<!-- Title: Eve shift. Thu Mar  7, 2002 -->
<!-- SciCo: Lina Galtieri -->
<!-- DAQAce: Yeongdae Shon -->
<!-- MonAce: Andy Hocker -->
<!-- CO:  Hartmut Stadie -->
<!-- OpManager: Bill Orejudos -->
<!-- Notes: -->
<pre></pre>


<!-- Date: Thu Mar 7 16:44:56 2002 -->
<pre>Supervised access in progress. To end at 20:30</pre>
<!-- Author: Lina -->


<!-- Date: Thu Mar 7 16:53:42 2002 -->
SM QIE Calibration 
  
<!-- Author: Y Shon :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=140539' target=_top>140539</a>) -->
<!-- Date: Thu Mar 7 17:00:32 2002 -->
I unsilenced all PSM alarms that Muge silenced on day shift. 
A lot of them had already been unsilenced.  By whom, I don't 
know.  Anyway, SVT guys joined in the game by causing an alarm 
on 2RR30C_1 without warning me.  I silenced it and made them 
promise to tell me when they were done working.
<!-- Author: Andy H. -->



<!-- Date: Thu Mar 7 17:06:44 2002 -->
<pre>CAl QIE with only WCAL's guided by expert</pre>
<!-- Author: Y Shon :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=140541' target=_top>140541</a>) -->
<!-- Comment: Thu Mar  7 17;37;44 comment by...rde -->
Downloaded endwall FRAN to get proper pedestals into wcal01 after channel swap during access.
<!-- Date: Thu Mar 7 17:16:55 2002 -->
Cal QIE, setup failed again because of Consummer Monotor Proxy problem. Paged Badgett, and he restart the proxy.
<!-- Author: Y Shon :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=140542' target=_top>140542</a>) -->


<!-- Date: Thu Mar 7 17:18:00 2002 -->
Cal QIE successful after restarting the consumer monitor proxy.
<!-- Author: Y Shon :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=140543' target=_top>140543</a>) -->


<!-- Date: Thu Mar 7 17:23:04 2002 -->
CHA ASD/TDC access work has been finished. <BR> 
LEDs are turned back on.
<!-- Author: Min-Jeong -->


<!-- Date: Thu Mar 7 17:28:19 2002 -->
CMP access work: Oscillations on South Wall stacks 29 and 39 fixed. S32L0 was dead, solved by replacing preamp card.   
Problem on North Wall S22L1 (low efficiency): swapped paddle card TDC slot 19 with 20.
<!-- Author: Lucio and Hyunsoo -->


<!-- Date: Thu Mar 7 17:30:53 2002 -->
<pre>SMX work 
 
Worked on single channel failures on W19,4W,1E,W18,W22 
 
Fixed crate problem with wacky peds on E3 
 
Swapped SMC card on pcal02, slot4 (box 6)</pre>
<!-- Author: byrum,lindgren,kuhlmann,nodulman -->


<!-- Date: Thu Mar 7 17:48:08 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=140546">140546</A>
 Activated at 2002.03.07 17:45:42
<!-- Author: RunControl -->
<!-- Comment: Thu Mar  7 17;52;56 comment by...Robin Erbacher -->
Took at spike run (140546) to check trigger rates, to make sure
that there were no loose trigger cables.  Average rate from
PreFred is 114 Hz, so everything looks good.  Hartmut will put trigMon plot into elog.
<!-- Date: Thu Mar 7 17:50:50 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=140546">140546</A>
 Terminated at 2002.03.07 17:49:59
<!-- Author: RunControl -->



<!-- Date: Thu Mar 7 17:52:36 2002 -->
<pre>Access to Alcove crates: 
1. power cycling fixed readout problem on the Roman Pot SMXR. 
2. installed a new (inverting) transition board for tower 
readout (tower #0-#18) in west MP. 
3. installed trigger patch panels in both crates to get MP L1 
trigger bits. Connected to ADMEM trigger outputs with short  
cables, and a long cable which is running up to L1 global crate. 
4. emptied a container for leak on west MP. 
</pre>
<!-- Author: K. Terashi -->



