<!-- Title: Day shift. Sat Feb  2, 2008 -->
<!-- SciCo: Ulrich Husemann -->
<!-- DAQAce: Eunsin Lee -->
<!-- MonAce: Hyunsu Lee (in training) -->
<!-- CO: Erik Brubaker -->
<!-- OpManager: JJ Schmidt -->
<!-- Notes: -->
We are at the tail end of store 5876. We have not been taking good data all store because of a problem with a COT power supply. The store will be dropped a little earlier than usual, at 9:00, followed by a 4-hour controlled access. Ops manager and ops heads are informed. We have a crew lined up to repair the power supply, Dervin Allen, Roberto Davilla, and Steve Hahn (plus any help they might need).


<!-- Date: Sat Feb 2 08:11:02.937 2008 -->
  Done TO. for crate/s:  
    b0pcal10. We shepherded it. 
<!-- Author: Eunsin/Hyunsu :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257114' target=_top>257114</a>) -->


<!-- Date: Sat Feb 2 08:19:13.935 2008 -->
L2 decision timeout
<!-- Author: Eunsin/Hyunsu :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257114' target=_top>257114</a>) -->


<!-- Date: Sat Feb 2 08:22:27.477 2008 -->
Dee called in. JJ and she suggest to offer this access to the ADMEM and CMP folks as well. We have paged both.
<!-- Author: Ulrich Husemann -->
<!-- Comment: Sat Feb  2 08;27;37 comment by...Ulrich -->
ADMEM pager carrier (Shalhout) calls back, will check with his colleagues.

<!-- Comment: Sat Feb  2 08;28;41 comment by...Ulrich -->
CMP pager carrier (Xiaojian) calls back. He cannot do the repair today.

<!-- Comment: Sat Feb  2 08;28;58 comment by...Ulrich -->
Mark Mattson will come in do to the ADMEM work.

<!-- Comment: Sat Feb  2 08;36;35 comment by...Ulrich -->
Checking the access requests (<a href="https://www-cdfonline.fnal.gov/internal/ops/access-requests/ongoing/AccessRequestTomcat-0094.html">#94</a> for ADMEM, <a href="https://www-cdfonline.fnal.gov/internal/ops/access-requests/ongoing/AccessRequestTomcat-0108.html">#108</a> for COT), we need to move both the north and the south walls, as well as the east IMU steel and north east CMX.


<!-- Comment: Sat Feb  2 12;05;35 comment by...m mattson -->
<br>Request 94 was muon (CMP), the ADMEM request was 97.
<!-- Date: Sat Feb 2 08:45:29.424 2008 -->
Although this data is bad due to compromised COT readout, it would be nice to understand the features we see in the XFT monitoring.  Could experts take a look at <a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2008&action=view&page=97&frame=2&anchor=00:44:07.997'>this entry</a> and comment?  
<!-- Author: Erik :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257114' target=_top>257114</a>) -->


<!-- Date: Sat Feb 2 08:52:20.903 2008 -->
MCR called. They will be terminating the store at 9:00 as expected. We will start bringing everything to End of Store shortly.
<!-- Author: Ulrich Husemann :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257114' target=_top>257114</a>) -->

<!-- Date: Sat Feb 2 08:53:46.617 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257114>257114</A>
 Terminated at 2008.02.02 08:52:58
<!-- Author: RunControl -->


<!-- Date: Sat Feb 2 08:53:47.267 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257114>257114</A>
TERMINATE: End of store
<!-- Author: Si/Elisabetta -->
<!-- Comment: Sat Feb  2 09;03;04 comment by...Ulrich Husemann -->
The store has been terminated.
<!-- Date: Sat Feb 2 08:57:19.413 2008 -->
We had another COT trip while ramping down HV for EOS. This time only SL8 tripped. We will notify Bob and Aseet about this by e-mail.
<!-- Author: Ulrich Husemann :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257114' target=_top>257114</a>) -->


<!-- Date: Sat Feb 2 09:00:13.070 2008 -->
We found b0pcal11 Error during end of run. We tried to have special request for pcal11 but, it was disable to do "control-halt-analyzer" in the KVM machine. We shepherded pcal11 and saved trace in the PC. 
<!-- Author: Eunsin/Hyunsu :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257114' target=_top>257114</a>) -->


<!-- Date: Sat Feb 2 09:06:29.618 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257115>257115</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 09:08:35.551 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257115>257115</A>
TERMINATE: done
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 09:08:36.622 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257115>257115</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Feb  2 10;21;54 comment by...Erik -->
OK, but still CEM 12E ch 0 dead.  Try again after repair work?

