<!-- Title: Day shift. Tue Sep 26, 2006 -->
<!-- SciCo: Joel Goldstein -->
<!-- DAQAce: Jason Slaunwhite -->
<!-- MonAce: Varsha Ramakrishnan -->
<!-- CO: Jeff Miles -->
<!-- OpManager: Koji Sato -->
<!-- Notes: -->
<pre>Data taking smoothly now, store 4975, lumi 61e30</pre>


<!-- Date: Tue Sep 26 08:19:01.841 2006 -->
Taking fcal01 to investigate problems seen on evening shift.
<!-- Author: convery -->
<!-- Comment: Tue Sep 26 08;27;03 comment by...convery -->
Asked cryo to reset power supply on the east alcove crate (somehow I was not able to do it myself from the control room).
<!-- Comment: Tue Sep 26 08;29;46 comment by...convery -->
still getting the error<br>
ADMEM_setCafeDelay geo32:/slot=16 VME bus error: old/target/new 0xff/0x5a/ff [4] 

<!-- Comment: Tue Sep 26 08;36;04 comment by...convery -->
Also bunch counter mismatches between slot 8 and 16.  I guess this is a problem with slot 16.  We will request an access to replace the admem.
<!-- Comment: Tue Sep 26 08;36;50 comment by...convery -->
<table border=0><td><tr><td bgcolor=#e0e030>Shift crew should continue to exclude fcal01 and to mark the miniplug bad.  thanks</td></tr></td></table>
<!-- Comment: Tue Sep 26 10;24;36 comment by...K. Terashi -->
The fcal01 crate was fixed and it can be included in daq.
<!-- Date: Tue Sep 26 08:21:22.162 2006 -->
<pre>  Done TO. for crate/s:  
    b0clc00</pre> 
 
<p>HRR fixed it</p>  
<!-- Author: Jason :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223642' target=_top>223642</a>) -->
<!-- Comment: Tue Sep 26 08;27;06 comment by...Jason -->
<p>Again! Hrr worked. Sheparded the crate to try and fix it. Shepard successul. We'll see when I get the next DTO.</p>
<!-- Date: Tue Sep 26 08:25:43.612 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-12163&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-12163' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-12164&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-12164' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-12165&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-12165' width='300'></a></center>
<!-- Author: Varsha -->




<!-- Date: Tue Sep 26 08:34:16.779 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223642>223642</A>
 Terminated at 2006.09.26 08:33:59
<!-- Author: RunControl -->


<!-- Date: Tue Sep 26 08:34:29.457 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223642>223642</A>
TERMINATE: quench
<!-- Author: jason -->




<!-- Date: Tue Sep 26 08:36:18.691 2006 -->
<table border=0><td><tr><td bgcolor=#e0e030>Second RF station trips giving "clean" abort. MCR have called RF experts to investigate.</td></tr></td></table>
<!-- Author: Joel -->


<!-- Date: Tue Sep 26 08:38:36.635 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223652>223652</A>
TERMINATE: end post quench checkout
<!-- Author: jason -->



<!-- Date: Tue Sep 26 08:39:13.388 2006 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2006&action=view&page=-12166&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-12166  width='300' align='top'></a></td><td>ABORT PLOT</td></tr></table>
<!-- Author: Varsha -->


<!-- Date: Tue Sep 26 08:42:33.635 2006 -->
<p><b>CLC01 returns</b> -- just prior to the RF trip CLC01 returned as a "stuck port" in RT server window. I'll check it later and see if it looks the same.</p>
<!-- Author: Jason :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223642' target=_top>223642</a>) -->


<!-- Date: Tue Sep 26 08:43:00.469 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-12167&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-12167' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-12168&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-12168' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-12169&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-12169' width='300'></a></center>
<!-- Author: Varsha -->

<!-- Date: Tue Sep 26 08:47:54.953 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223653>223653</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: jason -->


<!-- Date: Tue Sep 26 08:48:57.604 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223653>223653</A>
TERMINATE: done!!1!
<!-- Author: jason -->


<!-- Date: Tue Sep 26 08:48:58.259 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223653>223653</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Tue Sep 26 10;31;20 comment by...Jeff -->
Checks okay -- nevermind my pasting the <a href="http://www-cdfonline.fnal.gov/cgi/internal/elog/elog.pl?nb=2006&action=copyentry&page=712&time=10:27:53.665">table below</a>.
<!-- Date: Tue Sep 26 08:50:08.376 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223654>223654</A>
ACTIVATE: RunSet: CALIB_CLC_QIE 
<!-- Author: jason -->


<!-- Date: Tue Sep 26 08:51:01.043 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223654>223654</A>
CALIBRATION: <U><B>clcqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#clcqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Tue Sep 26 10;34;46 comment by...Jeff -->
checked okay
<!-- Date: Tue Sep 26 08:52:15.212 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223654>223654</A>
TERMINATE: done~!
<!-- Author: jason -->


<!-- Date: Tue Sep 26 08:53:30.635 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223655>223655</A>
ACTIVATE: RunSet: CALIB_QIE_FWD 
<!-- Author: jason -->


