<!-- Title: Eve shift. Sat Mar 16, 2002 -->
<!-- SciCo: Michael Schmidt -->
<!-- DAQAce: Muge Unel -->
<!-- MonAce: Henri Bachacou -->
<!-- CO: Franco Semeria -->
<!-- OpManager: Dave Ambrose -->
<!-- Notes: -->
<pre>Run quietly with partition 0 to get some data 
and monitoring information.  Call MCR at 6pm and 
see about schedule for next shot.  If still on 
schedule for ~8pm do silicon D mode and SMXR 
as well.</pre>


<!-- Date: Sat Mar 16 16:18:53 2002 -->
cot trip with rxpt error at the same time:<br> 
(MLE) b0dap59.fnal.gov:AWT-EventQueue-0:4:14:55 PM->Errors reported by the Trigg 
er Supervisor: 
  Partition Error XPT 
 
"If you think it's a problem with the Trigger Supervisor, think again!" 
 
(RC)   16:15:15 Pause -> PAUSED 
15'24" L2 Decision Timeout.[RXPT] 
(MLE) b0dap59.fnal.gov:AWT-EventQueue-0:4:15:24 PM->Errors reported by the Trigg 
er Supervisor: 
  Level 2 Decision Timeout 
 
"If you think it's a problem with the Trigger Supervisor, think again!"<br>event 13212
<!-- Author: muge :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141046' target=_top>141046</a>) -->


<!-- Date: Sat Mar 16 16:22:56 2002 -->
some numbers for the current run:<br> 
deadtime 0.7% 
L1a 1.6 khz 
l2a 100 hz 
l3a 12 hz
<!-- Author: muge :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141046' target=_top>141046</a>) -->


<!-- Date: Sat Mar 16 16:24:26 2002 -->
D0 called to inform us that their `pots' are in 
AND they are moving them closer.
<!-- Author: Michael Schmidt :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141046' target=_top>141046</a>) -->


<!-- Date: Sat Mar 16 16:25:14 2002 -->
<pre>All COT superlayers tripped (around event 13000 L3A). Turned them back on.</pre>
<!-- Author: Henri :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141046' target=_top>141046</a>) -->
<!-- Comment: Sat Mar 16 16;28;20 comment by...Michael Schmidt -->
<pre>The COT trip was well before the movement of the
D0 pots! </pre>


<!-- Date: Sat Mar 16 16:59:42 2002 -->
event 38992 cot trip. 
interesting thing:2nd time it comes with a l2decision error and it seems  fcal00.<pre> 
(MLE) b0dap59.fnal.gov:AWT-EventQueue-0:4:55:01 PM-> *** Received error message: 
 errmon Active HaltRecoverRun 1 crate/s: b0fcal00(26),  in error.[RXPT] 
(MLE) b0dap59.fnal.gov:AWT-EventQueue-0:4:55:04 PM->Errors reported by the Trigg 
er Supervisor: 
  Partition Error XPT 
 
"If you think it's a problem with the Trigger Supervisor, think again!" 
 
(RC)   16:55:8 Halt -> HALTED 
(MLE) b0l3gate2.fnal.gov:main:4:55:13 PM->Error on L3 node b0l3c08 (partition 0) 
 Sat Mar 16 16:55:11 2002 l3_node 
  in chain_loop (l3_chain.c:615) 
  @L3_INPUT_FAILURE:  chain 1 code Success (0 good, 131 bad) 
</pre> 
ablove is all is in errlog.  
<!-- Author: muge :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141046' target=_top>141046</a>) -->
<!-- Comment: Sat Mar 16 17;02;15 comment by...mku -->
i had auot hrr turned on.

<!-- Comment: Sat Mar 16 17;06;34 comment by...mku -->
no, the two error conditions do not seem to be the same (do they?),however, it's 2nd time the trip is accompanied by an error..


<!-- Date: Sat Mar 16 17:08:04 2002 -->
<pre>COT tripped again (run 141046, event 38992). 
 