<!-- Date: Sat Feb 2 09:09:32.279 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257116>257116</A>
ACTIVATE: RunSet: CALIB_CLC_QIE 
<!-- Author: Eunsin/Hyunsu -->



<!-- Date: Sat Feb 2 09:10:01.586 2008 -->
Detector hall wireless plugged in.
<!-- Author: m mattson (ADMEM) -->

<!-- Date: Sat Feb 2 09:10:47.983 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257116>257116</A>
TERMINATE: done
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 09:10:48.881 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257116>257116</A>
CALIBRATION: <U><B>clcqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#clcqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Feb  2 10;23;39 comment by...Erik -->
OK
<!-- Date: Sat Feb 2 09:13:29.128 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257117>257117</A>
ACTIVATE: RunSet: CALIB_QIE_FWD 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 09:13:30.638 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257117>257117</A>
TERMINATE: done
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 09:13:31.499 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257117>257117</A>
CALIBRATION: <U><B>bscqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#bscqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Feb  2 10;24;35 comment by...Erik -->
OK
<!-- Date: Sat Feb 2 09:17:25.409 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257118>257118</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL 
<!-- Author: Eunsin/Hyunsu -->



<!-- Date: Sat Feb 2 09:21:01.433 2008 -->
MCR called. The hall is ready for access. JJ is organizing the access. He has called in 6 people: Dervin Allen, Roberto Davilla, Steve Hahn, Sebastian Grinstein, Mark Mattson, Lou Morris. Jim Loskot (cryo tech) changed the collision hall fan settings for the access.
<!-- Author: Ulrich Husemann -->

<!-- Date: Sat Feb 2 09:23:02.549 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257118>257118</A>
TERMINATE: done
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 09:23:03.852 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257119>257119</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 09:25:20.264 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257119>257119</A>
TERMINATE: done
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 09:27:17.725 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257119>257119</A>
CALIBRATION: <U><B>pluglaser</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#pluglaser target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Feb  2 10;31;54 comment by...Erik -->
<p>Tower 11W wedge 17 channel 1 is quite low.  Not on reference plot.  This is in the CO fixlist as an "occasionally dead" tower in YMon plots.</p>
<a href='/cgi/elog/elog.pl?nb=2008&action=view&page=-702&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2008&action=view&page=-702' width='300'></a>

<!-- Date: Sat Feb 2 09:27:18.394 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257120>257120</A>
ACTIVATE: RunSet: BOTHPLUGS_MM 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 09:29:24.987 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257120>257120</A>
TERMINATE: done
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 09:34:56.671 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257121>257121</A>
ACTIVATE: RunSet: ACE_TOF_QIE_NEW 
<!-- Author: Eunsin/Hyunsu -->



<!-- Date: Sat Feb 2 09:34:56.944 2008 -->
Del Allspach called in. Bill Noe will come in and work on the solenoid again. The idea is to power down the solenoid, clean a valve and power it up again. We inform Sebastian, who is ok with the procedure. Jim Loskot will start powering down the solenoid now.
<!-- Author: Ulrich Husemann -->
<!-- Comment: Sat Feb  2 10;31;37 comment by...Ulrich -->
Jim informed us that the magnet is back to full power.
<!-- Date: Sat Feb 2 09:36:32.988 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257121>257121</A>
CALIBRATION: <U><B>tofqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#tofqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Feb  2 10;33;04 comment by...Erik -->
OK
<!-- Date: Sat Feb 2 09:37:01.965 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257121>257121</A>
TERMINATE: done
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 09:41:00.142 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257114>257114</A>
central: <tt>COT crates off due to fan failure</tt>
<!-- Author: Consumer Operator( Erik Brubaker) -->


<!-- Date: Sat Feb 2 09:41:01.952 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257114>257114</A>
trigger: <tt>Many problems related to missing COT crates in XFT; SVT affected through XFT.</tt>
<!-- Author: Consumer Operator( Erik Brubaker) -->


<!-- Date: Sat Feb 2 09:41:02.639 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257114>257114</A>
monitors: <tt>ObjectMon SVX do vs phi slide affected in first quadrant by COT/XFT/SVT issues.</tt>
<!-- Author: Consumer Operator( Erik Brubaker) -->


<!-- Date: Sat Feb 2 09:41:03.761 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257114>257114</A>
general: <tt>Lots of problems, all related to missing COT crates as far as I can tell.</tt>
<!-- Author: Consumer Operator( Erik Brubaker) -->



