<!-- Title: Owl shift. Fri Jun  6, 2008 -->
<!-- SciCo: Richard Hughes -->
<!-- DAQAce: Simone Pagan Griso -->
<!-- MonAce: &nbsp; -->
<!-- CO: Alberto Annovi -->
<!-- OpManager: Steve Hahn -->
<!-- Notes: -->
<pre>Should start shot setup soon!</pre>


<!-- Date: Fri Jun 6 00:31:08.266 2008 -->
MCR called, D0 1/2 silicon is off.  An expert is on the way to try and fix without an access, so shot setup is on hold.  If the expert cannot fix, there will be a ~1hr access.  
<!-- Author: Richard -->


<!-- Date: Fri Jun 6 00:44:38.821 2008 -->
1st floor PSM alarm is silenced. Everything is green inside but 1RR12D_1 that is masked. If we unsilence the all 1st floor alarms (1RR12D_! kept masked.. and silenced) we have PSM alarm anyway. I do not understand why. Steve will give a look tomorrow, we'll live with this for tonight. We also opened the 1st floor picture on another PC too (silicon VNODE2) just to be able to see if there are problems easier during this night.
<!-- Author: Simone :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263136' target=_top>263136</a>) -->


<!-- Date: Fri Jun 6 00:53:49.233 2008 -->
Consumers have been moved from Cosmics to Physics and back to Cosmics mode for testing to help remote CO in training...
<!-- Author: richard -->

<!-- Date: Fri Jun 6 01:09:54.251 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263136>263136</A>
 Terminated at 2008.06.06 01:08:18
<!-- Author: RunControl -->


<!-- Date: Fri Jun 6 01:09:55.429 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263136>263136</A>
TERMINATE: switching to l2torture
<!-- Author: Chang-Seong -->



<!-- Date: Fri Jun 6 01:12:47.042 2008 -->
MCR called, D0 will start an in about 10 minutes.  Will last 1 hour.  Then shot setup 30 minutes after that.
<!-- Author: richard -->


<!-- Date: Fri Jun 6 01:17:08.893 2008 -->
MCR says we will have quiet time for 1-2 hrs.  They will call before they put beam in the machine.  Will do some quiet time calibs.
<!-- Author: richard -->

<!-- Date: Fri Jun 6 01:19:36.731 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263138>263138</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 01:21:17.973 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263138>263138</A>
TERMINATE: done.
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 01:21:18.431 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263138>263138</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Jun  6 01;42;18 comment by...Alberto -->
calqie looks good. No new dead or hot channels.
<!-- Comment: Fri Jun  6 02;03;05 comment by...Alberto -->
Checked again with correct run number. No new dead or hot channels.
<!-- Date: Fri Jun 6 01:21:55.487 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263139>263139</A>
ACTIVATE: RunSet: CALIB_CLC_QIE 
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 01:28:59.864 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263139>263139</A>
CALIBRATION: <U><B>clcqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#clcqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Jun  6 02;06;29 comment by...Alberto -->
clcqie looks good
<!-- Date: Fri Jun 6 01:29:00.401 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263139>263139</A>
TERMINATE: done
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 01:31:38.869 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263141>263141</A>
ACTIVATE: RunSet: CALIB_QIE_FWD 
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 01:33:13.048 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263140>263140</A>
CALIBRATION: <U><B>bscqie</B></U> calibration FAILED, please investigate
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Jun  6 02;13;21 comment by...Alberto -->
bscqie run 263141 looks good.
<!-- Date: Fri Jun 6 01:33:14.953 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263141>263141</A>
TERMINATE: done
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 01:39:31.782 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263142>263142</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL 
<!-- Author: Chang-Seong -->



<!-- Date: Fri Jun 6 01:46:31.884 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263142>263142</A>
TERMINATE: done
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 01:47:34.430 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263143>263143</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 01:50:15.839 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263143>263143</A>
TERMINATE: done
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 01:50:16.850 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263143>263143</A>
CALIBRATION: <U><B>pluglaser</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#pluglaser target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Jun  6 02;18;02 comment by...Alberto -->
Pluglaser calibration is ok.
<!-- Date: Fri Jun 6 01:51:19.875 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263144>263144</A>
ACTIVATE: RunSet: BOTHPLUGS_MM 
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 01:53:40.109 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263144>263144</A>
TERMINATE: done
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 01:58:00.178 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263145>263145</A>
 Activated at 2008.06.06 01:56:43
<!-- Author: RunControl -->


