<!-- Title: 07:57:10  Mon Jun 12 2000 -->
<!-- Crewchief: R. Erbacher -->
<!-- Op1: &nbsp; -->
<!-- Op2: &nbsp; -->
<!-- Op3: &nbsp; -->
<!-- Op4: &nbsp; -->
<!-- Op5: &nbsp; -->
<!-- Op6: &nbsp; -->
<!-- Notes: -->
<h1> New Week </h1>


<!-- Date: Mon Jun 12 09:07:24 2000 -->
<HTML> 
<HEAD> 
   <META NAME="GENERATOR" CONTENT="Mozilla/4.05C-SGI [en] (X11; I; IRIX 6.5 IP32) [Netscape]"> 
</HEAD> 
<BODY> 
 
<H1> 
Weekly Maintenance Check:</H1> 
 
<UL> 
<LI> 
<B>b0ccal00: </B>Peds ok, Calibration problems.&nbsp; All zeroes except 
20,21 give strange values shown below.&nbsp; Reboot and all <B>ok</B>.</LI><br> 
<pre> 
* ADMEM=19 QIE=17 Failure 13: Slope too low                  (    0.0000) ( 0) ( 0) * 
* ADMEM=19 QIE=18 Failure 13: Slope too low                  (    0.0000) ( 0) ( 0) * 
* ADMEM=19 QIE=19 Failure 13: Slope too low                  (    0.0000) ( 0) ( 0) * 
* ADMEM=20 QIE= 0 Failure 13: Slope too low                  (-36973140302885660000000000000.0000) ( 0) ( 0) * 
* ADMEM=20 QIE= 1 Failure 13: Slope too low                  (-36973140302885660000000000000.0000) ( 0) ( 0) * 
</pre> 
 
<LI> 
<B>b0ccal01: </B>Floating point error.&nbsp; Rebooted crate and tried again.&nbsp; 
<B>Ok</B>.</LI> 
 
<LI> 
<B>b0ccal02: </B>All<B> ok</B>.&nbsp; Since the SW arch is not used in 
integration tests, these crates don't seem to need rebooting.&nbsp; What 
does integration test do to cause them all to need attention?</LI> 
 
<LI> 
<B>b0ccal03: </B>Ped problems: (18,9) gives zeroes (not rotation problem).&nbsp; 
Same as last week: transition board short. <B>Ok</B>.</LI> 
 
<LI> 
<B>b0ccal04: Ok</B>.</LI> 
 
<LI> 
<B>b0ccal05: Ok.</B></LI> 
 
<LI> 
<B>b0ccal06: </B>Peds all zero.&nbsp; Rebooted. Slot 18 ADMEM had blown 
F6 fuse.&nbsp; Replaced. <I>Same fuse blew again! </I>Replaced. It blew 
<I>again</I>. Pulled transition board, no effect.&nbsp; Put slot 17 admem 
in and fuse is ok.&nbsp; Put slot 18 admem into 17 and fuse blows. Bad 
admem (#50)... replaced it.</LI> 
 
<LI> 
<B>b0ccal07: </B>Same behavior as ccal00.&nbsp; Rebooted. <B>Ok</B> after 
that.</LI> 
 
<LI> 
<B>b0ccal08-09: </B>Off for cabling.</LI> 
 
<LI> 
<B>b0ccal10: </B>Peds all zero.&nbsp; Rebooted. <B>Ok</B> after that.</LI> 
 
<LI> 
<B>b0ccal11: </B>Short in transition board (18,18).&nbsp; Error (below) 
when calibrate by hand.&nbsp; Rebooted. <B>Ok</B> after that.</LI><br> 
<pre> 
Calibration started. 
NULL = 0 
Nevent = 20 Npoint = 20 Total Cards = 6 QIE = 0 to 19. 
...1 
val = 65535 200 
</pre> 
 
<LI> 
<B>b0ccal12: </B>Peds all zero.&nbsp; Rebooted.&nbsp; Ckcrate failed due 
to calibration warning (below) related to high peds. Downloaded anyway.&nbsp; 
<B>Ok</B>.</LI><br> 
<pre> 
*******************   Start of WARNING messages   ************** VALUE    CAP RANGE** 
* ADMEM=21 QIE= 7 Warning 11: Sigma(offset) big              (    1.8807) ( 0) ( 1) * 
*******************   End of WARNING messages   ************************************* 
</pre> 
 
<LI> 
<B>b0ccal13: </B>Peds all zero.&nbsp; Rebooted.&nbsp; <B>Ok</B> after that.</LI> 
 
<LI> 
<B>b0ccal14-15: </B>Off for cabling.</LI> 
</UL> 
 
</BODY> 
</HTML>
<!-- Author: R. Erbacher, Ph. Gris -->


<!-- Date: Mon Jun 12 09:24:32 2000 -->
<ul><li><b>b0ccal06</b></li></ul> 
<br> 
Configured the board and downloaded Et lookup table.  Peds ok.   
Calibration  
<br>Here is the new <b>Crate Map</b>: <br> 
<pre> 
 In Slot2   IDPROM = 0021 001 TRACER__V2.1A           
 In Slot7   IDPROM = 0.0............................. 
 In Slot8   IDPROM = 0.0............................. 
 In Slot16   IDPROM = 0089 003 ADMEM_V4.0          470 
 In Slot17   IDPROM = 0139 003 ADMEM_V4.0          470 
 In Slot18   IDPROM = 0162 003 ADMEM_V4.0          470 
 In Slot19   IDPROM = 0115 003 ADMEM_V4.0          470 
 In Slot20   IDPROM = 0146 003 ADMEM_V4.0          470 
 In Slot21   IDPROM = 0074 003 ADMEM_V4.0          470 
 
 
 
