<!-- Title: Day shift. Wed May 30, 2001 -->
<!-- SciCo: Tom LeCompte -->
<!-- DAQAce: Petra / Francesco/ Greg -->
<!-- MonAce: Sehban Kartal -->
<!-- CO: (none) -->
<!-- OpManager: Doug Benjamin -->
<!-- Notes: -->
<pre></pre>


<!-- Date: Wed May 30 08:23:44 2001 -->
Shift starts - we hand calorimeter to Rick. <br> 
Rich Stanek will cause some SVX alarms. <br> 
Andrew Martin is making some TSI mods, so he has all the trigger.
<!-- Author: Tom LeCompte -->


<!-- Date: Wed May 30 08:28:26 2001 -->
<pre>CMX wedges 16 and 17 east and west to standby  
because they tripped more than 3 times</pre>
<!-- Author: francesco -->
<!-- Comment: Wed May 30 08;38;20 comment by...TJL -->
<pre>Tom tried to turn them on after waiting 20 minutes, but they tripped again.  The problem looks real, so we paged Abraham.</pre>
<!-- Date: Wed May 30 08:35:33 2001 -->
<pre>the run 116258 we inherit from the previous shift  
ends with DONE timeout after 359931 events 
 
 
(MLE) b0htdc00:Messenger:8:20:57 AM->Runtime Error 49, Event 359931: No valid he 
ader for TDC in slot 8 
(MLE) b0tsi02:Messenger:8:20:55 AM->DONE timeout 
</pre>
<!-- Author: 4 aces toghether  :: (run <a href='/cgi-bin/runSummaryRun.cgi?116258' target=_top>116258</a>) -->



<!-- Date: Wed May 30 08:41:21 2001 -->
<pre>IFIX Workspace hangs again.  We restart it.</pre>
<!-- Author: TJL -->


<!-- Date: Wed May 30 08:49:40 2001 -->
Rob Roser calls and asks that the SW CMX miniskirt be turned off because he's working in that vicinity.  We turn off all of the CMX (because turning off just part of the muon system seems to induce IFIX workspace hangs.
<!-- Author: TJL -->
<!-- Comment: Wed May 30 12;07;45 comment by...TJL -->
<table border=0><td><tr><td bgcolor=#e0e030>Rob says this work will continue until 1600 or 1700.  Until then, the CMX needs to stay off.</td></tr></td></table>
<!-- Date: Wed May 30 08:54:55 2001 -->
<pre>CRYO tech called CR they are investigating a racks trip 
 
work on going with west svx cooling</pre>
<!-- Author: f -->


<!-- Date: Wed May 30 09:02:37 2001 -->
Rick releases the endwall; he still has the central and plug.
<!-- Author: TJL -->


<!-- Date: Wed May 30 09:09:44 2001 -->
<pre>I am about to turn off power on wcal06 and wcal02 in order to 
swap proms on TOF transition boards in those crates.</pre>
<!-- Author: M Jones -->


<!-- Date: Wed May 30 09:21:55 2001 -->
Rick releases everything.
<!-- Author: TJL -->


<!-- Date: Wed May 30 09:28:45 2001 -->
<pre>Power restored to wcal02 and wcal06.  Starting tests...</pre>
<!-- Author: M Jones -->


<!-- Date: Wed May 30 09:29:44 2001 -->
2 TSI's modified, and the fix seems to work.  Hooray!  (More details from the Yale group when they finish)
<!-- Author: TJL -->


<!-- Date: Wed May 30 09:53:59 2001 -->
<pre>Two proms replaced and the fix seems to work.  Ye-haw! 
b0wcal02 and b0wcal06 rebooted and released.</pre>
<!-- Author: M Jones -->


<!-- Date: Wed May 30 09:54:25 2001 -->
<pre>We have a request for a Rick Run.  (Owl apparently ran with the wrong configuration).  It looks like 11:00 or so there is a window between the WCAL and L1 work and the L2 work.</pre>
<!-- Author: TJL -->


<!-- Date: Wed May 30 10:13:21 2001 -->
We had received alarm msg from Silicon cooling SVX East side trip and L00 trip as well. We paged Rich Stanek who told us to reset the east side.  <!-- Author: Sehban&Francesco&Petra -->
<!-- Comment: Wed May 30 10;19;21 comment by...tjl -->
<pre>Petra resets it.</pre>

<!-- Comment: Wed May 30 10;26;51 comment by...4 aces = poker -->
<pre>diagnosis:

people working in the bore  
produced fake dew point trips </pre>


<!-- Date: Wed May 30 10:43:26 2001 -->
<pre>Jimmy  and shift log  reported problems with SMXR readout  
in ccal00 last night.   SMX Calibrations were performed around 
midnight showing no problem with ccal00, but a problem 
with the SMXR in slot 4 of ccal02 (West phi=7).    
 
Crawled up and checked the  SMD crate out.  Everything looked 
ok so took another calibration which showed No errors! 
 