<!-- Date: Fri Jun 6 01:58:01.550 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263145>263145</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [7,1023,706] 
<!-- Author: Chang-Seong -->




<!-- Date: Fri Jun 6 02:07:50.388 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263145>263145</A>
 Terminated at 2008.06.06 02:07:39
<!-- Author: RunControl -->


<!-- Date: Fri Jun 6 02:09:37.551 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263145>263145</A>
TERMINATE: done
<!-- Author: Chang-Seong -->
<!-- Comment: Fri Jun  6 02;09;58 comment by...Simone -->
FredL1A 46.2Hz. OK.

<!-- Date: Fri Jun 6 02:11:32.829 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263146>263146</A>
ACTIVATE: RunSet: ACE_TOF_QIE_NEW 
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 02:11:33.201 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263146>263146</A>
CALIBRATION: <U><B>tofqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#tofqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Jun  6 02;20;57 comment by...Alberto -->
tofqie is good.
<!-- Date: Fri Jun 6 02:11:34.036 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263146>263146</A>
TERMINATE: done
<!-- Author: Chang-Seong -->



<!-- Date: Fri Jun 6 02:16:47.967 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263147>263147</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 02:16:48.501 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263147>263147</A>
TERMINATE: done
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 02:17:32.083 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263147>263147</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Jun  6 02;28;01 comment by...Alberto -->
smx calibration is ok.
<!-- Date: Fri Jun 6 02:23:14.996 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263148>263148</A>
CALIBRATION: <U><B>cemled</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cemled target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Jun  6 02;42;24 comment by...Alberto -->
cemled look normal execpt a few channels with 0 ADC counts (see image below).
Is it normal to have a few channels with 0 ADC counts?
<!-- Date: Fri Jun 6 02:23:15.545 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263148>263148</A>
TERMINATE: done
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 02:23:36.722 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263148>263148</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 02:25:17.169 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263149>263149</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 02:25:18.832 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263149>263149</A>
TERMINATE: done
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 02:27:43.452 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263149>263149</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Jun  6 02;51;28 comment by...Alberto -->
<pre>The COT calibration is good. However, constants have been updated more than 7 times.
So I'm sending an email to Eiko as per instructions.

---&gt; Out of the last 10 calibrations, constants have been updated 8 times</pre>
<!-- Date: Fri Jun 6 02:30:11.778 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263150>263150</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 02:30:12.344 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263150>263150</A>
TERMINATE: done
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 02:30:13.912 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263150>263150</A>
CALIBRATION: <U><B>calxef</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calxef target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Jun  6 03;13;18 comment by...Alberto -->
<pre>We have a problem with 2 red wedges in CALXEF. This could be a know problem since Jan 27. See plot below.</pre>
<!-- Date: Fri Jun 6 02:36:35.158 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263151>263151</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 02:36:36.543 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263151>263151</A>
TERMINATE: done
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 02:36:37.696 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263152>263152</A>
ACTIVATE: RunSet: TOF_TAC_CALIB 
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 02:40:37.806 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263152>263152</A>
CALIBRATION: <U><B>toftac</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#toftac target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Jun  6 03;08;22 comment by...Alberto -->
toftac calibration is ok.
<!-- Date: Fri Jun 6 02:40:38.589 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263152>263152</A>
TERMINATE: done
<!-- Author: Chang-Seong -->



<!-- Date: Fri Jun 6 02:41:25.714 2008 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2008&action=view&page=-2752&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-2752  width='300' align='top'></a></td><td>List of channels with 0 ADC counts from CEMLED</td></tr></table>
<!-- Author: Alberto -->
<!-- Comment: Fri Jun  6 02;46;48 comment by...Alberto -->
Is it normal?
<!-- Date: Fri Jun 6 02:43:03.121 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263153>263153</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [19,1021,705] 
<!-- Author: Chang-Seong -->



