<!-- Title: Day shift. Fri Mar 16, 2007 -->
<!-- SciCo: K. Hara -->
<!-- DAQAce: Barbara Alvarez -->
<!-- MonAce: &nbsp; -->
<!-- CO: Howard Budd -->
<!-- OpManager: Max Goncharov -->
<!-- Notes: -->
<pre>colliding at 162E30 
</pre>


<!-- Date: Fri Mar 16 08:27:21.390 2007 -->
The scrapbook is not working i am going to do the plots for the last hour.
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->
<!-- Comment: Fri Mar 16 08;37;17 comment by...convery -->
The sequencer was stuck with ERR cdf_plot_start.  I restarted it.  It now says "Scraping is already complete.  Waiting for an hour since the last plot..."  We'll see what happens an hour from now.
<!-- Comment: Fri Mar 16 08;42;41 comment by...Michal -->
I restarted it also around 8:10 when it told me, that it is going to wait for an hour since the last plot... Maybe we have to get Steve Hahn to make his magic touch.
<!-- Comment: Fri Mar 16 11;42;59 comment by...Steve Hahn -->
<p>Roberto Dervin Jonathon and I have been working for about two
hours on PSM problems.  First we found that we are seeing the
same overshoots and spikes on the 2nd floor that we are seeing
on the 1st, but we only have errors on the 1st floor.  We had
several problems with the VOLTMAN PC, which meant we had to
reboot once.  Pilot error: we also accidentally rebooted CDFTEMP,
which meant PTM went grey in the global alarms along with PSM.
Since about 0130 last night, we have consistently been getting
PSM errors from 1RR12E_3, the middle of five CAMAC crates.  So,
Roberto and Dervin tried replacing the B&B (monitoring) box for
this crate.  Unfortunately, despite the B&B box testing OK, it
did not work with the CAMAC crate.  In fact, this killed the
entire readout daisy chain for all 1st floor crates.  In the
end, Roberto and Dervin replaced the original B&B box.  For
the moment, we are done, but nothing has been improved at this
point.</p>
<!-- Date: Fri Mar 16 08:31:08.843 2007 -->
<center><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-2839&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-2839' width='300'></a><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-2840&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-2840' width='300'></a><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-2841&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-2841' width='300'></a></center>
<!-- Author: barbara -->


<!-- Date: Fri Mar 16 08:31:59.184 2007 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2007&action=view&page=-2842&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2007&action=view&page=-2842  width='300' align='top'></a></td><td>pinkie (DVDD B0W2L0)</td></tr></table>
<!-- Author: hara -->




<!-- Date: Fri Mar 16 08:43:58.603 2007 -->
<pre>Done Timeout detected in CLC_00 crate. 
Auto HRR. 
</pre>
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->


<!-- Date: Fri Mar 16 08:52:02.077 2007 -->
BSU+CSP are red in High Voltage Summary Page 
paged expert - They know this since 3 days ago. We are expected to check the occupancy plots, and if they look normal we can ignore.
<!-- Author: hara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->
<!-- Comment: Fri Mar 16 08;57;37 comment by...barbara -->
I have checked the occupancy plots, but there is not occupancy plots for BSU. The other plots are fine!


<!-- Comment: Fri Mar 16 09;24;00 comment by...barbara -->
Suddenly, CSP and BSU went to green. 
<!-- Comment: Fri Mar 16 09;31;45 comment by...Akram -->
After some times communication between SSU&PC returned
<!-- Comment: Fri Mar 16 10;04;19 comment by...Phil -->
BSU occupancy is in the YMon Consumer plots
<!-- Comment: Fri Mar 16 10;05;34 comment by...Phil -->
Also I mentioned to Mikhal and at the 8am mtg, the most recent failures that caused bad data would all be flagged by PSM as the -5V power supplied died.
<!-- Date: Fri Mar 16 09:00:03.035 2007 -->
There are a few plots in the Tev elog regarding the losses this morning.  It's not clear to me if they are blaming the higher losses on tunes being in the wrong place. 
<br> 
08:32:33-   Tunes, both proton and pbar, and how they landed. We knobbed up the pbar tunes pretty quickly, but then needed to lower the proton tunes to get the proton halo losses down.  
 - prk  
<br>
<!-- Author: convery -->
<!-- Comment: Fri Mar 16 09;26;07 comment by...convery -->
The proton emittance also blew up during scraping.  It seems that both this and the high losses are blamed on an orbit smooth between stores, and also more pbars than the tune was set for.
<!-- Date: Fri Mar 16 09:05:18.862 2007 -->
<pre>Done Timeout detected in COT_02 crate. 
Auto HRR. 
</pre>
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->


