<!-- Title: Eve shift. Fri Feb 28, 2003 -->
<!-- SciCo: John Yoh -->
<!-- DAQAce: David Clark -->
<!-- MonAce: Alberto Annovi -->
<!-- CO: Andreas Korn,(Galyardt -->
<!-- OpManager: Nathan Eddy -->
<!-- Notes: -->
<pre>Waiting for beam (in Proton setup mode)</pre>


<!-- Date: Fri Feb 28 16:14:37 2003 -->
Short PSM alarm for endwall channel COSET_2 +5VDigital. Cleared by itself.
<!-- Author: Alberto -->


<!-- Date: Fri Feb 28 16:19:13 2003 -->
<pre>While I was checking C:TEVSYN for the shotsetup checklist,  
The corresponding line looked like this: 
 
-C:TEVSYN   CDF Tev....    .881   .881 MREV 
^^ is this minus sign of any meaning? I suppose it's not. 
</pre>
<!-- Author: Alberto -->
<!-- Comment: Sat Mar  1 08;13;29 comment by...Steve Hahn -->
Actually the minus sign before a device on a parameter page does have a meaning: it means you cannot control the value of that device from the console you are using.  In fact, C:TEVSYN can only be changed by the MCR.
<!-- Date: Fri Feb 28 16:37:15 2003 -->
<strong>Integrated dosage after Proton injection</strong><TABLE bgcolor=#cdc9a5> 
<TR><TH>Date</th><th>Time</th><th>BLM</th><th colspan=2>Dose</th></tr> 
<tr><td>2003.02.28</td><td>16:33:13</td><td>W Inner BLM</td><td>4.89</td><td>RADS</td></tr> 
<tr><td>2003.02.28</td><td>16:33:13</td><td>W Outer BLM</td><td>1.20</td><td>RADS</td></tr> 
<tr><td>2003.02.28</td><td>16:33:13</td><td>E Inner BLM</td><td>22.12</td><td>RADS</td></tr> 
<tr><td>2003.02.28</td><td>16:33:13</td><td>E Outer BLM</td><td>33.46</td><td>RADS</td></tr> 
</TABLE>  
<!-- Author: Alberto -->
<!-- Comment: Fri Feb 28 16;38;10 comment by...Alberto -->
Proton intensity is 8.9E12.


<!-- Date: Fri Feb 28 16:52:34 2003 -->
HEVB error: Corrupted data detected in several events all from same crate and channel. Here's info. for first one -  
<pre>Event 20438, crate 56, channel 7 has either bad Tracer ID or bad markers around Tracer word. 
 Tracer word event ID is 7, should be 6. 
Tracer word = 0x0fc70038 (I think). 
Channel data = 0x000001c8 0x0fc70038 0x434f5444 0x00140008 0x00000000 0x00000023 0x00000003 0x00000010 ... 
... 0x00000714 0x00000000 0x0000cbe3 0x00000000 0xaaaaaaaa 0x55555555 0x0fc70038 0x80000000 --></pre>
<!-- Author: David :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159588' target=_top>159588</a>) -->
<!-- Comment: Fri Feb 28 17;02;33 comment by...David -->
By the way, the several errors occurred in consecutive events.


<!-- Date: Fri Feb 28 17:30:22 2003 -->
short PSM alarm 1st floor channel 1RR18D_1 +5V digital
<!-- Author: Alberto -->
<!-- Comment: Fri Feb 28 19;06;55 comment by...Alberto -->
happened again for channel 1RR18D_1 -5.2V analog
<!-- Date: Fri Feb 28 17:57:00 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1653&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1653' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1654&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1654' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1655&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1655' width='300'></a></center>
<!-- Author: Andreas -->
<!-- Comment: Fri Feb 28 17;58;49 comment by...Andreas -->
On request from last shift.<br>
Checked CEMXEF Calibration from Run 159578.<br>
The CEMLED does NOT find any NEW dead channels.


