<!-- Title: Eve shift. Tue Oct 18, 2005 -->
<!-- SciCo: Bill Ashmanskas -->
<!-- DAQAce: Peter Dong -->
<!-- MonAce: Yuji Takeuchi -->
<!-- CO: Anna Sfyrla -->
<!-- OpManager: Guram Chlachidze / Bob Wagner -->
<!-- Notes: -->
We're down, down, down.  TOF calibrations in progress.  Access in progress.  Stack 18E10, no stash, no stacking while D:PMAG is being repaired.

<!-- Date: Tue Oct 18 16:32:09.232 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205765">205765</A>
TERMINATE: Finished successfully.
<!-- Author: Peter x2080 -->


<!-- Date: Tue Oct 18 16:33:05.815 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205766">205766</A>
ACTIVATE: RunSet: CAL_NOM 
<!-- Author: jdeng -->


<!-- Date: Tue Oct 18 16:35:54.427 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205766">205766</A>
TERMINATE: Done COT test: CAL_NOM.
<!-- Author: jdeng -->



<!-- Date: Tue Oct 18 16:37:52.070 2005 -->
TOF calibratons done.  Consumer operator dismissed.  
<!-- Author: Bill -->

<!-- Date: Tue Oct 18 16:39:57.983 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205767">205767</A>
ACTIVATE: RunSet: TOF_TAC_CALIB Per experts' suggestion, I'm going to run a few more of these to see if any of them fail.
<!-- Author: Peter x2080 -->


<!-- Date: Tue Oct 18 16:40:42.718 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205767">205767</A>
TERMINATE: No problems.
<!-- Author: Peter x2080 -->
<!-- Comment: Tue Oct 18 16;44;35 comment by...Peter Dong -->
Actually, this run seems to have failed.  It didn't show up at the CO's webpage and the log file shows a seg fault.
<!-- Comment: Tue Oct 18 16;50;50 comment by...Peter Dong -->
Rebooted calibration CSL per the little note above b0dap73.  Waiting to try again until after the online cluster reboot.
<!-- Date: Tue Oct 18 16:43:47.462 2005 -->
Online cluster reboot likely to happen soon ... fasten seatbelts.  
<!-- Author: Bill -->
<!-- Comment: Tue Oct 18 16;46;27 comment by...Bill -->
Reboot will happen at 5pm.



<!-- Comment: Tue Oct 18 17;03;50 comment by...Bill -->
Still trying to decide whether it will be 5pm or 5:30.  Probably 5:30 ....  TDC checkout still hoping to finish up.


<!-- Comment: Tue Oct 18 17;08;52 comment by...Bill -->
Some nodes are rebooting already; others will wait until 5:30.


<!-- Date: Tue Oct 18 16:51:13.531 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205769">205769</A>
ACTIVATE: RunSet: DAQTEST 
<!-- Author: jdeng -->


<!-- Date: Tue Oct 18 16:55:52.010 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205770">205770</A>
ACTIVATE: RunSet: ALL_COT  Trigger: PHYSICS_3_07 [2,725,597] 
<!-- Author: jdeng -->



<!-- Date: Tue Oct 18 17:01:31.411 2005 -->
I heard an elephant alarm.
<!-- Author: Yuji -->




<!-- Date: Tue Oct 18 17:32:17.500 2005 -->
Sarah and I did the controlled access to work on CMP TOP sections. We have done two things: 
 
1) replaced the preamp card for stack 60 which started an oscillation from Monday.  
 