<!-- Date: Fri Mar 16 09:08:04.363 2007 -->
<pre>Done Timeout detected in CLC_00 crate. 
Auto HRR. 
</pre>
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->

<!-- Date: Fri Mar 16 09:12:19.364 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236652>236652</A>
RUNSTATUS: <PRE>Marked Bad, explanation:
SVX 
ISL 
L00 
SVT 
</PRE>
<!-- Author: cdfscico -->





<!-- Date: Fri Mar 16 09:27:29.060 2007 -->
The scrapbook is working and the plots are ok.
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->



<!-- Date: Fri Mar 16 09:48:27.897 2007 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2007&action=view&page=-2843&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2007&action=view&page=-2843  width='300' align='top'></a></td><td>pinkie (DVDD B0W2L0), recovered</td></tr></table>
<!-- Author: hara -->


<!-- Date: Fri Mar 16 09:57:39.226 2007 -->
S. Hahn is going to reboot PSM computer.
<!-- Author: hara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->
<!-- Comment: Fri Mar 16 10;04;06 comment by...barbara -->
At the same time that PSM was red, we have also seen PTM red. Now PSM and PTM are coming back. 
<!-- Date: Fri Mar 16 10:00:09.260 2007 -->
Done Timeout detected in CLC_00 crate. 
Auto HRR.  
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->


<!-- Date: Fri Mar 16 10:02:07.870 2007 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2007&action=view&page=-2845&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2007&action=view&page=-2845  width='300' align='top'></a></td><td>pinkie, third time (DVDD B0W2L0)</td></tr></table>
<!-- Author: hara -->





<!-- Date: Fri Mar 16 10:09:48.384 2007 -->
Done Timeout detected in IMU_01 crate.  
Auto HRR. 
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->


<!-- Date: Fri Mar 16 10:22:49.299 2007 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2007&action=view&page=-2846&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2007&action=view&page=-2846  width='300' align='top'></a></td><td>pinkie 4th time (DVDD B0W2L0). The current value is very close to the limit (650).</td></tr></table>
<!-- Author: hara -->


<!-- Date: Fri Mar 16 10:27:39.508 2007 -->
<pre>*** Node b0cot13 in Error *** 
Last data more than 120s ago!  Node is not responsive 
Shepherd was successful.</pre>
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->


<!-- Date: Fri Mar 16 10:42:10.672 2007 -->
Done Timeout detected in HADRON_TIMING_00 crate 
Auto HRR
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->


<!-- Date: Fri Mar 16 10:43:45.859 2007 -->
Hourly plots checked, they are ok.
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->


<!-- Date: Fri Mar 16 11:01:26.378 2007 -->
<pre>Trigger Inhibit is set from the components: 
 VME Power  		[147] VME Crate Power Supplies from iFix 
</pre>
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->
<!-- Comment: Fri Mar 16 11;02;38 comment by...barbara -->
We are on halted state

<!-- Comment: Fri Mar 16 11;09;58 comment by...barbara -->
Taking data again, the inhibits have gone away.