QIE ID table: 
QIE    ADMEM: 16      17      18      19      20      21      
  0          5134    4370    2943     804    3627    2882 
  1           433    4420    1165     792    1237    2821 
  2          6573    4178    3781    4233    3361    2403 
  3          1055    4237    5196    4121    5290    4323 
  4           994    4103    6199     959    5420    4214 
  5          4263    4293    6300     815    3418    4480 
  6          4310    4198    6581     875    3310    4281 
  7           758    4507    6000    4225    5158    4388 
  8           791    4445    5918     716    3098    4240 
  9           998    4322    1886    4090    5946    4124 
 10          4426    4335    1742    4442    6318    4493 
 11           969    4369    1168    4430    3353    4065 
 12           693    4333    1407    4194    4615    4252 
 13           359    4360    1441    4114    4748     933 
 14          5773    4070    2963    5776    4663     948 
 15          6156    4253    1425    5783    5188     920 
 16          1147    4054    3016    4050    3288    4264 
 17          4391    4189    1440    4112    4785    4512 
 18          1140     913    4149    4089    3232     846 
 19          2285     901    1206    5788    5238    4340  
<!-- Author: Ph Gris, R. Erbacher -->


<!-- Date: Mon Jun 12 11:55:08 2000 -->
<HTML> 
<HEAD> 
   <META NAME="GENERATOR" CONTENT="Mozilla/4.05C-SGI [en] (X11; I; IRIX 6.5 IP32) [Netscape]"> 
</HEAD> 
<BODY> 
 
<UL> 
<LI> 
<B>b0ccal03:</B></LI> 
</UL> 
Pedestal problem for (18,9) : known transition board problem 
 
<P>Calibration problem: 
<BR>&nbsp;<B></B> 
 
<P>*******************&nbsp;&nbsp; Start of WARNING messages&nbsp;&nbsp; 
************** VALUE&nbsp;&nbsp;&nbsp; CAP RANGE** 
<BR>* ADMEM=20 QIE=14 Warning 11: Sigma(offset) big&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(&nbsp;&nbsp;&nbsp; 1.7060) ( 0) ( 1) * 
<BR>*******************&nbsp;&nbsp; End of WARNING messages&nbsp;&nbsp; 
************************************* 
 
<P>*******************&nbsp;&nbsp; Start of FAILURE messages&nbsp;&nbsp; 
************** VALUE&nbsp;&nbsp;&nbsp; CAP RANGE** 
<BR>* ADMEM=21 QIE=17 Failure 13: Slope too low&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(&nbsp;&nbsp;&nbsp; 0.3659) ( 1) ( 2) * 
<BR>*******************&nbsp;&nbsp; End of FAILURE messages&nbsp;&nbsp; 
************************************* 
 
<P>Problem in (20,14) is intermittent.&nbsp; Downloaded anyway.&nbsp; After 
downloading the FRAM <B>3 times </B>(slot 16 had 3 channels complaining) 
all is <B>ok</B>. 
</BODY> 
</HTML>
<!-- Author: Ph. Gris, R. Erbacher -->


<!-- Date: Mon Jun 12 12:30:12 2000 -->
<HTML> 
<HEAD> 
   <META NAME="GENERATOR" CONTENT="Mozilla/4.05C-SGI [en] (X11; I; IRIX 6.5 IP32) [Netscape]"> 
</HEAD> 
<BODY> 
 
<H1> 
Weekly Maintenance for Endwall</H1> 
 
<UL> 
<LI> 
<B>b0ecal00:&nbsp; </B>Peds ok.&nbsp; Calibration problems listed below.&nbsp; 
Downloaded FRAM anyway and all <B>ok</B>.</LI> 
 
<LI> 
<B>b0ecal01:&nbsp; </B>Ped problems and calibration problems, listed below.&nbsp; 
Downloaded FRAM. <B>Ok.</B></LI> 
 
<LI> 
<B>b0ecal02: </B>Failed calibrations (below).&nbsp; Pulled trans board 
for 16.&nbsp; Try again.&nbsp; Problem with (16,3) goes away, but nothing 
was seen in the pedestals!!&nbsp; Rebooted, rechecked.&nbsp; All <B>ok</B>.</LI> 
 
<LI> 
<B>b0ecal03: </B>Strangest symptoms yet (see below).&nbsp; Rebooted. Calibration 
problem.&nbsp; Downloaded anyway.&nbsp; <B>Ok</B>.</LI> 
 
<LI> 
<B>b0ecal04:&nbsp; Ok </B>after problems (below) and calibrated/downloaded 
by hand.</LI> 
 
<LI> 
<B>b0ecal05:&nbsp; </B>Calibration problems below.&nbsp; Downloaded anyways, 
and all <B>ok</B>.</LI> 
 
<LI> 
<B>b0ecal06:&nbsp; </B>Problems with peds and <I>different</I> problems 
with calibration (see below).&nbsp; Downloaded anyways.&nbsp; <B>Ok</B>.</LI> 
 
<LI> 
<B>b0ecal07:&nbsp; </B>Had to reboot. Failed calibrations listed below.&nbsp; 
Downloaded anyways and all <B>ok</B>.</LI> 
</UL> 
 
 
<P><B><U>Details of Problems with Endwall this week:</U></B> 
<UL> 
<LI> 
<B>b0ecal00:</B></LI> Calibration prob: <br> 
<pre> 
 
*******************   Start of FAILURE messages   ************** VALUE    CAP RANGE** 
* ADMEM=20 QIE= 6 Failure 14: Sigma(offset) too big          (   13.6844) ( 0) ( 0) * 
*******************   End of FAILURE messages   ************************************* 
</pre> 
 
<LI> 
<B>b0ecal01:</B></LI> Ped Prob:<br> 
<pre> 
*******************   Start of WARNING messages   ************** VALUE    CAP RANGE** 
* ADMEM=21 QIE= 4 Warning  4: Raw pedestal RMS big           (   12.1868) ( 3) ( 0) * 
*******************   End of WARNING messages   ************************************* 
</pre> 
Calib Prob:<br> 
<pre> 
*******************   Start of FAILURE messages   ************** VALUE    CAP RANGE** 
* ADMEM=20 QIE= 3 Failure 14: Sigma(offset) too big          (    9.4799) ( 0) ( 0) * 
*******************   End of FAILURE messages   ************************************* 
</pre> 
 
