<!-- Title: Day shift. Sun Dec 30, 2001 -->
<!-- SciCo: Larry N. -->
<!-- DAQAce: Jong-Young Chong -->
<!-- MonAce: Ayana! -->
<!-- CO: Mauro -->
<!-- OpManager: Kevin -->
<!-- Notes: -->
<pre>Continue no cpr, some muons, L=3, dead si ps running</pre>


<!-- Date: Sun Dec 30 08:39:37 2001 -->
<pre> current status 
 
 Rates 
 L1A 231 Hz 
 L2A 92  Hz 
 L3A 23  Hz 
 total deadtime ~.20% 
 
 Logger 5.09MB/s 
 
 B0 luminosity 2.9E30 
 delivered 249.7 nb-1 
 acquired  171.6 nb-1</pre>
<!-- Author: jy -->


<!-- Date: Sun Dec 30 08:46:56 2001 -->
Kevin shamed me into turning CPR on
<!-- Author: Larry :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135925' target=_top>135925</a>) -->


<!-- Date: Sun Dec 30 08:51:59 2001 -->
mauro notices problem with b2w7 in silicon monitoring.  This 
doesn't correspond to the fib that's off (fib00), at least not in the somewhat-outdated cabling summary. 
  
<!-- Author: ath -->


<!-- Date: Sun Dec 30 09:12:21 2001 -->
PEM HB alarm. H-C says wait 10m. 
  
<!-- Author: ath -->


<!-- Date: Sun Dec 30 09:22:21 2001 -->
PEM HB has cleared but still in alarm state.  Upstairs, comm. 
seems to be lost to 2 of the 6 west(?) power supplies.  This 
might be why the "bars" show 66.7% for the west(?) voltage. 
<br> 
Paging.
<!-- Author: ath -->

<!-- Date: Sun Dec 30 09:31:55 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135925">135925</A>
 Terminated at 2001.12.30 09:28:50
<!-- Author: RunControl -->



<!-- Date: Sun Dec 30 09:38:49 2001 -->
After 2 trips, set CPR to standby
<!-- Author: Larry -->


<!-- Date: Sun Dec 30 09:44:44 2001 -->
<pre>SiliMon Errors: 
"Error in Raw Monitor 
 All ADC zero in SB2W7L4NC0 to  5 
 All ADC zero in SB2W7L4PC5 and 6 
 Landau  bad  in SB2W7L4NC5 
 Landau  bad  in IB5W4L4NC3 
 page silicon!" 
 
Ayana says this crate has power on. 
 
Silicon paged.</pre>
<!-- Author: Mauro :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135925' target=_top>135925</a>) -->


<!-- Date: Sun Dec 30 09:45:40 2001 -->
 stopped the run and coldstart by request of silicon expert.
<!-- Author: jy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135925' target=_top>135925</a>) -->
<!-- Comment: Sun Dec 30 09;48;59 comment by...jy -->
 before finished this run, there was also DTO from fcal00, it didn't halt either. so reset it.


<!-- Date: Sun Dec 30 10:13:08 2001 -->
L3 transition failed during coldstart during quite long time. 
restarted L3 proxy. <p> 
but during partitioning after that, L3 is dead (red in RC) 
cleaned up EVB. but it didn't help.<p> 
paged L3.
<!-- Author: jy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135927' target=_top>135927</a>) -->
<!-- Comment: Sun Dec 30 11;26;07 comment by...Nuno/Boris -->
<pre>u01 had a 300M relay core file, no space left 
on the hard drive , removed it.
Killed all relay processes

EventFunnels get stuck in a bad state on the
outputs. We had to "killall -9 EventFunnel" 
everuwhere.

Cleaned up L3, restarted Level3 Proxy. 

</pre>
<!-- Date: Sun Dec 30 10:49:35 2001 -->
silicon: pink channels went green after coldstart.  few channels in the micron layers returned to pink, but we ignore this.
<!-- Author: ath -->


<!-- Date: Sun Dec 30 10:53:17 2001 -->
Plug monitoring: expert is looking into PEM/PHA monitoring problems (from italy; 21st century operations....) 5/6 channels West and 1/6 East complain "no Comm"; this is presumably noisy comm line and not caen error. expert might just turn off IFIX alarms for a while. 
  