<!-- Date: Thu Mar 7 17:54:06 2002 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2002&action=view&page=-1186&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2002&action=view&page=-1186  width='300' align='top'></td><td>spike run.</a></td></tr></table>
<!-- Author: Hartmut -->


<!-- Date: Thu Mar 7 18:25:35 2002 -->
CLC experts turned on HV.  Alarm, but that's normal according to 
them.
<!-- Author: Andy H. -->


<!-- Date: Thu Mar 7 18:26:35 2002 -->
Installed CAEN crate interlocks.  There is now a switch which 
can be used to turn off each crate separately. The switches 
are in the same rack as the 5V enable power supplies. 
 
  The interlock cable for SE Bot Bot failed.  This crate is 
not yet interlocked. 
 
 
  
<!-- Author: herndon -->


<!-- Date: Thu Mar 7 18:40:49 2002 -->
<pre>Summary of Muon FER work today :  
 
Installed SciMezz cards on BSU and TSU tdcs in b0imu00 and imu01.  
PALs exchanged on the transition modules for these tdcs and fiber run to the muon trigger crate. Hopefully BMU+BSU|TSU triggers are not far away! 
 
Pulled faulty tdcs in imu01 - slot14 (#218) and cmp00 slot 15 (#256). Replaced with ser#226 and #391 respectively.  
 
226 - has been known to cause low levels of BC errors in the past. The spare situation for ECL tdcs is that we have -1 at this point. Ive asked Arnd to mask off the run-halting on BC errors in imu crates for now.  
 
 </pre>
<!-- Author: eric & dan -->
<!-- Comment: Thu Mar  7 19;05;23 comment by...dan -->
Decided that my email to Arnd (with him being in another country) wasnt too useful, and I used the CMX miniskirt TDC in slot 14 of imu01

<!-- Date: Thu Mar 7 18:43:29 2002 -->
<pre>TDCtest on the cards pulled from muon system today. After 10 attemps - the behavior seen downstairs hasnt been duplicated. 
 
Furthermore, the diag crate seems to not want to follow the abort transition to start another run here is the error from terminal :  
Sent ack to Run Control: 3 CALIBRATION DONE SUCCESS  
WARNING: Lost connection to RTlm: error code = 514. 
Attempting to reconnect to RTlm. 
Connecting to &lt;tcp:localhost&gt; RTlm. 
Using tcp protocol. 
WARNING: Could only recv -1/8 first bytes from connection client:tcp:localhost:5111. 
Error was &lt;timeout reached&gt;. 
Could not connect to &lt;tcp:localhost&gt; RTlm. 
Skipping starting &lt;start_never:tcp:localhost&gt; RTlm. 
Cannot reconnect to RTlm. 
Cannot continue. 
 
 
giving up for now.....</pre>
<!-- Author: dan -->


<!-- Date: Thu Mar 7 18:46:54 2002 -->
<HR> 
HAD ASD/TDC access report: <BR>  
<HR> 
1. Nokolay replaced CEM/CHA PISA boxes on 6E. This seems to  reduce noise of 6E CHA ASD/TDC system as well, when checked  
noise with frequency meter after the work. Needs to be 
confirmed with data. <BR> <BR>   
 
2. 4 ASD boards of WHA which are not instrumented with new transition boards (0-5E,18-23E) were taken from the crates, 
and two of them were installed in CHA (12-14E, 18-20E).  
The rest of cards (2 from CHA and 2 from WHA) will be  
examined upstairs by Marco. <BR> <BR> 
 
3. CHA 0E-5L & 0E-5R PMT Bases replaced by Nikolay. This channel 
was always dead regardless of the replacement of ASDs. This  
is a study to figure out the problem of channels belong to  
same category. <BR> <BR> 
 
4. Swapped a CHA ASD board (21-23W) with a spare board to 
examine a dead channel in old card.<BR><BR> 
 
5. Swapped a PHA ASD board (18-20W) with a spare board to see 
how dead channels change. <BR> <BR> 
<!-- Author: Min-Jeong -->


<!-- Date: Thu Mar 7 19:00:59 2002 -->
<pre>Dosimetry: 
 
- Measured location of SVX power supplies (FIB racks) with 
  respect to beam pipe. 
 
