<!-- Title: Eve shift. Tue Sep 26, 2006 -->
<!-- SciCo: Itsuo Nakano -->
<!-- DAQAce: Robert Forrest -->
<!-- MonAce: Hugo Beauchemin -->
<!-- CO: Tracey Berry -->
<!-- OpManager: Kojo Sato -->
<!-- Notes: -->
<pre>No beam.  No estimation.</pre>

<!-- Date: Tue Sep 26 16:13:04.073 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223722>223722</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Tue Sep 26 16;47;45 comment by...Tracey -->
calibration ok
<!-- Date: Tue Sep 26 16:13:42.724 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223724>223724</A>
ACTIVATE: RunSet: COT_ACE 
<!-- Author: EIko X 2080 -->
<!-- Comment: Tue Sep 26 16;17;27 comment by...Shin-Shan Yu (Eiko) -->
<pre>The COT ASDQ calibration shows that there is no brand new hot or dead cell.</pre>
<!-- Date: Tue Sep 26 16:14:29.240 2006 -->
Search and secure of collision hall after supervised access is done.
<!-- Author: Koji -->



<!-- Date: Tue Sep 26 16:21:29.375 2006 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2006&action=view&page=-12178&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-12178  width='300' align='top'></a></td><td>PHA bad ped are in Channel 0 and 1 so this is ok. and 18 W, 11 is a known bad channel. 
Most of the other channels have sigma<10 so are ok for data-taking.</td></tr></table>
<!-- Author: Tracey -->

<!-- Date: Tue Sep 26 16:32:56.810 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223726>223726</A>
TERMINATE: Plug Sourcing, PEM 0.0, tubes 3 & 4
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 16:44:54.144 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223728>223728</A>
TERMINATE: Plug Sourcing, PEM 0.1, tubes 12 & 13
<!-- Author: Jason -->



<!-- Date: Tue Sep 26 16:57:12.768 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223729>223729</A>
TERMINATE: Plug Sourcing, PEM 1.0, tubes 21 & 22
<!-- Author: Jason -->


<!-- Date: Tue Sep 26 17:04:59.930 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223730>223730</A>
TERMINATE: Plug Sourcing, PEM 1.1, tubes 34 & 35
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 17:16:24.634 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223732>223732</A>
TERMINATE: Plug Sourcing, PEM 2.1, tubes 52 & 53
<!-- Author: Roberto -->



<!-- Date: Tue Sep 26 17:36:31.955 2006 -->
Quick cleanup of Level3 to exclude a couple of nodes that have been giving problems because their disks are full. We'll deal with them offline and put them back at a later stage.
<!-- Author: Conor :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223739' target=_top>223739</a>) -->


<!-- Date: Tue Sep 26 17:38:42.062 2006 -->
<pre>Plug Sourcing, PEM 2.0, tubes 43 & 44</pre>
<!-- Author: Alexei Sedov :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223731' target=_top>223731</a>) -->


<!-- Date: Tue Sep 26 17:48:23.337 2006 -->
Note for plug-sourcerors regarding Level3: I know you would prefer not to have to reset after every run, and that's okay with me. But it might so happen that the random 'dropping out' behaviour will affect both subfarms, and then you will no longer get any events. This will look like: both converters are all green, and their event counts will stop increasing. If this happens, you can recover by just doing  a reset.
<!-- Author: Conor :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223739' target=_top>223739</a>) -->

<!-- Date: Tue Sep 26 17:49:14.156 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223739>223739</A>
TERMINATE: Plug Sourcing, PEM 3.1, tubes 70 & 71
<!-- Author: Roberto -->
<!-- Comment: Tue Sep 26 17;51;37 comment by...Roberto -->
<pre>Plug Sourcing, PEM 3.0, tubes 61 & 62 run 223738</pre>

<!-- Date: Tue Sep 26 17:53:43.915 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223741>223741</A>
TERMINATE: Plug Sourcing, PEM 4.0, tubes 79 & 80
<!-- Author: Roberto -->
<!-- Comment: Tue Sep 26 18;38;01 comment by...Alex & Gene -->
<pre>wrong number of events</pre>
<!-- Date: Tue Sep 26 17:55:31.797 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223742>223742</A>
ACTIVATE: RunSet: PLUG_NEW  Trigger: COSMICS_NOTRACKS [9,731,632] 
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 18:01:48.812 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223742>223742</A>
TERMINATE: Plug Sourcing, PEM 4.1, tubes 92 & 93
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 18:09:59.751 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223744>223744</A>
TERMINATE: Plug Sourcing, PEM 5.0, tubes 101 & 102
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 18:21:42.225 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223748>223748</A>
TERMINATE: Plug Sourcing, PEM 5.1, tubes 110& 111
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 18:30:38.908 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223754>223754</A>
TERMINATE: Plug Sourcing, PEM 5.1, tubes 114 & 113
<!-- Author: Roberto -->
<!-- Comment: Tue Sep 26 18;33;26 comment by...Roberto -->
<pre>instead of PEM it is PHA</pre>