All Super Layers tripped: 
 
SL1 S9 
SL2 P12 
SL3 S0 
SL4 S6 
SL5 S2 
SL6 S8 
SL7 S7 
SL8 S9 
 
On the crate GUI, on the "2D plots", the following channels had OC trips: 
 
C1 2-3  (in the "x-y" coordinate ; axis have no titles) 
C2 1-1, 1-3 
C3 1-0, 3-1 
C4 2-6 
C5 0-5, 4-2 
 
It was about 40 minutes since last trip, and I did not see any beam losses when it happened, so I turned COT back on. 
 
SL2 did not come back up. The following channels indicated a SOC trip: SL2 S3, S7, S11 after I tried to bring all voltages up. 
 
At the second attempt, SL2 turned on and alarm went away. Resumed the run.</pre>
<!-- Author: Henri :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141046' target=_top>141046</a>) -->
<!-- Comment: Sat Mar 16 17;14;45 comment by...Henri -->
<pre>I talked to the COT expert on the phone: he said we should page him again if the COT trips twice within half-hour; else we can keep going, and turn it back on whenever it trips.</pre>

<!-- Date: Sat Mar 16 17:23:42 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141046">141046</A>
 Terminated at 2002.03.16 17:22:34
<!-- Author: RunControl -->



<!-- Date: Sat Mar 16 17:25:26 2002 -->
reformatters did it again:jumped to 15 % and no error popup for this. end the run and clean up the L3 farm to try to get back the subfarm 08 which has not been  getting events.
<!-- Author: muge :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141046' target=_top>141046</a>) -->


<!-- Date: Sat Mar 16 17:31:48 2002 -->
reformatters mainly seem to came from scpu 11 vrb 2 link 5 which is interpreted as fib_15, ch 5. we are repartitioning. did not clean up l3 farm yet.
<!-- Author: muge -->

<!-- Date: Sat Mar 16 17:34:23 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141047">141047</A>
 Activated at 2002.03.16 17:33:16
<!-- Author: RunControl -->
<!-- Comment: Sat Mar 16 17;37;13 comment by...muge -->
svt00 dome to.s will also l3 farm cleaning now
<!-- Comment: Sat Mar 16 17;39;46 comment by...mku -->
cleaned up l3 mon as well with hope to get the reform errors popup window working..
<!-- Date: Sat Mar 16 17:36:05 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141047">141047</A>
 Terminated at 2002.03.16 17:34:39
<!-- Author: RunControl -->




<!-- Date: Sat Mar 16 17:37:27 2002 -->
<pre>I just trimmed back SVT's list of masked ladders: 
 
SB3W0L2 - is a useful ladder, but used to give BC mismatches; 
          we'll see if they've gone away 
SB2W0L4 - drops readout, but not used by SVT in this wedge, 
          so no need to mask 
SB3W0L4 - DRO, unused by SVT tracking, no need to mask 
SB4W0L2 - DRO, now unintegrated, so SVT doesn't need to mask it 
SB1W1L4 - DRO, unused, no need to mask 
SB5W1L4 - DRO, unused, no need to mask 
SB4WAL3 - DRO, is a useful ladder, let's see if rate is low 
          enough to be acceptable 
 
SVT pager: 266-2706 
</pre>
<!-- Author: Bill Ashmanskas :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141047' target=_top>141047</a>) -->
<!-- Comment: Sat Mar 16 17;47;39 comment by...Bill -->
OK, since that led to frequent SVT sync errors in Wedge 0,
I remasked SB3W0L2 and SB4W0L2.  I'll see if I can determine
from the spy buffer data which of the two (or perhaps both)
was responsible.


<!-- Comment: Sat Mar 16 17;59;06 comment by...Bill -->
It was clearly SB3W0L2, because SB4W0L2 is not integrated.  The main benefit of above change (beyond reconfirming a known 
problem) is that SVT recovered SB4WA. I see good tracks coming
out of that wedge.

<!-- Comment: Sat Mar 16 20;30;32 comment by...Bill -->
The problem that caused the sync errors in Wedge 0 is clearly
seen in the offline data, and is
<a href=
'/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=103&frame=2&anchor=20:28:26'>
reported</a> 
here.  I suspect a bad FIB.

<!-- Comment: Sat Mar 16 20;44;53 comment by...Bill -->
I re-removed SB4W0L2 from the mask list, since it is masked
upstream, and there is no mechanism for it to have been causing the sync errors
earlier this evening.

