<!-- Title: Owl shift. Sun Feb 11, 2007 -->
<!-- SciCo: John Yoh -->
<!-- DAQAce: Andrew Kubik -->
<!-- MonAce: &nbsp; -->
<!-- CO: Simone DiDonato (at Pisa) -->
<!-- OpManager: Sasha Pronko -->
<!-- Notes: -->
<pre>Store 5221, run 233864 ongoing 
   Lum is 64.5</pre>


<!-- Date: Sun Feb 11 00:09:54.753 2007 -->
PSM alarm.  1RR12E_3 is culprit (Rob said this guy gave some issues during evening shift as well).  Channel 2 showed 0 volts, returned to -6V in about 10 seconds.  Caused a trigger inhibit (Muon Low Voltage from iFix).
<!-- Author: Andy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=233864' target=_top>233864</a>) -->
<!-- Comment: Sun Feb 11 01;05;27 comment by...Andy -->
again
<!-- Comment: Sun Feb 11 02;04;04 comment by...Andy -->
again.  iFix workspace crashed shortly after.  Restarted ok.
<!-- Comment: Sun Feb 11 03;11;55 comment by...Andy -->
again
<!-- Comment: Sun Feb 11 03;13;31 comment by...Andy -->
again
<!-- Comment: Sun Feb 11 03;19;02 comment by...Andy -->
again
<!-- Comment: Sun Feb 11 03;38;59 comment by...Andy -->
again
<!-- Comment: Sun Feb 11 03;39;58 comment by...Andy -->
again
<!-- Comment: Sun Feb 11 03;40;18 comment by...Andy -->
we are calling expert
<!-- Comment: Sun Feb 11 04;01;26 comment by...Andy -->
Happened again.  Power supply hardware pager didn't respond, but we were not completely sure this was the correct number.  We will try again.  
<!-- Comment: Sun Feb 11 04;09;36 comment by...Andy -->
again.  Pager still hasn't responded.  We will try calling Steve Hahn.
<!-- Comment: Sun Feb 11 04;09;58 comment by...Andy -->
again
<!-- Comment: Sun Feb 11 04;27;35 comment by...Andy -->
Steve says we should call Cryo and have them mask this crate.  We are doing this now.  He also asked that I go down and check this crate to see what it actually is for.
<!-- Comment: Sun Feb 11 04;37;50 comment by...Andy -->
It appears to be a CSX Power supply.  We haven't seemed to have any problems from CSX this morning though.  Right now, the crate is masked.
<!-- Comment: Sun Feb 11 04;46;14 comment by...Andy -->
Just a note, we masked the crate ourself.  For some reason, Cryo couldn't see it.  We gave Cryo notice of this action.
<!-- Comment: Sun Feb 11 07;13;30 comment by...Ulrich -->
If I remember correctly, you cannot permanently mask a crate in the PSM system from the Ace account. This was introduced to keep control over what is masked. Therefore you have to ask the cryo tech to do it.
<!-- Date: Sun Feb 11 00:33:46.910 2007 -->
Done TO for imu00.  Auto-HRR fixed.
<!-- Author: Andy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=233864' target=_top>233864</a>) -->


<!-- Date: Sun Feb 11 00:34:08.721 2007 -->
Done TO for htdc00.  Auto-HRR fixed.
<!-- Author: Andy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=233864' target=_top>233864</a>) -->
<!-- Comment: Sun Feb 11 06;51;42 comment by...Andy -->
again
<!-- Date: Sun Feb 11 00:37:19.178 2007 -->
Done TO for cot03.  Auto-HRR fixed.
<!-- Author: Andy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=233864' target=_top>233864</a>) -->


<!-- Date: Sun Feb 11 00:46:33.206 2007 -->
<center><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-1832&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-1832' width='300'></a><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-1833&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-1833' width='300'></a><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-1834&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-1834' width='300'></a></center>
<!-- Author: Andy -->


<!-- Date: Sun Feb 11 01:06:47.097 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=233864>233864</A>
general: <tt>CheckList done</tt>
<!-- Author: Consumer Operator( Simone Donati) -->



<!-- Date: Sun Feb 11 01:06:47.067 2007 -->
There was a Si alarm during the PSM alarm (see above).  Nothing turned red, all cooling LEDs on, nothing looks wrong in iMon, everything looks ok in real-time plots.  Not sure what caused the alarm.
<!-- Author: Andy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=233864' target=_top>233864</a>) -->
<!-- Comment: Sun Feb 11 01;14;38 comment by...Andy -->
COTCCE2 under ISL FLOW LOOPS in iFix ISL page went yellow briefly (gave value of 13.0).  Went back green in few seconds (now at value of 12.6)