<!-- Date: Sat Feb 2 09:41:06.806 2008 -->
When the north wall was being moved, we had an SVX trip in bulkhead 3, wedge 10, layer 4. This happens every time the north wall is moved. With my ex-silicon-expert hat on, I have silenced the iFix alarm and left the ladder off for now.
<!-- Author: Ulrilch Husemann -->
<!-- Comment: Sat Feb  2 09;50;02 comment by...Ulrich -->
Sebastian Carron (Silicon MPC) called about the trip (which he got on his alphanumeric pager). 
<!-- Comment: Sat Feb  2 10;13;44 comment by...Ulrich -->
Sebastian is here to hockerize a crate.
<!-- Date: Sat Feb 2 09:42:48.533 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257122>257122</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 09:43:13.956 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257122>257122</A>
TERMINATE: done
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 09:45:42.070 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257123>257123</A>
ACTIVATE: RunSet: TOF_TAC_CALIB 
<!-- Author: Eunsin/Hyunsu -->



<!-- Date: Sat Feb 2 09:48:40.517 2008 -->
Sebastian called from the collision hall. The repair work seems to proceed well. He asks us to inform MCR that we will most probably be done within 2.5 to 3 hours, which JJ just did.
<!-- Author: Ulrich Husemann -->


<!-- Date: Sat Feb 2 09:51:43.500 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257123>257123</A>
TERMINATE: done
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 09:51:44.537 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257123>257123</A>
CALIBRATION: <U><B>toftac</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#toftac target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Feb  2 10;33;49 comment by...Erik -->
OK
<!-- Date: Sat Feb 2 09:52:46.497 2008 -->
We got a PSM alarm from the arches, but couldn't find the cause. The alarm clear itself within seconds.
<!-- Author: Ulrich Husemann -->
<!-- Comment: Sat Feb  2 10;02;44 comment by...m mattson (ADMEM) -->
<br>
You have two possible suspects... We turned off ccal12 (SE corner), or the other group is dealing with CONET and turned off the alarm.
<!-- Comment: Sat Feb  2 10;21;31 comment by...Ulrich -->
True, CONET_1 was already in alarm, so it must be ccal12, which is indeed an arch crate (which I wasn't aware of).
<!-- Date: Sat Feb 2 09:54:51.574 2008 -->
<center><a href='/cgi/elog/elog.pl?nb=2008&action=view&page=-695&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2008&action=view&page=-695' width='300'></a><a href='/cgi/elog/elog.pl?nb=2008&action=view&page=-701&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2008&action=view&page=-701' width='300'></a></center>
<!-- Author: Erik -->
<!-- Comment: Sat Feb  2 10;00;52 comment by...Erik -->
<p>Some minor CES issues:</p>
<ul>
<li>Left is strips; this warm region in 21E seems to be relatively new.  Is it a problem?</li>
<li>Right is wires; the hot strip at 0E16 is not new but makes it hard to spot any other differences with respect to the reference.  Is there a way to fix or mask it?</li>
</ul>

<!-- Comment: Sat Feb  2 12;54;56 comment by...Larry N -->
1) warm band in average ph tends to be single event coherent noise, the warm counting inn 9E is a bit scarier.  Seems to come and go. 2)  And yes a mask would be wonderful, nicht wahr.
<!-- Date: Sat Feb 2 09:58:28.331 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257124>257124</A>
 Activated at 2008.02.02 09:58:24
<!-- Author: RunControl -->


<!-- Date: Sat Feb 2 09:58:32.328 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257124>257124</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [7,1023,706] 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 10:00:18.478 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257124>257124</A>
 Terminated at 2008.02.02 10:00:07
<!-- Author: RunControl -->


<!-- Date: Sat Feb 2 10:00:48.090 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257114>257114</A>
RUNSTATUS: <PRE>Marked Bad, explanation:
L1T 
L2T 
COT 
SVT 
</PRE>
One COT power supply broke at the beginning of the current store. 
As a consequence, b0cot10, b0cot11 and b0cot12 are off, but data was 
taken with all sub-detectors included. The missing parts of the COT
influence L1 and L2 triggers, including SVT.
<!-- Author: cdfscico -->