<!-- Date: Sat Mar 16 17:49:38 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141048">141048</A>
 Activated at 2002.03.16 17:48:17
<!-- Author: RunControl -->
<!-- Comment: Sat Mar 16 17;51;56 comment by...mku -->
subfarm 08 has still null inputs.
<!-- Date: Sat Mar 16 17:49:47 2002 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2002&action=view&page=-1400&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2002&action=view&page=-1400  width='300' align='top'></td><td>In IMU TSU Details graph, new dead channel in West plot</a></td></tr></table>
<!-- Author: Franco Semeria -->
<!-- Comment: Sat Mar 16 17;51;29 comment by...Franco -->
<pre>Run # 141046</pre>



<!-- Date: Sat Mar 16 17:55:55 2002 -->
<table border=0><td><tr><td bgcolor=#e0e030><pre>MCR plan:  DC beam studies (lens off) from 19:00 to 19:30; 
drop the store at 20:00; go immediately then to shot setup.</pre></td></tr></td></table>
<!-- Author: Michael Schmidt -->


<!-- Date: Sat Mar 16 17:58:01 2002 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2002&action=view&page=-1401&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2002&action=view&page=-1401  width='300' align='top'></td><td>Run # 141046. In IMU BSU Details graph, new noisy channel in East Rear plot</a></td></tr></table>
<!-- Author: Franco Semeria -->
<!-- Comment: Sat Mar 16 18;38;44 comment by...michael -->
<pre>BMU pager called.  Camille says we are stuck with the
new hot or dead single channels until and access.</pre>
<!-- Date: Sat Mar 16 18:01:27 2002 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2002&action=view&page=-1402&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2002&action=view&page=-1402  width='300' align='top'></td><td>L1 DCAS-ADMEM comparison: new error in lower left plot and some ofdiagonal point in upper right plot</a></td></tr></table>
<!-- Author: Franco Semeria -->


<!-- Date: Sat Mar 16 18:05:55 2002 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2002&action=view&page=-1403&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2002&action=view&page=-1403  width='300' align='top'></td><td>lower right plot is flat</a></td></tr></table>
<!-- Author: Franco Semeria -->


<!-- Date: Sat Mar 16 18:13:28 2002 -->
paged l3 about subfarm 08 and told him at 8pm the store will be dropped so he can attack the pr then.
<!-- Author: muge -->

<!-- Date: Sat Mar 16 18:27:24 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141048">141048</A>
 Terminated at 2002.03.16 18:22:50
<!-- Author: RunControl -->
<!-- Comment: Sat Mar 16 18;39;48 comment by...michael -->
<pre>Run terminated to get end of run calibrations in
before lens is turned off at 19:00</pre>
<!-- Date: Sat Mar 16 18:31:05 2002 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2002&action=view&page=-1405&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2002&action=view&page=-1405  width='300' align='top'></td><td>Two hot channels in top left plot</a></td></tr></table>
<!-- Author: Franco Semeria -->


<!-- Date: Sat Mar 16 18:37:09 2002 -->
Took SMX calibration with beam and without PCAL (CES, CPR, CCR off). 
 
Successful. No error.
<!-- Author: Henri :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141050' target=_top>141050</a>) -->




<!-- Date: Sat Mar 16 18:55:09 2002 -->
PSM alarm: 1RR18F again (see previous shift). Channel 3 is at 4.3V instead of 3.3V. Came back to normal after a few minutes (now 3.5V)
<!-- Author: Henri -->


<!-- Date: Sat Mar 16 18:55:22 2002 -->
silicon d-mode. beam solenoid on. cratematch.pl was ok. 141049 on 141051 off. at the end of 1st run (049) we got a csl error saying that previous run may not have ended and events are not logged. but the file made it to fcdfsgi2. in the 2nd run (051) l3 display and trigger rate mon were showing 0 l3a whereas l3 display was updating ok. during end xition we gor a bto due to buss error in hevb crate. but it went into idle state w/o pr. 
<!-- Author: muge :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141049' target=_top>141049</a>) -->
<!-- Comment: Sat Mar 16 19;01;30 comment by...muge -->
<pre>
36'9" b0dau32.fnal.gov:csl_mon_send:6:36:08 PM->Logger partition 0 cannot log this event.  It is from a p
revious run section.    This probably means that the previous run never ended.  Current run is 141049.
.
  -->  
 Additional Information: 

 Attention !!!. CSL_LOGGER_BAD_RS Error !!!. 
 Events in this run will NOT be written to disk !!!. 
 We RECOMMEND to END the RUN.
 