<!-- Date: Tue Sep 26 18:38:24.863 2006 -->
We lost the run summary web page - nothing except the header part was showing. 
<BR> 
Restarted tomcat and we are good now.  
<!-- Author: Koji / Robert -->

<!-- Date: Tue Sep 26 18:39:01.874 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223755>223755</A>
TERMINATE: Plug Sourcing, PHA 5.0, tubes 104 & 105
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 18:39:01.561 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223755>223755</A>
TERMINATE: Plug Sourcing, PHA 5.0, tubes 104 & 105
<!-- Author: Roberto -->



<!-- Date: Tue Sep 26 18:39:35.467 2006 -->
<pre>Plug Sourcing, PEM 3.0, tubes 61 & 62</pre>
<!-- Author: Alex :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223738' target=_top>223738</a>) -->


<!-- Date: Tue Sep 26 18:41:45.093 2006 -->
<p>Today we updated the SLAM firmware to version 2011/2111 to address a problem with the stereo slope matching.  Diagnostic tests show no errors. 
 
<p>More details can be found <a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=xft&action=view&page=30&frame=2&anchor=18:40:45'> 
here</a>. 
<br><br> 
  
<!-- Author: K.Lannon, J.Efron, J. Slaunwhite -->

<!-- Date: Tue Sep 26 18:43:53.094 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223760>223760</A>
ACTIVATE: RunSet: DAQTEST 
<!-- Author: nils -->


<!-- Date: Tue Sep 26 18:47:11.730 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223759>223759</A>
TERMINATE: Plug Sourcing, PHA 4.1, tubes 95 & 96
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 18:52:05.778 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223761>223761</A>
TERMINATE: Plug Sourcing, PHA 4.0, tubes 82 & 83
<!-- Author: Roberto -->
<!-- Comment: Tue Sep 26 18;53;18 comment by...Gene & Alex -->
<pre>run is bad: no events</pre>


<!-- Comment: Tue Sep 26 19;02;27 comment by...Roberto -->
<pre>null run</pre>
<!-- Date: Tue Sep 26 18:55:15.353 2006 -->
New version of XFT Finder firmware uploaded to FPGAs (v7r7) in XFT06, XFT07 and C3PO (new test crate). Will test for 30min of cosmics later on today before being included in data taking.
<!-- Author: Aron + Alison -->

<!-- Date: Tue Sep 26 19:01:52.726 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223763>223763</A>
TERMINATE: Plug Sourcing, PHA 4.0, tubes 82 & 83
<!-- Author: Roberto -->
<!-- Comment: Tue Sep 26 19;25;01 comment by...Alex -->
<pre>wrong number of events</pre>
<!-- Date: Tue Sep 26 19:04:21.868 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223764>223764</A>
TERMINATE: bad run
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 19:19:22.087 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223765>223765</A>
TERMINATE: Plug Sourcing, PHA 4.0, tubes 82 & 83
<!-- Author: Roberto -->



<!-- Date: Tue Sep 26 19:30:35.399 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223766>223766</A>
TERMINATE: Plug Sourcing, PEM 4.0, tubes 79 & 80
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 19:39:49.412 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223767>223767</A>
TERMINATE: Plug Sourcing, PHA 3.1 tubes 74 & 73
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 19:47:38.633 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223768>223768</A>
TERMINATE: Plug Sourcing, PHA 3.0, tubes 64&65
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 19:54:03.988 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223769>223769</A>
TERMINATE: Plug Sourcing, PHA 2.1, tubes 55 & 56
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 20:00:35.751 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223770>223770</A>
TERMINATE: Plug Sourcing, PHA 2.0, tubes 46 & 47
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 20:08:15.842 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223771>223771</A>
TERMINATE: Plug Sourcing, PHA 1.1, tubes 37 & 38
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 20:15:29.459 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223775>223775</A>
TERMINATE: Plug Sourcing, PHA 1.0, tubes 25 & 24
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 20:23:35.834 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223776>223776</A>
TERMINATE: Plug Sourcing, PHA 0.1, tubes 15 & 16
<!-- Author: Roberto -->



<!-- Date: Tue Sep 26 20:25:17.429 2006 -->
<pre>Replaced the tdc board in slot 11 of b0cmu01 with 
another spare to see if this reduces the rate of  
done timeouts from this crate.  Replaced board  
#115 with board #134.</pre>
<!-- Author: Eric James -->

<!-- Date: Tue Sep 26 20:31:38.547 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223779>223779</A>
TERMINATE: Plug Sourcing, PHA 0.0, tubes 6 & 7
<!-- Author: Roberto -->