<!-- Comment: Sun Feb 11 06;20;30 comment by...Ulrich -->
Did it say "Silicon Alarm"? This happens once in a while and appears to be a glitch in the alarm system. For a specific alarm from a pressure or flow in ISL would say "ISL alarm".
<!-- Comment: Sun Feb 11 06;23;20 comment by...Andy -->
Yeah, it did say 'Silicon Alarm'.  Good to know, Thanks!
<!-- Date: Sun Feb 11 01:31:10.018 2007 -->
BMU NW 276 Trip.  HV recovered.
<!-- Author: Andy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=233864' target=_top>233864</a>) -->


<!-- Date: Sun Feb 11 01:43:30.070 2007 -->
Keeping a close eye on CCE1SUP (Cooling Circuit East #1 Supply Flow) as instructed in day-shift elog.  Current value is 1.87, with a flat trend (looks fine!).
<!-- Author: Andy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=233864' target=_top>233864</a>) -->
<!-- Comment: Sun Feb 11 03;31;56 comment by...Andy -->
Current value 1.85.  Not noticeable trend.
<!-- Comment: Sun Feb 11 04;57;35 comment by...Andy -->
currently 1.84 (looks fine)
<!-- Comment: Sun Feb 11 07;10;53 comment by...Andy -->
currently 1.82 (looks fine)
<!-- Date: Sun Feb 11 02:05:58.481 2007 -->
<center><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-1835&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-1835' width='300'></a><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-1836&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-1836' width='300'></a><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-1837&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-1837' width='300'></a></center>
<!-- Author: Andy -->
<!-- Comment: Sun Feb 11 02;07;11 comment by...Andy -->
B0PAGC had a small rise, corresponding with a small decline in AGIGI2.  All within limits.


<!-- Date: Sun Feb 11 03:01:15.553 2007 -->
<center><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-1838&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-1838' width='300'></a><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-1839&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-1839' width='300'></a><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-1840&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-1840' width='300'></a></center>
<!-- Author: Andy -->
<!-- Comment: Sun Feb 11 03;01;39 comment by...Andy -->
Plots look fine.





<!-- Date: Sun Feb 11 03:22:01.214 2007 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2007&action=view&page=-1841&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2007&action=view&page=-1841  width='300' align='top'></a></td><td>AVDD B3W7L2 pinkie at 03:10 --recovered by HRR </td></tr></table>
<!-- Author: John Yoh -->


<!-- Date: Sun Feb 11 03:36:05.047 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=233864>233864</A>
general: <tt>CheckList done</tt>
<!-- Author: Consumer Operator( Simone Donati) -->






<!-- Date: Sun Feb 11 03:56:03.779 2007 -->
<center><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-1842&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-1842' width='300'></a><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-1843&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-1843' width='300'></a><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-1844&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-1844' width='300'></a></center>
<!-- Author: Andy -->
<!-- Comment: Sun Feb 11 03;56;26 comment by...Andy -->
Plots look fine.






<!-- Date: Sun Feb 11 04:28:58.850 2007 -->
TOF HV Trip.  Northwest and Northeast third were blue, and LV bar was missing from HV display.  Ramped it back up and are running again.
<!-- Author: Andy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=233864' target=_top>233864</a>) -->
<!-- Comment: Sun Feb 11 05;51;40 comment by...Andrew Kubic, John Yoh -->
This "trip??" exposed a really flawed TOF Diagonistic/HV page--
---the bar graph shows dark GREEN, with one bar at zero ???
---the message says " PMT 160 tripped"
---each 1/2 have 1/3 of the circle at dark blue (set to 
zero ???)

It took us several minutes to realize that the error message is wrong, and that we must raise the HV ON to set the 1/3 of the HV (which was dark blue) back up to ON. 