2) Tightened all screws for the accessible stacks. Two loose screws have been found and tightened, they are stack 18 and 69. There are a few missing screws found in stack 50, 51, 53, 59, 63. Stack 59 also has a small screw (there are a pair of them on the panel) being loose and can not be tightened. 
<!-- Author: Xiaojian, Sarah -->


<!-- Date: Tue Oct 18 17:40:33.022 2005 -->
<table border=0><td><tr><td bgcolor=#e0e030>We are ramping solenoids up</td></tr></td></table>
<!-- Author: Guram -->


<!-- Date: Tue Oct 18 17:55:05.307 2005 -->
Andy Hocker is done in the collision hall.  Rick is still down there as Rob Roser's buddy (closing muon wall).  
<!-- Author: Bill -->


<!-- Date: Tue Oct 18 17:59:19.569 2005 -->
Steve in cryo calls to say that solenoid is at full field.  
<!-- Author: Bill -->


<!-- Date: Tue Oct 18 18:31:27.483 2005 -->
<pre>ICICLE heartbeat on the global alarms window turned purple. 
Peter restarted ICICLE, but it said 'cannot connect to RTserver'. 
Peter confirmed RTserver was up and fine. 
</pre>
<!-- Author: Yuji and Peter -->
<!-- Comment: Tue Oct 18 18;38;29 comment by...yuji -->
It seemed to lose network drive (S:/).
If we clicked the drive S:, it said 'no sufficient resourse ...'.

<!-- Comment: Tue Oct 18 18;48;39 comment by...yuji -->
<pre>W. Badgett rebooted SOLENOID2.
After logged into the pc, we were still missing drive 'S:/'.
Steve called and he instructed me reconnect the network drive: 's:\\newcdf-fs1\mcs'.
Then everything seemed to work fine.</pre>
<!-- Date: Tue Oct 18 18:35:35.963 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205800">205800</A>
ACTIVATE: RunSet: ALL_COT  Trigger: PHYSICS_3_07 [2,725,597] 
<!-- Author: jdeng -->


<!-- Date: Tue Oct 18 18:37:08.702 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205801">205801</A>
ACTIVATE: RunSet: TOF_TAC_CALIB The reboot was delayed, so I'm giving it another shot.
<!-- Author: Peter x2080 -->


<!-- Date: Tue Oct 18 18:38:28.060 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205800">205800</A>
TERMINATE: Looks fine.
<!-- Author: jdeng -->




<!-- Date: Tue Oct 18 18:40:49.282 2005 -->
<TABLE WIDTH=70% BGCOLOR=#CFCFFF> 
<TR><TD> 
<FONT COLOR=#0F0FFF> 
The <I>fer</I> package for <U>F</U>ront<U>E</U>nd<U>R</U>eadout has been  
upped to version <B>v4_23</B> with the following change: 
<UL> 
<LI>Workaround for L2 DPS default to MIN prescale stuck problem due <BR> 
to sumL2Rate being zero, from Chun Lin (c_lin@fnal.gov) 
</UL> 
ScalerMonitor (among other java processes, incidentally) has been <BR> 
restarted to pick up the change. 
</FONT> 
</TR></TR> 
</TABLE> 
<P>  
<!-- Author: W.Badgett for Chun Lin -->

<!-- Date: Tue Oct 18 18:46:46.613 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205801">205801</A>
TERMINATE: See if this works.
<!-- Author: Peter x2080 -->
<!-- Comment: Tue Oct 18 18;50;28 comment by...Peter Dong -->
Doesn't make it into the database.

<!-- Date: Tue Oct 18 18:55:16.654 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205803">205803</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL 
<!-- Author: Peter x2080 -->


<!-- Date: Tue Oct 18 18:58:38.674 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205806">205806</A>
ACTIVATE: RunSet: ALL_COT  Trigger: PHYSICS_3_07 [2,725,597] 
<!-- Author: jdeng -->



<!-- Date: Tue Oct 18 18:58:38.413 2005 -->
<br><b>Diode Dosimeters:</b> 
<br>Measured I/V curves for the diode dosimeter located near the 
SVX detector. 
<br>
<!-- Author: A. Hocker, R.J. Tesarek -->

<!-- Date: Tue Oct 18 19:02:28.385 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205803">205803</A>
TERMINATE: Done.
<!-- Author: Peter x2080 -->


