<!-- Title: Owl shift. Fri Dec 21, 2001 -->
<!-- SciCo: Brenna Flaugher / Taku -->
<!-- DAQAce: Soushi Tsuno  -->
<!-- MonAce: Denys -->
<!-- CO: G. Latino -->
<!-- OpManager: Dave Ambrose -->
<!-- Notes: -->
<pre>Shot going in, protons loaded, solenoid up</pre>

<!-- Date: Fri Dec 21 00:14:37 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135434">135434</A>
 Activated at 2001.12.21 00:10:54
<!-- Author: RunControl -->



<!-- Date: Fri Dec 21 00:27:44 2001 -->
 
 
L00 just tripped. There were high losses in the detector (shot setup)<BR> 
The error message in the GUI window says <BR> 
ALARM: L00 B0 W4 L2 is reported OFF but low/high voltage is a bove 2/5 V<BR> 
Paged silicon, Andy called back and told me to check the full status of that wedge in L00 B0, and for channel two VBIAS voltage was 
shown as 512... Andy figured it's power supply problem, said he would contact more ppl and call back.   
<!-- Author: denys -->


<!-- Date: Fri Dec 21 00:28:43 2001 -->
<pre>During shot setup, we are taking the TT : TEST_PHYSICS_0_03[1,235] with SVT crates without collision. 
</pre>
<!-- Author: Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135434' target=_top>135434</a>) -->


<!-- Date: Fri Dec 21 00:29:04 2001 -->
Now, for the first time, SVT crate download is driven by 
contents of the hardware database.  I didn't intend to put 
in this change so close to the start of a store, but the 
shot came sooner than we had expected.  The new download 
seems to work, though it took a second try to handle 
correctly a special case for crate b0svt07. 
<p> 
Shortly after the first run in which the upgraded SVT  
crates were included, we got a done timeout from b0svt01. 
This has been happening now and then, and we're trying to 
see if it goes away when we turn off the on-crate monitoring 
software. 
<p> 
<b>If an SVT crate becomes unresponsive during a run, you 
may not need to end the run.  Try this first:</b> 
<p> 
<ul> 
<li>Reboot the crate (if vxboot can't log in, use setup reset_crate; reset_crate b0svtNN) 
<li>When the crate returns, click on that crate in the reply/ack window 
<li>Issue (to that crate only) RESET, then PARTITION, then CONFIGURE, then ACTIVATE 
<li>Do H/R/R 
<li>Cross your fingers 
</ul> 
<p>  
<!-- Author: Ashmanskas -->


<!-- Date: Fri Dec 21 00:33:10 2001 -->
lost our ACNET fast time plots, just as happened on Tuesday night's shot.  Called MCR, and they know about it.  Expert working on it.
<!-- Author: ambrose -->

<!-- Date: Fri Dec 21 00:38:50 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135434">135434</A>
 Terminated at 2001.12.21 00:35:42
<!-- Author: RunControl -->



<!-- Date: Fri Dec 21 00:44:25 2001 -->
Ended run to turn on all detector.
<!-- Author: Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135434' target=_top>135434</a>) -->


<!-- Date: Fri Dec 21 00:44:56 2001 -->
Talked to Andy again, he says they probably lost a power supply, i.e. they will need to take it out and do some surgery on it.<br> 
By Andy instruction I also acknowledged the alarm and reset a trip. However L00 is still red in iFix. 
<br> 
Andy said if same wedge in L00 shows same alarm again, acknowledge it but do not reset the trip.<br> 
When I asked about a proper way to deal with pink channel in IMON, he said to put elog entries only about those that do 
not go away after unmarking.
<!-- Author: denys -->


<!-- Date: Fri Dec 21 00:50:44 2001 -->
Begin Store 855  MCR never called us 
Instantaneous luminosity = 7.2 e30 
Acnet is hung, can not see loss plots 
Tev. Currents: tevpr=4145, tevpb=210  
<!-- Author: Brenna -->


<!-- Date: Fri Dec 21 00:52:15 2001 -->
During shot setup had a PHA HV alarm which eventually cleared. 
Then plug showermax had an HV alarm which cleared after about 10 min. 
L00 tripped and Ace consulted experts
<!-- Author: brenna -->


<!-- Date: Fri Dec 21 00:53:42 2001 -->
We do not have the plots of losses in ACNET again! 
This problem has happened  before and it seems like there 
is nothing we can do (problem with TeV clock and BSC clock)
<!-- Author: denys -->


<!-- Date: Fri Dec 21 00:53:57 2001 -->
All HV up except CMP - leave this off until losses are understood. 
SVX HV also up.
<!-- Author: Brenna :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135435' target=_top>135435</a>) -->


<!-- Date: Fri Dec 21 00:54:45 2001 -->
Silicon expert (Andy) guided me through turning the system on. Silicon is biased and ready to take data.
<!-- Author: denys -->


<!-- Date: Fri Dec 21 01:16:02 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-3038&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-3038  width='300' align='top'></td><td>SVRADs during the shot setup</a></td></tr></table>
<!-- Author: denys -->


<!-- Date: Fri Dec 21 01:17:25 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-3039&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-3039  width='300' align='top'></td><td>Losses during shot setup</a></td></tr></table>
<!-- Author: denys -->


<!-- Date: Fri Dec 21 01:21:39 2001 -->
<pre>We got these errors from b0fib00, b0fib04 and b0fib06 due to failing to turn SB4W5L2, SB3W6L4, SB1W2L1, SB5W0L2, and SB5W2L2 on. 
 
