<!-- Title: 01:30:45  Tue Nov  1 2005 -->
<!-- Crewchief: Dean -->
<!-- Op1: Jim H. - day -->
<!-- Op2: Bruce - night -->
<!-- Op3: &nbsp; -->
<!-- Op4: &nbsp; -->
<!-- Op5: &nbsp; -->
<!-- Op6: &nbsp; -->
<!-- Notes: -->
All system are running fine, the solenoid is powered to 4650 amps.


<!-- Date: Tue Nov 1 06:04:10 2005 -->
Historical Data is backed up to CDROM.
<!-- Author: Bill Noe -->


<!-- Date: Tue Nov 1 06:19:26 2005 -->
Cryo and gas readings are normal at this time. The CDF systems checklist and mechanical rooms are also normal at this time.
<!-- Author: Jim H. -->


<!-- Date: Tue Nov 1 13:41:09 2005 -->
We just received 2600 gallons of LN2.
<!-- Author: Jim H. -->


<!-- Date: Tue Nov 1 14:06:05 2005 -->
I took gas samples on trailer #321.
<!-- Author: KC -->


<!-- Date: Tue Nov 1 14:07:26 2005 -->
Rack alarm. Motor generator #3 glitch from disabling the silicon system.
<!-- Author: Jim H. -->


<!-- Date: Tue Nov 1 14:09:31 2005 -->
Silicon alarm. Both ISL nad SVX flows just tripped off.
<!-- Author: Jim H. -->
<!-- Comment: Tue Nov  1 14;13;58 comment by...Jim H. -->
Ulrich said the flows were tripped off for maintenance.


<!-- Date: Tue Nov 1 14:49:32 2005 -->
I talked to the sci-co Andy H. and he requested to fill the CLC west chamber. I filled it from 21.65 to 22.52psia.
<!-- Author: Jim H. -->


<!-- Date: Tue Nov 1 14:49:43 2005 -->
I replaced a 3 ft. 1/4 in. hose that went from VPla on the ISL manifold to the electrical pnuematic valve that controls the vacuum pump that was leaking.
<!-- Author: KC -->


<!-- Date: Tue Nov 1 15:00:10 2005 -->
Rack alarm. The silicon racks just tripped off. I'm not sure why at this time.
<!-- Author: Jim H. -->
<!-- Comment: Tue Nov  1 15;57;55 comment by...Jim H. -->
The silicon experts requested to reset the silicon racks. All 8 racks are back on.
<!-- Date: Tue Nov 1 15:55:24 2005 -->
Morris B. and Bob W. requested to increase Ar/O2 flow from .1 to .125scfh. Del A. increased the flow rate.
<!-- Author: Jim H. -->


<!-- Date: Tue Nov 1 15:56:09 2005 -->
Dee Hahn called and requested to put the HVAC on low for the access.
<!-- Author: Jim H. -->



<!-- Date: Tue Nov 1 16:01:39 2005 -->
I talked to Dell A. about the connector for corken compressor #1s heat exchanger. The plan is to repair it on thursday.
<!-- Author: Jim H. -->


<!-- Date: Tue Nov 1 16:45:34 2005 -->
The COT west racks were tripped off. The sci-co called and requested to reset them. All 4 racks are back on now.
<!-- Author: Jim H. -->


<!-- Date: Tue Nov 1 16:46:43 2005 -->
I just changed the east side gas shed pneumatic bottles. The west side is online now.
<!-- Author: Jim H. -->


<!-- Date: Tue Nov 1 17:51:04 2005 -->
Bruce on duty.
<!-- Author: Bruce -->


<!-- Date: Tue Nov 1 18:03:18 2005 -->
Gas and cryo readings taken all systems are running fine at this time.
<!-- Author: Bruce -->


<!-- Date: Tue Nov 1 18:32:54 2005 -->
HVAC is back in the full hvac mode the access is over.
<!-- Author: Bruce -->


<!-- Date: Tue Nov 1 19:00:30 2005 -->
Powered the west elcove as per the controls request.
<!-- Author: Bruce -->


<!-- Date: Tue Nov 1 21:23:56 2005 -->
Started mixing argon/ethane into tank #1 at this time.
<!-- Author: Bruce -->