 
<LI> 
<B>b0ecal02:</B></LI> Ped problems:<br> 
<pre> 
*******************   Start of FAILURE messages   ************** VALUE    CAP RANGE** 
* ADMEM=18 QIE= 0 Failure  6: Raw pedestal RMS TOO big       (   20.6404) ( 3) ( 0) * 
* ADMEM=18 QIE=11 Failure  6: Raw pedestal RMS TOO big       (   20.2229) ( 3) ( 0) * 
*******************   End of FAILURE messages   ************************************* 
</pre> 
Calibration probs:<br> 
<pre> 
*******************   Start of FAILURE messages   ************** VALUE    CAP RANGE** 
* ADMEM=16 QIE= 3 Failure 14: Sigma(offset) too big          (   13.2113) ( 0) ( 1) * 
*******************   End of FAILURE messages   ************************************* 
</pre> 
 
<LI> 
<B>b0ecal03:</B></LI> Strange ped probs before reboot:<br> 
<pre> 
 
*******************   Start of FAILURE messages   ************** VALUE    CAP RANGE** 
* ADMEM=16 QIE= 0 Failure 10: No fitting                     (    0.0000) ( 0) ( 0) * 
* ADMEM=16 QIE= 1 Failure 14: Sigma(offset) too big          (  212.3000) ( 0) ( 0) * 
* ADMEM=16 QIE= 2 Failure 14: Sigma(offset) too big          (  204.8500) ( 0) ( 0) * 
* ADMEM=16 QIE= 3 Failure 10: No fitting                     (    0.0000) ( 0) ( 0) * 
* ADMEM=16 QIE= 4 Failure 14: Sigma(offset) too big          (  252.5000) ( 0) ( 0) * 
* ADMEM=16 QIE= 5 Failure 10: No fitting                     (    0.0000) ( 0) ( 0) * 
* ADMEM=16 QIE= 6 Failure 14: Sigma(offset) too big          (  275.9500) ( 0) ( 0) * 
* ADMEM=16 QIE= 7 Failure 14: Sigma(offset) too big          (  266.8000) ( 0) ( 0) * 
... 
</pre> 
Calibration probs:<br> 
<pre> 
*******************   Start of FAILURE messages   ************** VALUE    CAP RANGE** 
* ADMEM=19 QIE= 6 Failure 14: Sigma(offset) too big          (   12.1319) ( 0) ( 0) * 
*******************   End of FAILURE messages   ************************************* 
</pre> 
 
<LI> 
<B>b0ecal04:</B></LI> Ped problems: <br> 
<pre> 
*******************   Start of FAILURE messages   ************** VALUE    CAP RANGE** 
* ADMEM=21 QIE= 6 Failure  6: Raw pedestal RMS TOO big       (   21.2380) ( 3) ( 0) * 
*******************   End of FAILURE messages   ************************************* 
</pre> 
Calibration problems:<br> 
<pre> 
 
*******************   Start of FAILURE messages   ************** VALUE    CAP RANGE** 
* ADMEM=16 QIE= 7 Failure 14: Sigma(offset) too big          (   18.2317) ( 0) ( 0) * 
* ADMEM=20 QIE= 8 Failure 10: No fitting                     (    0.0000) ( 0) ( 0) * 
*******************   End of FAILURE messages   ************************************* 
</pre> 
 
<LI> 
<B>b0ecal05:</B></LI> Calibration Problem in diff channel than ped warning:<br> 
<pre> 
*******************   Start of FAILURE messages   ************** VALUE    CAP RANGE** 
* ADMEM=18 QIE= 2 Failure 14: Sigma(offset) too big          (   51.8103) ( 0) ( 1) * 
*******************   End of FAILURE messages   ************************************* 
</pre> 
 
<LI> 
<B>b0ecal06:</B></LI> Calibration failures: <br> 
<pre>*******************   Start of FAILURE messages   ************** VALUE    CAP RANGE** 
* ADMEM=19 QIE= 4 Failure 14: Sigma(offset) too big          (    9.0099) ( 0) ( 0) * 
* ADMEM=19 QIE=10 Failure 14: Sigma(offset) too big          (    6.7960) ( 0) ( 0) * 
* ADMEM=21 QIE= 6 Failure 14: Sigma(offset) too big          (    9.9262) ( 0) ( 0) * 
*******************   End of FAILURE messages   ************************************* 
</pre> 
 
 
<LI> 
<B>b0ecal07:</B></LI> Calibration failures: <br> 
<pre> 
*******************   Start of FAILURE messages   ************** VALUE    CAP RANGE** 
* ADMEM=17 QIE= 6 Failure 14: Sigma(offset) too big          (    5.9936) ( 0) ( 0) * 
* ADMEM=20 QIE= 3 Failure 14: Sigma(offset) too big          (   10.7093) ( 0) ( 0) * 
*******************   End of FAILURE messages   ************************************* 
</pre> 
 
</UL> 
&nbsp; 
<BR>&nbsp; 
</BODY> 
</HTML>
<!-- Author: R. Erbacher, Ph. Gris -->


<!-- Date: Mon Jun 12 14:51:05 2000 -->
<br><b>Grounding, b0ecal00:</b> 
<br>Problems appear to be in the TRACER only.  Measure 7 ohms between HV cable shield and crate with transition boards plugged in.  Open (infinite resistance) when the transition boards are unplugged. 
<br> 
<br><b>Grounding, b0ecal01:</b> 
<br>Problems appear to be in the TRACER only.  Measure 6-7 ohms between HV cable shield and crate with transition boards plugged in.  Open (infinite resistance) when the transition boards are unplugged. 
<br> 
<br><b>Grounding, b0ecal03:</b> 
<br>Find problems in 3 wedges: 
<ul> 
 <li><i>Wedge-1:</i>  Measure 4.3 ohms between HV cable shield and rack ground.  Also find broken HV connector.</li> 
 <li><i>Wedge-2:</i>  Measure 4.3 ohms between HV cable shield and rack ground.</li> 
 <li><i>Wedge-4:</i>  Measure 2.8 ohms between HV cable shield and rack ground.</li> 
</ul>  
<!-- Author: N. Luzhetskiy, R. J. Tesarek -->


