<!-- Title: Eve shift. Wed Jul 11, 2001 -->
<!-- SciCo: Alan Sill -->
<!-- DAQAce: Alexei Sedov -->
<!-- MonAce: Andras Zsenei -->
<!-- CO: none -->
<!-- OpManager: Kevin Burkett -->
<!-- Notes: -->
<pre>Half of the collaboration is in the control room, breaking 
(fixing) things.  (See list from end of last shift.) 
Reintegrate when possible.</pre>


<!-- Date: Wed Jul 11 16:11:29 2001 -->
<pre>SVX, ISL and L00 all turned red. By the time the silicon 
expert came to the iFIX display from the other half of the 
control room the alarms have diappeared and we could not 
see anything abnormal on the silicon GUI!</pre>
<!-- Author: Andras -->
<!-- Comment: Wed Jul 11 16;18;36 comment by...Andras -->
<pre>SVX, ISL and L00 went red again on the Global Alarms page
(however, they stayed green on the High Voltage Summary...).
This time they are staying red, a silicon expert is studying 
the problem...</pre>


<!-- Comment: Wed Jul 11 16;31;58 comment by...Andras -->
<pre>Silicon expert told us that we will get an iFIX alarm (not 
heartbeat, but the SVX, ISL and L00 turning red) whenever 
the silicon GUI hangs (it is checking it in the background and
if there was no response for more than 2 minutes then it 
triggers the alarm).</pre>
<!-- Date: Wed Jul 11 16:21:16 2001 -->
Fix swapped CHA anode cables: 
<br> 
 Wedge 23-W:   5-R  ---> 4-R ---> 3-R ---> 5-R 
 <br> 
 Wedge 9-E:   3-L  <---> 4-L   
<br>
<!-- Author: fotis -->


<!-- Date: Wed Jul 11 16:24:40 2001 -->
<pre>The development of PSM monitoring is over for today. From now on take all the alarms seriously.</pre>
<!-- Author: Marcin -->



<!-- Date: Wed Jul 11 16:56:56 2001 -->
<pre>CMP Access Report: 
 
Today we were successful in restoring one dead channel on the  
South Wall of CMP (stack lucky 13), but unsure if we were successful 
in restoring another channel which was dead since it was not getting 
HV to the cathode. 
 
It has yet to be determined if layer 0 of South Wall stack 0 was 
intentionally or unintentionally jumpered out (there is no 
written record of this so I suspect the latter).  In any case,  
the cathode for this particular channel is now at standby since 
it is drawing slightly more current at full voltage than limits 
will permit.  I suspect that this excess current draw is due to  
the fact that this chamber has been jumpered out for several  
years and needs to burn off some crud.  If this is the case, the  
average current draw can be expected to slowly decrease in time  
and at some point in the near future will be at an acceptable level 
for our full voltage current limits. 
 
So, I'm leaving this section of the South Wall on STANDBY until 
further notice.  Should any section of the South Wall trip, the  
crew on shift at the moment should not bother trying to reset it. 
 
As for any other section of CMP (North, Top, Bottom) any trips are 
probably due to humidity levels in the collision hall and should 
reset at full voltage without incident.  This means, go ahead and 
reset any CMP trips with the exception of those on the South Wall.  
Thanks. </pre>
<!-- Author: Trevor  -->


<!-- Date: Wed Jul 11 17:46:33 2001 -->
QIE calibration for b0wcal03 (slots 5,6,8,9,10) and b0wcal07 (slots 5,6,7,8,9,10). Slot 7 of wcal03 fails calibration 
(self DAC calibration problem). 
Downloaded FRAM.
<!-- Author: Gervasio Gomez -->
<!-- Comment: Wed Jul 11 18;56;06 comment by...Ivan  -->
<pre>These are the TOF admems</pre>
<!-- Date: Wed Jul 11 17:47:15 2001 -->
I turned on the Central Hadron Laser. I need to fix couple  
of fibers on WHA. No effects on CHA since I disconnected the 
main fibers to CHA. There will be some .. "smell" on the 1st floor due to the laser's pump operation 
 
<!-- Author: fotis -->


<!-- Date: Wed Jul 11 18:39:58 2001 -->
<font size=+2>XFT missing pixels traced to cable swap:</font><p>  
 
We've seen a 60% drop in L1 track finding efficiency around 
phi=0.75-0.95 radians (43-54 degrees).  This has been there 
since May at least. 
 