<!-- Date: Tue Sep 26 08:54:22.745 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223655>223655</A>
CALIBRATION: <U><B>bscqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#bscqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Tue Sep 26 10;45;42 comment by...Jeff -->
The miniplug east signal towers are missing a cluster of 10 channels in the calibration. We could do the recovery procedure, which worked on Friday, but we've been excluding miniplug east, so I'll go see about that.  Everything else looks okay.
<!-- Date: Tue Sep 26 08:54:27.006 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223655>223655</A>
TERMINATE: done
<!-- Author: jason -->


<!-- Date: Tue Sep 26 09:01:13.849 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223656>223656</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL 
<!-- Author: jason -->


<!-- Date: Tue Sep 26 09:07:59.327 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223656>223656</A>
TERMINATE: ----=====:::: done ::::====------
<!-- Author: jason -->


<!-- Date: Tue Sep 26 09:09:15.841 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223657>223657</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: jason -->


<!-- Date: Tue Sep 26 09:11:33.803 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223657>223657</A>
TERMINATE: done
<!-- Author: jason -->


<!-- Date: Tue Sep 26 09:11:34.432 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223657>223657</A>
CALIBRATION: <U><B>pluglaser</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#pluglaser target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Tue Sep 26 10;47;39 comment by...Jeff -->
checks okay for the east towers. West laser is down until Friday.
<!-- Date: Tue Sep 26 09:12:38.115 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223658>223658</A>
ACTIVATE: RunSet: BOTHPLUGS_MM 
<!-- Author: jason -->


<!-- Date: Tue Sep 26 09:14:44.638 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223658>223658</A>
TERMINATE: done
<!-- Author: jason -->


<!-- Date: Tue Sep 26 09:20:49.657 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223659>223659</A>
 Activated at 2006.09.26 09:20:17
<!-- Author: RunControl -->


<!-- Date: Tue Sep 26 09:21:56.053 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223659>223659</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [7,792,632] 
<!-- Author: jason -->



<!-- Date: Tue Sep 26 09:25:29.713 2006 -->
<table border=0><td><tr><td bgcolor=#e0e030>MCR inform us that the 12 hour access is beginning now!</td></tr></td></table>
<!-- Author: Joel -->

<!-- Date: Tue Sep 26 09:29:41.202 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223659>223659</A>
 Terminated at 2006.09.26 09:29:23
<!-- Author: RunControl -->


<!-- Date: Tue Sep 26 09:30:36.109 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223659>223659</A>
TERMINATE: done with spike run
<!-- Author: jason -->



<!-- Date: Tue Sep 26 09:32:49.594 2006 -->
<P> 
Am working on AcnetMonitor and DownTimeCopy problems on  
b0server02;  ignore any ProcMon or other related alarms  
for now. 
<P>
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223659' target=_top>223659</a>) -->


<!-- Date: Tue Sep 26 09:33:56.179 2006 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2006&action=view&page=-12171&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-12171  width='300' align='top'></a></td><td>New dead PMT in WHA? (eta +8, phi 11)  Emailed experts.</td></tr></table>
<!-- Author: Jeff -->
<!-- Comment: Tue Sep 26 10;12;21 comment by...Jeff -->
As Larry points out, that is indeed eta +6, and the zero occupancy is very likely statistics
<!-- Date: Tue Sep 26 09:36:03.386 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223660>223660</A>
ACTIVATE: RunSet: ACE_TOF_QIE_NEW 
<!-- Author: jason -->


<!-- Date: Tue Sep 26 09:36:54.112 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223660>223660</A>
CALIBRATION: <U><B>tofqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#tofqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Tue Sep 26 10;50;19 comment by...Jeff -->
checked okay
<!-- Date: Tue Sep 26 09:37:11.487 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223660>223660</A>
TERMINATE: done
<!-- Author: jason -->



<!-- Date: Tue Sep 26 09:44:40.674 2006 -->
<p>Si Needs to use all Si crates + l3 + hardware evb+ csl ... I don't think I can run anything, actually. </p>
<!-- Author: Jason :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223660' target=_top>223660</a>) -->


<!-- Date: Tue Sep 26 09:46:46.211 2006 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2006&action=view&page=-12173&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-12173  width='300' align='top'></a></td><td>Somewhat high ADC counts in CES Wire West, phi wedge 15. </td></tr></table>
<!-- Author: Jeff -->
<!-- Comment: Tue Sep 26 10;46;17 comment by...Larry N -->
Coherent noise got out of hand in one event, no big deal!
<!-- Date: Tue Sep 26 10:00:40.658 2006 -->
<pre>The b0server02 machine runs the AcnetMonitor, DownTimeCopy  
and AcnetScaler applications, all of which use the AcNet  
java classes on the /cdf/acnet disk.  This disk is remotely  
mounted via a special network interface on b0server02 connected  
to the AcNet network.   Over the last week or so, these  
applications have add sporadic problems starting up and running. 
On the b0server02 system console, there were NFS errors, so  
I'm guessing there was some flakiness in the NFS mount  
of /cdf/acnet.    I have rebooted b0server02, and that  
seems to have cleared up the problem for now.   These applications  
have been restarted. 
</pre>
<!-- Author: badgett -->
<!-- Comment: Tue Sep 26 10;22;31 comment by...badgett -->
<table border=0><td><tr><td bgcolor=#e0e030>This should fix the problem where the DownTime logger 
entries from AcNet were not showing up on the Store 
summary plot.   Sorry about the delay, JJ!
</td></tr></td></table>
<!-- Date: Tue Sep 26 10:03:13.145 2006 -->
The XMon database and the <a href = 'https://www-cdfonline.fnal.gov/internal/mon/cdf_val/CrossSectionFits/main.html'>Official XMon Fits </a> web pages are updated with the new XMon fits generated using the stores: 4972 4970 4967 & 4963. The new XMon fits are based on the runs after the changes have been made in TL2D. Also these fits are generated using the modified fitting algorithm for the following changes. 1) Only those runs are used in fits for which there are more than 3 points. 2) The data points with zero cross sections are 'Excluded' from the fits. 3) If the first data point for every run is 20% below the other points, it is 'Excluded' from the fit. 4) All unused points are plotted in black color inverted triangles. 
 