<!-- Date: Fri Feb 28 18:12:22 2003 -->
<strong>Integrated dosage after antiprotons injection</strong><TABLE bgcolor=#cdc9a5> 
<TR><TH>Date</th><th>Time</th><th>BLM</th><th colspan=2>Dose</th></tr> 
<tr><td>2003.02.28</td><td>18:08:13</td><td>W Inner BLM</td><td>5.04</td><td>RADS</td></tr> 
<tr><td>2003.02.28</td><td>18:08:13</td><td>W Outer BLM</td><td>1.20</td><td>RADS</td></tr> 
<tr><td>2003.02.28</td><td>18:08:13</td><td>E Inner BLM</td><td>22.64</td><td>RADS</td></tr> 
<tr><td>2003.02.28</td><td>18:08:13</td><td>E Outer BLM</td><td>34.20</td><td>RADS</td></tr> 
</TABLE>  
<!-- Author: Alberto -->

<!-- Date: Fri Feb 28 18:15:33 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159588">159588</A>
TERMINATE: End long Shot Setup run.
<!-- Author: David -->


<!-- Date: Fri Feb 28 18:30:05 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159598">159598</A>
 Activated at 2003.02.28 18:29:05
<!-- Author: RunControl -->


<!-- Date: Fri Feb 28 18:30:06 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159598">159598</A>
ACTIVATE: Min Bias run
<!-- Author: David -->


<!-- Date: Fri Feb 28 18:38:13 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159598">159598</A>
 Terminated at 2003.02.28 18:38:10
<!-- Author: RunControl -->


<!-- Date: Fri Feb 28 18:38:14 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159598">159598</A>
TERMINATE: End Min Bias Run.
<!-- Author: David -->



<!-- Date: Fri Feb 28 18:42:07 2003 -->
<pre>CMU trip just after a spike in LOSTP. Channel TOP NE tripped. 
I turned it to standby and after 2 minutes back to on.</pre>
<!-- Author: Alberto -->
<!-- Comment: Fri Feb 28 19;30;29 comment by...Alberto -->
happened again
<!-- Comment: Fri Feb 28 20;54;27 comment by...Alberto -->
happened again
<!-- Date: Fri Feb 28 18:45:11 2003 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2003&action=view&page=-1656&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-1656  width='300' align='top'></td><td>Low occupancy in COT SL7, Cell 86.</a></td></tr></table>
<!-- Author: Jason & Andreas -->
<!-- Comment: Fri Feb 28 19;07;32 comment by...Jason & Andreas  -->
Kaori say's this is exactly as expected, this cell was disabled to recover SL7.
<!-- Date: Fri Feb 28 18:46:37 2003 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2003&action=view&page=-1657&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-1657  width='300' align='top'></td><td>Losses after scraping</a></td></tr></table>
<!-- Author: Alberto -->


<!-- Date: Fri Feb 28 18:48:32 2003 -->
Turning on silicon
<!-- Author: Alberto -->

<!-- Date: Fri Feb 28 18:53:50 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159603">159603</A>
 Activated at 2003.02.28 18:54:04
<!-- Author: RunControl -->



<!-- Date: Fri Feb 28 18:53:50 2003 -->
<pre>Some updates have been made in order to help RunControl  
to run on the RedHat 7.3.1 machines (kernel 2.4.18-10) with  
the JDK version v1_4_1-01 (Sun). 
 
If there is a significant amount of time (more than 2 hours) 
between stores this weekend, please try to use RunControl  
on b0dap73, per my previous instructions. 
 </pre>
<a href= 
'/cgi/elog/elog.pl?nb=2003&action=view&page=173&frame=2&anchor=16:00:21'> 
Click here for more instructions</a>
<!-- Author: W.Badgett -->

<!-- Date: Fri Feb 28 18:54:17 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159603">159603</A>
ACTIVATE: AAA_Current run.
<!-- Author: David -->



<!-- Date: Fri Feb 28 18:55:12 2003 -->
short PSM alarm for SVX_SET_1 channel -5.2V analog. 
It happened while turning on silicon.
<!-- Author: Alberto -->
<!-- Comment: Fri Feb 28 18;59;47 comment by...SCN -->
VME power supplies and ladder power supplies are completely 
decoupled (one is on 400 Hz, the other on 60 Hz) so I do
not really see a correlation-it is not that the VME boards 
require more current when the ladders are ON rather than
STANDBY... 