Kevin Burkett mentioned that there might be a cable swap 
somewhere between the COT and the TDC.  I looked at 
a couple of event displays and it's clear that in COT SL2, 
the first axial SL, the hits in cells 26 and 27 should be in 
cells 24 and 25 and vice versa. 
 
Could someone fix this before the access ends, please!<p>
 <a href= 
'/cgi/elog/det-system-elog.pl?nb=xft&action=view&page=19&frame=2&anchor=18:13:21' target=_blank> 
See this link for event display </a><p>  
<!-- Author: Evelyn Thomson -->



<!-- Date: Wed Jul 11 19:02:11 2001 -->
<!doctype html public "-//w3c//dtd html 4.0 transitional//en"> 
<html> 
<head> 
   <meta name="GENERATOR" content="Mozilla/4.76 [en] (X11; U; Linux 2.2.17-14 i686) [Netscape]"> 
</head> 
<body> 
<b>Jpython script and TAC TOF calibration</b><b></b> 
<p>Finally the Jpython script did work but in a lazy way. We have learned 
that: 
<p>- The sevb has to be started by hand before launching the script 
<br>- The sevb gets killed at the end of the script. 
<br>- Quite often the scripts hangs waiting from some "end of run" acknowledgment 
message. 
<br>   To solve this problem, one can send an "end" message using 
the local controller window 
<br>   to the  pulser crate and the wcal crates. 
<br>-There is a small bug in the script that set a wrong pulse amplitude 
value for the DDG. 
<br>  
<br>  
</body> 
</html>
<!-- Author: Ivan -->


<!-- Date: Wed Jul 11 19:04:05 2001 -->
Cryo techs call to say they will begin ramping the solenoid now.
<!-- Author: AFS -->
<!-- Comment: Wed Jul 11 19;22;24 comment by...AFS -->
... Inspecting via camera to make sure people are clear.  Rob Roser will go in to check for stray tools, warn people, etc.


<!-- Comment: Wed Jul 11 20;49;40 comment by...AFS -->
Actually starting now...
<!-- Date: Wed Jul 11 20:18:03 2001 -->
Niccolo Moggi has turned off plug crate 12 and disabled its alarms, so we will probably see some channels off in the plug during tonight's cosmic run.  He will put in a note when it is restored, probably late tomorrow.
<!-- Author: AFS -->


<!-- Date: Wed Jul 11 20:35:33 2001 -->
Problems w/ b0musc00 (Suspended job).  Trying to reboot it, because Bus reset didn't work.
<!-- Author: Alexei Sedov -->
<!-- Comment: Wed Jul 11 20;58;25 comment by...AFS -->
Control-x rebooted it, but Mon_III process still shows up as SUSPENDed. !!?
<!-- Comment: Wed Jul 11 21;13;20 comment by...AFS -->
REsetting the crate by pushing the button did not work either.  Ilya looked at the boot sequence after issuing another control-x and says it looks like a problem with the FER_DIR symbols during startup.  Paged Arnd and later, after waiting for a while w/ no answer, Frank Chlebana to see if either of them could help.
<!-- Comment: Wed Jul 11 21;17;38 comment by...Alexei -->
We suspect that the problem is in start up code, so we are trying to get sombody (Arnd, Bill).
<!-- Comment: Wed Jul 11 22;02;51 comment by...patrick -->
The startup procedure for this crate was modified
according to Bill Badgett's instructions so it
should now work. Please try again.

<!-- Date: Wed Jul 11 20:43:39 2001 -->
Turn WHA HV back on and the central laser off. 
<br> 
During this day work was done on the WHA source motors 
<br>  
for wedges 4-7 East (top four wedges) and 17-W. 
<br> 
A broken fiber on wedge 6-E was also fixed.  
<br>
<!-- Author: Nikolay, fabio and fotis -->






<!-- Date: Wed Jul 11 21:42:42 2001 -->
<pre>CLIST synchronization studies: 
 
 
I had the cal crates(L2, L1, CCal) with L2 processors most of today to study an error on the L2 Calorimeter Interface board (CLIST).  See L2 e-log for more detail.   
 
One change.  New firmware is now on CLIST to run overnight that hopes to avoid any problems associated with DDUN*'s noise (the MagicBus arbitration signal DDUN*).  If the CLIST holds up overnight, this firmware will remain, otherwise the old will be reinstalled.</pre>
<!-- Author: John Carlson -->



<!-- Date: Wed Jul 11 22:04:55 2001 -->
<pre>Magnet at full power so we are turning the COT and TOF on.</pre>
<!-- Author: Andras -->
<!-- Comment: Wed Jul 11 22;13;56 comment by...AFS -->
Checking w/ TOF people to see if they want to be on...


