<!-- Title: Owl shift. Thu Nov 29, 2001 -->
<!-- SciCo: Luciano Ristori -->
<!-- DAQAce: Marcin Wolters -->
<!-- MonAce: Ayana Holloway -->
<!-- CO: Jonas Rademacker -->
<!-- OpManager: Dave Ambrose -->
<!-- Notes: -->
<pre>Store 784 in progress. 
Luminosity is 1.10 E30. 
Taking data with PHISICS_0_02[4,216]</pre>


<!-- Date: Thu Nov 29 00:36:52 2001 -->
Hyunsoo has the CMU HV.  Since losses are calmer, bringing up cmx, bmu, cpr (these tripped on last shift during big losses.) 
<!-- Author: ath -->


<!-- Date: Thu Nov 29 00:45:12 2001 -->
<pre>Ces Wires, Ces Strips, CPR, CCR in West 7 and 8 are dead 
due to bad calibrations the last 2 days.  Standalone 
code confirms hardware problem in 7 West that appears 
to require access,  8 West needs more investigation. </pre>
<!-- Author: Steve Kuhlmann -->


<!-- Date: Thu Nov 29 00:55:07 2001 -->
L3 failed coldstart. CSL seems to have some errors: 
 
ace@b0dau32 21% cslcom check 
CDFVME_COMMON: Last updated by nahn on Thu Mar 23 13:03:15 CST 2000 
  
Number of CSL processes running             = 38 
  (0: OFF, >=7: ON, anything else: error) 
Number of shared memory segments            = 4 
  (0: OFF, 4: ON, anything else: error) 
Number of message queues                    = 6 
  (0: OFF, 6: ON, anything else: error) 
 
restarting CSL .. L3 proxy ...  
<!-- Author: Matt,Carsten,Marcin,Ilya,... :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=133666' target=_top>133666</a>) -->


<!-- Date: Thu Nov 29 01:01:01 2001 -->
HVMon++ shows CMU W3 anode is drawing 10 uA at 2.325kV. The calibration is wrong with that particular HV pod. It was showing 9 uA current draw when HV is off and HV value is at 0V. We'll have to replace the pod but will leave it in for now since it does not pose any operational problems. HV on W3 is turned back on.
<!-- Author: hyunsoo -->


<!-- Date: Thu Nov 29 01:02:21 2001 -->
<pre>Run 133667 started with trigger table PHYSICS_0_02 [4,216] </pre>
<!-- Author: M Wolter -->
<!-- Comment: Thu Nov 29 01;05;01 comment by...Matt -->
This run is marked as DAQ testing and the ignore inhibit is on while the losses are high and Steve and I are working on L2.  


<!-- Date: Thu Nov 29 01:17:26 2001 -->
Recent history: 
<dl> 
<li>MCR calls: they're going to scrape. 
<li>Proton loss spikes exceed 60kHz, CPR trips 
<li>Beam looks better 
<li>MCR calls: finished.  CPR back on 
</dl>
<!-- Author: ath :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=133667' target=_top>133667</a>) -->

<!-- Date: Thu Nov 29 01:22:50 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=133668">133668</A>  Activated at 2001.11.29 01:19:37
<!-- Author: Activation Notification -->
<!-- Comment: Thu Nov 29 01;30;38 comment by...Marcin, Matt -->
Now that losses are stable and we have a new ISOclique board working in L2, we are taking beam data again.  Physics_0_02[4,216] with L2 cutting.  After about 20k L1A (5.5k L3A) the rates are L1A = 46 Hz, L2A = 19 Hz, and L3A = 12 Hz.


<!-- Date: Thu Nov 29 01:54:37 2001 -->
<pre>Changed prescales at L1: 
 
L1 Trigger              old ps     new ps     old rate   new rate 
----------------------------------------------------------------- 
3  L1_JET10_&_CLC_PS50    50         2          0.5 Hz     10 Hz 
5  L1_JET5_&_CLC_PS20     20         2          12.0       125. 
7  L1_JET5_&_CLC_PS200    200        2          1.2        125. 
17 L1_JET10_PS10          10         1          6.1        61. 
19 L1_JET5_PS200          200        1          1.7        370. 
 
Rates are now: 370 Hz L1A, 170 Hz L2A, 170 Hz L3A with about 
0.9e30 inst lum and 17 MB/sec read out of CSL.  Deadtime is  
about 3%, most of which is readout.</pre>
<!-- Author: Matt :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=133668' target=_top>133668</a>) -->


<!-- Date: Thu Nov 29 02:07:49 2001 -->
At about 264k events, we had a L2 timeout with the following errors in the b0l2pr00 minicom:<p> 
ClistMon: Error: 1 != 0  Found a fatal error (FaultManager), busy-looping until L2 timeout.<p> 
The first HRR gave a new timeout from the L1Monitor code.  The second HRR got us running successfully.
<!-- Author: Matt :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=133668' target=_top>133668</a>) -->


