<!-- Title: Day shift. Sat Jul 27, 2002 -->
<!-- SciCo: melissa franklin -->
<!-- DAQAce: jim Kraus -->
<!-- MonAce: min suk kim -->
<!-- CO: art garfinkel -->
<!-- OpManager: arnd meyer -->
<!-- Notes: -->
<pre>no beam - expect tevatron studies for most of this shift</pre>

<!-- Date: Sat Jul 27 10:29:03 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148968">148968</A>
 Terminated at 2002.07.27 10:29:25
<!-- Author: RunControl -->


<!-- Date: Sat Jul 27 10:29:03 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148968">148968</A>
TERMINATE: Ending run to release Hadron timing and trigger crates.
<!-- Author: Jim Kraus -->


<!-- Date: Sat Jul 27 10:56:36 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148970">148970</A>
 Activated at 2002.07.27 10:57:04
<!-- Author: RunControl -->
<!-- Comment: Sat Jul 27 11;49;45 comment by...Jim Kraus -->
Taken with no L1 or L2 trigger crates.  Calib continuous and L2 Auto accept.
<!-- Date: Sat Jul 27 10:56:36 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148970">148970</A>
ACTIVATE: Cosmics run.
<!-- Author: Jim Kraus -->



<!-- Date: Sat Jul 27 14:48:13 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148970">148970</A>
 Terminated at 2002.07.27 14:48:49
<!-- Author: RunControl -->


<!-- Date: Sat Jul 27 14:48:33 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148970">148970</A>
TERMINATE: We ended the run to clean up the EVB because of an SCPU error.
<!-- Author: Jim Kraus -->


<!-- Date: Sat Jul 27 15:05:43 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148982">148982</A>
 Activated at 2002.07.27 15:06:26
<!-- Author: RunControl -->


<!-- Date: Sat Jul 27 15:06:00 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148982">148982</A>
 Terminated at 2002.07.27 15:06:52
<!-- Author: RunControl -->


<!-- Date: Sat Jul 27 15:06:09 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148982">148982</A>
TERMINATE: End to do calibrations.
<!-- Author: Jim Kraus -->


<!-- Date: Sat Jul 27 15:10:58 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148983">148983</A>
ACTIVATE: CLC QIE calibration.
<!-- Author: Jim Kraus -->


<!-- Date: Sat Jul 27 15:12:50 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148983">148983</A>
TERMINATE: <pre>QIE calibrations finished.  The following messages were recorded in the Error Display window:
(b0clc00) QIE calib: slot 04, qie 00:Warn  4: Raw ped RMS high   , val 16.5843, cap 0, rng0.
(b0clc00) QIE calib: slot 07, qie 13:Warn  4: Raw ped RMS high   , val 15.9345, cap 3, rng0.
(b0clc00) QIE calib: slot 09, qie 04:Warn  3: Raw ped low        , val 64.8750, cap 0, rng0.
(b0clc00) QIE calib: slot 09, qie 05:Warn  3: Raw ped low        , val 70.7850, cap 0, rng0.
(b0clc00) QIE calib: slot 09, qie 06:Warn  3: Raw ped low        , val 78.5850, cap 0, rng0.
(b0clc00) QIE calib: slot 09, qie 07:Warn  3: Raw ped low        , val 52.1650, cap 0, rng0.
(MLE) b0clc00:Readout_Calib:3:11:00 PM->ADMEM=9 QIE=10 Fail  5: Raw ped low         (26.555) (0) (0)
(b0clc00) QIE calib: slot 09, qie 10:Fail  5: Raw ped low        , val 26.5550, cap 0, rng0.
(b0clc00) QIE calib: slot 10, qie 02:Fail  6: Raw ped RMS high   , val 20.2012, cap 2, rng0.</pre>
<!-- Author: Jim Kraus -->


<!-- Date: Sat Jul 27 15:20:41 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148984">148984</A>
ACTIVATE: Muon calibration run.
<!-- Author: Jim Kraus -->