<!-- Date: Fri Jun 6 02:45:15.063 2008 -->
MCR called - will begin shot setup in about 25 minutes.  D0 is having problems and will probably request an 8hr access at the 9am meeting.
<!-- Author: richard -->
<!-- Comment: Fri Jun  6 07;34;27 comment by...phil -->
oi vey
<!-- Date: Fri Jun 6 03:12:02.884 2008 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2008&action=view&page=-2753&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-2753  width='300' align='top'></a></td><td>CALXEF shows two red wedges. 
They are NOT red in the reference plot. 
There is an entry with this problem in the CDF fixlist dated Jan 27. Do we still have this problem???</td></tr></table>
<!-- Author: Alberto -->


<!-- Date: Fri Jun 6 03:32:27.563 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263153>263153</A>
TERMINATE: shot setup.. switching to cosmic
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 03:44:48.753 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263154>263154</A>
 Activated at 2008.06.06 03:34:18
<!-- Author: RunControl -->


<!-- Date: Fri Jun 6 03:44:49.709 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263154>263154</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,1019,706] 
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 04:22:16.494 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263154>263154</A>
 Terminated at 2008.06.06 04:22:12
<!-- Author: RunControl -->



<!-- Date: Fri Jun 6 04:22:38.545 2008 -->
Final protons are being loaded.
<!-- Author: Simone :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263154' target=_top>263154</a>) -->

<!-- Date: Fri Jun 6 04:26:14.642 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263155>263155</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_5_01 [7,1088,721] 
<!-- Author: Chang-Seong -->



<!-- Date: Fri Jun 6 04:26:52.106 2008 -->
Done TO for crate cmu00, TDC slot 24.
<!-- Author: Simone :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263155' target=_top>263155</a>) -->


<!-- Date: Fri Jun 6 04:38:15.898 2008 -->
pbars in
<!-- Author: richard -->
<!-- Comment: Fri Jun  6 04;55;52 comment by...richard -->
this was the start of pbar injection
<!-- Date: Fri Jun 6 04:55:11.273 2008 -->
Ok, now ALL of the pbars are in
<!-- Author: richard -->



<!-- Date: Fri Jun 6 05:07:00.968 2008 -->
<table border=0><td><tr><td bgcolor=#e0e030>restarted scaler with with 'fer test' as Donatella requested.</td></tr></td></table>
<!-- Author: Simone :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263155' target=_top>263155</a>) -->


<!-- Date: Fri Jun 6 05:08:39.658 2008 -->
Failed to book all silicon crates. an instance of run control running on b0dasvx01 was taking them. I closed the partition and exited from that rc.
<!-- Author: Simone :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263156' target=_top>263156</a>) -->


<!-- Date: Fri Jun 6 05:17:41.589 2008 -->
Scraping is complete.  Store number 6200. 
 
TevPR 9579 E09 
TeVPB 2634 E09 
 
Initial Lum 299.47
<!-- Author: richard -->

<!-- Date: Fri Jun 6 05:24:40.150 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263157>263157</A>
TERMINATE: silicon not on
<!-- Author: Chang-Seong -->



<!-- Date: Fri Jun 6 05:32:26.280 2008 -->
Problem with silicon - cannot connect to RT server.  Running without silicon.....
<!-- Author: richard :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263158' target=_top>263158</a>) -->

<!-- Date: Fri Jun 6 05:34:00.031 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263158>263158</A>
 Activated at 2008.06.06 05:33:02
<!-- Author: RunControl -->


<!-- Date: Fri Jun 6 05:34:01.113 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263158>263158</A>
ACTIVATE: RunSet: AAA_NOSILICON  Trigger: PHYSICS_5_01 [7,1088,721] 
<!-- Author: Chang-Seong -->



<!-- Date: Fri Jun 6 05:38:07.445 2008 -->
L2 Decision TO. Auto HRR fixes it.
<!-- Author: Simone :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263158' target=_top>263158</a>) -->
<!-- Comment: Fri Jun  6 05;38;45 comment by...Simone -->
again

<!-- Comment: Fri Jun  6 05;40;20 comment by...Simone -->
again. RXPT errors LOOP. Had to issue manual HRR. it fixes it.