<!-- Author: ath -->
<!-- Comment: Sun Dec 30 11;30;26 comment by...ath -->
expert works her magic; plug monitoring is happy now.
<!-- Date: Sun Dec 30 11:12:56 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135928">135928</A>
 Activated at 2001.12.30 11:09:40
<!-- Author: RunControl -->


<!-- Date: Sun Dec 30 11:17:43 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135928">135928</A>
 Terminated at 2001.12.30 11:13:50
<!-- Author: RunControl -->


<!-- Date: Sun Dec 30 11:20:09 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135929">135929</A>
 Activated at 2001.12.30 11:16:38
<!-- Author: RunControl -->



<!-- Date: Sun Dec 30 11:23:31 2001 -->
 L3 crash seemed to be caused by one of the output node (b0l3u01) 
 That node's disk was full due to core  file.<BR> 
 now expert fixed the problem. <BR> 
 resume taking data.<BR> 
 at the restart. fcal00 stopped to respond again. so reset it.<BR> 
 at laat, we are taking data again.  
<!-- Author: jy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135929' target=_top>135929</a>) -->



<!-- Date: Sun Dec 30 11:34:39 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135929">135929</A>
 Terminated at 2001.12.30 11:29:55
<!-- Author: RunControl -->



<!-- Date: Sun Dec 30 11:43:37 2001 -->
<pre>I notice that in the last few days the rate into the CSL has 
sometimes been limited to 15 MB/sec (instead of the more usual 
20-22 MB/sec). This is probably because the CSL was 
writing to disk3 which is the slower, backup disk. 
I restarted the CSL with some debugging code to 
understand why the csl is writing to disk3 even though the 
main disks aren't full. 
</pre>
<!-- Author: tony vaiciulis :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135930' target=_top>135930</a>) -->

<!-- Date: Sun Dec 30 11:47:50 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135930">135930</A>
 Activated at 2001.12.30 11:44:13
<!-- Author: RunControl -->


<!-- Date: Sun Dec 30 13:00:30 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135930">135930</A>
 Terminated at 2001.12.30 12:56:30
<!-- Author: RunControl -->


<!-- Date: Sun Dec 30 13:09:04 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135931">135931</A>
 Activated at 2001.12.30 13:05:23
<!-- Author: RunControl -->



<!-- Date: Sun Dec 30 13:09:04 2001 -->
 changed trigger table to MINIMUM_DIAS_1[5,233]
<!-- Author: JY :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135931' target=_top>135931</a>) -->


<!-- Date: Sun Dec 30 13:11:50 2001 -->
Run out the last hour of the store on min bias
<!-- Author: Larry  :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135931' target=_top>135931</a>) -->

<!-- Date: Sun Dec 30 13:59:00 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135931">135931</A>
 Terminated at 2001.12.30 13:55:41
<!-- Author: RunControl -->



<!-- Date: Sun Dec 30 13:59:25 2001 -->
10 minute drop store warning
<!-- Author: Larry -->


<!-- Date: Sun Dec 30 14:01:58 2001 -->
 to prepare dump of the store. stopped the run
<!-- Author: jy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135931' target=_top>135931</a>) -->


<!-- Date: Sun Dec 30 14:07:40 2001 -->
 cosmic_notracks[3,233]
<!-- Author: jy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135932' target=_top>135932</a>) -->


<!-- Date: Sun Dec 30 14:10:49 2001 -->
MCR warns that they will bump the beam around some in an attempt to avoid quench on dump.  Good to be off, may be a while.
<!-- Author: Larry -->
<!-- Comment: Sun Dec 30 14;21;36 comment by...Larry -->
Yes, turn off solenoid to pump


<!-- Date: Sun Dec 30 14:29:13 2001 -->
took out silicons from the run
<!-- Author: jy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135934' target=_top>135934</a>) -->
<!-- Comment: Sun Dec 30 14;33;54 comment by...jy -->
due to many reformatter errors drom silicon unbiased.


<!-- Date: Sun Dec 30 14:43:14 2001 -->
Store 869 dropped
<!-- Author: Larry -->
<!-- Comment: Sun Dec 30 14;46;50 comment by...Larry -->
Quenched a4 and b0 quads
<!-- Date: Sun Dec 30 14:44:25 2001 -->
<center><a href='/cgi/elog/elog.pl?nb=2001&action=view&page=-3180&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2001&action=view&page=-3180' width='300'></a><a href='/cgi/elog/elog.pl?nb=2001&action=view&page=-3181&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2001&action=view&page=-3181' width='300'></a></center>
<!-- Author: monitoring ace -->



<!-- Date: Sun Dec 30 14:57:02 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-3182&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-3182  width='300' align='top'></td><td>complex plot showing how the beam was dumped.  </a></td></tr></table>
<!-- Author: ath -->
<!-- Comment: Sun Dec 30 15;03;33 comment by...ath -->
there's no imaginary axis, i just mean there's lots of
info plotted here.