<!-- Date: Tue Oct 18 19:05:00.137 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205807">205807</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: Peter x2080 -->


<!-- Date: Tue Oct 18 19:05:01.024 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205807">205807</A>
TERMINATE: Done.
<!-- Author: Peter x2080 -->


<!-- Date: Tue Oct 18 19:06:35.859 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205808">205808</A>
ACTIVATE: RunSet: BOTHPLUGS_MM 
<!-- Author: Peter x2080 -->


<!-- Date: Tue Oct 18 19:09:52.960 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205808">205808</A>
TERMINATE: End.
<!-- Author: Peter x2080 -->


<!-- Date: Tue Oct 18 19:10:55.767 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205810">205810</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Peter x2080 -->


<!-- Date: Tue Oct 18 19:10:56.851 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205810">205810</A>
TERMINATE: Done.
<!-- Author: Peter x2080 -->



<!-- Date: Tue Oct 18 19:13:04.039 2005 -->
We did some work on the COT crates today.  In the crates b0cot06, b0cot07, b0cot17, we replaced in total 4 XTC2 boards, which had failed on the detector.  We installed new transition boards for the XFT upgrade in the back of b0cot07.  During checkout we found a broken XTC1 in b0cot06, which we then successfully replaced.  During the final checkout we encountered done TO errors in b0cot15 (which we didn't touch).  We tried replacing one TDC that seemed to give problems, but it didn't help.  What helped in the end was to reseat the cards in the crate.  All COT crates should be in working condition now.
<!-- Author: Nils, Jianrong, Eric -->

<!-- Date: Tue Oct 18 19:16:05.473 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205811">205811</A>
ACTIVATE: RunSet: COTCALIB_ACE Last one didn't make it into the database.  Restarted calibration CSL and trying again.
<!-- Author: Peter x2080 -->
<!-- Comment: Tue Oct 18 19;18;39 comment by...Peter Dong -->
Actually, the last one did make it into the database, but somehow I couldn't get it to show up on the webpage.  (This is what happens when you send an Ace to do a CO's job.)  We can view the output still, it looks fine.
<!-- Date: Tue Oct 18 19:16:06.665 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205811">205811</A>
TERMINATE: Done.
<!-- Author: Peter x2080 -->



<!-- Date: Tue Oct 18 19:21:03.022 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205812">205812</A>
 Activated at 2005.10.18 19:20:40
<!-- Author: RunControl -->


<!-- Date: Tue Oct 18 19:21:55.202 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205812">205812</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [5,732,600] 
<!-- Author: Peter x2080 -->



<!-- Date: Tue Oct 18 19:23:43.439 2005 -->
Set BMU HV standby (was OFF except for SE 144 and 168)
<!-- Author: Yuji -->

<!-- Date: Tue Oct 18 19:28:52.543 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205812">205812</A>
 Terminated at 2005.10.18 19:28:29
<!-- Author: RunControl -->


<!-- Date: Tue Oct 18 19:28:53.150 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205812">205812</A>
TERMINATE: L1 accept rate: 66.3 Hz
<!-- Author: Peter x2080 -->


<!-- Date: Tue Oct 18 19:32:05.919 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205813">205813</A>
ACTIVATE: RunSet: TOF_TAC_CALIB Another try
<!-- Author: Peter x2080 -->



<!-- Date: Tue Oct 18 19:32:21.790 2005 -->
Set CPR HV standby (means OFF). It was ON, but there was no bar in HIGH VOLTAGE SUMMARY window. So HV ON might be fake. Anyway now it is OFF.
<!-- Author: Yuji -->


<!-- Date: Tue Oct 18 19:40:55.582 2005 -->
I called Markus to let him know that he can do his L3/EVB work any time now, if he likes.  He said he would probably come by very late this evening.
<!-- Author: Bill -->

<!-- Date: Tue Oct 18 19:41:50.564 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205813">205813</A>
TERMINATE: Done.
<!-- Author: Peter x2080 -->
<!-- Comment: Tue Oct 18 19;43;24 comment by...Peter Dong -->
This got into the database fine.