<!-- Comment: Fri Jun  6 05;40;42 comment by...Simone -->
again


<!-- Comment: Fri Jun  6 05;45;45 comment by...Simone -->
again
<!-- Comment: Fri Jun  6 05;47;21 comment by...Simone -->
again
<!-- Comment: Fri Jun  6 05;49;50 comment by...Simone -->
again
<!-- Comment: Fri Jun  6 05;52;37 comment by...Simone -->
again
<!-- Comment: Fri Jun  6 06;13;10 comment by...Simone -->
again
<!-- Comment: Fri Jun  6 06;15;36 comment by...Simone -->
again
<!-- Comment: Fri Jun  6 06;19;18 comment by...Simone -->
again
<!-- Comment: Fri Jun  6 06;22;38 comment by...Simone -->
again
<!-- Comment: Fri Jun  6 06;25;26 comment by...Simone -->
again
<!-- Comment: Fri Jun  6 06;31;47 comment by...Simone -->
again
<!-- Comment: Fri Jun  6 06;47;46 comment by...Simone -->
again
<!-- Comment: Fri Jun  6 07;12;28 comment by...Simone -->
again
<!-- Comment: Fri Jun  6 07;20;57 comment by...Simone -->
again
<!-- Date: Fri Jun 6 05:42:32.508 2008 -->
COT SL8 tripped. Paged COT expert. Turned on COT again. Now it's fine. We'll call the expert if it trips again.
<!-- Author: Simone :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263158' target=_top>263158</a>) -->





<!-- Date: Fri Jun 6 05:51:01.900 2008 -->
IMU red in DQMon.. Found 5 new bad channel in YMON/IMU/BMUCells_West_0. Asked the CO to xcheck.
<!-- Author: Simone :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263158' target=_top>263158</a>) -->
<!-- Comment: Fri Jun  6 05;51;29 comment by...Simone -->
now yello. Number of channel decreased to 4. 



<!-- Comment: Fri Jun  6 06;35;33 comment by...Simone -->
for the new run (263159) IMU is OK on DQMon. Zero bad channels found.
<!-- Date: Fri Jun 6 05:58:42.650 2008 -->
Trying to get a DQAQ expert to help with RT Server problem.  Paged DAQ.
<!-- Author: richard -->


<!-- Date: Fri Jun 6 06:01:48.939 2008 -->
<PRE> 
Short summary about silicon issues: 
 - after turnin clc on silicon pager carrier was not arrived yet. 
 - Spontaneous switch off of PS#9 while silicon in standby, while ramping the other HVs. SAR recovered succesfully 
 - when all HVs were ramping we paged silicon again. He called back when all HVs were up. he said 2 minutes to be there.  
 - We failed to turn on silicon because of RT server connection problems. Pages silicon SPL. 
 - Still working on that.. paged also DAQ. 
</PRE>
<!-- Author: Simone :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263158' target=_top>263158</a>) -->
<!-- Comment: Fri Jun  6 06;23;45 comment by...Sasha Pronko (Silicon MPC) -->
<pre>After firsdt reboot of svxinter, it was complaining about the RTServer. So, I asked ACE (Simone) to restart svxinter. He refused to do it saying that restarting RTserver would only cause more problems. So, we paged DAQ and eventually got hold of Bill Badgett who solved the problem. Silicon is included now. </pre>
<!-- Comment: Fri Jun  6 06;28;20 comment by...Simone -->
just to say things as _really_ happened: I di not refused. I've just asked if it was sure that is was the right thing to do, since i was seeing NO problems with RT server. He did not asnwer to that question. I also said that if Silicon SPL sasha was convinced that RT server was the problem I would have restarted it; infact i had a shell already opened ready to do it.
<!-- Comment: Fri Jun  6 06;39;05 comment by...richard -->
Things happened as the ACE described.  There was alot of confusion regarding what to do.  We *all* agreed (ace, scico, sasha pronko, silicon SPL) that the right thing to do was to page DAQ.  We did and the expert (Bill) solved the problem.  
<!-- Date: Fri Jun 6 06:02:59.541 2008 -->
<P> 
Presumably we had a stuck port with rtserver when svxinter was  
first reset;   i restarted rtserver and rebooted  
svxinter, and i think it's ok now... 
<P>
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263158' target=_top>263158</a>) -->

