<!-- Title: Eve shift. Sat Nov 17, 2001 -->
<!-- SciCo: S. Errede -->
<!-- DAQAce: S. Tsuno/D. Usynin -->
<!-- MonAce: &nbsp; -->
<!-- CO: &nbsp; -->
<!-- OpManager: Kevin Burkett -->
<!-- Notes: -->
<pre>Continue with ongoing cosmics run, then finish,  
then start another round of calibrations.  
People working in collision hall.</pre>


<!-- Date: Sat Nov 17 16:42:27 2001 -->
RXPT error seen in crate LEVEL2_CAL_04, 
we did HRR and the run is active, but we are not getting any  
events now. 
ending the run
<!-- Author: aces :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=132599' target=_top>132599</a>) -->
<!-- Comment: Sat Nov 17 16;44;10 comment by...aces -->
Run was ended with 51012 events


<!-- Date: Sat Nov 17 16:50:45 2001 -->
Abraham told us he doesnt need another cosmics run 
so we are going to try taking some calibrations now.
<!-- Author: aces -->


<!-- Date: Sat Nov 17 17:06:51 2001 -->
earlier Greg Feild had unmasked the CDF_ERROR line in b0l2cl04.  At his prompting I mased the error line again to see if the problems seen earlier will go away.  He and I will both be in tomorrow to do various work
<!-- Author: hray -->


<!-- Date: Sat Nov 17 17:07:45 2001 -->
<pre>Took some LED calibration data for CLC</pre>
<!-- Author: Ming and Valentin -->


<!-- Date: Sat Nov 17 17:11:22 2001 -->
Experts work of CLC done. 
CLC crates were returned.
<!-- Author: aces -->


<!-- Date: Sat Nov 17 17:21:30 2001 -->
We tried to take a dummy COT calibration but crate cot04 
failed to activate. 
Rebooting and resetting it didn't help so we are paging  
COT expert. 
<!-- Author: aces -->
<!-- Comment: Sat Nov 17 17;25;32 comment by...Denys/Soushi -->
Mike Cirby called back and said he was gonna come and look
at the problem, for now do not use cot04 crate if u wanna
include COT in any runs.


<!-- Date: Sat Nov 17 17:45:51 2001 -->
We did CLC QIE calibration.<br> 
Got the message about crate clc_00 missing cards in slots 16,17.<br> 
Calibration made it into the database.<br> 
We also got the following errors: 
<pre> 
CT: 11/17/2001, 17:32'53"  
Partition:7 Run number: 132624 (0x20610) 
(RC)   17:32:55 Config -> READY 
(RC)   17:33:0 Activate -> CALIBRATING 
(MLE) b0clc00:Messenger:5:33:36 PM->QIE calibration failed 
(MLE) b0clc00:Readout_Calib:5:33:36 PM->ADMEM=3 QIE=11 Fail 14: Sigma(off) big      (9.974411) (0) (0) 
</pre> 
 
And a lot of errors about sigma too big, more can be seen in the log file on bodap59 /cdf/code-common/cdfonline/errmon/log/errorfile132624.log  
<!-- Author: aces :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=132624' target=_top>132624</a>) -->


<!-- Date: Sat Nov 17 18:13:15 2001 -->
<pre>CMX HV of slot# 22,23 was down while taking Muon calibration in Run # 132629. 
 
We tryied again. 
We removed  
  CMX_01 TDC_CMX_16 in slot 8 
  CMX_01 TDC_CMX_17 in slot 9. 
 
Moun calib. done. 
Run # 132636. 
</pre>
<!-- Author: Denys, Soushi  :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=132636' target=_top>132636</a>) -->


<!-- Date: Sat Nov 17 18:14:45 2001 -->
After Moun calib., CMX HV of slot# 2,3 in west side was down.
<!-- Author: Denys and Soushi -->


<!-- Date: Sat Nov 17 18:22:48 2001 -->
Done with plug crates for today. 
Worked on mapping out the response of the Plug Laser 
LEDs and timed in a LN300 laser just in fro being refilled. 
Both East and West lasers are now LN300s
<!-- Author: Giovanni Pauletta -->


<!-- Date: Sat Nov 17 18:29:26 2001 -->
CMX HV of slot # 2,3 in west side triped again.
<!-- Author: aces -->


<!-- Date: Sat Nov 17 18:39:11 2001 -->
 With the central laser firing we Checked the outputs of all CHA <br> HAD ASDs. No dead channel was found and the noise rate is <br> less than 1 KHz. At this point any noisy or dead channels 
<br> should be due to either the decoder box/cable or due to  
<br> flaky TDC channels.   
<!-- Author: fabio and fotis -->


<!-- Date: Sat Nov 17 18:47:51 2001 -->
<pre>SMX calibration was done. 
Run # 132640 
 
but we got many error messages which is concerned with some slots of PCAL crate. 
 