<!-- Date: Sun Dec 30 15:07:56 2001 -->
CAL QIE calibration. appeared on db. following is error mesg.<BR> 
(MLE) b0puls01:Messenger:3:00:03 PM->QIE calibration failed  <BR> 
(b0puls01) QIE calib: slot 21, qie 00:Warn  3: Raw ped low        , val 148.9650, cap 0, rng0.<BR> 
(MLE) b0puls01:Readout_Calib:3:00:03 PM->ADMEM=21 QIE=4 Fail 10: No fitting          (0.0) (0) (0) <BR> 
(b0puls01) QIE calib: slot 21, qie 04:Fail 10: No fitting         , val 00.0000, cap 0, rng0. <BR> 
(MLE) b0ccal09:Readout_Calib:3:00:05 PM->ADMEM=19 QIE=6 Fail 14: Sigma(off) big      (1.4248966) (1) (0)<BR> 
(b0ccal09) QIE calib: slot 19, qie 06:Fail 14: Sigma(off) big     , val 01.4249, cap 1, rng0.  <BR> 
(MLE) b0pcal11:Readout_Calib:3:00:14 PM->ADMEM=17 QIE=3 Fail 14: Sigma(off) big      (1.1806831) (0) (0) <BR> 
(b0pcal11) QIE calib: slot 17, qie 03:Fail 14: Sigma(off) big     , val 01.1807, cap 0, rng0. <BR>
<!-- Author: jy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135935' target=_top>135935</a>) -->


<!-- Date: Sun Dec 30 15:10:35 2001 -->
SMX Calibration done. appeared in DB.
<!-- Author: jy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135936' target=_top>135936</a>) -->


<!-- Date: Sun Dec 30 15:15:21 2001 -->
CLC QIE calibration done. following is error mesg.<BR> 
(MLE) b0clc00:Messenger:5:19:10 AM->QIE calibration failed    <BR> 
(b0clc00) QIE calib: slot 03, qie 00:Warn 11: Sigma(off) high    , val 00.8020, cap 0, rng0.  <BR> 
(MLE) b0clc00:Readout_Calib:5:19:10 AM->ADMEM=4 QIE=2 Fail 14: Sigma(off) big      (18.417084) (0) (0) <BR> 
(b0clc00) QIE calib: slot 04, qie 02:Fail 14: Sigma(off) big     , val 18.4171, cap 0, rng0.<BR>  
<!-- Author: jy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135937' target=_top>135937</a>) -->
<!-- Comment: Sun Dec 30 15;17;31 comment by...jy -->
 appeared in DB.
<!-- Date: Sun Dec 30 15:17:10 2001 -->
TOF OIE Calibration done. appeared in DB.
<!-- Author: jy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135938' target=_top>135938</a>) -->
<!-- Comment: Sun Dec 30 15;20;20 comment by...jy -->
warning mesg.<BR>
(b0wcal02) QIE calib: slot 06, qie 08:Warn 10: Slope low          , val 03.0309, cap 0, rng0.<BR>
(b0wcal05) QIE calib: slot 07, qie 02:Warn 10: Slope low          , val 00.0258, cap 1, rng4.    <BR>    
<!-- Date: Sun Dec 30 15:17:34 2001 -->
The numbers: silicon rad during shift (all during dump/quench) 
<ul><li>blm 1 0.163 
<li>blm 2 0.017 
<li>blm 3 0.032 
<li>blm 4 0.014 
</ul> 
and our luminosity 
<ul> 
<li> total 60.1 nb-1 
<li> live  30.3 nb-1 
<li> efficient 50.4% 
</ul>
<!-- Author: ath -->


<!-- Date: Sun Dec 30 15:19:16 2001 -->
TOF TAC calibration done 7000 L3A  taken
<!-- Author: jy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135939' target=_top>135939</a>) -->



<!-- Date: Sun Dec 30 15:23:48 2001 -->
BSC QIE calibration done, appeard in DB
<!-- Author: jy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135940' target=_top>135940</a>) -->


<!-- Date: Sun Dec 30 15:26:44 2001 -->
COT Calibration. entry appeared in DB.
<!-- Author: jy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135941' target=_top>135941</a>) -->


<!-- Date: Sun Dec 30 15:30:46 2001 -->
Muon calibration, 282 L2A before <BR> 
29'43" 1 crate/s: b0imu01(81),  Done TO.[RXPT] 
<!-- Author: jy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135942' target=_top>135942</a>) -->