<!-- Date: Tue Sep 26 20:31:38.595 2006 -->
Talked with Jim at Cryo shift. 
The estimate for solenoid back up is 2-3 hours.
<!-- Author: Koji -->

<!-- Date: Tue Sep 26 20:41:14.893 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223780>223780</A>
TERMINATE: Plug Sourcing, PHA 0.0, tubes 1 & 2
<!-- Author: Roberto -->
<!-- Comment: Tue Sep 26 20;50;20 comment by...Roberto -->
<pre>PHA is PS</pre>
<!-- Date: Tue Sep 26 20:47:55.313 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223781>223781</A>
TERMINATE: Plug Sourcing, PS 0.1, tubes 10 & 11
<!-- Author: Roberto -->



<!-- Date: Tue Sep 26 20:56:23.058 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223782>223782</A>
TERMINATE: Plug Sourcing, PS 1.0, tubes 20 & 19
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 21:04:58.959 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223783>223783</A>
TERMINATE: Plug Sourcing, PS 1.1, tubes 32 & 33
<!-- Author: Roberto -->
<!-- Comment: Tue Sep 26 21;09;45 comment by...Alex & Gene -->
<pre>wrong number of events</pre>

<!-- Date: Tue Sep 26 21:11:36.534 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223784>223784</A>
TERMINATE: Plug Sourcing, PS 2.0, tubes 41 & 42
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 21:20:18.061 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223785>223785</A>
TERMINATE: Plug Sourcing, PS 1.1, tubes 32 & 33
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 21:28:19.509 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223786>223786</A>
TERMINATE: bad run
<!-- Author: Roberto -->



<!-- Date: Tue Sep 26 21:33:29.802 2006 -->
Talked with Jim at Cryo, the estimate would be around midnight for the whole solenoid turning on process.  He is going to call us back when he is ready. 
<BR> 
Called Marty at MCR as well. 
<br> 
D0 is out of collision hall and they are waiting for the D0 search&secure to be done - they also need some quiet time after that. 
<br> 
It is likely that we can enjoy this quiet time until around 23:00.  They are going to call us back before ending the quiet time anyway. 
<br> 
Also, they told me that the store establishment would most probably happen during the middle of the owl shift.  
<!-- Author: Koji -->

<!-- Date: Tue Sep 26 21:37:56.018 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223787>223787</A>
TERMINATE: Plug Sourcing, PS 2.1, tubes 51 & 50
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 21:45:32.464 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223788>223788</A>
TERMINATE: bad run
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 21:48:49.304 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223789>223789</A>
TERMINATE: bad run
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 22:01:56.982 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223790>223790</A>
TERMINATE: Plug Sourcing, PS 3.0, tubes 60 & 59
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 22:09:16.837 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223791>223791</A>
TERMINATE: Plug Sourcing, PS 3.1, tubes 68 & 69
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 22:13:33.864 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223793>223793</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 22:18:31.317 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223793>223793</A>
TERMINATE: Multiple DTO's from imu00, HRR didn't recover.
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 22:18:32.065 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223792>223792</A>
TERMINATE: Plug Sourcing, PS 4.0, tubes 78 & 79
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 22:18:33.205 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223794>223794</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 22:21:22.794 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223796>223796</A>
TERMINATE: muon calibration failed
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 22:23:29.180 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223797>223797</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 22:26:57.039 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223795>223795</A>
TERMINATE: Plug Sourcing, PS 4.1, tubes 90 & 91
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 22:34:22.108 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223798>223798</A>
TERMINATE: Plug Sourcing, PS 5.0, tubes 99 & 100
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 22:35:17.840 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223800>223800</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 22:38:47.162 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223801>223801</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 22:42:02.702 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223799>223799</A>
TERMINATE: Plug Sourcing, PS 5.1, tubes 108 & 109
<!-- Author: Roberto -->



<!-- Date: Tue Sep 26 22:43:26.025 2006 -->
Called MCR, Brian expects ~2hrs longer of quiet time.
<!-- Author: Koji -->
<!-- Comment: Tue Sep 26 22;50;15 comment by...Koji -->
Jim is going through the check list before the Solenoid power up.
(LLDEW is 26.93% now and the rate is 5%/hr, we will be ready to start ramping up solenoid within 1hr or so).