Then, we removed according to the previous instruction. 
(please see previous shift summary: Sat Nov. 17 16:01) 
 
Run #132641 
no error 
 
</pre>
<!-- Author: Denys and Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=132641' target=_top>132641</a>) -->


<!-- Date: Sat Nov 17 19:15:41 2001 -->
Performing TOF TAC calibration now, during the 
first try received Done Timeout and crate b0puls01 died. 
<br> 
unable to ping the crate, so we reset it, then reboot 
and it came back on. Actually only after the second reboot  
the partitioning succeeded.<br> 
 
After that the calibration finished successfully
<!-- Author: aces :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=132643' target=_top>132643</a>) -->


<!-- Date: Sat Nov 17 19:22:27 2001 -->
Taking ShowerMax Calibration now...<br> 
finished successfully
<!-- Author: aces :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=132644' target=_top>132644</a>) -->


<!-- Date: Sat Nov 17 19:25:22 2001 -->
Get COT O2 hi-level alarm again.  
People were working on COT a while ago. 
Not able to silence this alarm. Called 
Kevin Burkett, explained/discussed w/ him. 
Decided to disable COT gas monitoring until 
tomorrow, after gas has had some time to flow. 
Acknowledged alarm & disabled COT gas monitoring. 
<!-- Author: Steve Errede -->


<!-- Date: Sat Nov 17 19:41:57 2001 -->
At 19:00 we returned H.V. on Bottom CSP counters to nominal value
<!-- Author: Akram, Davit -->


<!-- Date: Sat Nov 17 19:48:21 2001 -->
We are taking Rick's Run now, when starting the run got the 
following error: <br> 
Done timeout 80000000 is not default value 8000000 for HardEvb 
<br> 
Busy timeout 80000000 is not default value 8000000 for HardEvb 
<br> 
 
Other than that the run went smoothly.<br> 
Fred L1A 156.1 Hz<br> 
Finished the run at 35910 events<br>  
<!-- Author: Denys / Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=132647' target=_top>132647</a>) -->


<!-- Date: Sat Nov 17 20:03:50 2001 -->
<pre>Just did CAL QIE calibration. Got some errors: 
(MLE) b0puls01:Messenger:7:59:08 PM-&gt;QIE calibration failed 
(MLE) b0puls01:Messenger:7:59:08 PM-&gt;QIE calibration failed 
(MLE) b0puls01:Readout_Calib:7:59:08 PM-&gt;ADMEM=21 QIE=0 Fail 10: No fitting          (0.0) (0) (0) 
(MLE) b0puls01:Readout_Calib:7:59:08 PM-&gt;ADMEM=21 QIE=0 Fail 10: No fitting          (0.0) (0) (0) 
 
other than that, calibration run ended successfully</pre>
<!-- Author: aces :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=132648' target=_top>132648</a>) -->


<!-- Date: Sat Nov 17 20:10:21 2001 -->
The COT was declared leak tight around noon today.  The SUVA was reconnected and the low voltage power was turned on around 17:00.  The COT low voltages and temperatures look ok.  Morris and I re-taped the baggies and started the nitrogen purge.  The baggie oxygen level was as expected (1.5%), so at least the baggies are presently ready for Plug closing.  Dave Ambrose will check for microcoax "plug-in" problems this evening.  SL5 and SL6 were run at 5% nominal high voltage and looked ok (shorts gone).  The chamber will be sufficiently flushed with nitrogen to test at full voltage late tomorrow morning.  Since the Plug closing is scheduled for 07:00, we prefer that the EAST Plug be closed, so that SL5 cells can still be disabled on the WEST end.
<!-- Author: R. Wagner -->


<!-- Date: Sat Nov 17 20:14:00 2001 -->
we took CAL LED calibration, finished with no  errors
<!-- Author: aces :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=132649' target=_top>132649</a>) -->