<!-- Date: Mon Jun 12 16:24:28 2000 -->
<HTML> 
<HEAD> 
   <META NAME="GENERATOR" CONTENT="Mozilla/4.05C-SGI [en] (X11; I; IRIX 6.5 IP32) [Netscape]"> 
</HEAD> 
<BODY> 
 
<UL> 
<LI> 
<B>SE arch, Calcard:</B></LI> 
</UL> 
Took calcards from b0ccal05 and b0ccal03 and put them in b0ccal13 and b0ccal12 
respectively. Tried some LED calibrations. Found that non of the transition 
boards for the calcard for b0ccal12 and b0ccal13 had fuses (15&nbsp; - 
1 A -&nbsp; fuses in total for each transition board). Put some in one 
board but run out of fuses. Thus took the calcard transition board of b0ccal05 
and put it in b0ccal12. Took an LED run: 
<BR>/mdc1/csl/00/A20000612152916.test 
<BR>All the concerning boxes (ie wedges 12,13,14,15,16,17 East) seem to 
flash. 
<BR>Checked, by flashing wedges one by one, that the cabling was ok (ie 
no transition boards swapping). 
<BR>Tried a Xenon calibration. 15E does not drive any current. 17E drives 
too much current (more than 0.8 mA which is our current limit). Saw a signal 
in the ADMEMs for 12E, 13E and 17E(??! The HV had tripped due to over current!!!). 
<BR>After investigations, found that 15E had no trigger nor HV cable plugged 
in. 14E had no trigger cable. 
</BODY> 
</HTML>
<!-- Author: Ph.Gris -->


<!-- Date: Mon Jun 12 18:22:58 2000 -->
<br><b>Calibration, b0ecal03:</b> 
<br>I ran a quick study to try and understand the problem with the endwall channels which don't calibrate correctly.  Below are some plots of RAW data taken with various DAC settings.  Figure 1 shows a good channel (slot=19,chan=5,range=0).  Figure 2 shows a problem channel (slot=19,chan=6,range=0).   
<br>  
<!-- Author: R.J. Tesarek -->


<!-- Date: Mon Jun 12 18:29:13 2000 -->
<table><tr><td><a href=/elog/notebooks/wedge/Mon_Jun_12_2000_182905.ps align=top><img src=/elog/gifs/ghost64.gif  width='56' align='top'></td><td>Figure 1:  Raw calibration data for crate=b0ecal03,  slot=19, chan=5.  Each plot consists of 1000 events in the given DAC setting.  This channel calibrates correctly.</a></td></tr></table>
<!-- Author: R.J. Tesare -->


<!-- Date: Mon Jun 12 18:37:18 2000 -->
<table><tr><td><a href=/elog/notebooks/wedge/Mon_Jun_12_2000_183712.ps align=top><img src=/elog/gifs/ghost64.gif  width='56' align='top'></td><td>Figure 2: Raw calibration data for crate=b0ecal03, slot=19, chan=5. Each plot consists of 1000 events in the given DAC setting. Note the wide "peak" distributions for all DAC  values except for the "pedestal" (DAC=0).  In addition,  note the "grass" in the distribution which appears to have a constant width.  This channel fails calibration in the  lower ranges.</a></td></tr></table>
<!-- Author: R.J. Tesarek -->


<!-- Date: Tue Jun 13 12:08:27 2000 -->
<HTML> 
<HEAD> 
   <META NAME="GENERATOR" CONTENT="Mozilla/4.05C-SGI [en] (X11; I; IRIX 6.5 IP32) [Netscape]"> 
</HEAD> 
<BODY> 
 
<UL> 
<LI> 
<B>SE arch, Calcard:</B></LI> 
</UL> 
Put 1 A fuses on the top two transition boards for the Calcard (b0ccal10 
and b0ccal11). Thanks to Fred who provided the fuses. 
<BR>Managed to flash with LEDs 3/4 of the SE arch (ie b0ccal10, b0ccal11 
and b0ccal13). One calcard is still missing. 
<BR>Did not manage to save the data: the CSL is still down (b0dau31 was 
down this morning for disks shuffling.... 
<BR>Checked the cabling for the top two crates. No obvious swapping observed. 
<BR>&nbsp; 
</BODY> 
</HTML>
<!-- Author: Ph.Gris -->


<!-- Date: Tue Jun 13 14:04:00 2000 -->
<HTML> 
<HEAD> 
   <META NAME="GENERATOR" CONTENT="Mozilla/4.05C-SGI [en] (X11; I; IRIX 6.5 IP32) [Netscape]"> 
</HEAD> 
<BODY> 
 
<UL> 
<LI> 
<B>SE arch, Calcard:</B></LI> 
</UL> 
Xenon calibration (b0ccal10, 11 and 13). Still no softevb to save the data!!!!!!!! 
<BR>Wedges that do not seem to flash: 
<BR>6E, 10E, 16E. 
<BR>17E still flashes even if it is off upstairs!!!!! 
<BR>After investigations: 
<BR>6E: no trigger cable at all. 
<BR>10E: no trigger cable plugged. 
<BR>16E: found that in fact the HV cable from wedge 17E was swapped with 
that of 16E. Thus what I thought was 16E (from the xenon HV point of view) 
was in fact 17E. Everything is thus compatible: 16E seems to drive too 
much current (bad xenon box). 
<BR>1:25pm.....the software event builder is back! We just lost 5 hours!!! 
Champagne!!! 
<BR>Made a Xenon run with b0ccal10, 11 and 12. The results are saved in: 
<BR>/mdc1/csl/00/A20000613132257.test 
<BR>It seems from all that that we have only one bad xenon box (16E). Nonetheless 
found 6E in overcurrent... 
 