<!-- Date: Sat Feb 2 10:00:49.867 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257124>257124</A>
TERMINATE: stop to have SMXR calib by requesting
<!-- Author: Eunsin/Hyunsu -->
<!-- Comment: Sat Feb  2 10;05;23 comment by...m mattson (ADMEM) -->
<br> L1 rate 48 Hz, so the L1 cables are on okay for ccal12.  A full spike run can be taken later.  Asking for a showermax calibration, as turning the power supply off and on sometimes upsets the SMX electronics.
<!-- Date: Sat Feb 2 10:05:23.325 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257125>257125</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Eunsin/Hyunsu -->



<!-- Date: Sat Feb 2 10:05:40.292 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257125>257125</A>
TERMINATE: done
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 10:09:02.413 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257125>257125</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Feb  2 10;15;50 comment by...Ulrich -->
Mark, are you checking the calibration results yourself?
<!-- Comment: Sat Feb  2 10;38;36 comment by...Erik -->
Checked later calib run below...
<!-- Date: Sat Feb 2 10:15:50.628 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257126>257126</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: Eunsin/Hyunsu -->



<!-- Date: Sat Feb 2 10:17:37.888 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257126>257126</A>
TERMINATE: done
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 10:17:38.311 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257126>257126</A>
CALIBRATION: <U><B>cemled</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cemled target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Feb  2 10;47;35 comment by...Erik -->
Checked later run below...
<!-- Date: Sat Feb 2 10:23:30.647 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257128>257128</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 10:23:31.997 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257128>257128</A>
TERMINATE: done
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 10:23:32.658 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257128>257128</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 10:23:33.257 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257128>257128</A>
TERMINATE: done
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 10:23:34.579 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257128>257128</A>
CALIBRATION: <U><B>calxef</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calxef target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Feb  2 10;47;53 comment by...Erik -->
Checked later run below...
<!-- Date: Sat Feb 2 10:24:27.249 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257129>257129</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Eunsin/Hyunsu -->



<!-- Date: Sat Feb 2 10:28:09.666 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257129>257129</A>
TERMINATE: done

<!-- Author: Eunsin/Hyunsu -->



<!-- Date: Sat Feb 2 10:28:09.429 2008 -->
Bob Wagner came in to reset the COT trip. He said that the voltage ramp needs more fine tuning for SL8 (which has the largest capacitance in the system).  
<p> 
Bob also pointed out that it is <b>very important to take COT calibrations after the repair</b>, as ADCs may have moved around.
<!-- Author: Ulrich Husemann -->
<!-- Comment: Sat Feb  2 10;51;29 comment by...Ulrich -->
TDCs that is.
<!-- Comment: Sat Feb  2 11;27;41 comment by...Ulrich -->
The calibrations check out ok, see below.
<!-- Date: Sat Feb 2 10:30:18.743 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257130>257130</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 10:30:19.860 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257130>257130</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Feb  2 10;38;21 comment by...Erik -->
OK
<!-- Date: Sat Feb 2 10:30:20.612 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257131>257131</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 10:31:26.564 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257131>257131</A>
CALIBRATION: <U><B>cemled</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cemled target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Feb  2 10;42;45 comment by...Erik -->
OK
<!-- Date: Sat Feb 2 10:31:37.645 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257131>257131</A>
TERMINATE: done
<!-- Author: Eunsin/Hyunsu -->






<!-- Date: Sat Feb 2 10:35:22.715 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257132>257132</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 10:35:57.340 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257132>257132</A>
CALIBRATION: <U><B>calxef</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calxef target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Feb  2 10;44;09 comment by...Erik -->
OK
<!-- Date: Sat Feb 2 10:35:58.488 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257132>257132</A>
TERMINATE: done
<!-- Author: Eunsin/Hyunsu -->




<!-- Date: Sat Feb 2 10:39:24.797 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257133>257133</A>
ACTIVATE: RunSet: ACE_QIE with fram download for ccal and pcal crates
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 10:40:45.658 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257133>257133</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->


<!-- Date: Sat Feb 2 10:40:53.295 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257133>257133</A>
TERMINATE: might be a problem with the CAFE we swapped in
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 10:42:35.406 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257134>257134</A>
ACTIVATE: RunSet: ACE_QIE again
<!-- Author: Eunsin/Hyunsu -->



<!-- Date: Sat Feb 2 10:42:47.053 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257134>257134</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->


<!-- Date: Sat Feb 2 10:43:40.202 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257134>257134</A>
TERMINATE: blah
<!-- Author: Eunsin/Hyunsu -->



<!-- Date: Sat Feb 2 10:45:46.582 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257135>257135</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 10:46:44.499 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257135>257135</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->