<!-- Date: Sat Jul 27 15:21:21 2002 -->
 PEM/PHA/PSH HB alarm. restarted HV monitoring on upstairs.
<!-- Author: Minsuk -->

<!-- Date: Sat Jul 27 15:22:38 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148984">148984</A>
TERMINATE: Muon calibration finished with no errors reported.
<!-- Author: Jim Kraus -->


<!-- Date: Sat Jul 27 15:27:23 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148986">148986</A>
ACTIVATE: BSC-Miniplug calibration run.
<!-- Author: Jim Kraus -->


<!-- Date: Sat Jul 27 15:40:56 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148986">148986</A>
TERMINATE: <pre>BSC-Miniplug calibraions finished;  we got the following messages in the Error Logger:

b0fcal00 :  Messenger : 3:26:52 PM
 -> QIE calibration failed

b0fcal00 :  Readout_Calib : 3:26:52 PM
 -> ADMEM=16 QIE=14 Fail 13: Slope low           (0.0010348854) (1) (0)

b0fcal01 :  Messenger : 3:26:56 PM
 -> QIE calibration failed

b0fcal01 :  Readout_Calib : 3:26:56 PM
 -> ADMEM=16 QIE=11 Fail 10: No fitting          (0.0) (0) (2)</pre>
<!-- Author: Jim Kraus -->
<!-- Comment: Sat Jul 27 15;45;17 comment by...Jim Kraus -->
<pre>Here are some addition messages from this run:


(b0clc00) QIE calib: slot 16, qie 01:Warn  8: Sigma(slp)/slp) big, val 00.0033, cap 1, rng0.
(b0clc00) QIE calib: slot 16, qie 02:Warn  8: Sigma(slp)/slp) big, val 00.0032, cap 0, rng0.
(b0clc00) QIE calib: slot 16, qie 04:Warn  8: Sigma(slp)/slp) big, val 00.0032, cap 0, rng0.
(b0clc00) QIE calib: slot 16, qie 05:Warn 11: Sigma(off) high    , val 01.6191, cap 0, rng0.
(b0clc00) QIE calib: slot 16, qie 05:Warn  8: Sigma(slp)/slp) big, val 00.0039, cap 0, rng0.
(b0clc00) QIE calib: slot 16, qie 06:Warn 11: Sigma(off) high    , val 01.6568, cap 0, rng0.
(b0clc00) QIE calib: slot 16, qie 06:Warn  8: Sigma(slp)/slp) big, val 00.0042, cap 0, rng0.
(b0clc00) QIE calib: slot 16, qie 07:Warn  8: Sigma(slp)/slp) big, val 00.0034, cap 0, rng0.
(b0clc00) QIE calib: slot 16, qie 07:Warn 11: Sigma(off) high    , val 01.5091, cap 2, rng0.
(b0clc00) QIE calib: slot 16, qie 08:Warn  8: Sigma(slp)/slp) big, val 00.0032, cap 0, rng0.
(b0clc00) QIE calib: slot 16, qie 09:Warn  8: Sigma(slp)/slp) big, val 00.0032, cap 0, rng0.
(b0clc00) QIE calib: slot 16, qie 12:Warn  8: Sigma(slp)/slp) big, val 00.0034, cap 0, rng0.
(b0clc00) QIE calib: slot 16, qie 13:Warn  8: Sigma(slp)/slp) big, val 00.0034, cap 0, rng0.
(b0clc00) QIE calib: slot 17, qie 00:Warn 11: Sigma(off) high    , val 02.5417, cap 0, rng0.
(b0clc00) QIE calib: slot 17, qie 00:Warn  8: Sigma(slp)/slp) big, val 00.0054, cap 0, rng0.
(b0clc00) QIE calib: slot 17, qie 01:Warn 11: Sigma(off) high    , val 02.7858, cap 0, rng0.
(b0clc00) QIE calib: slot 17, qie 01:Warn  8: Sigma(slp)/slp) big, val 00.0067, cap 0, rng0.
(b0clc00) QIE calib: slot 17, qie 04:Warn 11: Sigma(off) high    , val 02.1546, cap 0, rng0.
(b0clc00) QIE calib: slot 17, qie 04:Warn  8: Sigma(slp)/slp) big, val 00.0043, cap 0, rng0.
(b0clc00) QIE calib: slot 17, qie 05:Warn 11: Sigma(off) high    , val 02.5134, cap 0, rng0.
(b0clc00) QIE calib: slot 17, qie 05:Warn  8: Sigma(slp)/slp) big, val 00.0058, cap 0, rng0.
(b0clc00) QIE calib: slot 17, qie 08:Warn 11: Sigma(off) high    , val 02.2198, cap 0, rng0.
(b0clc00) QIE calib: slot 17, qie 08:Warn  8: Sigma(slp)/slp) big, val 00.0048, cap 0, rng0.
(b0clc00) QIE calib: slot 17, qie 09:Warn 11: Sigma(off) high    , val 02.4584, cap 0, rng0.
(b0clc00) QIE calib: slot 17, qie 09:Warn  8: Sigma(slp)/slp) big, val 00.0063, cap 0, rng0.
(MLE) b0fcal00:Messenger:3:26:52 PM->QIE calibration failed
(b0fcal00) QIE calib: slot 16, qie 06:Warn  9: Slope high         , val 03.6591, cap 0, rng0.
(b0fcal00) QIE calib: slot 14, qie 08:Warn  9: Slope high         , val 03.7343, cap 0, rng0.
(b0fcal00) QIE calib: slot 12, qie 01:Warn  9: Slope high         , val 03.6573, cap 0, rng0.
(b0fcal00) QIE calib: slot 10, qie 04:Warn  9: Slope high         , val 03.6661, cap 0, rng0.
(MLE) b0fcal00:Readout_Calib:3:26:52 PM->ADMEM=16 QIE=14 Fail 13: Slope low           (0.0010348854) (1) (0)
(b0fcal00) QIE calib: slot 16, qie 14:Fail 13: Slope low          , val 00.0010, cap 1, rng0.
(b0fcal00) QIE calib: slot 14, qie 10:Fail  5: Raw ped low        , val 00.0000, cap 0, rng0.
(b0fcal00) QIE calib: slot 06, qie 05:Fail 10: No fitting         , val 00.0000, cap 1, rng0.
(MLE) b0fcal01:Messenger:3:26:56 PM->QIE calibration failed