<P>LED run (b0ccal10, b0ccal11, b0ccal12): 
<BR>/mdc1/csl/00/A20000613135109.test 
<BR>All the corresponding boxes seem to flash. 
<BR>&nbsp; 
<BR>&nbsp; 
<BR>&nbsp; 
</BODY> 
</HTML>
<!-- Author: Ph.Gris -->


<!-- Date: Tue Jun 13 21:53:35 2000 -->
<ul><li><b>b0ecal03</b></li></ul> 
<br> I logged onto the crate and reconfigured the admems to be 
endwall admems, since one board was replaced and still had plug status.  
<!-- Author: R. Erbacher -->


<!-- Date: Wed Jun 14 11:10:57 2000 -->
<ul><li><b>b0ecal02</b></li></ul> 
<br> 
We found F3 fuses blown on the boards in slots 17-21 and an F1 fuse (10A) on the one in slot 17 blown.  Replaced.  However,  
we had to replace the admem in slot 17 (#313) because on replacing the fuse, the lead broke off and became lodged inside 
of the holder.  Cannot remove without removing the holder, we think.  Here is the new <b>Crate Map</b> for b0ecal02: 
<br> 
<pre> 
 In Slot2   IDPROM = 0136 001 TRACER__V2.1A           
 In Slot16   IDPROM = 0072 003 ADMEM_V4.0          470 
 In Slot17   IDPROM = 0242 003 ADMEM_V4.0          470 
 In Slot18   IDPROM = 0363 003 ADMEM_V4.0          460 
 In Slot19   IDPROM = 0346 003 ADMEM_V4.0          460 
 In Slot20   IDPROM = 0042 003 ADMEM_V4.0          470 
 In Slot21   IDPROM = 0331 003 ADMEM_V4.0          460 
 
 
 
QIE ID table: 
QIE    ADMEM: 16      17      18      19      20      21      
  0          2580    5812    2591    2177    2300    3574 
  1           848    5047    2205     419    2463    4477 
  2          4722    5022    2201    2450    2327     580 
  3          2208    5040    4905     477    2497     294 
  4          2431    5105    2280     245    2279     668 
  5          2130    5073    2379     317    2169    3936 
  6          2449    5837     961    2135    2472     528 
  7           272    5168    2458    2340    2585     539 
  8          2179    5001    2255    2224    2269    5945 
  9          2121    5889    2499    2483    2560    3674 
 10          2531    5050    2329    2341    3311    3594 
 11          2192    5025    2349    2265    2369     596 
 12             0       0       0       0       4       0 
 13           512       0       0       0       0       0 
 14             0       0       0       0       0       0 
 15             0       0       0       0   16384       0 
 16             0       0       0       0       0       0 
 17             0       0       0       0       0       0 
 18             0       0       0       0       0       0 
 19             0       0       0       0       0       0  
<!-- Author: R. Erbacher, D. Cambridge, N. Lutzhetskiy -->


<!-- Date: Wed Jun 14 11:21:02 2000 -->
<HTML> 
<HEAD> 
   <META NAME="GENERATOR" CONTENT="Mozilla/4.05C-SGI [en] (X11; I; IRIX 6.5 IP32) [Netscape]"> 
</HEAD> 
<BODY> 
 
<UL> 
<LI> 
<B>SE arch, Calcard:</B></LI> 
</UL> 
LED calibrations on b0ccal10, 11 and 13: 
<BR>/mdc1/csl/00/A20000614092802.test (TS=51) 
<BR>/mdc1/csl/00/A20000614093847.test (same timing) 
<BR>/mdc1/csl/00/A20000614094656.test(timing adjusted for b0ccal11: 4-4-4 
as trigger delays) 
<BR>/mdc1/csl/00/A20000614095924.test(timing adjusted for b0ccal11:5-4-5) 
<BR>/mdc1/csl/00/A20000614100847.test(same but TS=52) 
<BR>/mdc1/csl/00/A20000614100927.test(same but TS=51) 
<BR>/mdc1/csl/00/A20000614101450.test(same but TS=50) 
<BR>/mdc1/csl/00/A20000614102135.test(TS=51, trigger delays 11-11-11 for 
b0ccal11) 
<BR>/mdc1/csl/00/A20000614102732.test(same but trigger delays 12-12-12 
for b0ccal11) ----> timing seems ok. 
<BR>/mdc1/csl/00/A20000614104804.test(TS=51, trigger delays 5-5-5 and pipelength=37 
for b0ccal11) 
<BR>/mdc1/csl/00/A20000614105328.test(same but pipelength=39 for b0ccal11) 
--> timing seems ok. 
 
<P>LED calibrations with b0ccal10, 11 and 12: 
<BR>/mdc1/csl/00/A20000614110243.test: parameters: TS=51; trigger delays: 
5-5-5 (b0ccal11) 3-2-3 (b0ccal12); pipelength: (39,4) for b0ccal11; (38,4) 
for b0ccal12 
<BR>/mdc1/csl/00/A20000614111020.test (same as previously but 3-3-2 forb0ccal12) 
<BR>/mdc1/csl/00/A20000614111551.test (same but 3-3-3 for b0ccal02) 
 
<P>For all these previous measurements, b0ccal10 and b0ccal13 parameters 
were 3-3-3 for trigger delays and (38,4) for ADMEM parameters. 
<BR>&nbsp; 
<BR>&nbsp; 
</BODY> 
</HTML>
<!-- Author: Ph.Gris -->


<!-- Date: Wed Jun 14 11:29:12 2000 -->
<pre><ul><li><b>b0ecal02</b></li></ul> 
<br> 
After replacing the admem fuses and the board in slot 17 (having reconfigured and downloaded the Et lookup table), we ran a ckcrate on it and found a blown F4 in (19,0).  Replaced the CAFE card.  Here is the new <b>Crate Map</b>:<br> 
<pre> 
 In Slot2   IDPROM = 0136 001 TRACER__V2.1A           
 In Slot16   IDPROM = 0072 003 ADMEM_V4.0          470 
 In Slot17   IDPROM = 0242 003 ADMEM_V4.0          470 
 In Slot18   IDPROM = 0363 003 ADMEM_V4.0          460 
 In Slot19   IDPROM = 0346 003 ADMEM_V4.0          460 
 In Slot20   IDPROM = 0042 003 ADMEM_V4.0          470 
 In Slot21   IDPROM = 0331 003 ADMEM_V4.0          460 
 