<!-- Date: Sat Feb 2 10:47:00.259 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257135>257135</A>
TERMINATE: new CAFE card had good pedestal but bad calib slopes
<!-- Author: Eunsin/Hyunsu -->






<!-- Date: Sat Feb 2 10:53:11.986 2008 -->
Mark and Steve need to go back into the collision hall and swap in another CAFE card. The current card has good pedestals but bad slopes.
<!-- Author: Ulrich Husemann -->
<!-- Comment: Sat Feb  2 11;00;44 comment by...Ulrich -->
Steve called from the hall. They forgot to plug in the collision hall wireless again. Hyunsu just did that for them.
<!-- Comment: Sat Feb  2 11;11;48 comment by...m mattson (ADMEM) and s hahn -->
<br>Thank you, gentlemen.
<!-- Date: Sat Feb 2 10:54:23.453 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257137>257137</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Eunsin/Hyunsu -->
<!-- Comment: Sat Feb  2 10;56;08 comment by...Eunsin -->
During config memory (heap) corruption detected for crate cot10. So we shepherd it.
<!-- Date: Sat Feb 2 10:54:45.208 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257137>257137</A>
TERMINATE: done
<!-- Author: Eunsin/Hyunsu -->



<!-- Date: Sat Feb 2 10:55:15.508 2008 -->
Dervin called from the hall. The new power supply is in place. We have checked the voltage readings, and they are all within specifications. The fan status is OK. They will start closing up the detector, while we are taking a COT calibration run.
<!-- Author: Ulrich Husemann -->


<!-- Date: Sat Feb 2 10:57:22.535 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257137>257137</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Feb  2 11;00;12 comment by...Erik -->
OK


<!-- Comment: Sat Feb  2 11;03;24 comment by...Ulrich -->
We inform Dervin about the COT calibrations. They continue closing up the detector. The CMX arch is already in.

<!-- Date: Sat Feb 2 11:11:27.614 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257138>257138</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Eunsin/Hyunsu -->
<!-- Comment: Sat Feb  2 11;13;20 comment by...m mattson (ADMEM) and s hahn -->
<br>Checked pedestals AND calibration slopes from QIETEST program, and new CAFE card seems okay.  If SMX calib and spike run are okay, we will close up south wall and get out of the hall.
<!-- Date: Sat Feb 2 11:12:33.285 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257138>257138</A>
TERMINATE: done
<!-- Author: Eunsin/Hyunsu -->



<!-- Date: Sat Feb 2 11:14:27.126 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257138>257138</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Feb  2 11;15;19 comment by...m mattson (ADMEM) and s hahn -->
<br>Nothing from ccal12, so a good sign.
<!-- Date: Sat Feb 2 11:14:59.559 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257139>257139</A>
 Activated at 2008.02.02 11:14:42
<!-- Author: RunControl -->


<!-- Date: Sat Feb 2 11:15:00.274 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257139>257139</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [7,1023,706] 
<!-- Author: Eunsin/Hyunsu -->
<!-- Comment: Sat Feb  2 11;18;30 comment by...Eunsin/Hyunsu -->
L1A rate is 48.8Hz
<!-- Date: Sat Feb 2 11:17:25.753 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257139>257139</A>
 Terminated at 2008.02.02 11:16:26
<!-- Author: RunControl -->


<!-- Date: Sat Feb 2 11:17:26.455 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257139>257139</A>
TERMINATE: L1A rate is 48.8Hz
<!-- Author: Eunsin/Hyunsu -->




<!-- Date: Sat Feb 2 11:23:32.883 2008 -->
Geumbong Yu is here to perform a CSL fail-over test. Her work should not interfere with anything else.
<!-- Author: Ulrich Husemann -->
<!-- Comment: Sat Feb  2 11;31;29 comment by...Geumbong -->
This is CSL failover test. RestartCSL with broken connection to remote logger.  
<!-- Comment: Sat Feb  2 12;30;23 comment by...Geumbong -->
<b>CSL test was successful!</b>
<p>
New script can make CSL running even if some of csl remote loggers are unreachable, i.e. hardware problems. I'll leave the csl with new script for a couple of days and make it a default logger then. CSL operation page is modified for shift crew to use "setup csl -d" for this moment.  For the test detail, please see <a href=
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=cs-logger&action=view&page=17&frame=2&anchor=12:19:16'>
CSL e-log </a>. Thanks
<!-- Comment: Sat Feb  2 12;32;12 comment by...Geumbong -->
All the system used for CSL fail-over test are restored. 
<!-- Date: Sat Feb 2 11:31:29.604 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257140>257140</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: TEST_CSL_TORTURE [2,1018,714] 
<!-- Author: Eunsin/Hyunsu -->



<!-- Date: Sat Feb 2 11:34:27.899 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257140>257140</A>
TERMINATE: finish

<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 11:36:34.554 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257141>257141</A>
ACTIVATE: RunSet: ACE_QIE with fram download, all cal crates
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 11:37:45.790 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257141>257141</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Feb  2 11;42;01 comment by...Erik -->
OK.  CEM 12E all better!