How could 1/3 of the TOF HV be dark blue (did something set them
to zero ???)--and how could the bar graph still show up as
dark green ??? and yet the voice message and the run control
both said "TOF tripped" ????
<!-- Date: Sun Feb 11 04:39:45.287 2007 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2007&action=view&page=-1845&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2007&action=view&page=-1845  width='300' align='top'></a></td><td>Noticed that BIAS B4W2L4 (which cause pinkie last owl shift, and was masked afterwards) appeared to have a sharp apike this morning around 2:35 that almost reached 800.</td></tr></table>
<!-- Author: John Yoh -->




<!-- Date: Sun Feb 11 05:04:49.331 2007 -->
<center><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-1846&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-1846' width='300'></a><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-1847&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-1847' width='300'></a><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-1848&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-1848' width='300'></a></center>
<!-- Author: Andy -->
<!-- Comment: Sun Feb 11 05;05;07 comment by...Andy -->
Plots look fine.

<!-- Date: Sun Feb 11 05:47:04.065 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=233864>233864</A>
general: <tt> CheckList done</tt>
<!-- Author: Consumer Operator( Simone Donati) -->




<!-- Date: Sun Feb 11 05:58:33.556 2007 -->
<center><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-1849&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-1849' width='300'></a><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-1850&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-1850' width='300'></a><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-1851&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-1851' width='300'></a></center>
<!-- Author: Andy -->
<!-- Comment: Sun Feb 11 05;58;56 comment by...Andy -->
Plots look fine.





<!-- Date: Sun Feb 11 07:02:21.058 2007 -->
Done TO for imu01.  Auto-HRR fixed.
<!-- Author: Andy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=233864' target=_top>233864</a>) -->


<!-- Date: Sun Feb 11 07:08:59.177 2007 -->
<center><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-1852&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-1852' width='300'></a><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-1853&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-1853' width='300'></a><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-1854&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-1854' width='300'></a></center>
<!-- Author: Andy -->
<!-- Comment: Sun Feb 11 07;09;18 comment by...Andy -->
Plots look fine.




<!-- Date: Sun Feb 11 07:17:53.195 2007 -->
Done TO for pcal11.  Got stuck on HRR, shepherd successful.
<!-- Author: Andy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=233864' target=_top>233864</a>) -->

<!-- Date: Sun Feb 11 07:35:38.665 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=233864>233864</A>
general: <tt> CheckList done</tt>
<!-- Author: Consumer Operator( Simone Donati) -->


<!-- Date: Sun Feb 11 07:55:17.615 2007 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>PhysicsTable</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L3Accepts</TH>
<TH>DelivLumi nb<SUP>-1</SUP></TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN=233864 target=_runSummary>233864</A> x39188</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_4_01 [2,852,672]</FONT></TD>
<TD ALIGN=CENTER>17:48:47</TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>12:28:16</TD>
<TD ALIGN=RIGHT>   4,733,206</TD>   <TD ALIGN=RIGHT>  3,541.11</TD>
<TD ALIGN=RIGHT>  3,119.40</TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>07:55:01</TD>
<TD ALIGN=CENTER>12:28:16</TD>
<TD ALIGN=RIGHT>     4,733,206</TD> <TD ALIGN=RIGHT>  3,541.11</TD>
<TD ALIGN=RIGHT>  3,119.40</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Sun Feb 11 08:02:49.091 2007 -->
<font color=red>Shift Summary: </font><pre>00:00  Store 5221, run 233864 ongoing, lum 64.5 
 
01:05 PSM alarm, 1RR12E_3, again at 02:04--and 7 times 
      between 03:11 and 04:09; we paged expert at 218-8922 
      twice (no response)--called Steve Hahn--he suggested 
      masking 1RR12E_3 (appears to be a CSX PS), which we did 
 
01:06 Si alarm (brief excursion of CoTCCE2 under ISR FLOW LOOPs 
      briefly went yellow(13.0) but went green a few sec later 
 
01:31 BMU NW 276 tripped, recovered 
 
03:10 AVDD B3W7L2 pinkie--recovered by HRR 
 
04:28 TOF HV tripped announced on voice, and run control-- 
      Diagnostic page was confusing--and status appeared 
      to be screwed up (1/3 of PMT on both E and W were 
      dark blue --was it set to zero by something ??) and 
      yet the bar graph was dark green with left on at zero. 
      Clicking on the HV ON recovered the situation (though 
      the diagnotice error message was PMT 160 tripped ???) 
 
08:00 End shift with same store (5221) and run (233864) 
</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   233864
Delivered Luminosity   1405.3  
Acquired Luminosity    1259.8  
Efficiency             89.6
   
</td></tr></table></pre>
<!-- Author: John Yoh -->