- Investigated possible locations to place radiation dosimeters 
  (TLD's) for the measurement of the radiation field map of the  
  B0 collision hall.  
  Marked some of these locations on the north and south walls.</pre>
<!-- Author: Kostas Kordas -->



<!-- Date: Thu Mar 7 19:26:14 2002 -->
<pre>several works (QIE calib and DAQ test with new miniplug 
transition board) were done with fcal crates. Now they are free.</pre>
<!-- Author: K. Terashi -->


<!-- Date: Thu Mar 7 21:23:34 2002 -->
<pre>Supervised access is finished. Search and secure completed</pre>
<!-- Author: Lina -->
<!-- Comment: Tue Mar 12 10;45;20 comment by...T. Phillips -->
Counters installed for shielding test are desribed in <a href= /cgi/elog/det-system-elog.pl?nb=losses&action=view&page=16&scroll=true> radiation losses logbook. </a>
<!-- Date: Thu Mar 7 21:52:51 2002 -->
Laser run
<!-- Author: Y Shon :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=140562' target=_top>140562</a>) -->


<!-- Date: Thu Mar 7 22:14:53 2002 -->
<pre>QIE calibrations for the forward detectors: BSC, Miniplugs, Roman pots: 
 
The rc complained about  
CLC_00 missing card ACMEM_CLC_01 in slot 4 
CLC_00 missing card ACMEM_CLC_02 in slot 6 
CLC_00 missing card ACMEM_CLC_03 in slot 7 
CLC_00 missing card ACMEM_CLC_04 in slot 9 
CLC_00 missing card ACMEM_CLC_05 in slot 10, 
but the calibration seemed to have finished successfully. 
</pre>
<!-- Author: Y Shon :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=140565' target=_top>140565</a>) -->
<!-- Comment: Thu Mar  7 22;22;11 comment by...Y Shon -->
DBANA results are very similar to the standard plots.
<!-- Date: Thu Mar 7 22:18:48 2002 -->
Nikolay Luzhetskiy is here to do Wall calorimeter calibration. Magnetic field is up
<!-- Author: Lina -->


<!-- Date: Thu Mar 7 22:21:43 2002 -->
CMP HV has been acting weird all night.  Most of time it's 
ON, but will occasionally flash like it's ramping, and now it 
had a trip.  I put it to STBY and will keep an eye on it.
<!-- Author: Andy H. -->



<!-- Date: Thu Mar 7 22:44:09 2002 -->
<pre>Jason Nielsen is doing charge injection on L00 chips</pre>
<!-- Author: Lina -->


<!-- Date: Thu Mar 7 22:47:45 2002 -->
3 tripped channels on the TOF prompted me to page Gervasio. 
He was as puzzled as I was, since the PMTs were off to begin 
with.  He reset the trips.
<!-- Author: Andy H. -->


<!-- Date: Thu Mar 7 22:54:10 2002 -->
COT calibration
<!-- Author: Y Shon :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=140574' target=_top>140574</a>) -->


<!-- Date: Thu Mar 7 22:56:42 2002 -->
MUON calibration
<!-- Author: Y Shon :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=140576' target=_top>140576</a>) -->

<!-- Date: Thu Mar 7 23:55:58 2002 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>23:55:01</TD>                                                       
<TD ALIGN=CENTER>::</TD>                                                
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Fri Mar 8 00:11:49 2002 -->
<font color=red>Shift Summary: </font><pre>Access completed by 2100. Did calibrations of some systems. 
Nykolay did a Wall calibration, he is analysing the data to make sure it is OK, if not he wants to remake the calibration.  
Jason is doing the L00 charge injection. He will need some more time</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   140562-140576
Delivered Luminosity   0  
Acquired Luminosity    0  
Efficiency             0
   
</td></tr></table></pre>
<!-- Author: Lina -->