<!-- Comment: Wed Jul 11 22;40;48 comment by...AFS -->
Person answering TOF pager didn't know, so we leave them off...
<!-- Comment: Wed Jul 11 22;50;57 comment by...Andras -->
<pre>TOF experts arrived and we turned it on.</pre>
<!-- Date: Wed Jul 11 22:25:38 2001 -->
<pre>Trying to integrate system, 24.5K events in run 120020. All calorimeter, l1, l2, muon, cot & xft crates were included.</pre>
<!-- Author: Alexei Sedov :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=120020' target=_top>120020</a>) -->
<!-- Comment: Wed Jul 11 22;27;09 comment by...Alexei -->
All but b0musc00. Bill plan to solve the problem tomorrow.
<!-- Date: Wed Jul 11 22:27:03 2001 -->
<pre>Calibrated the chill silicon (now at -2.5) 
 
2000 nominal events (~10% reformatter errors) 
 
120011 DPS on 
120017 DPS off 
</pre>
<!-- Author: Lester -->
<!-- Comment: Wed Jul 11 22;29;04 comment by...Lester -->
silicon off, being left out tonight


<!-- Date: Wed Jul 11 22:31:33 2001 -->
Got integrated system together -- all except silicon and muon scintillator crate.  (See above.)  Immediately took it apart after taking a few thousand events in order to throw out CMX, IMU (per request of the trigger testers) and to give ShowerMax people 3 calorimeter crates (agreed to by L2 test people).  Running now with this reduced configuration.
<!-- Author: Alan Sill -->


<!-- Date: Wed Jul 11 22:32:50 2001 -->
<pre>New run (120021) was started w/o ccal2,3,4 (Karen Byrum) and CMX & IMU crates.</pre>
<!-- Author: Alexei Sedov :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=120021' target=_top>120021</a>) -->
<!-- Comment: Wed Jul 11 23;02;48 comment by...Alexei -->
All the time TOF was off ~55K events in run 120021. Now HV is turned on.


<!-- Date: Wed Jul 11 23:07:40 2001 -->
<pre>We are having trouble with the consumer displays that are  
supposed to run on b0dap55. When trying to start them from 
b0dap51 with the CO script  it says that it can't open display 
b0dap55.fnal.gov:0 Client is not authorized to connect to 
Server... Tryied xhost b0dap51 on b0dap55 and it did not help. 
Also we tried to log out and log back in and it was  
unsuccessfull as well...</pre>
<!-- Author: Andras -->


<!-- Date: Wed Jul 11 23:15:06 2001 -->
The shift crew is currently running for L2 studies.   We have three boards (L1, XTRPList, CList) working together in the crate without interference.  No L2 errors in Trigmon in the last hour. 
They will continue to take the run for us. 
<p> 
Details about L2 setup for this run are  
<a href= 
'/cgi/elog/det-system-elog.pl?nb=l2&action=view&page=12&frame=2&anchor=23:09:40'> 
in the L2 log </a>.  
<!-- Author: Matt, David, John, Stephen :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=120021' target=_top>120021</a>) -->


<!-- Date: Wed Jul 11 23:45:00 2001 -->
<pre>Run 120021 was stopped @ 130Kevents to integrate ccal2-4 back. 
Note : CLC is out (booked by partition 9) for all runs.</pre>
<!-- Author: Alexei Sedov :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=120021' target=_top>120021</a>) -->


<!-- Date: Wed Jul 11 23:53:01 2001 -->
<font color=red>Shift Summary: </font><pre>L2 tests and other expert work throughout shift.  Reintegrated 
near end of shift and took short run w/ all except silicon and 
b0musc00.  (Silicon was doing calibrations, and muon scintillator 
cate gave problems.)  Then ran L2 cosmic test run and allowed 
Shw Max to test w/ some cal crates.  Ended shift with new cosmic 
run going with all calorimeter crates included.  CMX, IMU, Si 
were excluded from cosmic test run to keep data volume down.</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   119995 - 120022
Delivered Luminosity   0  
Acquired Luminosity    0  
Efficiency             100%
   
</td></tr></table></pre>
<!-- Author: Alan Sill -->


<!-- Date: Wed Jul 11 23:53:43 2001 -->
14K events so far in run 120022.
<!-- Author: Alexei Sedov :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=120022' target=_top>120022</a>) -->