<!-- Title: 00:36:16  Thu May 17 2007 -->
<!-- Crewchief: Bruce / Jim L / Bruce -->
<!-- Op1: &nbsp; -->
<!-- Op2: &nbsp; -->
<!-- Op3: &nbsp; -->
<!-- Op4: &nbsp; -->
<!-- Op5: &nbsp; -->
<!-- Op6: &nbsp; -->
<!-- Notes: -->
Mechanical rooms are running fine no problems.Gas and cryo systems are running good.Regenerating cot cold trap #1.


<!-- Date: Thu May 17 05:25:37 2007 -->
Switched the nitrogen tank #18 vaporizor.
<!-- Author: Bruce -->


<!-- Date: Thu May 17 07:12:01 2007 -->
PSM ALARM -->1RR18G_2 --> unphysical values -->cleared in two minutes.
<!-- Author: Jim L -->


<!-- Date: Thu May 17 07:19:42 2007 -->
Fan rack satus: 1RR24G= Fan pack failure, 2RR22F= single failure(top is off), 2RR35I= single failure(bottom is off), and COSWT= single failure(low mid is off)
<!-- Author: Jim L -->


<!-- Date: Thu May 17 07:51:03 2007 -->
PSM ALARM --> East Arch #20(SMDCE_20)--> called the scico and he said that they were replacing it and DIDN'T CALL US !
<!-- Author: Jim L -->


<!-- Date: Thu May 17 08:14:59 2007 -->
FT-PCESUP alarm on low current -0.0 Ipm.
<!-- Author: KC -->
<!-- Comment: Thu May 17 14;00;07 comment by...huh? -->
? ? ? ? ?
<!-- Date: Thu May 17 08:18:06 2007 -->
SVX guys requested to reset the SVX set point to -3 C---> Done
<!-- Author: Jim L -->
<!-- Comment: Thu May 17 08;19;37 comment by...JIM l -->
SILICON ALARM--> chiller evap. temp on SVX ALARMed


<!-- Comment: Thu May 17 08;39;24 comment by...Jim L -->
It's out of alarm now.
<!-- Date: Thu May 17 08:30:22 2007 -->
Bob Wagner requested to put COT in recircutate mode---> Done
<!-- Author: Jim L -->


<!-- Date: Thu May 17 08:32:16 2007 -->
WATER ALARM ---> D.I. bottle reads -4.5 ---> this happened as the building emergency lights came ON. ????
<!-- Author: Jim L -->
<!-- Comment: Thu May 17 08;41;25 comment by...Jim L -->
Half of building power was turned off to change out a breaker for the kitchen.


<!-- Comment: Thu May 17 09;05;39 comment by...Jim L -->
Building power is restored and D.I is not in alarm.
<!-- Date: Thu May 17 08:54:38 2007 -->
Late note:  Readings and walk through completed--- still running.
<!-- Author: Jim L -->



<!-- Date: Thu May 17 09:25:33 2007 -->
SVX guys requested to reset the SVX set point back to -6 C---> Done
<!-- Author: Jim L -->


<!-- Date: Thu May 17 09:45:28 2007 -->
PSM ALARM --> 1RR21H_2 --> unphysical values--> cleared in two minutes.
<!-- Author: Jim L -->


<!-- Date: Thu May 17 09:46:21 2007 -->
Got go-a-head to start powering up the solenoid.
<!-- Author: Jim L -->
<!-- Comment: Thu May 17 11;05;45 comment by...Jim L -->
HIPOT was good.
<!-- Date: Thu May 17 09:58:22 2007 -->
ISL ALARMS --> All going cold.--> Scico said they are adjusting the set points.
<!-- Author: Jim L -->


<!-- Date: Thu May 17 10:12:09 2007 -->
Changed the COT check list item for proceedure #317 to #328
<!-- Author: Jim L -->


<!-- Date: Thu May 17 10:16:47 2007 -->
KC informed me he was going to change the ISL resin.
<!-- Author: Jim L -->
<!-- Comment: Thu May 17 11;04;09 comment by...Jim L -->
KC's all done with the resin change.
<!-- Date: Thu May 17 10:47:19 2007 -->
<table border=0><td><tr><td bgcolor=#e0e030>Solenoid is at full power.</td></tr></td></table>
<!-- Author: Jim L -->


<!-- Date: Thu May 17 10:52:18 2007 -->
PSM ALARM --> 1RR21E_2 -->channel1 went to 18 volts and chan. zero was unphysical value----> cleared two minutes later.
<!-- Author: Jim L -->


<!-- Date: Thu May 17 10:53:19 2007 -->
SVX guy asked to set SVX chiller temp to -10 C.--> done.
<!-- Author: Jim L -->




<!-- Date: Thu May 17 11:16:44 2007 -->
PSM ALARM ---> 1RR21_2--->unphysical value---> back in two
<!-- Author: Jim L -->
<!-- Comment: Thu May 17 11;17;59 comment by...Jim L -->
1RR21E_2


<!-- Date: Thu May 17 11:28:32 2007 -->
PSM ALARM--> 1RR21I_1 ---> unphysical value--> gone in 2
<!-- Author: Jim L -->


<!-- Date: Thu May 17 11:29:35 2007 -->
Dee requested to put the fans on FULL in the collition hall.--> DONE.
<!-- Author: Jim L -->


<!-- Date: Thu May 17 13:02:45 2007 -->
PSM ALARM -->1RR18G_2--> cleared in 2
<!-- Author: Jim L -->


<!-- Date: Thu May 17 13:05:01 2007 -->
PSM ALARM --> CLEARED in 2
<!-- Author: Jim L -->
<!-- Comment: Thu May 17 13;05;38 comment by...oops -->
1RR21I_1



<!-- Date: Thu May 17 14:02:42 2007 -->
PSM --> 1RR12C_4---> Communication ERROR --> back in 2
<!-- Author: Jim L -->


<!-- Date: Thu May 17 15:02:25 2007 -->
HVAC ALARM --> ODH head in col. hall southwest cieling reads 17.84 % o2 --> investigating
<!-- Author: Jim L -->
<!-- Comment: Thu May 17 15;23;19 comment by...Jim L -->
After checking the cameras in the c-hall and verifying that we were not loosing any gas from the cryo systems, Del and I put the troubled head in By-pass. Will contact George Wyatt in the morning, and keep an eye on it all night.


<!-- Date: Thu May 17 15:24:49 2007 -->
PSM ALARM --> 1RR12D_2 --> comm.errors --> back in 2
<!-- Author: Jim L -->


<!-- Date: Thu May 17 15:56:32 2007 -->
PSM ALARMS --> 1RR18D+1 and 1RR18C_1 ---> comm.ERRORS-->back in 2
<!-- Author: Jim L -->


<!-- Date: Thu May 17 15:58:56 2007 -->
PSM ALARM 1RR21E_2 --> comm error-->b n 2
<!-- Author: Jim L -->
<!-- Comment: Thu May 17 17;27;25 comment by...oops -->
Unphysical value error


<!-- Date: Thu May 17 17:51:23 2007 -->
Bruce on duty.
<!-- Author: Bruce -->


<!-- Date: Thu May 17 18:01:20 2007 -->
Gas and cryo readings taken everything is running fine,mechanical rooms are running fine no problems at this time.
<!-- Author: Bruce -->


<!-- Date: Thu May 17 22:52:21 2007 -->
Mechanical room,gas and cryo systems are running fine no problems this evening so far.
<!-- Author: Bruce -->