<!-- Date: Thu Nov 29 02:41:04 2001 -->
<h4> Level 3 Jet Rates and 3TeV Calorimter Tower</h4> 
We have been wondering about the high L3 jet rates observed with the new executable for some time until Luciano came up with an explanation:<BR> For some time, that 
calorimeter has been signalling a 3TeV tower in every event. This is a hardware 
problem, and the corresponding piece of hardware has been masked out as 
far as Level 2 is concerned, but Level 3 still sees it.  
<!-- Author: Jonas :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=1333668' target=_top>1333668</a>) -->
<!-- Comment: Thu Nov 29 02;41;38 comment by...Jonas -->
The runnumber is 133668


<!-- Date: Thu Nov 29 03:17:28 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-2686&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-2686  width='300' align='top'></td><td>Acnet stuff: B0L(ive)LUM ok, B0T(otal)LUM ok, but B0T(otal)LIV(e) not OK.  I convinced acnet to make an integrated luminosity and integrated live luminosity plot that looks reasonable anyway. More details when I figure out what units I've actually set for the integral....</a></td></tr></table>
<!-- Author: ath -->
<!-- Comment: Thu Nov 29 08;09;21 comment by...ATH -->
Read the y-axis for the integral as ten inv. nanobarns (not the legend.)
<!-- Date: Thu Nov 29 03:34:11 2001 -->
We've decided not to wake up CLC people over this integrated lum problem (since we're not sure it's the CLC crate.)  We know the live gate and the integrated luminosity are ok, so this might just be a cable thing? 
  
<!-- Author: owl people -->
<!-- Comment: Thu Nov 29 03;37;27 comment by...ath -->
although the CLC does look rather suspicious: click on a run number to verify that 

CLC01 has
<font color=red>nothing</font> in the slot list.




<!-- Comment: Thu Nov 29 10;17;56 comment by...Arnd -->
This is OK, nothing is read out from clc01 through the central DAQ
<!-- Date: Thu Nov 29 04:10:14 2001 -->
MCR called: they will fly wires several times to diagnose a problem.  
<!-- Author: ath -->

<!-- Date: Thu Nov 29 06:00:37 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=133668">133668</A>  Terminated at 2001.11.29 05:57:06
<!-- Author: Termination Notification -->



<!-- Date: Thu Nov 29 06:01:43 2001 -->
MCR calls to tell us they are going to do electrostatic separator scans.  
<!-- Author: Luciano :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=133668' target=_top>133668</a>) -->


<!-- Date: Thu Nov 29 06:07:49 2001 -->
Run 133668 was ended with about 2.2 million events.
<!-- Author: Luciano -->

<!-- Date: Thu Nov 29 06:09:28 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=133669">133669</A>  Activated at 2001.11.29 06:06:11
<!-- Author: Activation Notification -->
<!-- Comment: Thu Nov 29 07;39;20 comment by...Carsten -->
Processors 019-034 turned yellow (old).


<!-- Date: Thu Nov 29 07:56:21 2001 -->
<font color=red>Shift Summary: </font><pre>	Shift Summary 
	Owl - 11/29/01 
 
Start of shift conditions: 
   Store 784 in progress.  
   Luminosity is 1.10 E30.  
   Taking data with PHISICS_0_02[4,216] 
 
- 00:45 Steve Kuhlmann finds problems in CES: 
        - 7W requires access 
        - 8W needs more investigation. 
 
- 01:01 Hyunsoo finds problem in CMU HV. It turns out to be 
        a problem in the monitoring (bad pod will have to be replaced) 
 
- 01:22 Start run 133668 with L2 cutting. 
        Trigger: PHYSICS_0_02[4,216] + changed prescales 
        Rates: 370 Hz L1A, 170 Hz L2A, 170 Hz L3A  
        with about 0.9e30 inst lum. 
        L3 rates are funny due to bad Admem in WHA. 
        Deadtime is about 3%, most of which is readout. 
 
- 06:07 run 133668 ended with ~2.2 million events. 
        MCR calls they are going to perform separator scans, 
        they do not know when they will dump the beam. 
 
- 07:20 They are done with separator scans.</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   133666-133669
Delivered Luminosity   54.6  
Acquired Luminosity    ???  
Efficiency             ???
   
</td></tr></table></pre>
<!-- Author: Luciano -->
<!-- Comment: Thu Nov 29 08;09;15 comment by...Carsten -->
The Delivered Luminosity during this shift was 23.7nb-1. (54.6nb-1 is for the store 784)



<!-- Date: Thu Nov 29 08:19:19 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-2690&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-2690  width='300' align='top'></td><td>The integrated live lum is still kludged.  losses, SVX losses, luminosity for shift (plus a little overlap.)</a></td></tr></table><font color=red><cite>(entry outside this shift's time range )</cite></font>
<!-- Author: ath -->


<!-- Date: Thu Nov 29 08:20:15 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-2691&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-2691  width='300' align='top'></td><td>SVX rad</a></td></tr></table><font color=red><cite>(entry outside this shift's time range )</cite></font>
<!-- Author: ath -->


<!-- Date: Thu Nov 29 08:22:27 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-2693&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-2693  width='300' align='top'></td><td>losses, note spike that ate ~1/2 the luminosity.</a></td></tr></table><font color=red><cite>(entry outside this shift's time range )</cite></font>
<!-- Author: ath -->

<!-- Date: Thu Nov 29 08:28:48 2001 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=133669">133669</A>  Terminated at 2001.11.29 08:26:01
<!-- Author: Termination Notification -->