 CSL was NOT receiving events from the correct run section.
 If this occurs at the start of a run, 
 the previous run was NOT succesfully ended by Level3.
 Try starting a new run.
 If this error repeats, contact Level3 farm people and notify
 them that EORs are NOT being sent to the CSL.

 If the error occurs during the course of a run, please send 
 a description of the problem to CSL people.
</pre>
this happened at the end of silicon run 141049.

<!-- Date: Sat Mar 16 19:04:54 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141053">141053</A>
 Activated at 2002.03.16 19:03:19
<!-- Author: RunControl -->
<!-- Comment: Sat Mar 16 19;46;30 comment by...muge -->
in this run we got l3 gold and also orange popup csl error. from errlog<pre>
6'48" 3 crate/s: b0cot16(64), b0cot18(66), b0svx02(112),  Done TO.6 crate/s: b0l1cl00(128), b0l1cl01(129)
, b0l1cl02(130), b0l1cl03(131), b0l1cl04(132), b0l1cl05(133),  in error.[RXPT]triggered readout sent even
ts mismatch[EOR]</pre>
<!-- Date: Sat Mar 16 19:06:46 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141053">141053</A>
 Terminated at 2002.03.16 19:04:53
<!-- Author: RunControl -->


<!-- Date: Sat Mar 16 19:34:35 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141055">141055</A>
 Activated at 2002.03.16 19:32:39
<!-- Author: RunControl -->
<!-- Comment: Sat Mar 16 19;35;56 comment by...muge -->
l3 and csl seem to be working all right now..


<!-- Date: Sat Mar 16 19:37:10 2002 -->
i was on the phoine with Ben, csl expert. he said he will try to  diagnose the problems and will get back to us.
<!-- Author: muge -->

<!-- Date: Sat Mar 16 19:45:52 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141055">141055</A>
 Terminated at 2002.03.16 19:44:21
<!-- Author: RunControl -->




<!-- Date: Sat Mar 16 19:54:22 2002 -->
<pre>Stopped running to bring the detector voltages 
to standby.  MCR notified. 
 
Begin calibrations as appropriate.</pre>
<!-- Author: michael -->


<!-- Date: Sat Mar 16 19:59:01 2002 -->
Muon Calibration: no error. 
<!-- Author: Henri :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141057' target=_top>141057</a>) -->


<!-- Date: Sat Mar 16 19:59:20 2002 -->
cot calib. no errors in log.
<!-- Author: muge :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141056' target=_top>141056</a>) -->
<!-- Comment: Sat Mar 16 20;34;45 comment by...muge -->
dbana says the timing shift is -0.07 as compared with the previous bad cotchanto calib run.
<!-- Comment: Sat Mar 16 23;13;20 comment by...muge -->
note runDBANA did not work on b0dap58. we used the setup procedure on calib webpage (source ~cdfsoft .. etc..)but did not put any version number since Hanri claimed it will pick up the current one. This is true for all  dbana diagnoses.
<!-- Comment: Sat Mar 16 23;34;05 comment by...muge -->
aces: please see the entry from Vaia down below.
<!-- Date: Sat Mar 16 19:59:48 2002 -->
tof tac calib. no errors.
<!-- Author: muge :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141058' target=_top>141058</a>) -->


<!-- Date: Sat Mar 16 20:10:01 2002 -->
Cal QIE: Successful. No error except: 
 