<!-- Comment: Fri Mar 16 12;02;24 comment by...convery -->
Perhaps we should have masked the PSM inhibit during this work.
<!-- Date: Fri Mar 16 11:15:14.865 2007 -->
Although the consumers were OK and the displays were OK 
concon claimed they were bad. We restarted concon and it 
claimed the consumers were OK.
<!-- Author: Howard Budd :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->


<!-- Date: Fri Mar 16 11:16:56.617 2007 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2007&action=view&page=-2847&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2007&action=view&page=-2847  width='300' align='top'></a></td><td>pinkie, 4th time (DVDD B0W2L0)</td></tr></table>
<!-- Author: hara -->
<!-- Comment: Fri Mar 16 11;17;28 comment by...hara -->
sorry, 5th time


<!-- Date: Fri Mar 16 11:24:24.111 2007 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2007&action=view&page=-2848&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2007&action=view&page=-2848  width='300' align='top'></a></td><td>pinkie, 6th time (DVDD, B0W2L0)</td></tr></table>
<!-- Author: hara -->


<!-- Date: Fri Mar 16 11:25:46.926 2007 -->
concon decided all the monitors were bad again, 
we restarted on of the displays and then it claimed 
everything was OK
<!-- Author: Howard Budd -->
<!-- Comment: Fri Mar 16 13;14;30 comment by...hara -->
send e-mail to Naganoma to notice this at 11:50
<!-- Date: Fri Mar 16 11:27:44.940 2007 -->
Hourly plots checked, they are ok. 
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->


<!-- Date: Fri Mar 16 11:28:16.264 2007 -->
Done Timeout detected in CLC_00 crate. Auto HRR.  
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->



<!-- Date: Fri Mar 16 11:45:23.286 2007 -->
<pre>We are having several "high deadtimes", this problem has been seen  &lt;a href='/cgi/elog/elog.pl?nb=2007&action=view&page=204&frame=2&anchor=21:53:58.638'&gt;before&lt;/a&gt;</pre>
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->
<!-- Comment: Fri Mar 16 15;49;46 comment by...barbara -->
This entry is wrong, look 2 entried down for the good one. I am sorry
<!-- Date: Fri Mar 16 11:45:26.241 2007 -->
<p>Oops, accidentally put entry about PSM work as a comment  
to an earlier entry.  It's located  
<a href= 
'/cgi/elog/elog.pl?nb=2007&action=view&page=224&frame=2&anchor=08:27:21.390'> 
here</a>.</p>  
<!-- Author: Steve Hahn :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->


<!-- Date: Fri Mar 16 11:45:59.255 2007 -->
I will try it again: 
We are having several "high deadtimes", this problem has been seen  <a 
href='/cgi/elog/elog.pl?nb=2007&action=view&page=204&frame=2&anchor=21:53:58.638'>before</a>
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->


<!-- Date: Fri Mar 16 11:58:45.475 2007 -->
Done Timeout detected in IMU_00 crate. Auto HRR.  
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->



<!-- Date: Fri Mar 16 12:09:44.163 2007 -->
<pre> 
 Attention!!!. FERML_HIGH_DEADTIME !!!  
 - Trigger/DAQ Deadtime is too high (at 31.240328 %).  
 - Look for source of hot triggers.  
 - Contact a trigger expert for further assistance.  
</pre>
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->


<!-- Date: Fri Mar 16 12:24:50.186 2007 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2007&action=view&page=-2849&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2007&action=view&page=-2849  width='300' align='top'></a></td><td>The CMU plot is showing red, with a spike at spike at Wedge 14</td></tr></table>
<!-- Author: Howard Budd -->


<!-- Date: Fri Mar 16 12:35:40.963 2007 -->
<center><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-2850&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-2850' width='300'></a><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-2851&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-2851' width='300'></a></center>
<!-- Author: jdl -->
<!-- Comment: Fri Mar 16 12;42;23 comment by...jdl -->
Shown here are the deadtime and the rate for the L2_TWO_CMU1.5_PT1.5 trigger for the current run.  The spikes in the trigger rate are caused by noise storms in CMU14W.  When the trigger rate spikes, we get large deadtime.  There are three possible courses of action:
<ul>
<li>Mask off by hand dimuon and muon+gap triggers that are on DPS and lumi enable during storms
<li>Disable 14W by removing cables from TDC, rendering the data bad for CMU
<li>Live with brief periods of high deadtime
</ul>
My recommendation is to live with the problem unless the noise stays.  
<!-- Comment: Fri Mar 16 12;54;07 comment by...Phil -->
The noise is coming and going this morning. I saw one exceeding brief burst (blink and you missed it) and one of a little long duration (minutes). There obviously are some I didn't see. But it doesn't seem to be continuous as it was the other morning.
<!-- Date: Fri Mar 16 12:36:34.956 2007 -->
Hourly plots checked, they are ok. 
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->