<!-- Comment: Fri Feb 28 19;55;12 comment by...Alberto -->
happened again.
<!-- Comment: Fri Feb 28 19;57;02 comment by...Alberto -->
happened again
<!-- Comment: Fri Feb 28 20;43;57 comment by...Alberto -->
happened again.
<!-- Comment: Fri Feb 28 20;55;33 comment by...Alberto -->
<a href='/cgi/elog/elog.pl?nb=2003&action=view&page=177&frame=2&anchor=20:53:53'>This is the plot for the last 3 hours </a> 
<!-- Comment: Fri Feb 28 21;14;23 comment by...Alberto -->
happened again.
<!-- Comment: Fri Feb 28 21;53;53 comment by...Alberto -->
happened again. 
<!-- Comment: Fri Feb 28 22;32;19 comment by...Alberto -->
happened again.
<!-- Comment: Fri Feb 28 23;12;03 comment by...Alberto -->
happened again. 
<!-- Comment: Fri Feb 28 23;34;30 comment by...Alberto -->
happened again. 
<!-- Date: Fri Feb 28 19:00:57 2003 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2003&action=view&page=-1658&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-1658  width='300' align='top'></td><td>BEST SVT COVERAGE EVER since the beginning of RUN2</a></td></tr></table>
<!-- Author: THE SILICON GROUP -->


<!-- Date: Fri Feb 28 19:03:27 2003 -->
Store 2285 : 
 
18:10	Pbars in --60 so-so groups, 3 wimpy ones 
18:29	Scraping finished 
18:30 	CLC HV up --LUm is 13.6 E30 
18:45 or so --lost about 10% of lum due to separator spark; 
	Lum now 12.2 and stable
<!-- Author: John Yoh -->


<!-- Date: Fri Feb 28 19:03:30 2003 -->
Done Timeout in WCAL00: 
<PRE>Event 51006: Bunchcounters in slot 5 (BC=62) and slot 7 (BC=71) disagree (at least)</PRE>
<!-- Author: David :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159603' target=_top>159603</a>) -->




<!-- Date: Fri Feb 28 19:16:28 2003 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2003&action=view&page=-1659&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-1659  width='300' align='top'></td><td>Losses while TevMon complained twice for spikes</a></td></tr></table>
<!-- Author: Alberto -->
<!-- Comment: Fri Feb 28 19;32;26 comment by...Jason Nielsen -->
This was the new version of TevMon (with faster updating) which
gave errors.  The old version (running concurrently) was silent.

<!-- Date: Fri Feb 28 19:25:38 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159603">159603</A>
 Terminated at 2003.02.28 19:25:28
<!-- Author: RunControl -->


<!-- Date: Fri Feb 28 19:26:50 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159603">159603</A>
TERMINATE: End run.
<!-- Author: David -->



<!-- Date: Fri Feb 28 19:31:21 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159605">159605</A>
ACTIVATE: Silicon D-Mode Calibration - DPS ON.
<!-- Author: David -->




<!-- Date: Fri Feb 28 19:34:07 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1660&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1660' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1661&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1661' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1662&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1662' width='300'></a></center>
<!-- Author: Alberto -->


<!-- Date: Fri Feb 28 19:34:59 2003 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2003&action=view&page=-1663&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-1663  width='300' align='top'></td><td>L2 TriggerMonitor CLIST energy word validation: <BR> entries to the right of spike at 1 in both bottom plots <BR> disagree with comments on reference plot. </a></td></tr></table>
<!-- Author: Jason & Andreas -->
<!-- Comment: Fri Feb 28 19;40;08 comment by...Jason -->
<pre>This plot was taken from Run 159603 (physics run).</pre>
<!-- Comment: Fri Feb 28 19;44;25 comment by...Andreas -->
Called Heather Ray. She say's this is OK as long as top plot shows no errors above 0.01.
<!-- Comment: Fri Feb 28 20;23;52 comment by...Jason -->
This effect went away in run 159607.
<!-- Date: Fri Feb 28 19:37:07 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159605">159605</A>
TERMINATE: End Sil D-Mode.
<!-- Author: David -->


<!-- Date: Fri Feb 28 19:39:40 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159606">159606</A>
ACTIVATE: Silicon D-Mode calibration - DPS OFF.
<!-- Author: David -->




<!-- Date: Fri Feb 28 19:46:05 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159606">159606</A>
TERMINATE: End Silicon D-Mode Calibration.
<!-- Author: David -->