 
Partition:0 Run number: 135437 (0x2110d) 
(MLE) b0fib06:Messenger:1:08:38 AM-&gt;Error Initializing HDI Slot 11 Chan 7: SVX B4W5 
(MLE) b0fib06:Messenger:1:08:38 AM-&gt;Statechange failed 
(MLE) b0fib04:Messenger:1:08:39 AM-&gt;Error Initializing HDI Slot 10 Chan 7: SVX B3Wb 
(MLE) b0fcal00:Messenger:1:08:39 AM-&gt;ADMEM in slot 6 not properly configured 
(MLE) b0fcal01:Messenger:1:08:34 AM-&gt;ADMEM in slot 16 not properly configured 
(MLE) b0fib00:Messenger:1:08:42 AM-&gt;Error Initializing HDI Slot 13 Chan 1: SVX B1W2 
(MLE) b0fib04:Messenger:1:08:39 AM-&gt;Error Initializing HDI Slot 17 Chan 2: SVX B5W0 
(MLE) b0fcal00:Messenger:1:08:39 AM-&gt;ADMEM in slot 16 not properly configured 
(MLE) b0fcal01:Messenger:1:08:34 AM-&gt;ADMEM in slot 14 not properly configured 
(MLE) b0fib00:Messenger:1:08:42 AM-&gt;Statechange failed 
(MLE) b0fib04:Messenger:1:08:39 AM-&gt;Error Initializing HDI Slot 18 Chan 2: SVX B5W2 
(MLE) b0fcal00:Messenger:1:08:39 AM-&gt;ADMEM in slot 14 not properly configured 
(MLE) b0fcal01:Messenger:1:08:34 AM-&gt;ADMEM in slot 12 not properly configured 
(MLE) b0fib04:Messenger:1:08:39 AM-&gt;Statechange failed 
(MLE) b0fcal00:Messenger:1:08:39 AM-&gt;ADMEM in slot 12 not properly configured 
(MLE) b0fcal01:Messenger:1:08:34 AM-&gt;ADMEM in slot 10 not properly configured 
(MLE) b0fcal00:Messenger:1:08:39 AM-&gt;ADMEM in slot 10 not properly configured 
(MLE) b0fcal01:Messenger:1:08:34 AM-&gt;ADMEM in slot 8 not properly configured 
(MLE) b0fcal00:Messenger:1:08:39 AM-&gt;ADMEM in slot 8 not properly configured 
(MLE) b0fcal01:Messenger:1:08:35 AM-&gt;ADMEM in slot 6 not properly configured 
 
</pre>
<!-- Author: Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135437' target=_top>135437</a>) -->


<!-- Date: Fri Dec 21 01:22:33 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-3041&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-3041  width='300' align='top'></td><td>close up of losses during the time when L00 tripped (around 00:07)</a></td></tr></table>
<!-- Author: denys -->

<!-- Date: Fri Dec 21 01:38:33 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135439">135439</A>
 Activated at 2001.12.21 01:34:45
<!-- Author: RunControl -->



<!-- Date: Fri Dec 21 01:40:34 2001 -->
<pre>Now, I are running with silicon!! 
Run # 135439</pre>
<!-- Author: Soushi  :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135439' target=_top>135439</a>) -->


<!-- Date: Fri Dec 21 01:43:57 2001 -->
getting PSH alarms every now and then, different crates all the time, and the alarms clear themselves after a few minutes
<!-- Author: denys -->


<!-- Date: Fri Dec 21 01:44:25 2001 -->
100% reformatter errors now
<!-- Author: brenna :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135439' target=_top>135439</a>) -->
<!-- Comment: Fri Dec 21 01;59;31 comment by...denys -->
This is because SVX tripped (B1 W2 L1)<br>
Andy is trying to figure out what's up with all the trips
<!-- Comment: Fri Dec 21 06;55;14 comment by...SCN -->
That damn SOC!
<!-- Comment: Fri Dec 21 06;56;34 comment by...Hasty  -->
Whoops, take it back.  That ain't the SOC.
<!-- Date: Fri Dec 21 01:59:18 2001 -->
<pre>Reformatter error 100% from FIB05, channel 1. 
Paged Andy, He suggest to call DAQ experts. 
Sorry to trouble you. 
 
