .01 to .08 uA at 1900V.
*****