<!-- Date: Fri Feb 28 19:54:34 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159607">159607</A>
 Activated at 2003.02.28 19:53:54
<!-- Author: RunControl -->


<!-- Date: Fri Feb 28 19:54:35 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159607">159607</A>
ACTIVATE: AAA_NOSILICON run.
<!-- Author: David -->





<!-- Date: Fri Feb 28 20:03:00 2003 -->
<pre>It looks like someone reseted the SVXRAD? variables. 
They now are zero.</pre>
<!-- Author: Alberto -->
<!-- Comment: Sat Mar  1 08;17;37 comment by...Steve Hahn -->
The SVRAD integrated doses are reset to zero twice by the MCR: once before shot setup, and once when scraping is done.  This is because there are different integrated doses alarm levels for these two periods.
<!-- Date: Fri Feb 28 20:18:49 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159607">159607</A>
 Terminated at 2003.02.28 20:19:05
<!-- Author: RunControl -->


<!-- Date: Fri Feb 28 20:19:21 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159607">159607</A>
TERMINATE: End No silicon run.
<!-- Author: David -->



<!-- Date: Fri Feb 28 20:21:08 2003 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2003&action=view&page=-1664&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-1664  width='300' align='top'></td><td>Looks like 8 dead channels in IMU TSU East plot.</a></td></tr></table>
<!-- Author: Andreas & Jason -->
<!-- Comment: Fri Feb 28 20;21;53 comment by...Jason -->
The run number for this plot is 159607.




<!-- Comment: Mon Mar  3 10;29;17 comment by...Camille Ginsburg -->
<pre>It looks like north-east CCU-A, top board, failed.
Strange pattern.</pre>
<!-- Date: Fri Feb 28 20:53:53 2003 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2003&action=view&page=-1665&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-1665  width='300' align='top'></td><td>SVX_SET_1 -5V analog status</a></td></tr></table>
<!-- Author: Alberto -->




<!-- Date: Fri Feb 28 21:02:59 2003 -->
<table border=0><td><tr><td bgcolor=#e0e030>The new (test version) TEVMon process continues to give Silicon Danger alarms whenever there is a spike above 25Khz even when the silicon is out of the run and on standby.</td></tr></td></table>
<!-- Author: David -->
<!-- Comment: Fri Feb 28 22;36;51 comment by...Gino -->
<table border=0><td><tr><td bgcolor=#e0e030><pre>TevMon is a monitor of the beam conditions. It does NOT know 
if the silicon in integrated or not in data taking.</pre></td></tr></td></table>
<!-- Date: Fri Feb 28 21:05:00 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1667&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1667' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1668&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1668' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1669&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1669' width='300'></a></center>
<!-- Author: Alberto -->

<!-- Date: Fri Feb 28 21:05:47 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159608">159608</A>
 Activated at 2003.02.28 21:05:21
<!-- Author: RunControl -->


<!-- Date: Fri Feb 28 21:05:52 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159608">159608</A>
ACTIVATE: XFT Efficiency test run, 2 miss on all 4 axial super layers.
<!-- Author: David -->


<!-- Date: Fri Feb 28 21:05:52 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159608">159608</A>
ACTIVATE: XFT Efficiency test run, 2 miss on all 4 axial super layers.
<!-- Author: David -->




<!-- Date: Fri Feb 28 22:01:53 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159608">159608</A>
 Terminated at 2003.02.28 22:01:17
<!-- Author: RunControl -->


<!-- Date: Fri Feb 28 22:02:16 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159608">159608</A>
TERMINATE: End XFT studies run.
<!-- Author: David -->



<!-- Date: Fri Feb 28 22:08:00 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1670&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1670' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1671&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1671' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1672&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1672' width='300'></a></center>
<!-- Author: Alberto -->


<!-- Date: Fri Feb 28 22:17:56 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1673&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1673' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1674&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1674' width='300'></a></center>
<!-- Author: Andreas & Jason -->
<!-- Comment: Fri Feb 28 22;22;37 comment by...Andreas & Jason -->
CLAD & CLTD Occupancy plots from LumMon for run 159608. <BR>
Possible bad channels in CLAD West Layer 3; CLTD West <BR>
	Layers 1 (channel 5), 2 (channel 20), & 3 (channels 35 & 37).