The error messages are like 
 
(MLE) b0l3gate2.fnal.gov:main:1:37:48 AM-&gt;Error on L3 node b0l3094 (partition 0) Fri Dec 21 01:37:47 2001 l3_no 
de  
  in refoInt_reformatProc (l3_refevt.c:610) 
  @L3_REFORMAT_ERROR 
 
RAWREF Error - UTILSILDEC - with code 38 -- START 
 Line -  311 in: src/library/rr_rawevt.c 
 Buffer position: 41551 
 Error count: 1 of 1 
 Reject rate: 50.000 percent 
 
 Detector ID not known (yet?). 
 FIB ids found: 
  0xe1204dfc  0xc4c4c4c4  0xe1224dfc  0xe1234dfc  0xe1244dfc  0x00000000  0x00000000  0x00000000  0x00000000  0 
x00000000 
 
 Error location code: 0x0000b110 
 
 
 Crate Id?: 0x00000000 
 
 Error during scanning MINI structure! 
 Last Unit prob. worked on (all counts from 0): 
  SCPU: 11 
  VRB : 1 
  LINK: 1 
  MINI: 0 
 
 Buffer (41531 until posi+2=41553) 
 0x0725f324  0x07270126  0x032cf328  0xf32e062d  0x0330082f  0x05360135  
 0xf338f337  0x023a0639  0x063c043b  0xf33e073d  0x06450144  0x014b0446  
 0xf34d064c  0x0662f361  0x06640463  0x0866f365  0xf37a0167  0xf37c057b  
 0xc1c1027f  0x00000000  0xc4c4c4c4  0x00000000  0xfc4d22e1  
 
......... 
</pre>
<!-- Author: Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135439' target=_top>135439</a>) -->
<!-- Comment: Fri Dec 21 02;17;52 comment by...ADF -->
FIB05 a red herring.  This reformatter error is because
of the Bias OverCurrent trip of e121, aka SB1W2 L1.  
The trip was reset, curing reformatter errors.
<!-- Date: Fri Dec 21 02:03:12 2001 -->
heartbeat alarm for PEM, PHA and PSH
<!-- Author: denys -->
<!-- Comment: Fri Dec 21 04;17;33 comment by...denys -->
oh yeah i forgot to mention that the heartbeat was recovered eventually
<!-- Date: Fri Dec 21 02:09:29 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135439">135439</A>
 Terminated at 2001.12.21 02:05:08
<!-- Author: RunControl -->


<!-- Date: Fri Dec 21 02:13:46 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135440">135440</A>
 Activated at 2001.12.21 02:10:52
<!-- Author: RunControl -->




<!-- Date: Fri Dec 21 02:17:59 2001 -->
Brenna told me that Jonathan Lius called and said he discovered that CMX was oscillating, and maybe producing fake triggers. He asked us to turn off wedges 11,21 west side and 7,22 east side.<br> 
So I just did that, also paged CMX to notify them about it
<!-- Author: denys -->


<!-- Date: Fri Dec 21 02:24:04 2001 -->
Trip in COT superlayer 4, channel S1.<br> 
I am resetting the trip
<!-- Author: denys -->

<!-- Date: Fri Dec 21 02:25:18 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135440">135440</A>
 Terminated at 2001.12.21 02:21:58
<!-- Author: RunControl -->



<!-- Date: Fri Dec 21 02:25:19 2001 -->
<pre>New run #135440 TEST_PHYSICS_0_03[1,235] 
Tev luminosity : 6.4e30 
L1 accept : 134 
L2 accept : 128 
L3 accept :  32</pre>
<!-- Author: Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135440' target=_top>135440</a>) -->


<!-- Date: Fri Dec 21 02:26:08 2001 -->
Ended run. we'll go new trigger table.
<!-- Author: Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135440' target=_top>135440</a>) -->

<!-- Date: Fri Dec 21 02:35:22 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135441">135441</A>
 Activated at 2001.12.21 02:31:44
<!-- Author: RunControl -->



<!-- Date: Fri Dec 21 02:36:18 2001 -->
unmarking all channels in IMON...
<!-- Author: denys -->


<!-- Date: Fri Dec 21 02:37:53 2001 -->
<pre>Start new run with new trigger table. 
PHYSICS_0_02[7,234] 
 
no error!! 
 
Tev luminosity : 6.3e30 
L1 accept :  420 
L2 accept :  370 
L3 accept :   51 
Fred L1A  : 1400 
current dead time : 70 
reformatter error rate : ~3 %</pre>
<!-- Author: Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135441' target=_top>135441</a>) -->


<!-- Date: Fri Dec 21 02:47:35 2001 -->
<pre>quite a  few channels in SVX are pink (overcurrent) and unmarking 
them doesnt help. The channes are: 
for pink in BIAS: 
B0W9L2 B1W10L4  B1W11L4 B2W7L4 B3w11L2 B4w5l2  
b5w0l2 b5w2l2 b5w2l4 b5w6l4 
pink in DVDD: 
b3w3l1 
pink in AVDD: 
b1w11l2 b1w11L4 b3w0L4 b4w5L1</pre>
<!-- Author: denys -->
<!-- Comment: Fri Dec 21 02;59;20 comment by...denys -->
I tried unmark all again, and all pink in AVDD and DVDD went away, BIAS is still there.
I noticed that all the pink channels are the ones where Imax was set to very high values (2000-5000 uA) so that may be the reason for the state of the channels. <br>
Also for all these channels nominal current is exactly 0...
<!-- Date: Fri Dec 21 02:49:16 2001 -->
<pre>Done time out from b0svt03. 
Followed by Bill's instruction, restart again.</pre>
<!-- Author: Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135441' target=_top>135441</a>) -->


<!-- Date: Fri Dec 21 03:11:54 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135441">135441</A>
 Terminated at 2001.12.21 03:07:58
<!-- Author: RunControl -->