<!-- Date: Fri Mar 16 13:03:15.273 2007 -->
Done Timeout detected in COT_07 crate. Auto HRR.  
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->


<!-- Date: Fri Mar 16 13:06:04.268 2007 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2007&action=view&page=-2852&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2007&action=view&page=-2852  width='300' align='top'></a></td><td>The Xmon plots with the CMU in them seem so be higher 
than the fit for this run. THe plot shown is an example</td></tr></table>
<!-- Author: Howard Budd -->

<!-- Date: Fri Mar 16 13:06:54.103 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653>236653</A>
muons: <tt>CMU 14W is has higher occupancy</tt>
<!-- Author: Consumer Operator( howard Budd) -->


<!-- Date: Fri Mar 16 13:06:55.657 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653>236653</A>
general: <tt>Check list done </tt>
<!-- Author: Consumer Operator( howard Budd) -->




<!-- Date: Fri Mar 16 13:24:53.780 2007 -->
<pre>TeVMon warning set!  
Lens2 has gone down from 10.5 to 2. 
Backing pump has tripped off. </pre>
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->
<!-- Comment: Fri Mar 16 13;26;55 comment by...hara -->
The problem is vacuum pump


<!-- Comment: Fri Mar 16 13;33;26 comment by...barbara -->
Lens 2 has came back. I will add the plot down
<!-- Comment: Fri Mar 16 13;34;02 comment by...hara -->
MCR called: lens is back and working.
<!-- Date: Fri Mar 16 13:32:34.657 2007 -->
Hourly plots checked, they are ok.  
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->




<!-- Date: Fri Mar 16 13:35:56.322 2007 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2007&action=view&page=-2853&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2007&action=view&page=-2853  width='300' align='top'></a></td><td>Lens 2 plot history</td></tr></table>
<!-- Author: barbara -->


<!-- Date: Fri Mar 16 13:36:32.759 2007 -->
Done Timeout detected in COT_07 crate. Auto HRR.  
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->


<!-- Date: Fri Mar 16 13:41:45.524 2007 -->
Done Timeout detected in IMU_00 crate. Auto HRR.  
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->


<!-- Date: Fri Mar 16 13:42:29.093 2007 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2007&action=view&page=-2854&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2007&action=view&page=-2855  width='300' align='top'></a></td><td>The noisy wedge 14W cmu wedge we were seeing now seems to be OK</td></tr></table>
<!-- Author: Howard Budd -->


<!-- Date: Fri Mar 16 13:49:28.351 2007 -->
PSM alarm. Comunication error for 1RR12E_3
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->


<!-- Date: Fri Mar 16 13:57:45.374 2007 -->
Done Timeout detected in CLC_00 crate. Auto HRR.  
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->


<!-- Date: Fri Mar 16 14:14:49.119 2007 -->
*** Node b0pcal10 in Error *** 
Last data more than 120s ago!  Node is not responsive 
Shepherd was successful!
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->


<!-- Date: Fri Mar 16 14:28:47.719 2007 -->
I have checked the hourly plots. They are fine.
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->


<!-- Date: Fri Mar 16 15:05:49.247 2007 -->
<pre>Done Timeout detected in FCAL_01 crate. Auto HRR. 
Again: Done Timeout detected in FCAL_01 crate. Shepherd was successful!</pre>
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->


<!-- Date: Fri Mar 16 15:06:41.348 2007 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2007&action=view&page=-2856&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2007&action=view&page=-2856  width='300' align='top'></a></td><td>CES East Phi Wedge 13 seems to be noisy</td></tr></table>
<!-- Author: Howard Budd -->