<!-- Date: Tue Sep 26 22:58:30.025 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223804>223804</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 23:01:35.932 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223804>223804</A>
TERMINATE: imu00 DTO, shepherding happened before the run and didn't work (again, like previous couple runs).
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 23:01:56.543 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223803>223803</A>
TERMINATE: Plug Sourcing, SMX 1.1, tubes 88 & 89
<!-- Author: Roberto -->



<!-- Date: Tue Sep 26 23:05:11.928 2006 -->
<pre>RTSERVER error.(several error alarms from 21:00 to 23:00 
connection failed with clc01</pre>
<!-- Author: Roberto :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223805' target=_top>223805</a>) -->
<!-- Comment: Tue Sep 26 23;05;54 comment by...Roberto -->
<pre>new error,now with imu00</pre>

<!-- Date: Tue Sep 26 23:08:03.702 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223805>223805</A>
TERMINATE: bad run
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 23:08:04.124 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223806>223806</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Roberto -->
<!-- Comment: Tue Sep 26 23;15;04 comment by...Koji -->
<pre>We have been having sticky DTO's in the last several muon calibrations.
Already talked with Eric James on the phone, he is going to take a look.
Meanwhile, we have been trying a little with this crate, no success yet...
We have tried shepherding the crate followed by run restarting.
In this last try, I tried reset_crate followed by shepherding.
The following is the Errorlog.

Code: 0x4710800000011  Mnemonic: FERML_VME_BUS_ERROR
Node Name: b0imu00; Node Id: 131.225.237.164
Process: Messenger; PID: 15493616; Time: 11:08:12 PM
Code Line: 859; Routine: FER_mess.c
Parameters: TDC_readyForReadoutNoWait: slot=14 VME bus error reading TDC status register [9] 
</pre>
<!-- Date: Tue Sep 26 23:11:18.580 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223806>223806</A>
TERMINATE: Tried reset_crate + shepherding before running.
DTO imu00.
<!-- Author: Roberto -->



<!-- Date: Tue Sep 26 23:20:30.217 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223808>223808</A>
TERMINATE: Plug Sourcing, SMX 1.0, tubes 86 & 87
<!-- Author: Roberto -->



<!-- Date: Tue Sep 26 23:25:23.172 2006 -->
Jim came upstairs to tell that he finished his solenoid checklist, when solenoid alarm started - he ran downstairs and is working on it.
<!-- Author: Koji -->

<!-- Date: Tue Sep 26 23:31:41.041 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223810>223810</A>
TERMINATE: tried rebooting from vxlogin. doesn't work, same bus error on slot 14 again.
<!-- Author: Roberto -->



<!-- Date: Tue Sep 26 23:31:43.655 2006 -->
<pre>MCR called.  They have a problem (proton injection).  They'll access to Tevatron.  It'll take several hours.  We won't have beam for several hours. 
</pre>
<!-- Author: Itsuo -->

<!-- Date: Tue Sep 26 23:33:51.136 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223809>223809</A>
TERMINATE: Plug Sourcing, SMX 0.1, tubes 30 & 31
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 23:38:23.539 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223812>223812</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 23:43:57.582 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223811>223811</A>
TERMINATE: Plug Sourcing, SMX 0.0, tubes 28 & 29
<!-- Author: Roberto -->


<!-- Date: Tue Sep 26 23:47:21.961 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223814>223814</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Roberto -->



<!-- Date: Tue Sep 26 23:48:14.127 2006 -->
<pre>Since there will be no beam for several hours, released CO.</pre>
<!-- Author: Itsuo -->

<!-- Date: Tue Sep 26 23:55:12.246 2006 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>PhysicsTable</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L3Accepts</TH>
<TH>DelivLumi nb<SUP>-1</SUP></TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>23:55:01</TD>
<TD ALIGN=CENTER>::</TD>
<TD ALIGN=RIGHT></TD>               <TD ALIGN=RIGHT></TD>
<TD ALIGN=RIGHT></TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->


<!-- Date: Tue Sep 26 23:59:01.071 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223816>223816</A>
 Activated at 2006.09.26 23:57:30
<!-- Author: RunControl -->



<!-- Date: Tue Sep 26 23:59:08.485 2006 -->
<font color=red>Shift Summary: </font><pre>Since we had no beam, we ran Plug Sourcing, Calibrations e.t.c. 
We won't have beam for several hours due to Tevatron access.</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   
Delivered Luminosity   0  
Acquired Luminosity    0  
Efficiency             0
   
</td></tr></table></pre>
<!-- Author: Itsuo -->


<!-- Date: Tue Sep 26 23:59:20.658 2006 -->
<pre> Attention!!! L2 Decision Timeout ERROR !!.  
  
 If L2 Timeouts persist, please page L2  
 
Issued manual HRR.</pre>
<!-- Author: Ford :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223816' target=_top>223816</a>) -->
<!-- Comment: Wed Sep 27 00;05;37 comment by...Ford -->
<pre>Happened again ~5 min later. Issued another HRR.</pre>
<!-- Comment: Wed Sep 27 00;08;09 comment by...Ford -->
<pre>And again just now</pre>
<!-- Date: Wed Sep 27 00:01:17.692 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=223816>223816</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,730,639] 
<!-- Author: Ford -->