<!-- Date: Tue Oct 18 19:45:13.988 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205815">205815</A>
ACTIVATE: RunSet: TOF_TAC_CALIB Just to make sure.
<!-- Author: Peter x2080 -->



<!-- Date: Tue Oct 18 19:45:21.593 2005 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2005&action=view&page=-18816&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2005&action=view&page=-18817  width='300' align='top'></a></td><td>To the left is a picture of RunControl when after it has  
gone into "PanicMode" over the Trigger Rates being over  
the Silicon Safety limits.   Note that the Run transition  
(and Activate) are not available, while the End transition  
is still an option.  The "Panic" facility is not limited  
to the Silicon Safety, and can be extended to other  
error conditions if needed.   (Click on image to get a  
clearer view!)</td></tr></table>
<!-- Author: W.Badgett -->

<!-- Date: Tue Oct 18 19:54:53.323 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205815">205815</A>
TERMINATE: Done.
<!-- Author: Peter x2080 -->
<!-- Comment: Tue Oct 18 19;56;00 comment by...Peter Dong -->
This one also got into the database fine.  One more successful try and I'm declaring it temporarily fixed.

<!-- Date: Tue Oct 18 19:56:21.931 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205816">205816</A>
ACTIVATE: RunSet: TOF_TAC_CALIB Last try.
<!-- Author: Peter x2080 -->


<!-- Date: Tue Oct 18 19:58:06.837 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205816">205816</A>
TERMINATE: Done -- too fast.  Perhaps I started this run too soon after the last.
<!-- Author: Peter x2080 -->


<!-- Date: Tue Oct 18 20:08:45.792 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205817">205817</A>
ACTIVATE: RunSet: TOF_TAC_CALIB 
<!-- Author: Peter x2080 -->


<!-- Date: Tue Oct 18 20:08:47.034 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205817">205817</A>
TERMINATE: Done.
<!-- Author: Peter x2080 -->



<!-- Date: Tue Oct 18 20:11:07.601 2005 -->
All the TOF TAC calibrations I tried got into the database, except one, I think because I started it before the last process stopped.  For now the calibration seems to be working fine, at least since my last reboot of the calibration CSL.
<!-- Author: Peter Dong -->

<!-- Date: Tue Oct 18 20:12:36.130 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205819">205819</A>
 Activated at 2005.10.18 20:11:35
<!-- Author: RunControl -->


<!-- Date: Tue Oct 18 20:12:37.319 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205819">205819</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,730,598] 
<!-- Author: Peter x2080 -->



<!-- Date: Tue Oct 18 20:24:52.778 2005 -->
DAQ ace dismissed.  
<!-- Author: Bill -->


<!-- Date: Tue Oct 18 20:50:47.543 2005 -->
Got TM2 Events Pending TO error.
<!-- Author: Yuji :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205819' target=_top>205819</a>) -->
<!-- Comment: Tue Oct 18 20;51;32 comment by...Yuji -->
Auto HRRic.

<!-- Comment: Tue Oct 18 21;15;04 comment by...Yuji -->
Again.

<!-- Comment: Tue Oct 18 21;22;26 comment by...Yuji -->
Again.

<!-- Comment: Tue Oct 18 21;29;45 comment by...Yuji -->
Again.

<!-- Comment: Tue Oct 18 21;33;37 comment by...Yuji -->
Again.

<!-- Comment: Tue Oct 18 22;03;07 comment by...Yuji -->
Again.