 </pre>
<!-- Author: byrum,lindgren -->


<!-- Date: Wed May 30 10:46:05 2001 -->
<pre>We try to do the Rick run, but we found that Karen Byrum had grabbed CCAL02 without telling anyone.  We get her to release it, and do the run.</pre>
<!-- Author: TJL -->


<!-- Date: Wed May 30 11:27:59 2001 -->
<table border=0><td><tr><td bgcolor=#e0e030><pre>Run 116265 : We managed the run by configuration of rick run. 
	     But we have to ignor inhibit to run.</pre></td></tr></td></table>
<!-- Author: Frencesco & Sehban :: (run <a href='/cgi-bin/runSummaryRun.cgi?116265' target=_top>116265</a>) -->


<!-- Date: Wed May 30 11:29:38 2001 -->
<pre>run control crash 
 
2001.05.30 11:24:28 Configurable action detected: physicsTableName INTEGRATION_SUMET_2 
2001.05.30 11:24:28 Configurable action detected: l3TagSet 130 
2001.05.30 11:24:29 runedit.RunTypeEditor [12] 
2001.05.30 11:24:34 runedit.RunTypeEditor [12] 
2001.05.30 11:24:36 runedit.RunTypeEditor [12] 
2001.05.30 11:24:36 runedit.RunTypeEditor [13] 
2001.05.30 11:24:36 rc.GUI [12] 
2001.05.30 11:24:36 rc.GUI [13] 
2001.05.30 11:24:36 rc.GUI [14] 
2001.05.30 11:24:36 rc.GUI [25] 
2001.05.30 11:24:36 rc.GUI [26] 
2001.05.30 11:24:36 rc.GUI [27] 
2001.05.30 11:24:36 rc.GUI [28] 
2001.05.30 11:25:01 Enabling FSM 'Reset Sub State Machine' 
2001.05.30 11:25:01 FSM 'Reset Sub State Machine' has finished successfully 
SIGSEGV 11 (*) segmentation violation 
        stackpointer=0xbedc1af4 
Writing java dump to javacore21021.991239908.txt...  OK 
SIGABRT 6 (*) abort process 
        stackpointer=0xbedc1598 
 
</pre>
<!-- Author: francesco sehban -->


<!-- Date: Wed May 30 11:42:47 2001 -->
Rich Stanek calls and says SVX cooling work will stop for 
about 45 minutes.
<!-- Author: TJL -->


<!-- Date: Wed May 30 11:49:36 2001 -->
<table border=0><td><tr><td bgcolor=#e0e030>All TS modules at B0 have been updated to prevent more 
than one from using either the TS scaler or L2 Processor 
protocol bussed lines on the TS J3 backplane. 
 
The spare TS is in partition 1; Andrew will investigate 
the previously reported troubles for this module when 
running fixed period calibrations.  Perhaps the shift 
crew can help on this. 
 
We have also swapped in the Global Level 2 module ID 1 
(with the VMW chipset upgrade) and reinstalled the 
front panel cables. 
</td></tr></td></table>
<!-- Author: Michael Schmidt -->



<!-- Date: Wed May 30 12:30:42 2001 -->
<pre>run 116266: this is supposed to be a regular rick run 
	    we ended this run after 310296 events  
	    BUT (does someone change the trig.table ?) 
	    we got only SUM_ET_5[1] trigger with a  
	    rate of ~ 170 Hz (with all the cal crates  
	    included)</pre>
<!-- Author: francesco & sehban :: (run <a href='/cgi-bin/runSummaryRun.cgi?116266' target=_top>116266</a>) -->
<!-- Comment: Wed May 30 12;33;20 comment by...FRANCESCO -->
<pre>the run config file was : ACE_CALORIMETER_MINIMAL_2[2,126]</pre>

<!-- Comment: Wed May 30 12;37;11 comment by...f. -->
<pre>ALSO: the inhibit ignore flag had to be set in order to run</pre>


<!-- Date: Wed May 30 12:41:42 2001 -->
<pre>run 116267: same thing but with ACE_CALORIMETER_MINIMAL_2[2,130] 
	    same rate SUM_ET_5[1] ~ 170 Hz</pre>
<!-- Author: francesco :: (run <a href='/cgi-bin/runSummaryRun.cgi?116267' target=_top>116267</a>) -->


<!-- Date: Wed May 30 12:52:32 2001 -->
HVMON down for 10 minutes.
<!-- Author: TJL -->


<!-- Date: Wed May 30 13:17:37 2001 -->
<pre>Hvmon system was temporarily short down and a logical error in VB script of CPRpressureVoltage.grf was fixed.</pre>
<!-- Author: Yeongdae Shon -->


<!-- Date: Wed May 30 13:56:01 2001 -->
Short cosmic run taken for Monica in order to test L2. 
Took 10000 events with everything except IMU. 
SUMET_2[130].
<!-- Author: Petra :: (run <a href='/cgi-bin/runSummaryRun.cgi?116277' target=_top>116277</a>) -->


<!-- Date: Wed May 30 15:10:52 2001 -->
<pre>Several Done timeouts from CCAL crates, rebooting doesn't help  
and HRR doesn't help either. Usually it stopped after 3, 6, 9 ...  
events. RC expert paged.</pre>
<!-- Author: Petra -->
<!-- Comment: Wed May 30 15;48;58 comment by...Arnd -->
<pre>The DONE timeout was set to a value which is sometimes too short
for the software event builder (8000000, ~1s). Increasing by
a factor of 10 fixed the problem.

Someone must have picked an old configuration that contained the
shorter timeout.
</pre>


<!-- Date: Wed May 30 15:50:24 2001 -->
<pre>CMP Access Report: 
 
We connected a custom HV jumper to three layers of NW stack 42.  This 
enabled us to jumper out the single chamber of stack 42 which has 
the internal HV problem.  HV is up and holding on the three  
connected layers, and signals looked okay for these chambers as well. 
 
Tony is trying once again to fix the gas leaks in NW stacks 7 and 9.</pre>
<!-- Author: Trevor, Kevin and Tony -->


<!-- Date: Wed May 30 15:51:49 2001 -->
<pre>Annoyed by the weird behaviour of run control menus? Then quit 
run control, setup fer, and restart run control. Bill reverted 
to the old java version, and I updated fer. 
</pre>
<!-- Author: Arnd -->


<!-- Date: Wed May 30 15:59:47 2001 -->
<font color=red>Shift Summary: </font><pre>L1 testing by Greg is about complete. The Level 2 people are getting ready to take it over.</pre> 
<!-- Author: TLC/JP -->