<!-- Date: Fri Dec 21 03:22:32 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135442">135442</A>
 Activated at 2001.12.21 03:19:26
<!-- Author: RunControl -->


<!-- Date: Fri Dec 21 03:25:27 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135442">135442</A>
 Terminated at 2001.12.21 03:21:39
<!-- Author: RunControl -->



<!-- Date: Fri Dec 21 03:25:31 2001 -->
<pre>Run #135441 was ended due to checking too high dead time. 
We turn back old trigger table to check that. 
PHYSICS_0_02[6,231] 
Still current dead time is too high : 70 %. 
</pre>
<!-- Author: Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135442' target=_top>135442</a>) -->
<!-- Comment: Fri Dec 21 03;47;53 comment by...masa -->
<pre>L1_CMX6_PT8_PS trigger rate is still very high (800 Hz). 
This trigger is autoaccepted at L2 with no prescale.
DAQ deadtime can be caused by his. Changing either 
L1 or L2 prescale to 10 will fix the problem</pre>
<!-- Date: Fri Dec 21 03:36:22 2001 -->
Clean up EVB/L3 display.
<!-- Author: Soushi -->

<!-- Date: Fri Dec 21 03:39:46 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135443">135443</A>
 Activated at 2001.12.21 03:36:44
<!-- Author: RunControl -->



<!-- Date: Fri Dec 21 03:42:35 2001 -->
<pre>Try to take old physics run after clean up EVD/L3 display. 
PHYSICS_0_02[6,230] 
 
Still too high dead time : 70 %</pre>
<!-- Author: Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135443' target=_top>135443</a>) -->


<!-- Date: Fri Dec 21 04:07:03 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135443">135443</A>
 Terminated at 2001.12.21 04:03:50
<!-- Author: RunControl -->



<!-- Date: Fri Dec 21 04:09:20 2001 -->
BMU wedge 0 on the east side tripped. Resetting the trip
<!-- Author: denys -->
<!-- Comment: Fri Dec 21 04;23;49 comment by...denys -->
another trip in the same spot. Will wait a few minutes before resetting it. (muon HV page says the longer you wait the more stable the detector will be :) 
<!-- Date: Fri Dec 21 04:14:52 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135444">135444</A>
 Activated at 2001.12.21 04:11:45
<!-- Author: RunControl -->


<!-- Date: Fri Dec 21 04:17:33 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135444">135444</A>
 Terminated at 2001.12.21 04:14:15
<!-- Author: RunControl -->




<!-- Date: Fri Dec 21 04:27:06 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135445">135445</A>
 Activated at 2001.12.21 04:24:12
<!-- Author: RunControl -->



<!-- Date: Fri Dec 21 04:27:07 2001 -->
Retreated abck to 6,30 trigger table (it worked a few days ago) 
but deadtime still 70%.
<!-- Author: Brenna -->


<!-- Date: Fri Dec 21 04:29:27 2001 -->
To figure out the source of this dead time, we removed VRB_SVX_02 crate. But things did not improve. Bring back this crate again. 
Try new run.
<!-- Author: Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135444' target=_top>135444</a>) -->
<!-- Comment: Fri Dec 21 08;25;39 comment by...SCN -->
<h3>Don't ever do this!</h3>
<ul>
<li> Although VRB_SVX_02 is often the crate which first <I>reports</I> a problem, it is very rarely the cause of the 
problem.  It reports any trigger problem from any of the 9 Silicon crates.  Not only that, it is often the first to report a problem related to the EVB, which can happen in any of the 14 
EVB (VRB) crates.
<li> Not only this, but SVX_02 contains the one and only 
Silicon Readout Controller (SRC) so if it is not in the 
run, no Silicon crate (Fibs, Vrbs) will function correctly.
I expect you learned this the hard way.
</ul>
<!-- Date: Fri Dec 21 04:31:34 2001 -->
Start new run. PHYSICS_0_02[6,230] 
Now, we are looking TrigMon and YMon display.
<!-- Author: Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135445' target=_top>135445</a>) -->


<!-- Date: Fri Dec 21 04:41:47 2001 -->
<center><a href='/cgi/elog/elog.pl?nb=2001&action=view&page=-3042&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2001&action=view&page=-3042' width='300'></a><a href='/cgi/elog/elog.pl?nb=2001&action=view&page=-3043&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2001&action=view&page=-3043' width='300'></a></center>
<!-- Author: Giuseppe -->
<!-- Comment: Fri Dec 21 04;45;30 comment by...Giuseppe -->
TrigMon is giving these NOT "standard" plots.
Expert (Heather) paged: it's probably due to a mismatch between
different versions for the L2 executable and for TrigMon. 
Matt Worcest mailed.
<!-- Comment: Fri Dec 21 08;02;44 comment by...Matt -->
Good work catching this.  The Cluster Validation error is due to a change in TL2D format, which TrigMon now handles properly (see below).  The Magic Bus plot is actually correct, but we noticed that the reference histogram is wrong!  We have updated the plot in the CO reference binder.
<!-- Date: Fri Dec 21 04:43:49 2001 -->
turned CMX to standby per expert's request.<br> 
Also another one of those weird PEM alarms
<!-- Author: denys -->