<!-- Comment: Sat Feb  2 11;57;07 comment by...m mattson, s hahn -->
<br>Swapped in a new CAFE card for ccal12 slot 16 ch 0.  The pedestals were okay, but after we left the hall (for the first time) found out that the slopes were zero, probably a bad charge injection circuit.  Went in a second time, new CAFE card looks good for both pedestal and calib slopes.
<!-- Date: Sat Feb 2 11:47:34.878 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257141>257141</A>
TERMINATE: done
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 11:49:14.165 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257142>257142</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: TEST_CSL_TORTURE [2,1018,714] 
<!-- Author: Eunsin/Hyunsu -->



<!-- Date: Sat Feb 2 11:50:18.604 2008 -->
We have officially finished our access about 10 minutes ago. D0 is done as well. 
<!-- Author: Ulrich Husemann -->
<!-- Comment: Sat Feb  2 12;06;41 comment by...m mattson -->
<br>Marked request 97 (ADMEM) as finished.  Updated reference values for the cal calibration.  Hall wireless unplugged.
<!-- Date: Sat Feb 2 11:59:57.069 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257142>257142</A>
TERMINATE: done
<!-- Author: Eunsin/Hyunsu -->



<!-- Date: Sat Feb 2 12:03:08.363 2008 -->
<h3>COT Power Supply Repair Successful</h3> 
We opened up East IMU, North East CMX, and the North muon wall. We replaced the CONET_1 power supply. Thanks Roberto and Lou! At the same time, Sebastian was trained in moving muons walls.
<!-- Author: Dervin Allen -->




<!-- Date: Sat Feb 2 12:07:36.884 2008 -->
Good work. Thanks everyone.
<!-- Author: Phil -->

<!-- Date: Sat Feb 2 12:11:50.126 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257144>257144</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: TEST_CSL_TORTURE [2,1018,714] 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 12:14:17.160 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257144>257144</A>
TERMINATE: done CSL test
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 12:19:08.007 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257145>257145</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: TEST_CSL_TORTURE [2,1018,714] 
<!-- Author: Eunsin/Hyunsu -->



<!-- Date: Sat Feb 2 12:20:52.426 2008 -->
We have contacted MCR about the plan for the Tevatron turn-on. They have just started with C4 lead work, which they will finish in about 30 minutes. After that they need another 30 minutes to turn on the TeV and 60 minutes for an extensive dry squeeze. So we should have another two hours of quiet time.  
 
<p> 
We have contacted Alexei Sedov, who will come in and do a plug sourcing run. He instructed us to ramp the plug HV and wait until he comes in. 
<!-- Author: Ulrich Husemann -->
<!-- Comment: Sat Feb  2 14;18;53 comment by...Sebastian -->
Thanks to Alexei for coming in to do this.
<!-- Date: Sat Feb 2 12:27:07.821 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257146>257146</A>
 Activated at 2008.02.02 12:26:14
<!-- Author: RunControl -->


<!-- Date: Sat Feb 2 12:30:13.861 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257146>257146</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,1019,706] 
<!-- Author: Eunsin/Hyunsu -->





<!-- Date: Sat Feb 2 12:49:45.966 2008 -->
Solenoid status is on. It is information for plug sourcing
<!-- Author: Eunsin/Hyunsu :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257146' target=_top>257146</a>) -->



<!-- Date: Sat Feb 2 13:00:43.889 2008 -->
From the MCR e-log: 
<pre> 
12:43:05-   The TH vacuum work has hit a setback. A the solenoid at the vacuum station blew apart. This adds 1.5 hours to the TeV access. - DJC 
</pre> 
So it seems we will have plenty of time for the plug sourcing today (Alexei arrived about 5 minutes ago).
<!-- Author: Ulrich Husemann -->

<!-- Date: Sat Feb 2 13:05:29.567 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257146>257146</A>
 Terminated at 2008.02.02 13:05:18
<!-- Author: RunControl -->