QIE ID table: 
QIE    ADMEM: 16      17      18      19      20      21      
  0          2580    5812    2591    1169    2300    3574 
  1           848    5047    2205     419    2463    4477 
  2          4722    5022    2201    2450    2327     580 
  3          2208    5040    4905     477    2497     294 
  4          2431    5105    2280     245    2279     668 
  5          2130    5073    2379     317    2169    3936 
  6          2449    5837     961    2135    2472     528 
  7           272    5168    2458    2340    2585     539 
  8          2179    5001    2255    2224    2269    5945 
  9          2121    5889    2499    2483    2560    3674 
 10          2531    5050    2329    2341    3311    3594 
 11          2192    5025    2349    2265    2369     596 
<pre> 
</pre>
<!-- Author: D. Cambridge, R. Erbacher, N. Luzhetskiy -->


<!-- Date: Wed Jun 14 14:16:15 2000 -->
<HTML> 
<HEAD> 
   <META NAME="GENERATOR" CONTENT="Mozilla/4.05C-SGI [en] (X11; I; IRIX 6.5 IP32) [Netscape]"> 
</HEAD> 
<BODY> 
 
<UL> 
<LI> 
<B>Se arch, Calcard:</B></LI> 
</UL> 
Xenon calibrations and timing adjustments: 
<BR>b0ccal10, 11 and 12. 
 
<P>/mdc1/csl/00/A20000614113919.test with the following set of parameters: 
 
<P>TS=51 
 
<P>&nbsp;&nbsp; crate&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; trigger 
delays&nbsp;&nbsp;&nbsp;&nbsp; ADMEM config. 
 
<P>b0ccal10&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
3-2-3&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(38,4) 
<BR>b0ccal11&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
3-2-3&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(38,4) 
<BR>b0ccal12&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
3-2-3&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(38,4) 
 
<P>/mdc1/csl/00/A20000614123904.test with: 
 
<P>TS=51 
 
<P>crate&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; trigger delays&nbsp;&nbsp;&nbsp;&nbsp; 
ADMEM config. 
 
<P>b0ccal10&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
4-4-4&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(38,4) --> timing ok. 
<BR>b0ccal11&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
4-4-4&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(38,4) --> timing not ok. 
<BR>b0ccal12&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
3-3-3&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(38,4)--> timing ok. 
 
<P>Wedges 8 and 11 did not flash for this run. 
 
<P>/mdc1/csl/00/A20000614124900.test with: 
 
<P>TS=51 
 
<P>crate&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; trigger delays&nbsp;&nbsp;&nbsp;&nbsp; 
ADMEM config. 
 
<P>b0ccal10&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
4-4-4&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(38,4) --> timing ok. 
<BR>b0ccal11&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
4-4-4&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(39,4)--> timing not ok. 
<BR>b0ccal12&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
3-3-3&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(38,4)--> timing ok 
 
<P>Wedges 6, 8 and 11 did not flash. 
 
<P>/mdc1/csl/00/A20000614125614.test with 
 
<P>TS=51 
 
<P>crate&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; trigger delays&nbsp;&nbsp;&nbsp;&nbsp; 
ADMEM config. 
 
<P>b0ccal10&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
4-4-4&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(38,4) --> timing ok. 
<BR>b0ccal11&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
5-5-5&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(39,4)--> timing not ok. 
<BR>b0ccal12&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
3-3-3&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(38,4)--> timing ok 
 
<P>Wedges 6, 8 and 11 did not flash. 
 
<P>/mdc1/csl/00/A20000614131036.test with: 
 
<P>TS=51 
 
<P>crate&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; trigger delays&nbsp;&nbsp;&nbsp;&nbsp; 
ADMEM config. 
 
<P>b0ccal10&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
4-4-4&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(38,4) --> timing ok. 
<BR>b0ccal11&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
7-7-7&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(38,4)--> timing not ok. 
<BR>b0ccal12&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
3-3-3&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(38,4)--> timing ok 
 
<P>Wedges 6, 8 and 11 did not flash. 
 
<P>Found that wedges 6,8 and 11 had tripped. 
<BR>An interesting feature for 6: as the run is going on, the driven current 
is bigger and bigger up to the trip limit. Nothing special seen for 8 and 
11. 
 
<P>/mdc1/csl/00/A20000614140950.test with: 
 
<P>TS=51 
 
<P>crate&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; trigger delays&nbsp;&nbsp;&nbsp;&nbsp; 
ADMEM config. 
 
<P>b0ccal10&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
4-4-4&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(38,4) --> timing ok. 
<BR>b0ccal11&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
3-3-3&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(40,4) -->&nbsp; timing&nbsp; ok. 
<BR>b0ccal12&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
3-3-3&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(38,4) -->&nbsp; timing ok 
<BR>&nbsp; 
<BR>&nbsp; 
</BODY> 
</HTML>
<!-- Author: Ph.Gris -->


<!-- Date: Wed Jun 14 16:24:18 2000 -->
<HTML> 
<HEAD> 
   <META NAME="GENERATOR" CONTENT="Mozilla/4.05C-SGI [en] (X11; I; IRIX 6.5 IP32) [Netscape]"> 
</HEAD> 
<BODY> 
 
<UL> 
<LI> 
<B>SE arch, Calcard:</B></LI> 
</UL> 
Xenon calibrations on b0ccal10, 11 and 13: 
 
<P>/mdc1/csl/00/A20000614161631.test with the parameters: 
 
<P>TS=51 
 
<P>&nbsp;&nbsp;&nbsp; crate&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; trigger delays&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
ADMEM parameters 
<BR>b0ccal10&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 4-4-4&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(38,4) 
<BR>b0ccal11&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 3-3-3&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(40,4) 
<BR>b0ccal13&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 3-2-3&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(38,4) --> timing ok. 
 