(MLE) b0ccal06:Readout_Calib:8:08:24 PM->ADMEM=16 QIE=18 Fail 14: Sigma(off) big 
      (1.6423285) (0) (1)  
<!-- Author: Henri :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141060' target=_top>141060</a>) -->
<!-- Comment: Sat Mar 16 22;23;09 comment by...Henri -->
Noticed there are many new bad channels: their mean values are all on the lower side of the limit (most of them are just at the limit). Paged Vaia.
<!-- Date: Sat Mar 16 20:10:14 2002 -->
clc qie done.hv on no beam. <br> 
(MLE) b0clc00:Readout_Calib:3:13:18 PM->ADMEM=4 QIE=0 Fail 14: Sigma(off) big      (12.868423) (0) (0)
<!-- Author: muge :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141059' target=_top>141059</a>) -->
<!-- Comment: Sat Mar 16 20;31;32 comment by...muge -->
forgot to note:crate was reset.
<!-- Date: Sat Mar 16 20:16:07 2002 -->
roman pot qie calib done. no errors. pes on. 
<!-- Author: muge :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141062' target=_top>141062</a>) -->
<!-- Comment: Sat Mar 16 21;44;59 comment by...mku -->
made it to db.
<!-- Date: Sat Mar 16 20:19:59 2002 -->
bsc calib done. no errors. 
<!-- Author: muge :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141063' target=_top>141063</a>) -->
<!-- Comment: Sat Mar 16 21;49;07 comment by...muge -->
channel 2e2 had a higher rms as compared to ref plot (~0.05 vs .14)
<!-- Date: Sat Mar 16 20:28:52 2002 -->
miniplug calib done. hv off. 
<pre> 
(MLE) b0fcal00:Messenger:8:21:48 PM->QIE calibration failed 
(MLE) b0fcal00:Readout_Calib:8:21:48 PM->ADMEM=6 QIE=0 Fail 14: Sigma(off) big      (1.482114) (0) (0) 
(MLE) b0fcal01:Messenger:8:21:46 PM->QIE calibration failed 
(MLE) b0fcal01:Readout_Calib:8:21:46 PM->ADMEM=16 QIE=9 Fail 14: Sigma(off) big      (1.1754962) (0) (2) </pre>
<!-- Author: muge :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141065' target=_top>141065</a>) -->





<!-- Date: Sat Mar 16 20:36:49 2002 -->
tof qie done. no errors.
<!-- Author: muge :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141067' target=_top>141067</a>) -->
<!-- Comment: Sat Mar 16 22;08;03 comment by...muge -->
results look similar to 140983
<!-- Date: Sat Mar 16 20:43:23 2002 -->
<pre>L3 expert paged to check out the stalled subfarm.</pre>
<!-- Author: michael -->
<!-- Comment: Sat Mar 16 21;24;50 comment by...michael -->
<pre>Tried again; may have used the wrong number.  Subfarm in question is c08.</pre>
<!-- Date: Sat Mar 16 20:43:31 2002 -->
smx qie done. no errors. 
<!-- Author: muge :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141068' target=_top>141068</a>) -->
<!-- Comment: Sat Mar 16 20;44;28 comment by...muge -->
ces cpr ccr were on stby.



<!-- Date: Sat Mar 16 20:55:46 2002 -->
2nd attempt for a smxqie with ces/cpr/ccr off. but we had the pilot bunch in! so this is bad..
<!-- Author: muge :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141069' target=_top>141069</a>) -->


<!-- Date: Sat Mar 16 20:57:24 2002 -->
team work is good! we did not have time for Rick's run only. i will try to check the dbana results..
<!-- Author: muge -->


<!-- Date: Sat Mar 16 21:00:51 2002 -->
Ben, csl expert, got back to us after diagnosing the error. He says except a few events at the end of run 141049 the run was ok. apparently it was due to a timing error at the end of run....
<!-- Author: muge -->


<!-- Date: Sat Mar 16 21:08:28 2002 -->
<pre>Shot setup underway.  Solenoid is ready to energize.</pre>
<!-- Author: michael -->


<!-- Date: Sat Mar 16 21:13:46 2002 -->
trying cosmics. l3 went pink at coldstart. checked proxy.working. 
trying coldstart again.
<!-- Author: muge :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141071' target=_top>141071</a>) -->


<!-- Date: Sat Mar 16 21:17:28 2002 -->
<pre>Looks like protons are going in and we ask for 
the solenoid to be brought up.</pre>
<!-- Author: michael -->



<!-- Date: Sat Mar 16 21:29:25 2002 -->
<pre>Final proton bunches are being loaded. 
 