<!-- Date: Tue Oct 18 22:03:47.744 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205819">205819</A>
 Terminated at 2005.10.18 22:03:18
<!-- Author: RunControl -->


<!-- Date: Tue Oct 18 22:03:48.357 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205819">205819</A>
TERMINATE: Try to Clean up EVB
<!-- Author: Peter x2080 -->



<!-- Date: Tue Oct 18 22:10:53.729 2005 -->
<pre>Cleaned up EVB (from Ace control release 2.17) returned the followings: 
 
10:04:27 BEGIN: Reboot SM, SCPUs and restart EVB proxy 
Looking for suspended processes on SM and SCPUs 
b0eb10: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb11: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb12: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb13: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb14: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb15: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb16: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb17: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb18: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb19: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb20: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb21: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb22: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb23: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb24: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb25: I can't log in. The CPU closed the rlogin connection, don't know why. 
EVB proxy stop: Killed: ZMConverter, zwgc, EvbProxy 
Rebooting Scanner Manager and SCPUs. 
b0eb10: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb11: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb12: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb13: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb14: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb15: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb16: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb17: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb18: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb19: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb20: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb21: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb22: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb23: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb24: I can't log in. The CPU closed the rlogin connection, don't know why. 
b0eb25: I can't log in. The CPU closed the rlogin connection, don't know why. 
Check a Merlin display for the default partition to see if all the CPUs 
listed above finish booting in the next minute or two. If they don't, 
try pushing the Reset buttons on the CPUs that didn't make it. 
EVB proxy restart: OK 
10:08:50 END: Reboot SM, SCPUs and restart EVB proxy 
</pre>
<!-- Author: Yuji -->
<!-- Comment: Tue Oct 18 22;13;54 comment by...Yuji -->
This ace control is not usual one...?

<!-- Comment: Tue Oct 18 22;18;37 comment by...Yuji -->
Ok. I started Ace control (release 3.0). And cleanup was successfully.

<!-- Date: Tue Oct 18 22:22:08.425 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205820">205820</A>
 Activated at 2005.10.18 22:20:55
<!-- Author: RunControl -->


<!-- Date: Tue Oct 18 22:22:09.206 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205820">205820</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,730,598] 
<!-- Author: Peter x2080 -->



<!-- Date: Tue Oct 18 23:03:24.473 2005 -->
Got TM2 Events Pending TO Error.
<!-- Author: Yuji :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205820' target=_top>205820</a>) -->
<!-- Comment: Tue Oct 18 23;05;34 comment by...Yuji -->
Auto HRR worked fine.


<!-- Comment: Tue Oct 18 23;15;26 comment by...Yuji -->
Again. The effect of EVB cleanup ran out?
<!-- Comment: Tue Oct 18 23;25;17 comment by...Yuji -->
Again. Auto HRRific.
<!-- Date: Tue Oct 18 23:06:21.133 2005 -->
MCR called us. They will inject protons shortly.
<!-- Author: Yuji -->
<!-- Comment: Tue Oct 18 23;09;13 comment by...Yuji -->
All HVs are in End-of-store positons.


<!-- Comment: Tue Oct 18 23;22;12 comment by...Yuji -->
Sebastian finished his silicon work around 23:00.
<!-- Date: Tue Oct 18 23:13:20.021 2005 -->
We got an usual random alarm. This time it was three times beep.
<!-- Author: Yuji :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205820' target=_top>205820</a>) -->



<!-- Date: Tue Oct 18 23:20:26.304 2005 -->
IMU BMU Work : We first attacked the oscillation in SE near stack 150. Replaced stack # 150 preamp. The oscillation appears to have been reduced but not eliminated. It is possible this chamber is sparking.  
 
After replacement however, SE144 high voltage failed to come up. After skipping over the 150, 151 jumpered pair, the HV still failed at low voltages. We then powered stacks 144-149 with supply 144, and 150-155 with the supply for 156. Both sectors came up to voltage. I felt confident the problem was not in the chambers at that point, so returned the supply to the default config.  
 