<!-- Date: Fri Dec 21 05:02:38 2001 -->
<center><a href='/cgi/elog/elog.pl?nb=2001&action=view&page=-3045&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2001&action=view&page=-3045' width='300'></a><a href='/cgi/elog/elog.pl?nb=2001&action=view&page=-3046&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2001&action=view&page=-3046' width='300'></a></center>
<!-- Author: Brenna -->


<!-- Date: Fri Dec 21 05:03:40 2001 -->
THis looks like the source of our trigger problems: 
CMX_west occupancy is ~ 20/bin, CMX East occupancy is ~5000-6000 in many bins.
<!-- Author: Brenna -->


<!-- Date: Fri Dec 21 05:05:30 2001 -->
Eric James suggests we put CMX to standby.  We do, but the rates are unchanged! 
Eric suggests we power cycle the crate since the so many channels on east side are high he doesn't think it is a trigger card problem.
<!-- Author: Brenna -->

<!-- Date: Fri Dec 21 05:07:10 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135445">135445</A>
 Terminated at 2001.12.21 05:03:38
<!-- Author: RunControl -->


<!-- Date: Fri Dec 21 05:17:37 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135446">135446</A>
 Activated at 2001.12.21 05:14:41
<!-- Author: RunControl -->


<!-- Date: Fri Dec 21 05:20:12 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135446">135446</A>
 Terminated at 2001.12.21 05:17:20
<!-- Author: RunControl -->


<!-- Date: Fri Dec 21 05:23:21 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135447">135447</A>
 Activated at 2001.12.21 05:20:14
<!-- Author: RunControl -->


<!-- Date: Fri Dec 21 05:33:55 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135447">135447</A>
 Terminated at 2001.12.21 05:30:06
<!-- Author: RunControl -->



<!-- Date: Fri Dec 21 05:40:24 2001 -->
Remove B0CMX01 crate, then restart again.
<!-- Author: Soushi -->
<!-- Comment: Fri Dec 21 05;42;05 comment by...Soushi -->
We could not "coldsatrt" due to something failure from b0cmx01 crate.


<!-- Date: Fri Dec 21 05:42:19 2001 -->
After CMX crate was power cycled  run control could not initialize it.  Ace tried 3 times but no luck.  Paged Eric again. 
No error message is appearing in error window so he is coming in. 
(~1hour)  In the mean time we will try running without that crate. If that doesn't work Eric suggested power cycling the crate again.
<!-- Author: Brenna -->

<!-- Date: Fri Dec 21 05:42:55 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135448">135448</A>
 Activated at 2001.12.21 05:39:33
<!-- Author: RunControl -->



<!-- Date: Fri Dec 21 05:49:56 2001 -->
<pre>Take a new run without CMX. 
 
The dead time is drasticaly decreased to ~0.3. 
The source of this high deadtime was coming from CMX. 
We continue this run for a while. 
 
L1 accept : 320 
L2 accept : 158 
L3 accept :  53 
Fred L1A  : 320 
current dead time : 0.33</pre>
<!-- Author: Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135448' target=_top>135448</a>) -->

<!-- Date: Fri Dec 21 06:24:12 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135448">135448</A>
 Terminated at 2001.12.21 06:20:49
<!-- Author: RunControl -->


<!-- Date: Fri Dec 21 06:31:13 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135449">135449</A>
 Activated at 2001.12.21 06:28:12
<!-- Author: RunControl -->



<!-- Date: Fri Dec 21 06:35:04 2001 -->
<pre>Start TEST_PHYSICS_0_03[1,235] without b0cmx01. 
 
Run # 135449 
L1 accept : 252 
L2 accept : 195 
L3 accept :  95 
Fred L1A  : 321 
current dead time : 30 % 
 
L3 monitor shows alomost green.</pre>
<!-- Author: Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135449' target=_top>135449</a>) -->

<!-- Date: Fri Dec 21 06:45:34 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135449">135449</A>
 Terminated at 2001.12.21 06:42:33
<!-- Author: RunControl -->


<!-- Date: Fri Dec 21 06:54:37 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135450">135450</A>
 Activated at 2001.12.21 06:51:34
<!-- Author: RunControl -->





<!-- Date: Fri Dec 21 07:02:37 2001 -->
Start new trigger table. PHYSICS_0_02[7,234]. 
Now, we are including CMX crate, that is, default one. 
 
Run # 135450 
Tev luminosity : 4.9e30 
L1 accept : 320 
L2 accept : 156 
L3 accept :  50 
Fred L1A  : 320 
current dead time : 0.4 %
<!-- Author: Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135450' target=_top>135450</a>) -->

<!-- Date: Fri Dec 21 07:04:00 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135450">135450</A>
 Terminated at 2001.12.21 07:00:03
<!-- Author: RunControl -->



<!-- Date: Fri Dec 21 07:04:01 2001 -->
during this run, CMX turned on.
<!-- Author: Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135450' target=_top>135450</a>) -->

<!-- Date: Fri Dec 21 07:09:32 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135451">135451</A>
 Activated at 2001.12.21 07:05:15
<!-- Author: RunControl -->



<!-- Date: Fri Dec 21 07:12:22 2001 -->
<pre>Start again. to take good run. 
CMX turned ON. 
 