BLMs: 
 
W inner : .65 
W outer : .16 
E inner : 1.7 
E outer : 1.4</pre>
<!-- Author: Henri -->
<!-- Comment: Sat Mar 16 21;31;31 comment by...Henri -->
It looks like protons were lost while I was writing; a new final proton loading is starting.
<!-- Date: Sat Mar 16 21:31:02 2002 -->
<pre>Protons 3/4 in and now appear to be lost.  These were 
apparently for tuning.  Final protons going in now.</pre>
<!-- Author: michael -->



<!-- Date: Sat Mar 16 21:43:09 2002 -->
started a phys run daq testing but asa i activate i got csl error. reset and start over.. 
<br>39'10" triggered readout sent events mismatch[EOR]b0dau32.fnal.gov:csl_mon_send:9:39:10 PM->Logger partit 
ion 0 cannot log this event. Run 141073 not accepted. Last run = 141073.
<!-- Author: muge :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141073' target=_top>141073</a>) -->


<!-- Date: Sat Mar 16 21:44:53 2002 -->
<pre>2 BSC alarms within 10 minutes on iFix. Went away by themselves immediately. </pre>
<!-- Author: Henri -->



<!-- Date: Sat Mar 16 21:47:15 2002 -->
<pre>Protons in; paging Ming (CLC) for another special scraping run.</pre>
<!-- Author: michael -->
<!-- Comment: Sat Mar 16 21;53;56 comment by...michael -->
<pre>Ming is here; a trial run is attempted.</pre>

<!-- Date: Sat Mar 16 21:59:00 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141075">141075</A>
 Activated at 2002.03.16 21:57:59
<!-- Author: RunControl -->



<!-- Date: Sat Mar 16 21:59:50 2002 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2002&action=view&page=-1406&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2002&action=view&page=-1406  width='300' align='top'></td><td>smx new bad channles</a></td></tr></table>
<!-- Author: muge -->

<!-- Date: Sat Mar 16 22:04:33 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141075">141075</A>
 Terminated at 2002.03.16 22:03:25
<!-- Author: RunControl -->



<!-- Date: Sat Mar 16 22:05:13 2002 -->
Song is here to do clc tests..
<!-- Author: muge :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141075' target=_top>141075</a>) -->


<!-- Date: Sat Mar 16 22:05:54 2002 -->
<pre>Solenoid is up.</pre>
<!-- Author: michael -->


<!-- Date: Sat Mar 16 22:11:25 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141076">141076</A>
 Activated at 2002.03.16 22:10:17
<!-- Author: RunControl -->



<!-- Date: Sat Mar 16 22:11:25 2002 -->
<pre>Anti-protons have been loaded (325E9 total).</pre>
<!-- Author: Henri -->

<!-- Date: Sat Mar 16 22:14:10 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141076">141076</A>
 Terminated at 2002.03.16 22:13:15
<!-- Author: RunControl -->
<!-- Comment: Sat Mar 16 22;23;47 comment by...Ming -->
<pre>Started this run (using trigger table zero_bias_ps1k[2,232])
when Tevatron began scrapping. Re-adjust the prescale factor
to 5003 =&gt; taking data at about 350 Hz.
Unfortunately Tevatron quenched almost right away after
the run had started. Will try again to take this run during
next store.</pre>
<!-- Date: Sat Mar 16 22:21:23 2002 -->
<table border=0><td><tr><td bgcolor=#e0e030>Quench during scraping.  We were ready otherwise. 
Ming didn't get his run (it was started).  Waiting 
for further word from MCR.  Will do Rick's run for 
consolation. 
 