We then attempted to replace preamp in NW 26, which had severe oscillation. Turns out both 26, and 27 had severe oscillation, with time running out I unplugged the low voltage supply to both   preamps, so an alert CO will see a two chamber hole in the BMU west, around phi 26,27 - no need to page for this.  
 
Upstairs I investigated the HV supply. SE 144 was tripping upon ramp at low votlage (200V or less). No excessive current draw. On the back of the CAEN pod, I swapped 144 and 168 lines, and they both held full voltage for a while. This may cause a problem in the future, and I am suspicious this is a bad CAEN supply.  
  
<!-- Author: Sunny C. and Dan Cyr -->



<!-- Date: Tue Oct 18 23:29:28.950 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205820">205820</A>
 Terminated at 2005.10.18 23:28:30
<!-- Author: RunControl -->


<!-- Date: Tue Oct 18 23:29:29.930 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205820">205820</A>
TERMINATE: Switch to L2 toture since some beams are injected.
<!-- Author: Peter x2080 -->



<!-- Date: Tue Oct 18 23:31:58.213 2005 -->
We will switch to L2 torture since we had some beam in tevatron now. Also issued EVB cleanup.
<!-- Author: Yuji -->

<!-- Date: Tue Oct 18 23:34:07.323 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205821">205821</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [16,729,599] 
<!-- Author: Peter x2080 -->


<!-- Date: Tue Oct 18 23:57:09.439 2005 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>Physics able</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L1Accepts</TH>
<TH>L2Accepts</TH>
<TH>L3Accepts</TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>23:55:01</TD>
<TD ALIGN=CENTER>::</TD>
<TD ALIGN=RIGHT></TD>
<TD ALIGN=RIGHT></TD>
<TD ALIGN=RIGHT></TD>
<TD ALIGN=RIGHT></TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Tue Oct 18 23:59:37.913 2005 -->
<font color=red>Shift Summary: </font><pre>No data! 
</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   n/a
Delivered Luminosity   0  
Acquired Luminosity    0  
Efficiency             NaN
   
</td></tr></table></pre>
<!-- Author: Bill Ashmanskas -->


<!-- Date: Wed Oct 19 00:52:55.115 2005 -->
Began shift with 18E10 in Accumulator, no Recycler stash, no stacking due to PMAG 
swap-out and Tevatron F sector vacuum work, CDF in controlled access to collision 
hall.  Access tasks included CEM PMT, SVX, COT07, Roman pots, BMU, CMP stack 60.  Access ended at 19:00. 
<p> 
Online cluster reboot was considered for 17:00, but was decided not to do it until 
TDC checkout work was complete.  In the end, it was postponed until 08:00 Weds. 
<p> 
Markus will probably come in on owl shift to do some EVB/L3 work.  He wants 
<!-- Author: Bill -->


<!-- Date: Wed Oct 19 00:55:54.694 2005 -->
<pre>Evening shift summary: 
 
Began shift with 18E10 in Accumulator, no Recycler stash, no stacking due to PMAG 
swap-out and Tevatron F sector vacuum work, CDF in controlled access to collision 
hall.  Access tasks included CEM PMT, SVX, COT07, Roman pots, BMU, CMP stack 60.  Access ended at 19:00. 
 
Online cluster reboot was considered for 17:00, but was decided not to do it until 
TDC checkout work was complete.  In the end, it was postponed until 08:00 Weds. 
 
Markus will probably come in on owl shift to do some EVB/L3 work.  He wants 'control room quiet time' (not Tevatron quiet time). 
 
CO and DAQ ace were sent home early. 
 
F sector access finished up around 21:00, and Tevatron injected protons about 23:00 for studies.  Antiproton source pulsed magnet work was still in progress 
at the end of shift, so still no stacking.  Also, the small Accumulator stack was  
dropped for a rings access.  So we're still at least a shift away from the next store. 
</pre> <font color=red><cite>(entry outside this shift's time range )</cite></font>
<!-- Author: Bill -->