<!-- Date: Fri Jun 6 06:07:07.478 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263158>263158</A>
TERMINATE: put silicon back
<!-- Author: Chang-Seong -->


<!-- Date: Fri Jun 6 06:08:05.986 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263158>263158</A>
 Terminated at 2008.06.06 06:05:01
<!-- Author: RunControl -->


<!-- Date: Fri Jun 6 06:09:08.429 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263159>263159</A>
 Activated at 2008.06.06 06:08:30
<!-- Author: RunControl -->


<!-- Date: Fri Jun 6 06:09:09.755 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263159>263159</A>
ACTIVATE: RunSet: AAA_CURRENT  Trigger: PHYSICS_5_01 [7,1088,721] 
<!-- Author: Chang-Seong -->





<!-- Date: Fri Jun 6 06:18:27.581 2008 -->
<table><tr><td><a href=/elog/notebooks/2008/Fri_Jun__6_2008_Owl.061821.ps align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-2755  align='top'></a></td><td>There is a know problem for PPR since May 3. 
The problems are two strips of consecutive deak channels (see plot).</td></tr></table>
<!-- Author: Alberto -->


<!-- Date: Fri Jun 6 06:18:38.351 2008 -->
HBM alarm in iFix. Acknowledged a pop-up in the ifix server window. waiting to see if it clears.
<!-- Author: Simone :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263159' target=_top>263159</a>) -->
<!-- Comment: Fri Jun  6 06;25;45 comment by...Simone -->
cleared
<!-- Date: Fri Jun 6 06:24:59.490 2008 -->
<center><a href='/cgi/elog/elog.pl?nb=2008&action=view&page=-2759&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2008&action=view&page=-2759' width='300'></a><a href='/cgi/elog/elog.pl?nb=2008&action=view&page=-2758&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2008&action=view&page=-2758' width='300'></a></center>
<!-- Author: Simone -->
<!-- Comment: Fri Jun  6 06;25;15 comment by...Simone -->
shot setup 6200






<!-- Date: Fri Jun 6 06:34:38.976 2008 -->
Done TO for crate b0pcal04. HRR not helping. After the first one, 'task suspended' error on that crate. Shepherding the crate solves the problem.
<!-- Author: Simone :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263159' target=_top>263159</a>) -->



<!-- Date: Fri Jun 6 06:37:12.502 2008 -->
FrontEnd Error: SVT_TRACKING_00. Auto + manual HRR fixes it.
<!-- Author: Simone :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263159' target=_top>263159</a>) -->



<!-- Date: Fri Jun 6 06:41:49.516 2008 -->
<table><tr><td><a href=/elog/notebooks/2008/Fri_Jun__6_2008_Owl.064142.ps align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-2761  align='top'></a></td><td>Large spikes in XFT finder. Marked good because threshold for bad is more than twice the baseline. However, this is very close to twice the baseline.</td></tr></table>
<!-- Author: Alberto -->


<!-- Date: Fri Jun 6 06:43:26.954 2008 -->
 
I have checked the outputs on 1RR12D-1 and is within range. 
 
I also looked at the PSM Computer on 3rd floor and everything is green. Although the entire 1st floor remains silenced. 
  
   
<!-- Author: Roberto Davila :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263159' target=_top>263159</a>) -->


<!-- Date: Fri Jun 6 06:46:40.081 2008 -->
<table><tr><td><a href=/elog/notebooks/2008/Fri_Jun__6_2008_Owl.064635.ps align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-2763  align='top'></a></td><td>Error from XFT finder. This seems to be an know problem since November 2007.</td></tr></table>
<!-- Author: Alberto -->



<!-- Date: Fri Jun 6 06:52:06.304 2008 -->
Done TO for crate b0pcal05. Stuck in halt transition. First shepherd failed (heap corruption). Secondo shepherd was succesful.
<!-- Author: Simone :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263159' target=_top>263159</a>) -->

<!-- Date: Fri Jun 6 07:01:47.568 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263158>263158</A>
si: <tt>Silicon OFF</tt>
<!-- Author: Consumer Operator( Alberto Annovi) -->