<P>The only thing is that the xenon box on wedge 16 is dead. This gives 
us 2 bad boxes (in addition to wedge 6 which drives too much current). 
<BR>&nbsp; 
</BODY> 
</HTML>
<!-- Author: Ph.Gris -->


<!-- Date: Wed Jun 14 17:00:41 2000 -->
<br><b>blown fuse, b0ccal06:</b> 
<br>Discovered that ADMEM slot=18 (id=162) had a blown f2 fuse 
(+5V digital).  Replaced fuse and installed into crate.  Smelled smoke and discovered that the digital lights (power LED and ready LED) were not on.  Removed the ADMEM and discovered that capacitor C350 on the back of the ADMEM had fried.  Swapped out 
ADMEM for a new one.  On closer inspection, the PC board is warped.  
<p>Below is the new crate map. 
<pre> 
 In Slot2   IDPROM = 0021 001 TRACER__V2.1A           
 
 In Slot7   IDPROM = 0.0............................. 
 In Slot8   IDPROM = 0.0...8.................S....... 
 
 In Slot16   IDPROM = 0089 003 ADMEM_V4.0          470 
 In Slot17   IDPROM = 0139 003 ADMEM_V4.0          470 
 In Slot18   IDPROM = 0124 003 ADMEM_V4.0          470 
 In Slot19   IDPROM = 0115 003 ADMEM_V4.0          470 
 In Slot20   IDPROM = 0146 003 ADMEM_V4.0          470 
 In Slot21   IDPROM = 0074 003 ADMEM_V4.0          470 
 
QIE ID table: 
QIE    ADMEM: 16      17      18      19      20      21      
  0          5134    4370    2077     804    3627    2882 
  1           433    4420    5861     792    1237    2821 
  2          6573    4178    6021    4233    3361    2403 
  3          1055    4237    6180    4121    5290    4323 
  4           994    4103    6083     959    5420    4214 
  5          4263    4293    5823     815    3418    4480 
  6          4310    4198    2078     875    3310    4281 
  7           758    4507    6071    4225    5158    4388 
  8           791    4445    5171     716    3098    4240 
  9           998    4322    5091    4090    5946    4124 
 10          4426    4335    6133    4442    6318    4493 
 11           969    4369    6162    4430    3353    4065 
 12           693    4333    1818    4194    4615    4252 
 13           359    4360    6160    4114    4748     933 
 14          5773    4070    1827    5776    4663     948 
 15          6156    4253    6472    5783    5188     920 
 16          1147    4054    4268    4050    3288    4264 
 17          4391    4189    6223    4112    4785    4512 
 18          1140     913    2090    4089    3232     846 
 19          2285     901    6295    5788    5238    4340 
</pre>
<!-- Author: R.J. Tesarek, C. Johnson -->


<!-- Date: Wed Jun 14 21:46:27 2000 -->
<br><b>Calibration problems in b0ecal03:</b> 
Revisit the problems observed earlier in b0ecal03, slot=19, chan=6.  First, noted that the pedestal width for this channel  
approximately 100 ADMEM counts (this feature is new today).  Then try the following tests: 
<ul> 
  <li>Normal conditions ==><i> reproduce problem observed earlier</i></li> 
  <li>Unplug transition board in slot 19 ==><i> problem vanishes and the channels appears to calibrate normally.</i></li> 
  <li>Turn off HV for upper Pisa box (HV to offending PMT) ==> <i>problem persists</i></li> 
  <li>HV on, swap CAFE cards for channels 6 <--> 7 ==> <i>problem vanishes for both channels</i></li> 
  <li>Swap CAFE cards back ==><i> problem re-appears exactly as before</i></li> 
  <li>Replace CAFE card ==><i> problem vanishes</i></li> 
</ul> 
<b>Conclusion: </b>  The problem observed with this channel appears to be a combined problem with the CAFE card and the specific ADMEM channel.  For tonight, I've replaced the CAFE 
card in this channel, but we should probably swap the CAFE card back into the same ADMEM channel, reproduce the effect and then 
return the ADMEM to Lee and Charlie or Lou to debug the problem more fully. 
<p>I will work on a test to identify this kind of problem and add this to our usual test procedure in ckcrate.xpt. 
<br>
<!-- Author: R.J. Tesarek -->


<!-- Date: Thu Jun 15 09:43:31 2000 -->
<!doctype html public "-//w3c//dtd html 4.0 transitional//en"> 
<html> 
<head> 
   <meta name="GENERATOR" content="Mozilla/4.5 [en] (X11; I; IRIX64 6.2 IP25) [Netscape]"> 
</head> 
<body> 
 
<ul> 
<li> 
<b>SE arch, Calcard:</b></li> 
</ul> 
LED calibrations for b0ccal10, 11 and 13: 
<br>/mdc1/csl/00/A20000615082901.test with the following parameters: 
<p>TS=51 
<p>&nbsp;&nbsp;&nbsp;&nbsp; crate&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; trigger 
delays&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; ADMEM parameters 
<p>&nbsp;b0ccal10&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 3-3-3&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(38,4) 
<br>&nbsp;b0ccal11&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5-5-5&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(39,4) 
<br>&nbsp;b0ccal13&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 3-3-3&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
(38,4) 
</body> 
</html>
<!-- Author: Ph.Gris -->