For comparison, the old fits (before 09/01/2006) are available at  the following links: 
<p> 
<a href = 'https://www-cdfonline.fnal.gov/internal/mon/cdf_val/CrossSectionFits/official/L1old/_all.html'>L1 Old </a> 
</p> 
<p> 
<a href = 'https://www-cdfonline.fnal.gov/internal/mon/cdf_val/CrossSectionFits/official/L2old/_all.html'>L2 Old </a>  
</p> 
<p> 
<a href =  
'https://www-cdfonline.fnal.gov/internal/mon/cdf_val/CrossSectionFits/official/L3old/_all.html'>L3 Old </a> 
</p> 
 
 
  
<!-- Author: Nagesh :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223659' target=_top>223659</a>) -->
<!-- Comment: Tue Sep 26 10;11;57 comment by...Nagesh -->
More on point 3): In the actual algorithm, any point if it is way below (more than 20%) the other points on Cross section values, it is ignored. Usually the first point at the starting of every run has low cross section.
<!-- Date: Tue Sep 26 10:04:29.376 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223642>223642</A>
cal: <tt>Possible new dead PMT in WHA East -- but the run was low/medium statistics so maybe the occupancy just wasn't high enough.</tt>
<!-- Author: Consumer Operator( Jeff Miles ) -->
<!-- Comment: Tue Sep 26 10;54;17 comment by...Jeff -->
that was indeed the case.
<!-- Date: Tue Sep 26 10:04:31.320 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223642>223642</A>
smx: <tt>~32 channels have high ADC counts in CES Wire West -- not red-hot, but high. Marked okay, but asking experts for advice.</tt>
<!-- Author: Consumer Operator( Jeff Miles ) -->


<!-- Date: Tue Sep 26 10:04:32.418 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223642>223642</A>
muons: <tt>CMP part B has some new empty channels, but this is surely low statistics.</tt>
<!-- Author: Consumer Operator( Jeff Miles ) -->


<!-- Date: Tue Sep 26 10:04:33.179 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223642>223642</A>
monitors: <tt>Low statistics failures in EM/HAD TDC timing checks. Okay.</tt>
<!-- Author: Consumer Operator( Jeff Miles ) -->


<!-- Date: Tue Sep 26 10:04:36.798 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223634>223634</A>
RUNSTATUS: <PRE>Marked Bad, explanation:
Miniplug east side appears empty
</PRE>
<!-- Author: cdfscico -->




<!-- Date: Tue Sep 26 10:13:47.746 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223637>223637</A>
RUNSTATUS: <PRE>Marked Bad, explanation:
East miniplug not in run
</PRE>
<!-- Author: cdfscico -->


<!-- Date: Tue Sep 26 10:17:36.400 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223638>223638</A>
RUNSTATUS: <PRE>Marked Bad, explanation:
East Miniplug out of run
</PRE>
<!-- Author: cdfscico -->


<!-- Date: Tue Sep 26 10:21:49.031 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223641>223641</A>
RUNSTATUS: <PRE>Marked Bad, explanation:
East miniplug out of run
</PRE>
<!-- Author: cdfscico -->



<!-- Date: Tue Sep 26 10:23:23.688 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223669>223669</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_4_00 [7,833,649] 
<!-- Author: badgett -->


<!-- Date: Tue Sep 26 10:24:02.337 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223642>223642</A>
RUNSTATUS: <PRE>Marked Bad, explanation:
East miniplug out of run
</PRE>
<!-- Author: cdfscico -->




<!-- Date: Tue Sep 26 10:27:53.665 2006 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2006&action=view&page=-12175&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-12175  width='300' align='top'></a></td><td>calqie calibration has PHA wedge 20W full of zero slopes... a  few of these channels have bad pedestal means, but not most</td></tr></table>
<!-- Author: Jeff -->
<!-- Comment: Tue Sep 26 10;32;38 comment by...Jeff -->
Just realized these zero slopes are all PHA channel 0, which we are instructed to ignore... sorry.
<!-- Comment: Tue Sep 26 15;44;30 comment by...m mattson (ADMEM) -->
Its good that you noticed this, AND that you found the relevant instructions.
<!-- Date: Tue Sep 26 10:28:39.625 2006 -->
The problem on fcal01 crate was due to corrupted software on the ADMEM slot 16. The board was reconfigured using qietest code. Tested the crate with qie calibration and daq test runs and the crate looks ok. Shift crew: please include the fcal01 crate in L2 torture or cosmics, and if it is ok, then put it in data taking with beam. thanks.
<!-- Author: K. Terashi -->