<!-- Date: Fri Jun 6 07:01:48.695 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263158>263158</A>
central: <tt>COT plots are RED but satisfy 'no gaps>8 cells" condition.<br>
The occupancy is just out of the green band.</tt>
<!-- Author: Consumer Operator( Alberto Annovi) -->


<!-- Date: Fri Jun 6 07:01:49.016 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263158>263158</A>
pr: <tt>Still know problem (kwon since May 3) with dead channels on PPRTowerCanvas.</tt>
<!-- Author: Consumer Operator( Alberto Annovi) -->


<!-- Date: Fri Jun 6 07:01:49.868 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263158>263158</A>
general: <tt>Check list is good. Found several know problems, some of them are known since months ago....</tt>
<!-- Author: Consumer Operator( Alberto Annovi) -->


<!-- Date: Fri Jun 6 07:01:50.420 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263158>263158</A>
muons: <tt>BMUCOCellsEast is now better than reference, i.e. less dead channels.<br>
BMUCOCellsWest is good, but shows a single new dead (or low stat.) channel.</tt>
<!-- Author: Consumer Operator( Alberto Annovi) -->


<!-- Date: Fri Jun 6 07:01:51.908 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263158>263158</A>
trigger: <tt>Problem known since Apr 4 in IMU HAT Canvas 1.4.7.1. </tt>
<!-- Author: Consumer Operator( Alberto Annovi) -->



<!-- Date: Fri Jun 6 07:02:21.577 2008 -->
SVX, ISL iFix alarms. PS#15 spontaneous switch off. SAR did NOT start the recovery. It says it's enabled but the monitor windows (the one like imon) is all white. So I powered back on (black columns on imon.. so it was really "just" a spontaneous switch off) manually the wedges L00/W0, L00/W4, L00/W5, ISLB3W2, ISLB5W4. 
 This fixes everything.  
<!-- Author: Simone :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263159' target=_top>263159</a>) -->


<!-- Date: Fri Jun 6 07:08:05.523 2008 -->
<table><tr><td><a href=/elog/notebooks/2008/Fri_Jun__6_2008_Owl.070801.ps align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-2765  align='top'></a></td><td>Problems with L00 for run 263159. I do not see this in the CDF fix list.</td></tr></table>
<!-- Author: Alberto -->
<!-- Comment: Fri Jun  6 07;13;48 comment by...Alberto -->
Now it is fine (see comment from Simone coming soon)


<!-- Comment: Fri Jun  6 07;21;47 comment by...Simone -->
maybe this was an hint of the problem we had some minutes later (the PS#15 switch off). But i hope that some expoert can comment on this.

<!-- Comment: Fri Jun  6 07;23;32 comment by...Simone -->
The plot was made at 656. Exactly when the problem occurred.

<!-- Comment: Fri Jun  6 09;04;21 comment by...Sasha Pronko (Silicon MPC) -->
Thanks for posting this plot into e-log. If you see anything like this for an extended period of time ~5-10 min. Please ask ACE to do an HRR. If it doesn't go away after an HRR, please page Silicon. 

<!-- Comment: Fri Jun  6 09;06;40 comment by...sasha -->
in respond to Simone's comment. Yes, it could have been related to a problem with PS15.
<!-- Date: Fri Jun 6 07:31:51.731 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263159>263159</A>
general: <tt>Check list is good. All problems are known problems.</tt>
<!-- Author: Consumer Operator( Alberto Annovi) -->