<!-- Date: Sat Nov 17 20:18:14 2001 -->
<pre>Looking at the Error log file from run 132646, there was a problem reading out the header from the cards in slot 7 & 8 in crate b0cot05.  This then led to bunch counter mismatches.  Not sure what would have caused this problem, but trying the run again to see what happens.  As well, looking at the problems that had come from b0cot04, the crate seems to be acting the same way that it had prior to replacing the TRACER.  I'm going to have to go into the pit to look at this tomorrow. </pre>
<!-- Author: Kirby -->


<!-- Date: Sat Nov 17 20:18:34 2001 -->
COT calibrations kept failing, we paged Mike and he is here 
now working on it. Note to aces again: do not include 
cot04 into any runs untill further notice.
<!-- Author: Denys / Soushi -->
<!-- Comment: Sat Nov 17 20;22;14 comment by...aces -->
<table border=0><td><tr><td bgcolor=#e0e030>Turns out COT05 also has a problem, so both
cot05 and cot04 should be removed from any runs for now.</td></tr></td></table>


<!-- Date: Sat Nov 17 20:23:27 2001 -->
Woohoo! Finally successfully completed the COT 
calibration without the crates 4 and 5.<br> 
Going to try taking cosmics now.
<!-- Author: Denys/ Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=132651' target=_top>132651</a>) -->


<!-- Date: Sat Nov 17 21:10:08 2001 -->
When trying to start cosmics run, we received error message from TSI when trying to ACTIVATE the run. 
<br> 
RXPT monitor shows an error in pcal09 (in both realtime error and latched error windows)<br> 
So we try to start the run again after removing pcal09, let's  
see if it works.
<!-- Author: aces -->


<!-- Date: Sat Nov 17 21:12:57 2001 -->
CMX HV was down in the slot # 11, 12 in the west side 
while setting runconfigration of cosmic run. 
we paged expert.
<!-- Author: Denys & Soushi -->
<!-- Comment: Sat Nov 17 21;17;29 comment by...aces -->
Pager responded and told us to go downstairs to muon3
computer and turn off that particular wedge (11-12) where
the alarm was set (hardware trip alarm, by the way).
<br>
so we did just that, alarm is gone.


<!-- Date: Sat Nov 17 21:20:07 2001 -->
We just started a cosmics run without crates cot04, cot05,pcal09 
<br> 
Average L3 accept rate is 22Hz
<!-- Author: aces :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=132655' target=_top>132655</a>) -->


<!-- Date: Sat Nov 17 21:54:53 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-2648&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-2648  width='300' align='top'></td><td>Raw hits in the COT show some noisy channels</a></td></tr></table>
<!-- Author: aces -->