<!-- Date: Tue Sep 26 10:48:55.951 2006 -->
<h3>Silicon noise-bias scan in progress</h3> 
 
<p>A silicon noise-bias scan is in progress. We should be done before the plug sourcing starts.</p> 
  
<!-- Author: Ankush -->
<!-- Comment: Tue Sep 26 15;32;48 comment by...Ankush -->
<p>The bias-scan was completed just after 2pm.</p>
<!-- Date: Tue Sep 26 10:50:39.824 2006 -->
<table border=0><td><tr><td bgcolor=#e0e030><pre> 
restarted the rtserver. 
 
</pre></td></tr></td></table>
<!-- Author: Sasha Golossanov -->
<!-- Comment: Tue Sep 26 11;03;32 comment by...Sasha Golossanov -->
<pre>
the upgraded mv5500 has been tested and programmed to be b0cot08; it is ready to be installed.



</pre>
<!-- Date: Tue Sep 26 11:03:10.935 2006 -->
CMX HV is now under expert control - don't change unless instructed.  For the record NE 18-19+20 are being ramped down in prep for work to be done there.
<!-- Author: Dale Stentz -->
<!-- Comment: Tue Sep 26 11;29;52 comment by...Dale -->
SW CMX arch opened without any hasle (as expected).  Closing will be a bit more interesting.
<!-- Comment: Tue Sep 26 12;06;16 comment by...Dale and Phil -->
These wedges are now at full HV to test the switch we made.  We'll keep it this way until the end of the access.  Summary of CMX work to be entered at a later time.
<!-- Comment: Tue Sep 26 15;26;25 comment by...Dale, Phil, and Andy -->
CMX HV control is returned back to the control of the shift crew.  HV was left at stand-by but taking a cosmic run with HV on while we have quite time would be helpful.  (<i>summary to be linked below</i>)
<!-- Comment: Tue Sep 26 20;09;51 comment by...Dale, Phil, and Andy -->
better late then never...
<h3>CMX access summary:</h3>
We reconfigured the HV in two places in CMX.  First the NE CMX wedges 18-20 was arranged to its default configuration (away from <a href=
'/cgi/elog/elog.pl?nb=2006&action=view&page=481&frame=2&anchor=10:22:29.698'>
this</a>).  The trips in this area might be affected for the better but the work really just helps to narrow down to a smaller section.<br><br>
The second item was to do a cable swap at the NW wedges 21 and 22-23.  These wedges have been effectively "double swapped" to test whether the trips in the area are due to the cable (unlikely) or if it is truly a problem at the chamber level.<br><br>
Both changes are transparent to the shift crew and require no special instructions.  Further details can be found in the CMX e-log.
<!-- Date: Tue Sep 26 11:13:36.564 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=218174>218174</A>
RUNSTATUS: <PRE>Marked Bad, explanation:
No silicon in run.
CO found problems with everything else
</PRE>
<!-- Author: cdfscico -->



<!-- Date: Tue Sep 26 11:17:45.227 2006 -->
<table border=0><td><tr><td bgcolor=#e0e030><P> 
<H4>Jason Hedden will be patching the b0server00...04 and  
b0server06...11 machines now.  Expect short downtimes. 
</H4> 
<P> 
</td></tr></td></table>
<!-- Author: badgett -->

<!-- Date: Tue Sep 26 11:23:36.070 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=218176>218176</A>
RUNSTATUS: <PRE>Marked Bad, explanation:
Silicon out of run; CO found problems with other components
</PRE>
<!-- Author: cdfscico -->


<!-- Date: Tue Sep 26 11:25:57.390 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=218178>218178</A>
RUNSTATUS: <PRE>Marked Bad, explanation:
Silicon out of run; CO marked many other components bad
</PRE>
<!-- Author: cdfscico -->



<!-- Date: Tue Sep 26 11:28:00.681 2006 -->
<pre>The MVME5500 crate processor in crate LEVEL2_PULSAR_02 (b0l2pu02) 
has been swapped for a newly modified MVME5500 with the  
Motorola ECO11973 repair.  As noted in previous swaps, this  
modification fixes a particular CPU hang-up problem. 
 
       FNAL Number 
Old    570357 
New    570652 
 
 
 </pre>
<!-- Author: badgett -->



<!-- Date: Tue Sep 26 11:48:44.872 2006 -->
<table border=0><td><tr><td bgcolor=#e0e030><pre> 
replaced CMU card in b0cot08.  testing... 
 
</pre></td></tr></td></table>
<!-- Author: Sasha Golossanov -->
<!-- Comment: Tue Sep 26 14;31;06 comment by...Sasha Golossanov -->
<pre>
appears to work, but wasn't able to run COT calibrations
due to "other problems, B.Badgett". 