Run # 135451 
L1 accept : 320 
L2 accept : 150 
L3 accept :  50 
Fred L1A  : 324 
current dead time : 0.2</pre>
<!-- Author: Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135451' target=_top>135451</a>) -->

<!-- Date: Fri Dec 21 07:18:13 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135451">135451</A>
 Terminated at 2001.12.21 07:14:54
<!-- Author: RunControl -->


<!-- Date: Fri Dec 21 07:20:36 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135452">135452</A>
 Activated at 2001.12.21 07:17:34
<!-- Author: RunControl -->



<!-- Date: Fri Dec 21 07:24:08 2001 -->
<pre>End run due to this errors many times. 
We could not take data. 
 
(RC)   7:14:27 Halt -&gt; HALTED 
(RC)   7:14:34 Recover -&gt; RECOVERED 
(RC)   7:14:41 Run -&gt; ACTIVE 
(MLE) b0l2de00:Messenger:7:14:42 AM-&gt;CListMonitor: Sync error in L2 pipeline, event 53010 
(MLE) b0l2de00:Messenger:7:14:46 AM-&gt;Alpha in slot 3 is no longer processing at spot 8 
(MLE) b0l2de00:Messenger:7:14:46 AM-&gt;CListMonitor: Sync error in L2 pipeline, event 53010 
(MLE) b0dap59.fnal.gov:AWT-EventQueue-0:7:14:51 AM-&gt;Errors reported by the Trigger Supervisor: 
  Level 2 Decision Timeout 
 
"If you think it's a problem with the Trigger Supervisor, think again!" 
 
 
</pre>
<!-- Author: Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135451' target=_top>135451</a>) -->
<!-- Comment: Fri Dec 21 08;02;15 comment by...hray -->
We are also seeing errors with ALpha not processing in spot 61 as well
<!-- Date: Fri Dec 21 07:24:40 2001 -->
Take new run # 135452.
<!-- Author: Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135452' target=_top>135452</a>) -->


<!-- Date: Fri Dec 21 07:41:11 2001 -->
<center><a href='/cgi/elog/elog.pl?nb=2001&action=view&page=-3048&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2001&action=view&page=-3048' width='300'></a><a href='/cgi/elog/elog.pl?nb=2001&action=view&page=-3047&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2001&action=view&page=-3047' width='300'></a></center>
<!-- Author: denys -->
<!-- Comment: Fri Dec 21 07;43;20 comment by...denys -->
luminosities and losses for this shift. No comment on our efficiency...
<!-- Date: Fri Dec 21 07:43:19 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135452">135452</A>
 Terminated at 2001.12.21 07:40:19
<!-- Author: RunControl -->




<!-- Date: Fri Dec 21 07:45:24 2001 -->
ended run due to the same error.
<!-- Author: Soushi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135453' target=_top>135453</a>) -->

<!-- Date: Fri Dec 21 07:48:40 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135453">135453</A>
 Activated at 2001.12.21 07:45:02
<!-- Author: RunControl -->



<!-- Date: Fri Dec 21 07:52:03 2001 -->
<font color=red>Shift Summary: </font><pre>Owl Shift Summary Dec.21 
======================== 
0)shot going in, protons loaded, Solnoid ramping 
 
1) L00 and PHA alarm PSM alarms.  L00 is really tripped (paged Andy) 
PHA looks ok and error message has time stamp 8:01??? 
 
2) PHA alarm clears after ~10 minutes and then plug shower max 
goes red: time stamp 00:17 but it looks ok.  After ~10 min. it clears. 
3) Acnet dies, MCR notified, and they get it going after the shot is in. 
4) Only indication that scraping complete is svx rad mon  
are reset, no call from MCR! 
5) HV all up, except silicon - expert on phone guiding Ace. 
6) SVX ready, Run control can't get 3 SVX crates to work, call Andy again. 
7) Eventually Andy and Aces get those errors cleared and start run -&gt; 100% 
reformatter errors!  Call Andy again. 
8) Why does Live luminosity scaler run when we are dead? 
9) Johnathan calls up to tell us that CMX is oscillating and we should use 
trigmon to find the bad wedges and turn them off and page the expert. 
Trigmon plots indicate that wedges  11 and 21 on west and 7 and 22 on East are 
very high compared to the neighbors.  Ace goes downstairs to turn them off. 
11) Ace starts new run, 135440 to see if he and Andy have fixed reformater 
errors and if CMX rates are down. Looks like reformatter errors are gone, but  
L1 accept rate is still 30k hz and we are 90 % dead.  Trigmon doesn't indicate 
any high rates on summary pages? We do find a hot tower in CEM that seems to 
have been there on Dec. 3rd.  Also find that diagonal plot has an off diagonal 
entry.  
12) Decide to switch to standard trigger table to see if that clears up problems. 
But, alas deadtime is 70%, but L1 accept rate is only 1300 hz??? 
Deadtime is BUSY/DT.  L3 is about half and half light blue and dark blue. 
13)Heather respnds to page.  Off diagonal plot is due to different versions 
of L2 exe and trigmon. It is OK for now.  She has us look Logger rate = 10Mb/sec 
which is OK.  She suggests going back to older trigger table (last one that was 
actually run with beam!!!) [6, 231] Still dead.  Ace will clean up event builder 
and try [6,230]. 
14) Still 70% dead.  Lummon has no indications of what the normal  
trigger rates should be.  After much consultation 
with Level 3 we conclude that 374 hz into L3 is too high so there must be 
something wrong with the detector. 
15) Dig around lummon rates find CMX-CMU rate is 280hz.  CMX_pt8 is 600hz. 
I'm guessing these are too high!  Page L1 muon trigger.  What are the rates 
supposed to be? 
16) Eric James calls back and walks us through to some CMX plots. 
CMX_west occupancy is ~ 20/bin, CMX East occupancy is ~5000-6000 in many bins. 
We set CMX east to Standby and rates are unchanged. 
Eric suggests we power cycle the crate since the so manny channels on  
east side are high. 
17) After CMX crate was power cycled run control could not initialize it.  
Ace tried 3 times but no luck. Paged Eric again. No error message is  
appearing in error window so he is coming in. (~1hour) In the mean time  
we will try running without that crate. If that doesn't work Eric suggested  
power cycling the crate again.  
18) Dropping crate worked!  Deadtime with physics 6,230 is 1%! 
19) 30 min smooth running - end and try test physics table again. 
Rates ok but Level 3 turns green. 
20) Eric comes in and power cycles the crate again.  We start a new run  
with physics {7,234} and include the crate.   
Rates ok (CMX is at standby).  Turn up voltage and rates are 
still ok.  End run and start 1st good run with CMX of the night. 135452! 
21) Heather et al show up to take system away. 
 