<!-- Date: Sun Dec 30 15:37:57 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135943">135943</A>
 Activated at 2001.12.30 15:33:12
<!-- Author: RunControl -->


<!-- Date: Sun Dec 30 15:43:40 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135943">135943</A>
 Terminated at 2001.12.30 15:40:35
<!-- Author: RunControl -->



<!-- Date: Sun Dec 30 15:43:40 2001 -->
PMT spike run,  30k events taken<BR> 
L1 rate is 75Hz
<!-- Author: jy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135943' target=_top>135943</a>) -->


<!-- Date: Sun Dec 30 15:47:53 2001 -->
CEM LED calibration done. appeared in DB
<!-- Author: jy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135944' target=_top>135944</a>) -->

<!-- Date: Sun Dec 30 15:58:24 2001 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135925>
135925</A> 
x212F5</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_0_03 [1,238]</TD>                                           
<TD ALIGN=CENTER>00:43:28</TD>                                          
<TD ALIGN=CENTER>09:28:50</TD>                                          
<TD ALIGN=CENTER>08:21:12</TD>                                          
<TD ALIGN=RIGHT>      8,642,892</TD>                                    
<TD ALIGN=RIGHT>     4,060,750</TD>                                     
<TD ALIGN=RIGHT>     706,387</TD>                                       
<TD ALIGN=RIGHT>  120.745</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135929>
135929</A> 
x212F9</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_0_03 [1,238]</TD>                                           
<TD ALIGN=CENTER>11:16:38</TD>                                          
<TD ALIGN=CENTER>11:29:55</TD>                                          
<TD ALIGN=CENTER>00:13:10</TD>                                          
<TD ALIGN=RIGHT>        149,311</TD>                                    
<TD ALIGN=RIGHT>        64,478</TD>                                     
<TD ALIGN=RIGHT>      11,771</TD>                                       
<TD ALIGN=RIGHT>    1.828</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135930>
135930</A> 
x212FA</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_0_03 [1,238]</TD>                                           
<TD ALIGN=CENTER>11:44:13</TD>                                          
<TD ALIGN=CENTER>12:56:30</TD>                                          
<TD ALIGN=CENTER>01:11:12</TD>                                          
<TD ALIGN=RIGHT>        769,461</TD>                                    
<TD ALIGN=RIGHT>       331,240</TD>                                     
<TD ALIGN=RIGHT>      60,140</TD>                                       
<TD ALIGN=RIGHT>    9.733</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135931>
135931</A> 
x212FB</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>MINIMUM_BIAS_1 [5,233]</TD>                                         
<TD ALIGN=CENTER>13:05:23</TD>                                          
<TD ALIGN=CENTER>13:55:41</TD>                                          
<TD ALIGN=CENTER>00:25:08</TD>                                          
<TD ALIGN=RIGHT>        320,957</TD>                                    
<TD ALIGN=RIGHT>       320,957</TD>                                     
<TD ALIGN=RIGHT>     290,028</TD>                                       
<TD ALIGN=RIGHT>    3.195</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>15:53:01</TD>                                                       
<TD ALIGN=CENTER>10:10:43</TD>                                          
<TD ALIGN=RIGHT>       9,882,621</TD>                                   
<TD ALIGN=RIGHT>      4,777,425</TD>                                    
<TD ALIGN=RIGHT>     1,068,326</TD>                                     
<TD ALIGN=RIGHT>  135.501</TD>                                          
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Sun Dec 30 16:03:23 2001 -->
CEM XEF calibration done - appeared in DB
<!-- Author: jy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135945' target=_top>135945</a>) -->


<!-- Date: Sun Dec 30 16:04:39 2001 -->
<font color=red>Shift Summary: </font><pre>8:00 	Continue running, no CPR, some CMX, CMP 
 
9:45	Sillimon problem - experts reqest cold start 
	PEM/PHA communication problem, get budget code, call 
	   Italy, they are on it 
 
10:10	L3 problems, experts come in, see report above 
11:20	Back on 
 
11:44	Rony V. reports CSL was using slow disk, start new code 
	with debug features 
 
13:00	Go to minbias for the last hour of the store 
 
13:59	10 minute warning -&gt; turn off 
 
14:40	attempts to avoid quinch on drop failed 
	-&gt; calibrations, pumping 
 
15:30	Mary B trying to figure out si errors</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   135925-43
Delivered Luminosity   60  
Acquired Luminosity    30  
Efficiency             50%
   
</td></tr></table></pre>
<!-- Author: Larry -->