<!-- Date: Fri Jun 6 07:34:39.907 2008 -->
<table><tr><td><a href=/elog/notebooks/2008/Fri_Jun__6_2008_Owl.073430.ps align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-2767  align='top'></a></td><td>Plot shows a know problem with XTRP. Problem is known since last december and the reference plot must be updated.</td></tr></table>
<!-- Author: Alberto -->
<!-- Comment: Fri Jun  6 08;41;59 comment by...Olga Norniella -->
Today (if we have enough time between stores) we (Kevin and I) are going to try to fix this problem. I changed the data board (#11) on Wednesday but the problem is still there. If we don't solve the problem I will change the reference plot.
<!-- Date: Fri Jun 6 07:46:37.463 2008 -->
<table><tr><td><a href=/elog/notebooks/2008/Fri_Jun__6_2008_Owl.074634.ps align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-2769  align='top'></a></td><td>XTC slot efficiency slightly below expected value, but well above the safety limit of 80%. I'm sending email to experts as indicated in the instructions.</td></tr></table>
<!-- Author: Alberto -->
<!-- Comment: Fri Jun  6 09;06;21 comment by...Heather -->
Thanks for posting.  Data are fine.  This TDC-XTC2 was swapped out yesterday.  I have updated the CotT0 timing offsets we use for the monitoring (from calibration run 263149), so it should look better in the next run when they are picked up.  At this point, I think it is a monitoring issue, not a hardware issue.
<!-- Date: Fri Jun 6 07:56:31.557 2008 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>PhysicsTable</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L3Accepts</TH>
<TH>DelivLumi nb<SUP>-1</SUP></TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN=263156 target=_runSummary>263156</A> x403F4</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_5_01 [7,1088,721]</FONT></TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>05:18:22</TD>
<TD ALIGN=CENTER>00:00:00</TD>
<TD ALIGN=RIGHT></TD>               <TD ALIGN=RIGHT></TD>
<TD ALIGN=RIGHT></TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>0</TD></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN=263158 target=_runSummary>263158</A> x403F6</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_5_01 [7,1088,721]</FONT></TD>
<TD ALIGN=CENTER>05:33:02</TD>
<TD ALIGN=CENTER>06:05:01</TD>
<TD ALIGN=CENTER>00:27:18</TD>
<TD ALIGN=RIGHT>     256,758</TD>   <TD ALIGN=RIGHT>    508.16</TD>
<TD ALIGN=RIGHT>    427.27</TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN=263159 target=_runSummary>263159</A> x403F7</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_5_01 [7,1088,721]</FONT></TD>
<TD ALIGN=CENTER>06:08:30</TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>01:28:35</TD>
<TD ALIGN=RIGHT>     738,351</TD>   <TD ALIGN=RIGHT>  1,336.13</TD>
<TD ALIGN=RIGHT>  1,114.68</TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>07:55:01</TD>
<TD ALIGN=CENTER>01:55:53</TD>
<TD ALIGN=RIGHT>       995,109</TD> <TD ALIGN=RIGHT>  1,844.29</TD>
<TD ALIGN=RIGHT>  1,541.95</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->


<!-- Date: Fri Jun 6 07:58:20.263 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263158>263158</A>
RUNSTATUS: <PRE>Marked Bad, explanation: not powered
SVX 
ISL 
L00 
SVT 
</PRE>
<!-- Author: cdfscico -->



<!-- Date: Fri Jun 6 08:00:19.200 2008 -->
Started shift in shot setup.  Held there for an hour or so.   
 
D0 needed 1hr access to check silicon problem.  Did not fix - they need an 8hr access to address. 
 
Restarted shot setup around 3am. 
 
Major problem with silicon at start of run: cannot connect to RT server.....   eventually decide to page DAQ then decide to restart RT server AND restart svxinter 
 
Summary 
======= 
Delivered luminosity:	2249.1 nb-1 
Acquired luminosity:	1588.4 nb-1 
Efficiency:	70.6  
<!-- Author: richard -->


<!-- Date: Fri Jun 6 08:05:31.661 2008 -->
<table><tr><td><a href=/elog/notebooks/2008/Fri_Jun__6_2008_Day.080527.ps align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-2771  align='top'></a></td><td>IMU HAT Canvas is red due to a known problem.</td></tr></table>
<!-- Author: Alberto -->


<!-- Date: Fri Jun 6 08:19:55.764 2008 -->
<pre>Stuck Cellid S/B2/W4/L4/C0-6 .  
Auto HRR fixed</pre> <font color=red><cite>(entry outside this shift's time range )</cite></font>
<!-- Author: Will -->



<!-- Date: Fri Jun 6 08:42:55.638 2008 -->
<pre>Done Timeout detected in crate CLC_02. 
During Auto HRR b0pcal05 did not allow transition, shepherded + HRR fixed</pre> <font color=red><cite>(entry outside this shift's time range )</cite></font>
<!-- Author: Will -->