</pre>
<!-- Date: Tue Sep 26 11:51:59.477 2006 -->
OS Updates are complete, all CDF Online systems are up to date. b0server05 still needs to be rebooted, since it is currently in use. Bill will let me know when I can reboot.
<!-- Author: Jason Hedden -->



<!-- Date: Tue Sep 26 12:08:32.424 2006 -->
<table border=0><td><tr><td bgcolor=#e0e030><P> 
The b0server00..04 and 06..11 machines have been <BR> 
patched and rebooted by Jason Hedden.  We still <BR> 
need to do <B>b0server05</B> which runs the <B>rtserver</B><BR> 
Tentatively, we plan to reboot in two hours (14:00 or so) 
<P> 
</td></tr></td></table>
<!-- Author: badgett -->


<!-- Date: Tue Sep 26 12:12:54.790 2006 -->
<pre>The MVME5500 crate processor in crate COT_08 (b0cot08)  
has been swapped for a newly modified MVME5500 with the   
Motorola ECO11973 repair.  As noted in previous swaps, this   
modification fixes a particular CPU hang-up problem.  
 
       FNAL Number  
Old    568798   
New    570328 
 
NB: We have NOT been able to test this crate, as the power  
to all of COT crates was cut soon after the swap (Dervin, et al) 
 
</pre>
<!-- Author: badgett, sasha golossanov -->


<!-- Date: Tue Sep 26 12:27:27.081 2006 -->
<p><b>L3 Manager problems</b></p> 
<p>L3 manger died during the <a href= 
'/cgi/elog/elog.pl?nb=2006&action=view&page=712&frame=2&anchor=12:08:32.424'> 
update/reboot of b0server04 </a>. The <a href="http://www-cdfonline.fnal.gov/ops/ace2help/l3Manager/">instructions  page</a> was outdated and we could not resolve the problem ourselves. Badgett updated the page to relect the new PC that the server runs on. Following the updated instrcutions we restarted the level 3 manager.</p>
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 12:28:54.942 2006 -->
<p>The whole COT tripped. We don't know why. I haven't brought it back up yet because we think we want to investigate the source of the trip.</p>
<!-- Author: Jason -->
<!-- Comment: Tue Sep 26 12;39;09 comment by...Jason -->
<p>Someone is working on the COT and we expect it to have tripped. We'll bring it up later</p>
<!-- Date: Tue Sep 26 12:33:56.455 2006 -->
<p>Donatella has requested that I run shot setup or cosmics -- anything that puts data through the system so that she can run tests. I'm not going to do a run now because I don't want to risk bumping into the Si tests. I'll make another post when I manage to run.</p>
<!-- Author: Jason -->

<!-- Date: Tue Sep 26 12:39:04.389 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=218223>218223</A>
si: <tt>Silicon was not included</tt>
<!-- Author: Consumer Operator( Jeff Miles ) -->


<!-- Date: Tue Sep 26 12:39:05.771 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=218223>218223</A>
central: <tt>No COT occupancy plots - as if no ROOT file were there (but there was no such complaint).</tt>
<!-- Author: Consumer Operator( Jeff Miles ) -->


<!-- Date: Tue Sep 26 12:39:06.055 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=218223>218223</A>
muons: <tt>Numerous unfilled plots (as though the consumers hadn't stored them) for all muon detectors.</tt>
<!-- Author: Consumer Operator( Jeff Miles ) -->


<!-- Date: Tue Sep 26 12:39:07.916 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=218223>218223</A>
trigger: <tt>L2 pulsar slide indicates high error rate for CES.</tt>
<!-- Author: Consumer Operator( Jeff Miles ) -->


<!-- Date: Tue Sep 26 12:39:08.395 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=218223>218223</A>
monitors: <tt>The lack of plots seems to indicate that the consumers were not all functioning properly.</tt>
<!-- Author: Consumer Operator( Jeff Miles ) -->


<!-- Date: Tue Sep 26 12:39:09.730 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=218223>218223</A>
general: <tt>I'm running the checklist on this older run (June '06) as an example of past runs with missing bits or which were not otherwise marked by the SciCo. &nbsp;Many aspects of the data seem incomplete, in that there are missing plots for the COT and muon systems, and there were L2 errors. There are too many suspect figures to paste them all to the e-log. I have only marked "OK" what was demonstrably good.</tt>
<!-- Author: Consumer Operator( Jeff Miles ) -->




<!-- Date: Tue Sep 26 13:31:00.363 2006 -->
Short access report: I power cycled Bottom CSP CCU's to restore their response. Response is good now.
<!-- Author: Oleg Pukhov -->


<!-- Date: Tue Sep 26 13:39:04.117 2006 -->
<pre>The following TDCs are not communicative: 
 
Crate   Slot 
b0cot04 18 
b0cot09 2 
b0cot09 15 
 
Sci-Co is calling the TDC pager.</pre>
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223684' target=_top>223684</a>) -->
<!-- Comment: Tue Sep 26 13;45;37 comment by...Jason -->
<p>Power cycled b0cot04, waiting for Nils to reseat a board in b0cot09 slot 15</p>