<!-- Date: Sat Nov 17 22:53:21 2001 -->
Dave came in and said he would like to take COT to do 
some work on it.<br> 
We stopped the cosmics run at 124862 events.
<!-- Author: aces :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=132655' target=_top>132655</a>) -->
<!-- Comment: Sat Nov 17 22;58;07 comment by...aces -->
During the run, we had the following errors in error logger:
(MLE) b0cot03:Messenger:9:46:51 PM->Runtime Error 1, Event 39823: data truncated
(MLE) b0cot01:Messenger:10:13:53 PM->Runtime Error 1, Event 75070: data truncated
(MLE) b0cot02:Messenger:10:13:53 PM->Runtime Error 1, Event 75070: data truncated
(MLE) b0cot09:Messenger:10:13:55 PM->Runtime Error 1, Event 75070: data truncated
(MLE) b0cot08:Messenger:10:13:53 PM->Runtime Error 1, Event 75070: data truncated
(MLE) b0cot10:Messenger:10:13:55 PM->Runtime Error 1, Event 75070: data truncated
(MLE) b0cot00:Messenger:10:13:52 PM->Runtime Error 1, Event 75070: data truncated
(MLE) b0cot11:Messenger:10:13:53 PM->Runtime Error 1, Event 75070: data truncated
(MLE) b0cot12:Messenger:10:13:53 PM->Runtime Error 1, Event 75070: data truncated
(MLE) b0cot09:Messenger:10:16:37 PM->Runtime Error 2, Event 78599: data truncated
(MLE) b0cot00:Messenger:10:16:35 PM->Runtime Error 2, Event 78599: data truncated
(MLE) b0cot08:Messenger:10:16:35 PM->Runtime Error 2, Event 78599: data truncated
(MLE) b0cot01:Messenger:10:16:36 PM->Runtime Error 2, Event 78599: data truncated
(MLE) b0cot11:Messenger:10:16:35 PM->Runtime Error 2, Event 78599: data truncated
(MLE) b0cot02:Messenger:10:16:36 PM->Runtime Error 2, Event 78599: data truncated
(MLE) b0cot12:Messenger:10:16:36 PM->Runtime Error 2, Event 78599: data truncated
(MLE) b0cot10:Messenger:10:33:08 PM->Runtime Error 2, Event 100140: data truncated
(MLE) b0cot11:Messenger:10:33:06 PM->Runtime Error 3, Event 100140: data truncated
(MLE) b0cot01:Messenger:10:33:07 PM->Runtime Error 3, Event 100140: data truncated
(MLE) b0cot12:Messenger:10:33:07 PM->Runtime Error 3, Event 100140: data truncated
(MLE) b0cot12:Messenger:10:34:51 PM->Runtime Error 4, Event 102421: data truncated
(MLE) b0cot11:Messenger:10:34:51 PM->Runtime Error 4, Event 102421: data truncated
(MLE) b0cot09:Messenger:10:34:53 PM->Runtime Error 3, Event 102421: data truncated
(MLE) b0cot08:Messenger:10:34:51 PM->Runtime Error 3, Event 102421: data truncated
(MLE) b0cot02:Messenger:10:34:51 PM->Runtime Error 3, Event 102421: data truncated
(MLE) b0cot01:Messenger:10:34:51 PM->Runtime Error 4, Event 102421: data truncated
(MLE) b0cot00:Messenger:10:34:50 PM->Runtime Error 3, Event 102421: data truncated
(MLE) b0cot08:Messenger:10:47:46 PM->Runtime Error 4, Event 119282: data truncated
(MLE) b0cot11:Messenger:10:47:46 PM->Runtime Error 5, Event 119282: data truncated
(MLE) b0cot02:Messenger:10:47:47 PM->Runtime Error 4, Event 119282: data truncated
(MLE) b0cot01:Messenger:10:47:47 PM->Runtime Error 5, Event 119282: data truncated
(MLE) b0cot12:Messenger:10:47:47 PM->Runtime Error 5, Event 119282: data truncated
(MLE) b0cot00:Messenger:10:47:46 PM->Runtime Error 4, Event 119282: data truncated
(MLE) b0cot09:Messenger:10:47:49 PM->Runtime Error 4, Event 119282: data truncated
CT: 11/17/2001, 22:52'9"
52'9" 6 crate/s: b0l1cl00(128), b0l1cl01(129), b0l1cl02(130), b0l1cl03(131), b0l1cl04(132), b0l1cl05(133),  in
error.[RXPT]
(RC)   22:52:59 End -> IDLE
(EOR) End of Run report
(EOR) b0tsi00: triggered 124903/readout 124903/sent 124903 events.
(EOR) b0ccal00: triggered 124903/readout 124903/sent 124903 events.



<!-- Date: Sat Nov 17 23:00:49 2001 -->
<pre>We start cosmic run without COT. 
Run # 132656</pre>
<!-- Author: Denys & Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=132656' target=_top>132656</a>) -->


<!-- Date: Sat Nov 17 23:32:50 2001 -->
In current & previous cosmic_notrack runs, 
we looked at many monitoring plots & DF display.  
Saw some hot channels on various detectors, e.g. 
CEM PMT occ/event shows East (eta,phi) = (5,5R) is hot 
and West (eta,phi) = (0,10R) is warm.  
WHA PMT occ/event shows East (eta,phi) = (9,8R) is hot. 
CMP <widths> for Top, Layer 3, Stacks 66-69, 71-76 high.
<!-- Author: S. Errede :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=132655' target=_top>132655</a>) -->


<!-- Date: Sun Nov 18 00:00:58 2001 -->
<font color=red>Shift Summary: </font><pre>End eve. shift - still running cosmic_notracks, no COT. 
</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   123599-132656
Delivered Luminosity   0  
Acquired Luminosity    0  
Efficiency             -
   
</td></tr></table></pre>
<!-- Author: Steve Errede -->


<!-- Date: Sun Nov 18 00:02:40 2001 -->
<table border=0><td><tr><td bgcolor=#e0e030>Just wanna highlight for the next aces that when u  
include PCAL crates in the runs, you should edit  
the crates and remove slots:<br> 
pcal10 - slot 4<br> 
pcal08 - slot 6 
</td></tr></td></table>
<!-- Author: aces -->