(b0fcal01) QIE calib: slot 16, qie 00:Warn  2: Raw ped high       , val 451.3100, cap 0, rng0.
(b0fcal01) QIE calib: slot 16, qie 00:Warn  8: Sigma(slp)/slp) big, val 00.0031, cap 1, rng0.
(b0fcal01) QIE calib: slot 16, qie 01:Warn  9: Slope high         , val 03.6518, cap 0, rng0.
(b0fcal01) QIE calib: slot 16, qie 04:Warn  8: Sigma(slp)/slp) big, val 00.0033, cap 0, rng0.
(b0fcal01) QIE calib: slot 16, qie 05:Warn  9: Slope high         , val 03.7090, cap 0, rng0.
(b0fcal01) QIE calib: slot 16, qie 05:Warn  8: Sigma(slp)/slp) big, val 00.0032, cap 0, rng0.
(b0fcal01) QIE calib: slot 16, qie 06:Warn  2: Raw ped high       , val 467.3500, cap 0, rng0.
(b0fcal01) QIE calib: slot 16, qie 09:Warn  8: Sigma(slp)/slp) big, val 00.0033, cap 1, rng0.
(b0fcal01) QIE calib: slot 16, qie 18:Warn  9: Slope high         , val 00.0300, cap 0, rng4.
(b0fcal01) QIE calib: slot 14, qie 00:Warn  8: Sigma(slp)/slp) big, val 00.0034, cap 0, rng0.
(b0fcal01) QIE calib: slot 14, qie 00:Warn  2: Raw ped high       , val 452.7337, cap 3, rng0.
(b0fcal01) QIE calib: slot 14, qie 01:Warn  9: Slope high         , val 03.6530, cap 0, rng0.
(b0fcal01) QIE calib: slot 14, qie 01:Warn  8: Sigma(slp)/slp) big, val 00.0034, cap 0, rng0.
(b0fcal01) QIE calib: slot 14, qie 04:Warn  9: Slope high         , val 03.6702, cap 0, rng0.
(b0fcal01) QIE calib: slot 14, qie 04:Warn  8: Sigma(slp)/slp) big, val 00.0033, cap 0, rng0.
(b0fcal01) QIE calib: slot 14, qie 05:Warn  8: Sigma(slp)/slp) big, val 00.0031, cap 1, rng0.
(b0fcal01) QIE calib: slot 14, qie 10:Warn  8: Sigma(slp)/slp) big, val 00.0031, cap 0, rng0.
(b0fcal01) QIE calib: slot 14, qie 14:Warn  8: Sigma(slp)/slp) big, val 00.0031, cap 1, rng0.
(b0fcal01) QIE calib: slot 14, qie 15:Warn  8: Sigma(slp)/slp) big, val 00.0037, cap 0, rng0.
(b0fcal01) QIE calib: slot 14, qie 17:Warn  9: Slope high         , val 03.7216, cap 0, rng0.
(b0fcal01) QIE calib: slot 14, qie 18:Warn  8: Sigma(slp)/slp) big, val 00.0030, cap 1, rng0.
(b0fcal01) QIE calib: slot 14, qie 19:Warn  9: Slope high         , val 03.6603, cap 0, rng0.
(b0fcal01) QIE calib: slot 14, qie 19:Warn  8: Sigma(slp)/slp) big, val 00.0037, cap 0, rng0.
(b0fcal01) QIE calib: slot 12, qie 01:Warn  8: Sigma(slp)/slp) big, val 00.0040, cap 0, rng0.
(b0fcal01) QIE calib: slot 12, qie 03:Warn  8: Sigma(slp)/slp) big, val 00.0037, cap 0, rng0.
(b0fcal01) QIE calib: slot 12, qie 07:Warn  8: Sigma(slp)/slp) big, val 00.0033, cap 0, rng0.
(b0fcal01) QIE calib: slot 12, qie 09:Warn  8: Sigma(slp)/slp) big, val 00.0036, cap 0, rng0.
(b0fcal01) QIE calib: slot 12, qie 11:Warn  8: Sigma(slp)/slp) big, val 00.0036, cap 0, rng0.
(b0fcal01) QIE calib: slot 12, qie 13:Warn  8: Sigma(slp)/slp) big, val 00.0040, cap 0, rng0.
(b0fcal01) QIE calib: slot 12, qie 13:Warn 11: Sigma(off) high    , val 01.6020, cap 0, rng3.
(b0fcal01) QIE calib: slot 12, qie 15:Warn  8: Sigma(slp)/slp) big, val 00.0039, cap 0, rng0.
(b0fcal01) QIE calib: slot 12, qie 15:Warn 11: Sigma(off) high    , val 01.5321, cap 1, rng0.
(b0fcal01) QIE calib: slot 12, qie 19:Warn 11: Sigma(off) high    , val 01.6611, cap 0, rng1.
(b0fcal01) QIE calib: slot 10, qie 03:Warn  8: Sigma(slp)/slp) big, val 00.0030, cap 3, rng0.