<!-- Comment: Tue Sep 26 13;58;27 comment by...Jason -->
<p>Nils reseated the board, running calib again</p>
<!-- Date: Tue Sep 26 13:46:59.428 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223685>223685</A>
ACTIVATE: RunSet: COTCALIB_ACE testing 
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 13:47:00.771 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223685>223685</A>
TERMINATE: done w/ COT calib --- cot09 excluded in this calib
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 13:47:03.113 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223684>223684</A>
CALIBRATION: <U><B>cot</B></U> calibration FAILED, please investigate
<!-- Author: CDFCALIB -->



<!-- Date: Tue Sep 26 13:59:54.634 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223686>223686</A>
TERMINATE: end , fail config
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 14:01:56.617 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223687>223687</A>
TERMINATE: end, did not config, problem with cot09 slot2
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 14:08:10.848 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223689>223689</A>
ACTIVATE: RunSet: COTCALIB_ACE all crates this time
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 14:08:12.470 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223689>223689</A>
TERMINATE: done, all crates this time
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 14:08:57.801 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223686>223686</A>
CALIBRATION: <U><B>cot</B></U> calibration FAILED, please investigate
<!-- Author: CDFCALIB -->


<!-- Date: Tue Sep 26 14:18:36.074 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223693>223693</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Tue Sep 26 16;03;46 comment by...Tracey Berry -->
Calibration ok
<!-- Date: Tue Sep 26 14:18:38.142 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223694>223694</A>
TERMINATE: done
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 14:18:39.227 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223693>223693</A>
TERMINATE: done
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 14:18:40.567 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223693>223693</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 14:21:04.486 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223695>223695</A>
ACTIVATE: RunSet: CAL_NOM trying non-standard cal to see if we can fix cot problems
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 14:21:05.844 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223695>223695</A>
TERMINATE: done
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 14:22:02.586 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223696>223696</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 14:23:10.010 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223696>223696</A>
TERMINATE: done
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 14:23:11.427 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223696>223696</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Tue Sep 26 16;17;23 comment by...Tracey -->
will add plot to e-log see below.
<!-- Date: Tue Sep 26 14:30:09.955 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223698>223698</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Tue Sep 26 16;25;33 comment by...Tracey -->
calibration ok
<!-- Date: Tue Sep 26 14:30:10.804 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223698>223698</A>
ACTIVATE: RunSet: COTCALIB_ACE ---------------====again
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 14:30:11.232 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223698>223698</A>
TERMINATE: done
<!-- Author: Jason -->




<!-- Date: Tue Sep 26 14:32:34.374 2006 -->
<table border=0><td><tr><td bgcolor=#e0e030><pre> 
will be using b0pcal01 to test SMXR calibrations routine. 
please do not include this crate in your partition. 
 
 
</pre></td></tr></td></table>
<!-- Author: Sasha Golossanov :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223698' target=_top>223698</a>) -->
<!-- Comment: Tue Sep 26 14;48;37 comment by...Sasha Golossanov -->
<table border=0><td><tr><td bgcolor=#e0e030><pre>
will grab the b0pcal02 as well.

</pre></td></tr></td></table>
<!-- Date: Tue Sep 26 14:35:29.886 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223700>223700</A>
ACTIVATE: RunSet: CAL_NOM 
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 14:36:19.440 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223700>223700</A>
TERMINATE: done, special run for NIls
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 14:37:37.420 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223701>223701</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 14:37:58.461 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223701>223701</A>
TERMINATE: done
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 14:40:29.449 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223701>223701</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->



<!-- Date: Tue Sep 26 14:40:49.721 2006 -->
<p>It looks like many of the COT calibration problems were due to a low voltage trip for the whole west wall.. we reset the trips and now the calibration looks better.</p>
<!-- Author: Jason :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223701' target=_top>223701</a>) -->

<!-- Date: Tue Sep 26 14:42:39.895 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223702>223702</A>
ACTIVATE: RunSet: COT_ACE 
<!-- Author: Eiko X 2080 -->



<!-- Date: Tue Sep 26 14:44:59.887 2006 -->
<p>Went up in notch with Del Allspach to check CHA PMT 06W T4R 
which shows up in YMON occupancy plot. Found that SMB feedthru 
on PMT base had broken off in the SMB connector on the anode 
cable.  This is a long tedious job to fix; ideally, would like a 
real 12 hour access.</p>
<!-- Author: Steve Hahn :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223701' target=_top>223701</a>) -->


<!-- Date: Tue Sep 26 14:50:57.270 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223703>223703</A>
ACTIVATE: RunSet: SMQIECALIB LUT storage test...
<!-- Author: ag -->



<!-- Date: Tue Sep 26 14:51:34.708 2006 -->
There are a lot of bad channels on this 
<a href= 
'/cgi/elog/elog.pl?nb=2006&action=view&page=712&frame=2&anchor=14:23:11.427'> 
 cal_qie</a>  
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 14:56:18.258 2006 -->
<table border=0><td><tr><td bgcolor=#e0e030><P> 
b0server05 and rtserver are being rebooted right now! 
<P> 
Please pause your DAQ RunControl activities. 
<P></td></tr></td></table>
<!-- Author: badgett -->