<!-- Date: Sat Feb 2 13:05:30.537 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257146>257146</A>
TERMINATE: done
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 13:18:18.743 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257147>257147</A>
TERMINATE: Plug Sourcing, tubes 3&4, PEM 0.0
<!-- Author: Eunsin/Hyunsu -->
<!-- Comment: Sat Feb  2 19;45;45 comment by...Willis -->
Solenoid is ON.
<!-- Date: Sat Feb 2 13:20:26.741 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257148>257148</A>
ACTIVATE: RunSet: PLUG_NEW  Trigger: COSMICS_NOTRACKS [9,982,707] 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 13:27:56.482 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257148>257148</A>
TERMINATE: Plug Sourcing, tubes 12&13, PEM 0.1
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 13:30:30.002 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257149>257149</A>
ACTIVATE: RunSet: PLUG_NEW  Trigger: COSMICS_NOTRACKS [9,982,707] 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 13:35:35.454 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257149>257149</A>
TERMINATE: Plug Sourcing, tubes 21&22, PEM 1.0
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 13:38:02.291 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257150>257150</A>
ACTIVATE: RunSet: PLUG_NEW  Trigger: COSMICS_NOTRACKS [9,982,707] 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 13:45:13.743 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257150>257150</A>
TERMINATE: Plug Sourcing, tubes 34&35, PEM 1.1
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 13:47:34.619 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257151>257151</A>
ACTIVATE: RunSet: PLUG_NEW  Trigger: COSMICS_NOTRACKS [9,982,707] 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 13:53:19.704 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257151>257151</A>
TERMINATE: Plug Sourcing, tubes 43&44, PEM 2.0
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 13:56:03.595 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257152>257152</A>
ACTIVATE: RunSet: PLUG_NEW  Trigger: COSMICS_NOTRACKS [9,982,707] 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 14:02:33.766 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257152>257152</A>
TERMINATE: Plug Sourcing, tubes 52&53, PEM 2.1
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 14:04:57.015 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257153>257153</A>
ACTIVATE: RunSet: PLUG_NEW  Trigger: COSMICS_NOTRACKS [9,982,707] 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 14:11:29.660 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257153>257153</A>
TERMINATE: Plug Sourcing, tubes 61&62, PEM 3.0
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 14:15:39.161 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257154>257154</A>
ACTIVATE: RunSet: PLUG_NEW  Trigger: COSMICS_NOTRACKS [9,982,707] 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 14:18:53.023 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257154>257154</A>
TERMINATE: Plug Sourcing, tubes 70&71, PEM 3.1
<!-- Author: Eunsin/Hyunsu -->



<!-- Date: Sat Feb 2 14:20:51.175 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257155>257155</A>
ACTIVATE: RunSet: PLUG_NEW  Trigger: COSMICS_NOTRACKS [9,982,707] 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 14:27:32.940 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257155>257155</A>
TERMINATE: Plug Sourcing, tubes 79&80, PEM 4.0
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 14:29:45.304 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257156>257156</A>
ACTIVATE: RunSet: PLUG_NEW  Trigger: COSMICS_NOTRACKS [9,982,707] 
<!-- Author: Eunsin/Hyunsu -->



<!-- Date: Sat Feb 2 14:33:34.985 2008 -->
Elisabette is ill and will not be on overlap shift this evening or Sunday evening. 
<!-- Author: jj -->

<!-- Date: Sat Feb 2 14:37:16.584 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257156>257156</A>
TERMINATE: Plug Sourcing, tubes 92&93, PEM 4.1
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 14:38:04.530 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257157>257157</A>
ACTIVATE: RunSet: PLUG_NEW  Trigger: COSMICS_NOTRACKS [9,982,707] 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 14:45:00.440 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257157>257157</A>
TERMINATE: Plug Sourcing, tubes 101&102, PEM 5.0
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 14:47:59.886 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257158>257158</A>
ACTIVATE: RunSet: PLUG_NEW  Trigger: COSMICS_NOTRACKS [9,982,707] 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 14:54:38.553 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257158>257158</A>
TERMINATE: Plug Sourcing, tubes 110&111, PEM 5.1
<!-- Author: Eunsin/Hyunsu -->