(b0fcal01) QIE calib: slot 10, qie 05:Warn  8: Sigma(slp)/slp) big, val 00.0031, cap 0, rng0.
(b0fcal01) QIE calib: slot 10, qie 10:Warn  8: Sigma(slp)/slp) big, val 00.0032, cap 0, rng0.
(b0fcal01) QIE calib: slot 10, qie 10:Warn  9: Slope high         , val 00.4563, cap 0, rng2.
(b0fcal01) QIE calib: slot 10, qie 11:Warn  2: Raw ped high       , val 461.2400, cap 0, rng0.
(b0fcal01) QIE calib: slot 10, qie 16:Warn  9: Slope high         , val 03.6519, cap 0, rng0.
(b0fcal01) QIE calib: slot 10, qie 19:Warn  9: Slope high         , val 03.6550, cap 0, rng0.
(b0fcal01) QIE calib: slot 08, qie 00:Warn  8: Sigma(slp)/slp) big, val 00.0033, cap 0, rng0.
(b0fcal01) QIE calib: slot 08, qie 00:Warn  2: Raw ped high       , val 474.0500, cap 0, rng0.
(b0fcal01) QIE calib: slot 08, qie 10:Warn  8: Sigma(slp)/slp) big, val 00.0033, cap 0, rng0.
(b0fcal01) QIE calib: slot 08, qie 12:Warn  2: Raw ped high       , val 454.9550, cap 0, rng0.
(b0fcal01) QIE calib: slot 08, qie 15:Warn  2: Raw ped high       , val 474.0550, cap 0, rng0.
(b0fcal01) QIE calib: slot 06, qie 08:Warn  8: Sigma(slp)/slp) big, val 00.0034, cap 0, rng0.
(b0fcal01) QIE calib: slot 06, qie 09:Warn  8: Sigma(slp)/slp) big, val 00.0030, cap 0, rng0.
(b0fcal01) QIE calib: slot 06, qie 13:Warn  8: Sigma(slp)/slp) big, val 00.0034, cap 0, rng0.
(b0fcal01) QIE calib: slot 06, qie 14:Warn  8: Sigma(slp)/slp) big, val 00.0030, cap 3, rng0.
(b0fcal01) QIE calib: slot 06, qie 17:Warn  8: Sigma(slp)/slp) big, val 00.0031, cap 0, rng0.
(b0fcal01) QIE calib: slot 06, qie 18:Warn 11: Sigma(off) high    , val 01.7394, cap 0, rng0.
(b0fcal01) QIE calib: slot 06, qie 18:Warn  8: Sigma(slp)/slp) big, val 00.0045, cap 0, rng0.
(MLE) b0fcal01:Readout_Calib:3:26:56 PM->ADMEM=16 QIE=11 Fail 10: No fitting          (0.0) (0) (2)
(b0fcal01) QIE calib: slot 16, qie 11:Fail 10: No fitting         , val 00.0000, cap 0, rng2.
(b0fcal01) QIE calib: slot 12, qie 00:Fail  3: Too many warnings  , val 03.0000, cap 0, rng0.
(b0fcal01) QIE calib: slot 12, qie 05:Fail  3: Too many warnings  , val 03.0000, cap 0, rng0.
(b0fcal01) QIE calib: slot 12, qie 12:Fail  4: Raw ped high       , val 512.3300, cap 0, rng0.
(b0fcal01) QIE calib: slot 06, qie 16:Fail 14: Sigma(off) big     , val 05.8958, cap 0, rng0.</pre>

<!-- Date: Sat Jul 27 15:51:21 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148992">148992</A>
ACTIVATE: COT calibration run.
<!-- Author: Jim Kraus -->


<!-- Date: Sat Jul 27 15:55:35 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148992">148992</A>
TERMINATE: COT calibration run ended with no errors.
<!-- Author: Jim Kraus -->


<!-- Date: Sat Jul 27 15:55:35 2002 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>15:55:02</TD>                                                       
<TD ALIGN=CENTER>::</TD>                                                
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Sat Jul 27 15:55:35 2002 -->
<font color=red>Shift Summary: </font><pre>no beam - accelerator studies - laser calibrations on the central calorimeter 
tof studies 
</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   
Delivered Luminosity   0  
Acquired Luminosity    0  
Efficiency             ?
   
</td></tr></table></pre>
<!-- Author: melissa franklin -->