<!-- Date: Tue Sep 26 15:03:51.964 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223703>223703</A>
TERMINATE: b0server05 was rebooted.  start over....
<!-- Author: ag -->


<!-- Date: Tue Sep 26 15:08:49.338 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223704>223704</A>
ACTIVATE: RunSet: SMQIECALIB more testing...
<!-- Author: ag -->


<!-- Date: Tue Sep 26 15:10:31.634 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223705>223705</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_4_00 [7,833,649] for donatella
<!-- Author: Jason -->



<!-- Date: Tue Sep 26 15:10:34.957 2006 -->
<p>Doing AAA_SHOTSETUP for Donatella, no L3 used in this run</p>
<!-- Author: Jason :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223705' target=_top>223705</a>) -->

<!-- Date: Tue Sep 26 15:14:22.228 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223704>223704</A>
TERMINATE: done for now
<!-- Author: ag -->



<!-- Date: Tue Sep 26 15:14:28.275 2006 -->
<pre>COT ASDQ calibration after the fix of west side LV: 
 
All the cells look fine. But I will take another calibration after  
the access finishes. 
</pre>
<!-- Author: Shin-Shan Yu -->

<!-- Date: Tue Sep 26 15:15:46.138 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223705>223705</A>
TERMINATE: ending to run w/ ignore l2 problems
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 15:16:31.023 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223707>223707</A>
ACTIVATE: RunSet: SMQIECALIB code debugging 01
<!-- Author: ag -->


<!-- Date: Tue Sep 26 15:18:03.605 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223706>223706</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_4_00 [7,833,649] 
<!-- Author: Jason -->



<!-- Date: Tue Sep 26 15:18:24.337 2006 -->
XFT 06 07 dONE TO
<!-- Author: JASON :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223706' target=_top>223706</a>) -->


<!-- Date: Tue Sep 26 15:19:07.624 2006 -->
PSM complained that the -5V supply for the bottom CMP ASD crate was 0. Dale and I reclaimed our keys and made a quick dash to the hall to check it out. It had been switched off. I turned it back on. Good thing we caught it before the hall closed, or I'd really have been POd.
<!-- Author: Phil Schlabach -->


<!-- Date: Tue Sep 26 15:21:37.334 2006 -->
<P> 
Have promoted a new version of <i>fer</I> FrontEndReadout <BR> 
product to version <I>v5_10</I>, with the changes: 
<UL> 
<LI>Set the task priority of the auxlillary rtlm_session  
task on b0clc01 to 189, in the hopes that the task will  
be able to send its keep-alive to rtserver, and not induce  
a stuck port on the rtserver. 
<LI>Initialize the TRIG_imp.c VISION handles at the  
beginning of CONFIG (previously missing) 
<LI>Ditto for TESTCLOCK_imp.c 
<LI>Fix an addressing bug in XFT.c XFTSF_writeFiberMaskRegisters() function that was over-writing  
the dead wire registers. 
<LI>Add l2Readout flag in the XftFinderListElement class for  
the T2FD bank.  (not completely implemented yet) 
</UL> 
Only the CDF front-end crates have been rebooted. 
<P>
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223705' target=_top>223705</a>) -->

<!-- Date: Tue Sep 26 15:25:47.753 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223706>223706</A>
TERMINATE: done
<!-- Author: Jason -->



<!-- Date: Tue Sep 26 15:27:09.213 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223707>223707</A>
TERMINATE: will need more debugging.
<!-- Author: ag -->


<!-- Date: Tue Sep 26 15:28:58.846 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223714>223714</A>
ACTIVATE: RunSet: SMQIECALIB code debugging 02
<!-- Author: ag -->


<!-- Date: Tue Sep 26 15:30:07.552 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223714>223714</A>
TERMINATE: done for tonight.
<!-- Author: ag -->



<!-- Date: Tue Sep 26 15:30:07.271 2006 -->
BMU Access report:  Coming down from Madison didn't leave us much time to investigate reoccurring oscillations.  Instead, we eliminated tiny wanna-be oscillations (one in east and one in west).  Also we put NW54 LV back, because we suspect that the culprit is NW55 not 54 -- NW55 is still unplugged.
<!-- Author: Woo-Hyun & Yeongdae & Varsha -->



<!-- Date: Tue Sep 26 15:36:28.556 2006 -->
<pre>Debugging the problem with channel RX9 on vrb12; The new test crate b03cpo is connected to this channel and fails "Start Test" during vrbtest. 
(1) yesterday we swapped b03cpo with b0svt08 (RX8, vrb12): problem stays with RX9 channel. 
(2) Today we swapped the Tart card #11 with the spares #1 and #3: same problem with RX9. 
(3) We also swapped the vrb with spare one: same problem with RX9. 
 