<!-- Date: Sat Feb 2 15:21:00.702 2008 -->
We have called MCR asking for an update on the Tevatron situation. As of now their repair work is finished and everybody is out of the tunnel: 
<pre> 
15:05:58-   Work is completed on the C47 spool. The upstream and downstream as-found resistances were 4 and 23 micro-Ohms respectively. We cleaned and replated both interfaces; found a cracked compression washer on the downstream lead. Post repair resistances were 2 and 1 micro-Ohms respectively. Also "Biddled" the D13 spool; as found resistances were 10 and 3 micro-Ohms. I opted not to service these joints. - brooker 
</pre> 
So the TeV turn-on will commence now, with 6 ramps and 1 hour of dry squeeze. We should have another two hours before shot setup.
<!-- Author: Ulrich Husemann -->

<!-- Date: Sat Feb 2 15:28:29.364 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257159>257159</A>
TERMINATE: Plug Sourcing, tubes 113&114, PHA 5.1
<!-- Author: Eunsin/Hyunsu -->



<!-- Date: Sat Feb 2 15:32:39.398 2008 -->
The plug sourcing ran into problems. One wire seems to be in an undefined position. According to Alexei and Willis, this does not cause problems in data taking, but should be fixed during the next access, also because it may be a radiation hazard, depending on where the source got stuck. Willis will submit the request. We will continue the plug sourcing for the rest of the quiet time.
<!-- Author: Ulrich Husemann -->

<!-- Date: Sat Feb 2 15:36:57.681 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257161>257161</A>
TERMINATE: Plug Sourcing, tubes 104&105, PHA 5.0
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 15:45:32.529 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257162>257162</A>
TERMINATE: Plug Sourcing, tubes 95&96, PHA 4.1
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 15:47:18.708 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257164>257164</A>
ACTIVATE: RunSet: PLUG_NEW  Trigger: COSMICS_NOTRACKS [9,982,707] 
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 15:53:31.548 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257164>257164</A>
TERMINATE: Plug Sourcing, tubes 82&83, PHA 4.0
<!-- Author: Eunsin/Hyunsu -->


<!-- Date: Sat Feb 2 15:55:17.527 2008 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>PhysicsTable</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L3Accepts</TH>
<TH>DelivLumi nb<SUP>-1</SUP></TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN=257114 target=_runSummary>257114</A> x3EC5A</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_5_01 [3,1016,713]</FONT></TD>
<TD ALIGN=CENTER>21:01:08</TD>
<TD ALIGN=CENTER>08:52:58</TD>
<TD ALIGN=CENTER>10:10:34</TD>
<TD ALIGN=RIGHT>   3,884,110</TD>   <TD ALIGN=RIGHT>  4,206.96</TD>
<TD ALIGN=RIGHT>  3,582.53</TD>
<TD ALIGN=CENTER>0</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>15:55:00</TD>
<TD ALIGN=CENTER>10:10:34</TD>
<TD ALIGN=RIGHT>     3,884,110</TD> <TD ALIGN=RIGHT>  4,206.96</TD>
<TD ALIGN=RIGHT>  3,582.53</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Sat Feb 2 15:59:28.858 2008 -->
<font color=red>Shift Summary: </font><pre>Smooth data-taking with compromised COT (broken power supply in 
b0cot12: b0cot10-12 off as a consequence). Access to repair COT 
power supply and CAFE card successful. Plug sourcing. 
 
9:00  Store 5876 terminated 
      COT trips again while ramping down 
 
9:30  CDF controlled access starts 
      Collision hall work: COT power supply swap, ADMEM CAFE card swap 
      Solenoid work: ramp down solenoid and clean a valve 
 
10:45 The new COT power supply is in place 
      The new ADMEM CAFE card does not work, needs to be swapped again 
 
11:15 CSL fail-over test 
 
11:40 Controlled access finished 
      The new COT calibrations check out fine 
      The replacement of the ADMEM card work fine 
 
12:15 The CDF and D0 accesses are over, but there is still an ongoing 
      Tevatron repair -&gt; use 2+ hours of quiet time for plug sourcing 
 
13:15 Plug sourcing starts 
      Tevatron repair work is delayed by at least 1.5 hours 
 
15:15 Plug sourcing problem: one source stuck 
 
15:20 Tevatron work finished, starting Tevatron turn-on 
</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   257114
Delivered Luminosity   219.3  
Acquired Luminosity    197.0  
Efficiency             89.8
   
</td></tr></table></pre>
<!-- Author: Ulrilch Husemann -->

<!-- Date: Sat Feb 2 16:01:54.782 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257165>257165</A>
TERMINATE: Plug Sourcing, tubes 73&74, PHA 3.1
<!-- Author: Eunsin/Hyunsu -->