<!-- Date: Fri Mar 16 15:08:07.695 2007 -->
Done Timeout detected in CLC_00 crate. Auto HRR.  
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->


<!-- Date: Fri Mar 16 15:14:26.110 2007 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2007&action=view&page=-2857&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2007&action=view&page=-2857  width='300' align='top'></a></td><td>We get an error in DAQErrMon </td></tr></table>
<!-- Author: Howard Budd -->
<!-- Comment: Fri Mar 16 15;19;47 comment by...barbara -->
Now, this plots looks fine.

<!-- Comment: Fri Mar 16 15;34;56 comment by...barbara -->
We have talked with the daq expert,Sasha; if this happens again we must call him again. If this error just happens once this is no real problem.


<!-- Date: Fri Mar 16 15:35:21.484 2007 -->
I have checked the hourly plots. They are fine.
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->


<!-- Date: Fri Mar 16 15:37:59.864 2007 -->
EVB2B is red. HRR has fixed it!
<!-- Author: barbara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653' target=_top>236653</a>) -->


<!-- Date: Fri Mar 16 15:43:34.707 2007 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2007&action=view&page=-2858&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2007&action=view&page=-2858  width='300' align='top'></a></td><td>pinkie (DVDD, B0W2L0) again</td></tr></table>
<!-- Author: hara -->

<!-- Date: Fri Mar 16 15:44:04.332 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653>236653</A>
smx: <tt>CES East Wedge 13 seems to be a bit noisy</tt>
<!-- Author: Consumer Operator( howard Budd) -->


<!-- Date: Fri Mar 16 15:44:05.893 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653>236653</A>
muons: <tt>CMU 14E seems to be back as being noisy</tt>
<!-- Author: Consumer Operator( howard Budd) -->


<!-- Date: Fri Mar 16 15:44:06.705 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236653>236653</A>
general: <tt>check list done</tt>
<!-- Author: Consumer Operator( howard Budd) -->


<!-- Date: Fri Mar 16 15:49:46.481 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=236698>236698</A>
ACTIVATE: RunSet: DAQTEST cd ~cdf
<!-- Author: sartori -->



<!-- Date: Fri Mar 16 15:55:05.916 2007 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>PhysicsTable</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L3Accepts</TH>
<TH>DelivLumi nb<SUP>-1</SUP></TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN=236653 target=_runSummary>236653</A> x39C6D</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_4_01 [5,866,678]</FONT></TD>
<TD ALIGN=CENTER>07:58:09</TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>07:03:51</TD>
<TD ALIGN=RIGHT>   3,091,364</TD>   <TD ALIGN=RIGHT>  3,117.54</TD>
<TD ALIGN=RIGHT>  2,783.47</TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>15:55:00</TD>
<TD ALIGN=CENTER>07:03:51</TD>
<TD ALIGN=RIGHT>     3,091,364</TD> <TD ALIGN=RIGHT>  3,117.54</TD>
<TD ALIGN=RIGHT>  2,783.47</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Fri Mar 16 16:00:41.660 2007 -->
<font color=red>Shift Summary: </font><pre>shift started with colliding beam at 162E30 (store#5284) 
 
LOSTP, ~10kHz at beginning of shot, went down to 0.8 kHz level in 1/2 hr 
 
7 times pinkie alarm for the same ch (DVDD B0W2L0) 
 
work on PSM problem for 2hr (Steve and others)-rebooting VOLTMAN/replacing B&B cards - data taking halted for 7 min during this work 
 
deadtime reached ~20% for short period but time to time: CMU14W noise is producing high rate for L2_TWO_CMU1.5_PT1.5 
 
vac. pump of electron lens2 tripped, setting TeVMon at WARNING. back in 10 min.</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   236653
Delivered Luminosity   3100.66  
Acquired Luminosity    2769.60  
Efficiency             89.3
   
</td></tr></table></pre>
<!-- Author: hara -->