 
  
</pre>
<!-- Author: M. Datta -->


<!-- Date: Tue Sep 26 15:36:48.659 2006 -->
<h3>Logic Analyser behind the TS has been disconnected</h3> 
 
<p>After a month or so, the L1 Bursts have not returned.  So I have disconnected the Logic Analyser connected to the scalars and TS backplane.</p> 
 
<p>I have left the connector for the logic analyser attached to the TS Backplane in case it is needed in the future.</p> 
 
<p>This should now free up the space behind the TS crates.</p>
<!-- Author: Ankush -->


<!-- Date: Tue Sep 26 15:39:14.900 2006 -->
We are done with the supervised access.
<!-- Author: Koji -->


<!-- Date: Tue Sep 26 15:40:32.677 2006 -->
<table border=0><td><tr><td bgcolor=#e0e030><pre> 
done with SMXR calbrations testing, for tonight. 
b0pcal01 and b0pcal02 were returned back to the ACE. 
 
 
 </pre></td></tr></td></table>
<!-- Author: Sasha Golossanov :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223718' target=_top>223718</a>) -->


<!-- Date: Tue Sep 26 15:44:03.967 2006 -->
<table border=0><td><tr><td bgcolor=#e0e030><P> 
<H3>The <U>b0pcal01</U> serial port seems to be broken. 
Please call Jonathan Lewis</h3> 
<P> 
</td></tr></td></table>
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223718' target=_top>223718</a>) -->



<!-- Date: Tue Sep 26 15:44:37.093 2006 -->
The Silicon interlock code has been changed without problems. The detector is back in STANDBY. For details see the <a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=320&frame=2&anchor=14:26:18'> 
Silicon e-log</a>.
<!-- Author: Ulrich -->

<!-- Date: Tue Sep 26 15:48:32.827 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223718>223718</A>
TERMINATE: end, could not config 
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 15:49:03.600 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=218400>218400</A>
muons: <tt>Hard to tell whether the wedges are dead or the histogram is incompletely filled -- my guess is the latter. &nbsp;Also, no BSU or TSU scint numbers.</tt>
<!-- Author: Consumer Operator( Jeff Miles ) -->


<!-- Date: Tue Sep 26 15:49:04.549 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=218400>218400</A>
trigger: <tt>High L2 error rate for muons.</tt>
<!-- Author: Consumer Operator( Jeff Miles ) -->


<!-- Date: Tue Sep 26 15:49:05.194 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=218400>218400</A>
general: <tt>Checking this old run because it had missing bits. &nbsp;Seems to generally have some missing monitoring info for the muons, and the L2 trigger definitely had errors.</tt>
<!-- Author: Consumer Operator( Jeff Miles ) -->



<!-- Date: Tue Sep 26 15:51:07.430 2006 -->
<P> 
We had to power cycle b0pcal01 and b0pcal02 after the SMXR  
tests earlier, in order to get them to boot.   Please  
remember to do an SM QIE Calibration tonight, before  
the plug source calibrations. 
<P> 
  
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223719' target=_top>223719</a>) -->

<!-- Date: Tue Sep 26 15:52:19.759 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223719>223719</A>
TERMINATE: csl error
<!-- Author: Jason -->
<!-- Comment: Tue Sep 26 16;03;35 comment by...badgett -->
Restarted Main CSL from the ProcMon;   i had never 
seen it tried before, but it actually worked!
<!-- Date: Tue Sep 26 15:55:10.383 2006 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>PhysicsTable</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L3Accepts</TH>
<TH>DelivLumi nb<SUP>-1</SUP></TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN=223642 target=_runSummary>223642</A> x3699A</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_4_00 [7,833,649]</FONT></TD>
<TD ALIGN=CENTER>06:29:34</TD>
<TD ALIGN=CENTER>08:33:59</TD>
<TD ALIGN=CENTER>01:38:24</TD>
<TD ALIGN=RIGHT>     504,263</TD>   <TD ALIGN=RIGHT>    479.88</TD>
<TD ALIGN=RIGHT>    376.63</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>15:55:00</TD>
<TD ALIGN=CENTER>01:38:24</TD>
<TD ALIGN=RIGHT>       504,263</TD> <TD ALIGN=RIGHT>    479.88</TD>
<TD ALIGN=RIGHT>    376.63</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->


<!-- Date: Tue Sep 26 15:57:03.869 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223720>223720</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_4_00 [7,833,649] 
<!-- Author: Jason -->



<!-- Date: Tue Sep 26 16:02:05.666 2006 -->
<P> 
We're now running AAA_SHUTDOWN with no obvious problems,  
and all of the CDF readout crates.   Things look good after  
the access. 
<P> 
XFT have left the test patterns in place, so L1A is 23kHz,  
but it is all rejected at L2. 
<P>
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223720' target=_top>223720</a>) -->


<!-- Date: Tue Sep 26 16:03:25.966 2006 -->
<font color=red>Shift Summary: </font><pre>Took data with store 4975 before beam was lost due to RF trip. Abort seems to have been clean. Took quiet time calibrations. 
 
Tevatron decided to go into long access. First CDF personel into the collision hall at about 10:30. Silicon running bias scan. 
 
As shift ends, plug sourcing is beginning and search and secure underway. No estimate yet for machine/D0 secure or next shot. 
 
 
 
</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   223642
Delivered Luminosity   118.3  
Acquired Luminosity    102.2  
Efficiency             86.4
   
</td></tr></table></pre>
<!-- Author: Joel -->



<!-- Date: Tue Sep 26 16:05:36.142 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223720>223720</A>
TERMINATE: releasing crates 
<!-- Author: Jason -->