</td></tr></td></table>
<!-- Author: michael -->


<!-- Date: Sat Mar 16 22:23:05 2002 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2002&action=view&page=-1407&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2002&action=view&page=-1407  width='300' align='top'></td><td>led difference plot</a></td></tr></table>
<!-- Author: muge -->




<!-- Date: Sat Mar 16 22:25:25 2002 -->
<pre>Also took the following calibrations earlier tonight (during quiet time): 
 
- Cal LED :   run 141060 no error 
- Cal XENON:  run 141064 no error 
- Plug Laser: run 141066 no error </pre>
<!-- Author: Henri -->

<!-- Date: Sat Mar 16 22:32:15 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141077">141077</A>
 Activated at 2002.03.16 22:31:19
<!-- Author: RunControl -->


<!-- Date: Sat Mar 16 22:45:22 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141077">141077</A>
 Terminated at 2002.03.16 22:44:23
<!-- Author: RunControl -->



<!-- Date: Sat Mar 16 22:47:44 2002 -->
Rick's run.  l1 rate is 80 hz. took 69k events to collect mot=re stat. for ymon (consumers started a little bit late and CO has been monitoring them) 
<!-- Author: muge :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141077' target=_top>141077</a>) -->


<!-- Date: Sat Mar 16 22:50:58 2002 -->
Vaia is here and looking at the results of cal qie calib.
<!-- Author: muge -->


<!-- Date: Sat Mar 16 22:55:19 2002 -->
<pre>New plan (from Dave): 
 
MCR will give us at least 30 minutes quiet time and a 
10 minute warning.  While magnet is up we will get the 
silicon back up and take D mode again. 
 
After this we will bring the magnet down. 
 
And we will repeat or do any other needed calibrations 
(perhaps with quiet time). 
 
MCR may need to do some studies later, but it is 
unknown yet. 
 
Stacking has recommenced.</pre>
<!-- Author: michael -->



<!-- Date: Sat Mar 16 23:14:27 2002 -->
silicon dps-on calin is paused due to a trip in a svx ladder. Hanri is on the phone with expert.
<!-- Author: muge :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141078' target=_top>141078</a>) -->


<!-- Date: Sat Mar 16 23:29:29 2002 -->
<table border=0><td><tr><td bgcolor=#e0e030><pre>OK... 
Due to shared libraries problem the ~cdfcalib/runDBANA 
version is not running. I notified Jack Cranshaw about 
that via e-mail. 
The way to run DBANA now, until that problem is fixed is 
 
source ~cdfsoft/cdf2.cshrc 
setup cdfsoft2 development 
DBANA 
 
We will try to have more than one person able to put the correct 
version of DBANA behind the ~cdfcalib/runDBANA script 
</pre></td></tr></td></table>
<!-- Author: vaia -->
<!-- Comment: Sun Mar 17 14;03;58 comment by...jcranshaw -->
<pre>The problem was that DBANA was based on 4.4.0int2 
which got removed. I've sent a mail to cdf code management 
to clear things up.

The runDBANA script is fine now. Please go back to using it. 
For the moment, it is still pointing to development, but the 
shift crew should just use the script.</pre>
<!-- Comment: Sun Mar 17 14;04;17 comment by...jcranshaw -->
<pre>&lt;pre&gt;The problem was that DBANA was based on 4.4.0int2 
which got removed. I've sent a mail to cdf code management 
to clear things up.

The runDBANA script is fine now. Please go back to using it. 
For the moment, it is still pointing to development, but the 
shift crew should just use the script.&lt;/pre&gt;</pre>
<!-- Date: Sat Mar 16 23:37:26 2002 -->
another attempt to do silicon calib without the offending ladder this time. got a pink fib00: hdi slot 13 chan 9.
<!-- Author: muge :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141079' target=_top>141079</a>) -->
<!-- Comment: Sat Mar 16 23;38;59 comment by...mku -->
ok, standalone config did not wokr.abort altogether.

<!-- Comment: Sat Mar 16 23;46;43 comment by...mku -->
apparently the channel corresponds to  the offending ladder