<!-- Date: Fri Feb 28 22:27:23 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159609">159609</A>
 Activated at 2003.02.28 22:26:59
<!-- Author: RunControl -->


<!-- Date: Fri Feb 28 22:29:20 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159609">159609</A>
ACTIVATE: XFT Test run, 1-miss design in COT SL2,4 and 2-miss design in COT SL6,8
<!-- Author: David -->





<!-- Date: Fri Feb 28 22:49:35 2003 -->
Heartbeat alarm from CSP, BSU, TSU, CSW, CSX, MSK. 
It clear by itself after a couple of minutes.
<!-- Author: Alberto -->


<!-- Date: Fri Feb 28 22:53:12 2003 -->
HEVB detected corrupted data in several consecutive events (as seen earlier in this shift) from COT07: 
<pre>SCPU-P0-E-TracerEventId: Event 78812, crate 56, channel 7 has either bad Tracer ID or bad markers around Tracer word. 
 Tracer word event ID is 13, should be 12. 
Tracer word = 0x542d0038 (I think). 
Channel data = 0x00000319 0x542d0038 0x434f5444 0x00140008 0x00000000 0x00000174 0x00000003 0x00000010 ... 
... 0x00000ee4 0x00000000 0x0001b3ac 0x00000000 0xaaaaaaaa 0x55555555 0x542d0038 0x80000000 -->   
. 
. 
</pre>and several more.
<!-- Author: David :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159609' target=_top>159609</a>) -->


<!-- Date: Fri Feb 28 23:09:58 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1675&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1675' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1676&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1676' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-1677&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-1677' width='300'></a></center>
<!-- Author: Alberto -->


<!-- Date: Fri Feb 28 23:27:53 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159609">159609</A>
 Terminated at 2003.02.28 23:27:22
<!-- Author: RunControl -->


<!-- Date: Fri Feb 28 23:27:54 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159609">159609</A>
TERMINATE: End XFT efficiency test.
<!-- Author: David -->



<!-- Date: Fri Feb 28 23:40:46 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159609">159609</A>
RUNSTATUS: L1T_STATUS 
SVX_STATUS 
ISL_STATUS 
L00_STATUS 
SVT_STATUS 
<!-- Author: cdfscico -->


<!-- Date: Fri Feb 28 23:40:47 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159608">159608</A>
RUNSTATUS: L1T_STATUS 
SVX_STATUS 
ISL_STATUS 
L00_STATUS 
SVT_STATUS 
<!-- Author: cdfscico -->


<!-- Date: Fri Feb 28 23:42:05 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159607">159607</A>
RUNSTATUS: SVX_STATUS 
ISL_STATUS 
L00_STATUS 
SVT_STATUS 
<!-- Author: cdfscico -->