</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   135434-135452
Delivered Luminosity   154.6  
Acquired Luminosity    36.73  
Efficiency             24
   
</td></tr></table></pre>
<!-- Author: Brenna -->


<!-- Date: Fri Dec 21 07:55:00 2001 -->
<center><a href='/cgi/elog/elog.pl?nb=2001&action=view&page=-3049&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2001&action=view&page=-3049' width='300'></a><a href='/cgi/elog/elog.pl?nb=2001&action=view&page=-3050&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2001&action=view&page=-3050' width='300'></a></center>
<!-- Author: Matt -->
<!-- Comment: Fri Dec 21 07;59;31 comment by...Matt -->
The key L2 plots from TrigMon.  As reported <a href=
'/cgi/elog/elog.pl?nb=2001&action=view&page=924&frame=2&anchor=04:41:47'>above</a>, these plots showed errors.  This was due to a change in the TL2D format, which TrigMon now handles properly, as seen in these plots.  The new TrigMon has been built, but not yet declared current in the control room.  That should be done today.
<!-- Date: Fri Dec 21 07:56:43 2001 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135434>
135434</A> 
x2110A</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>TEST_PHYSICS_0_03 [1,235]</TD>                                      
<TD ALIGN=CENTER>00:10:54</TD>                                          
<TD ALIGN=CENTER>00:35:42</TD>                                          
<TD ALIGN=CENTER>00:13:18</TD>                                          
<TD ALIGN=RIGHT>        141,188</TD>                                    
<TD ALIGN=RIGHT>       110,302</TD>                                     
<TD ALIGN=RIGHT>      42,720</TD>                                       
<TD ALIGN=RIGHT>    2.131</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135439>
135439</A> 
x2110F</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>TEST_PHYSICS_0_03 [1,235]</TD>                                      
<TD ALIGN=CENTER>01:34:45</TD>                                          
<TD ALIGN=CENTER>02:05:08</TD>                                          
<TD ALIGN=CENTER>00:00:19</TD>                                          
<TD ALIGN=RIGHT>         30,468</TD>                                    
<TD ALIGN=RIGHT>        28,768</TD>                                     
<TD ALIGN=RIGHT>           0</TD>                                       
<TD ALIGN=RIGHT>    0.110</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>0</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135440>
135440</A> 
x21110</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>TEST_PHYSICS_0_03 [1,235]</TD>                                      
<TD ALIGN=CENTER>02:10:52</TD>                                          
<TD ALIGN=CENTER>02:21:58</TD>                                          
<TD ALIGN=CENTER>00:00:41</TD>                                          
<TD ALIGN=RIGHT>         60,878</TD>                                    
<TD ALIGN=RIGHT>        57,249</TD>                                     
<TD ALIGN=RIGHT>      10,583</TD>                                       
<TD ALIGN=RIGHT>    0.278</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>0</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135441>
135441</A> 
x21111</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_0_02 [7,234]</TD>                                           
<TD ALIGN=CENTER>02:31:44</TD>                                          
<TD ALIGN=CENTER>03:07:58</TD>                                          
<TD ALIGN=CENTER>00:08:49</TD>                                          
<TD ALIGN=RIGHT>        732,089</TD>                                    
<TD ALIGN=RIGHT>       636,465</TD>                                     
<TD ALIGN=RIGHT>      87,979</TD>                                       
<TD ALIGN=RIGHT>    3.087</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135442>
135442</A> 
x21112</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_0_02 [6,231]</TD>                                           
<TD ALIGN=CENTER>03:19:26</TD>                                          
<TD ALIGN=CENTER>03:21:39</TD>                                          
<TD ALIGN=CENTER>00:00:39</TD>                                          
<TD ALIGN=RIGHT>         52,537</TD>                                    
<TD ALIGN=RIGHT>        45,663</TD>                                     
<TD ALIGN=RIGHT>       6,324</TD>                                       
<TD ALIGN=RIGHT>    0.172</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>0</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135443>
135443</A> 
x21113</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_0_02 [6,230]</TD>                                           
<TD ALIGN=CENTER>03:36:44</TD>                                          
<TD ALIGN=CENTER>04:03:50</TD>                                          
<TD ALIGN=CENTER>00:08:07</TD>                                          
<TD ALIGN=RIGHT>        643,545</TD>                                    
<TD ALIGN=RIGHT>       559,602</TD>                                     
<TD ALIGN=RIGHT>      78,527</TD>                                       
<TD ALIGN=RIGHT>    2.758</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135445>
135445</A> 