<!-- Date: Sat Mar 16 23:55:58 2002 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141046>
141046</A> 
x226F6</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_1_01 [2,270]</TD>                                           
<TD ALIGN=CENTER>15:56:29</TD>                                          
<TD ALIGN=CENTER>17:22:34</TD>                                          
<TD ALIGN=CENTER>01:10:11</TD>                                          
<TD ALIGN=RIGHT>      6,726,260</TD>                                    
<TD ALIGN=RIGHT>       412,368</TD>                                     
<TD ALIGN=RIGHT>      45,070</TD>                                       
<TD ALIGN=RIGHT>   18.355</TD>                                          
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141048>
141048</A> 
x226F8</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_1_01 [2,270]</TD>                                           
<TD ALIGN=CENTER>17:48:17</TD>                                          
<TD ALIGN=CENTER>18:22:50</TD>                                          
<TD ALIGN=CENTER>00:34:11</TD>                                          
<TD ALIGN=RIGHT>      3,063,423</TD>                                    
<TD ALIGN=RIGHT>       189,667</TD>                                     
<TD ALIGN=RIGHT>      24,778</TD>                                       
<TD ALIGN=RIGHT>    8.292</TD>                                          
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141055>
141055</A> 
x226FF</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_1_01 [2,270]</TD>                                           
<TD ALIGN=CENTER>19:32:39</TD>                                          
<TD ALIGN=CENTER>19:44:21</TD>                                          
<TD ALIGN=CENTER>00:11:29</TD>                                          
<TD ALIGN=RIGHT>        947,555</TD>                                    
<TD ALIGN=RIGHT>        58,800</TD>                                     
<TD ALIGN=RIGHT>       7,937</TD>                                       
<TD ALIGN=RIGHT>    2.493</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>23:55:01</TD>                                                       
<TD ALIGN=CENTER>01:55:52</TD>                                          
<TD ALIGN=RIGHT>      10,737,238</TD>                                   
<TD ALIGN=RIGHT>        660,835</TD>                                    
<TD ALIGN=RIGHT>        77,785</TD>                                     
<TD ALIGN=RIGHT>   29.140</TD>                                          
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Sun Mar 17 00:06:48 2002 -->
Running silicon calibration DPS-off 
<!-- Author: mario :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141082' target=_top>141082</a>) -->
<!-- Comment: Sun Mar 17 00;07;46 comment by...Mario -->
We got CSL error suggesting end of run. The calibration is 
already finished....let repeat it them.
<!-- Date: Sun Mar 17 00:07:00 2002 -->
<font color=red>Shift Summary: </font><pre>Inherit store from day shift, ran relatively smoothly until 
trying to come back for data after taking the D mode silicon 
runs at the end of the store.  Troubles with CSL, possibly 
caused/aggrevated by the D mode run. 
 
Made good preparations for a big store which was lost during 
first part of scraping. 
 
Calibrations are pretty much up to date but should be checked 
with DBANA development version. 
 
End shift retaking D mode with magnet up.  Need to then 
put silicon back to standby and power down the solenoid magnet.</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   141046-141055
Delivered Luminosity   63  
Acquired Luminosity    32  
Efficiency             0.51
   
</td></tr></table></pre>
<!-- Author: michael -->
<!-- Comment: Sun Mar 17 00;11;36 comment by...michael -->
<table border=0><td><tr><td bgcolor=#e0e030><pre>One last message to pass on: day shift should try again
to contact L3 hardware experts to look into L3 subfarm
c08.</pre></td></tr></td></table>
<!-- Comment: Sun Mar 17 00;28;01 comment by...michael -->
<pre>And I forgot to add to page Ming for another special
run with CLC during scraping.  He needs to be here
to set the prescales where he wants them and should
be contacted when the final protons are in.</pre>
<!-- Date: Sun Mar 17 00:09:56 2002 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2002&action=view&page=-1408&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2002&action=view&page=-1408  width='300' align='top'></td><td>Shift lum</a></td></tr></table>
<!-- Author: Henri -->



<!-- Date: Sun Mar 17 00:16:56 2002 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2002&action=view&page=-1409&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2002&action=view&page=-1409  width='300' align='top'></td><td>others</a></td></tr></table><font color=red><cite>(entry outside this shift's time range )</cite></font>
<!-- Author: Henri -->


<!-- Date: Sun Mar 17 00:17:39 2002 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2002&action=view&page=-1410&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2002&action=view&page=-1410  width='300' align='top'></td><td>one more</a></td></tr></table><font color=red><cite>(entry outside this shift's time range )</cite></font>
<!-- Author: Henri -->


<!-- Date: Sun Mar 17 00:18:16 2002 -->
Second run of silicon calibration DPS-OFF  is done<font color=red><cite>(entry outside this shift's time range )</cite></font>
<!-- Author: Mario -->