<!-- Date: Fri Feb 28 23:55:01 2003 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159598>
159598</A> 
x26F6E</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>MINIMUM_BIAS_1 [9,251,318]</TD>                                     
<TD ALIGN=CENTER>18:29:05</TD>                                          
<TD ALIGN=CENTER>18:38:10</TD>                                          
<TD ALIGN=CENTER>00:09:01</TD>                                          
<TD ALIGN=RIGHT>        334,230</TD>                                    
<TD ALIGN=RIGHT>        33,383</TD>                                     
<TD ALIGN=RIGHT>      33,383</TD>                                       
<TD ALIGN=RIGHT>    6.967</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>0</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159603>
159603</A> 
x26F73</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_1_04 [4,255,357]</TD>                                       
<TD ALIGN=CENTER>18:54:04</TD>                                          
<TD ALIGN=CENTER>19:25:28</TD>                                          
<TD ALIGN=CENTER>00:30:28</TD>                                          
<TD ALIGN=RIGHT>     14,462,212</TD>                                    
<TD ALIGN=RIGHT>       189,508</TD>                                     
<TD ALIGN=RIGHT>      62,810</TD>                                       
<TD ALIGN=RIGHT>   21.153</TD>                                          
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159607>
159607</A> 
x26F77</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_1_04 [4,255,357]</TD>                                       
<TD ALIGN=CENTER>19:53:54</TD>                                          
<TD ALIGN=CENTER>20:19:05</TD>                                          
<TD ALIGN=CENTER>00:25:07</TD>                                          
<TD ALIGN=RIGHT>     11,040,054</TD>                                    
<TD ALIGN=RIGHT>       111,532</TD>                                     
<TD ALIGN=RIGHT>      27,215</TD>                                       
<TD ALIGN=RIGHT>   16.100</TD>                                          
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159608>
159608</A> 
x26F78</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_1_04 [4,255,357]</TD>                                       
<TD ALIGN=CENTER>21:05:21</TD>                                          
<TD ALIGN=CENTER>22:01:17</TD>                                          
<TD ALIGN=CENTER>00:55:52</TD>                                          
<TD ALIGN=RIGHT>     30,989,627</TD>                                    
<TD ALIGN=RIGHT>       260,073</TD>                                     
<TD ALIGN=RIGHT>      59,023</TD>                                       
<TD ALIGN=RIGHT>   32.618</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159609>
159609</A> 
x26F79</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_1_04 [4,255,357]</TD>                                       
<TD ALIGN=CENTER>22:26:59</TD>                                          
<TD ALIGN=CENTER>23:27:22</TD>                                          
<TD ALIGN=CENTER>01:00:06</TD>                                          
<TD ALIGN=RIGHT>     26,393,840</TD>                                    
<TD ALIGN=RIGHT>       244,512</TD>                                     
<TD ALIGN=RIGHT>      59,754</TD>                                       
<TD ALIGN=RIGHT>   32.365</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>23:55:01</TD>                                                       
<TD ALIGN=CENTER>03:00:36</TD>                                          
<TD ALIGN=RIGHT>      83,219,963</TD>                                   
<TD ALIGN=RIGHT>        839,008</TD>                                    
<TD ALIGN=RIGHT>       242,185</TD>                                     
<TD ALIGN=RIGHT>  109.205</TD>                                          
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Fri Feb 28 23:57:11 2003 -->
Called MCR--telling them that we have finished our studies--MCR says that there is no plans to do any TEV studies--they will 
keep the store until the stack (currently 22) get reasonable 
(100 ma ?? which would be at least 9AM) --so, we keep on taking 
AAA_NOSILICON --current lum is 8.7 E30
<!-- Author: John Yoh -->


<!-- Date: Sat Mar 1 00:08:43 2003 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2003&action=view&page=-1678&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-1678  width='300' align='top'></td><td>Shift luminosity report</a></td></tr></table>
<!-- Author: Alberto -->


<!-- Date: Sat Mar 1 00:11:46 2003 -->
<font color=red>Shift Summary: </font><pre>16:00	Start shift on Proton setup 
17:00	MCR having trouble injecting Pbars 
18:10	All Pbars in (6 so-so groups, 3 wimpy ones) 
	D0 Lum is 12.6 E30; scraping 
18:29	Scraping finished 
18:30	Store 2285  --initial CDF Lum is 13.6 E30 
18:32	Taking Min-bias data 
18:47	Separator spark result in 12% loss in Lum. 
18:54	Turn on SVX--Physics run 159603 
19:00+	Several D-mode errors due to spikes; LostP now 
	  at close to 20K and spikey. Talk with SVX expert 
19:31   Silicon D-mode calibration 
19:45	Finished Silicon D-mode calib; turn SVX to standby 
	  due to high and spikey LOSTP losses 
19:45	Called MCR to have them tune down the losses; slight 
	  improvements (steady slight lowering of losses) 
19:54	Run 159607 AAA_NOSILICON 
21:05	Run 159608 XFT efficiency test run 
22;27	Run 159609 XFT Test run again 
23:50 	Run 159610 back to AAA_NOSILICON (lum 8.7E30)</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   159598, 1569607-10
Delivered Luminosity   201.8  
Acquired Luminosity    115,8  
Efficiency             57%
   
</td></tr></table></pre>
<!-- Author: John Yoh -->

<!-- Date: Sat Mar 1 00:24:34 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=159610">159610</A>
 Activated at 2003.03.01 00:24:44
<!-- Author: RunControl -->



<!-- Date: Sat Mar 1 08:07:56 2003 -->
Saw that b0fib06 was red on VxWorks, so I decided to reboot the crate. There are suspended processes.
<!-- Author: Sal -->
<!-- Comment: Sat Mar  1 08;09;02 comment by...Sal -->
'tis green again.