x21115</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_0_02 [6,230]</TD>                                           
<TD ALIGN=CENTER>04:24:12</TD>                                          
<TD ALIGN=CENTER>05:03:38</TD>                                          
<TD ALIGN=CENTER>00:09:35</TD>                                          
<TD ALIGN=RIGHT>        661,004</TD>                                    
<TD ALIGN=RIGHT>       565,419</TD>                                     
<TD ALIGN=RIGHT>      81,504</TD>                                       
<TD ALIGN=RIGHT>    3.078</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135448>
135448</A> 
x21118</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_0_02 [6,230]</TD>                                           
<TD ALIGN=CENTER>05:39:33</TD>                                          
<TD ALIGN=CENTER>06:20:49</TD>                                          
<TD ALIGN=CENTER>00:40:40</TD>                                          
<TD ALIGN=RIGHT>        784,994</TD>                                    
<TD ALIGN=RIGHT>       362,347</TD>                                     
<TD ALIGN=RIGHT>     128,836</TD>                                       
<TD ALIGN=RIGHT>   12.197</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135449>
135449</A> 
x21119</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>TEST_PHYSICS_0_03 [1,235]</TD>                                      
<TD ALIGN=CENTER>06:28:12</TD>                                          
<TD ALIGN=CENTER>06:42:33</TD>                                          
<TD ALIGN=CENTER>00:09:54</TD>                                          
<TD ALIGN=RIGHT>        194,522</TD>                                    
<TD ALIGN=RIGHT>       150,703</TD>                                     
<TD ALIGN=RIGHT>      70,425</TD>                                       
<TD ALIGN=RIGHT>    2.808</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135450>
135450</A> 
x2111A</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_0_02 [7,234]</TD>                                           
<TD ALIGN=CENTER>06:51:34</TD>                                          
<TD ALIGN=CENTER>07:00:03</TD>                                          
<TD ALIGN=CENTER>00:08:21</TD>                                          
<TD ALIGN=RIGHT>        157,928</TD>                                    
<TD ALIGN=RIGHT>        74,687</TD>                                     
<TD ALIGN=RIGHT>      25,168</TD>                                       
<TD ALIGN=RIGHT>    2.220</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135451>
135451</A> 
x2111B</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_0_02 [7,234]</TD>                                           
<TD ALIGN=CENTER>07:05:15</TD>                                          
<TD ALIGN=CENTER>07:14:54</TD>                                          
<TD ALIGN=CENTER>00:05:50</TD>                                          
<TD ALIGN=RIGHT>        110,790</TD>                                    
<TD ALIGN=RIGHT>        53,012</TD>                                     
<TD ALIGN=RIGHT>      42,901</TD>                                       
<TD ALIGN=RIGHT>    1.668</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135452>
135452</A> 
x2111C</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_0_02 [7,234]</TD>                                           
<TD ALIGN=CENTER>07:17:34</TD>                                          
<TD ALIGN=CENTER>07:40:19</TD>                                          
<TD ALIGN=CENTER>00:17:31</TD>                                          
<TD ALIGN=RIGHT>        328,376</TD>                                    
<TD ALIGN=RIGHT>       156,684</TD>                                     
<TD ALIGN=RIGHT>      94,823</TD>                                       
<TD ALIGN=RIGHT>    4.929</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135453>
135453</A> 
x2111D</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_0_02 [7,234]</TD>                                           
<TD ALIGN=CENTER>07:45:02</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>00:06:31</TD>                                          
<TD ALIGN=RIGHT>        120,224</TD>                                    
<TD ALIGN=RIGHT>        57,325</TD>                                     
<TD ALIGN=RIGHT>     115,678</TD>                                       
<TD ALIGN=RIGHT>    1.889</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>07:53:01</TD>                                                       
<TD ALIGN=CENTER>02:10:19</TD>                                          
<TD ALIGN=RIGHT>       4,018,543</TD>                                   
<TD ALIGN=RIGHT>      2,858,226</TD>                                    
<TD ALIGN=RIGHT>       785,468</TD>                                     
<TD ALIGN=RIGHT>   37.326</TD>                                          
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->





<!-- Date: Fri Dec 21 08:23:01 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135453">135453</A>
 Terminated at 2001.12.21 08:19:04
<!-- Author: RunControl -->



<!-- Date: Fri Dec 21 08:27:47 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=135454">135454</A>
 Activated at 2001.12.21 08:24:29
<!-- Author: RunControl -->