<!-- Date: Thu Jun 15 14:08:24 2000 -->
<br><b>NW EndWall (crates b0ecal00, b0ecal03):</b> 
<br>Replaced TRACERs in both endwall crates for models with the connectors disconnected electrically from the front panels.  This 
appears to have worked correctly for b0ecal00 using Nikolai's modified TRACER, but not for the TRACER to replace that in b0ecal03 (modified by Lee Scott).  The later TRACER (S/N=165) exhibited the following symptoms: 
<pre> 
 ADMEM  QIE  CAP #EVENTS AVE_RANGE  MEAN   RMS  
   16    0    0    1500   0.0        0.0   0.0 
   16    0    1     no events 
   16    0    2     no events 
   16    0    3     500   0.0      193.6   9.2 
   16    1    0    1500   0.0        0.0   0.0 
   16    1    1     no events 
   16    1    2     no events 
   16    1    3     500   0.0      172.0  10.7 
</pre> 
This condition did not exist in the TRACER it was to replace. 
<br>below are the new crate maps for the crates: 
<br><b>b0ecal00:</b> 
<pre> 
 In Slot2   IDPROM = 0090 001 TRACER__V2.1A           
 
 In Slot16   IDPROM = 0207 003 ADMEM_V4.0          470 
 In Slot17   IDPROM = 0090 003 ADMEM_V4.0          470 
 In Slot18   IDPROM = 0111 003 ADMEM_V4.0          470 
 In Slot19   IDPROM = 0068 003 ADMEM_V4.0          470 
 In Slot20   IDPROM = 0128 003 ADMEM_V4.0          470 
 In Slot21   IDPROM = 0121 003 ADMEM_V4.0          470 
 
QIE ID table: 
QIE    ADMEM: 16      17      18      19      20      21      
  0          3706    2139    2202     311    2223    5141 
  1           254    2154    2405    3614    2259    2283 
  2           630    2415    2333    3622    2362    2495 
  3           503    2444     483     444    2370    4805 
  4           509     472    2180    2411    2501    6352 
  5           436    2193     348     292    2452    2217 
  6           241     306     219     269    2481    2292 
  7          3885    2150    2323     321    2476    2423 
  8           582    2253     455     312    2186    2496 
  9           621     293    2302     431    2537    2256 
 10           460    2361    2335     307    2284    2376 
 11           422    2498     392     264    2445    2500 
 12             0       0       0       0      11       0 
 13             0       0       0       0   61440       0 
 14             0       0       0       0   32768       0 
 15             0       0       0       0       0       0 
 16             0       0       0       0       0     768 
 17             0       0       0       0       0       0 
 18             2       0       0       0   24582       0 
 19             0       0       0       0    8192       0 
</pre> 
<br><b>b0ecal03:</b> 
<pre> 
 In Slot2   IDPROM = 0083 001 TRACER__V2.1A           
 
 In Slot16   IDPROM = 0312 003 ADMEM_V4.0          470 
 In Slot17   IDPROM = 0315 003 ADMEM_V4.0          470 
 In Slot18   IDPROM = 0176 003 ADMEM_V4.0          470 
 In Slot19   IDPROM = 0311 003 ADMEM_V4.0          460 
 In Slot20   IDPROM = 0318 003 ADMEM_V4.0          470 
 In Slot21   IDPROM = 0345 003 ADMEM_V4.0          460 
 
QIE ID table: 
QIE    ADMEM: 16      17      18      19      20      21      
  0          3734     222    3449     349    4184    1642 
  1          3847     597    3214     634    1743    6059 
  2           542     660    3416     243    3593    5063 
  3           497     508    6089     446     610    5858 
  4          3910    3700    5265    2330    3689    3691 
  5           499    3642    3471    2162    3587    6520 
  6           530     356    3305    4841    3798    3790 
  7          3823     516    4629    2486    3880     507 
  8           417    3623    3238    3899     470    3718 
  9          3597    3654    5200    3904     585    4868 
 10          1099     289    4710    4028    3983     650 
 11           624     569    5281    3931    3635    3705 
 12             0       0       0       0       0       0 
 13             0       0       0       0       0       0 
 14             0       0       0       0       0       0 
 15             0       0       0       0       0       0 
 16             0       0       0       0       0       0 
 17             0       0       0       0       2       0 
 18             0       0       0       0       0       0 
 19             0       0       0       0       0       0 
</pre>
<!-- Author: R.J. Tesarek -->


<!-- Date: Thu Jun 15 14:16:37 2000 -->
<pre><br><b>S.E. EndWall:</b> 
<br>One of the technicians drove a lift into the Pisa box (HV distribution) for wedges 12-15.  In order to avoid HV accidents, I've disconnected the HV for the entire SE EndWall. 
<br></pre>
<!-- Author: R.J. Tesarek -->


<!-- Date: Thu Jun 15 15:29:56 2000 -->
<ul><b>Cal Crate Reboots</b></ul> 
<br>During last night's integration tests I was able to reproduce the problem in which the calibrations fail (are left in a strange state) until we reboot the crate by doing a run using runcontrol 
and then issuing an abort.  Jim Patrick and I looked at the processor to see if we could determine what was wrong.  We couldn't find anything.  The next plan is to try to run and then exit gracefully and not gracefully to see what happens.  Also, I would like to see if beginning another run using runcontrol resets the crate so that it calibrates properly from runcontrol, or if it is necessary to reboot from there as well.
<!-- Author: R. Erbacher -->


<!-- Date: Fri Jun 16 14:11:29 2000 -->
<HTML> 
<HEAD> 
   <META NAME="GENERATOR" CONTENT="Mozilla/4.05C-SGI [en] (X11; I; IRIX 6.5 IP32) [Netscape]"> 
</HEAD> 
<BODY> 
 
<UL> 
<LI> 
<B>SE arch, Calcard:</B></LI> 
</UL> 
Made one EM calibration run (b0ccal10, 11 and 13): 
<BR>/b0dau32/csl/00/A20000616082913.test 
 
<P>Wanted to calibrate the hadron calorimeter. Could not set correctly 
the HV (Run I values). After two hours of investigations: found a hadron 
HV cable plugged in an EM slot (bottom box). 
<BR>Managed to start a hadron run (HV set). Then had problems with wedges 
15 and 12. Wanted to investigate: switched on the lights under the detector. 
The arch tripped at the same time. Took one hour to recover. 
</BODY> 
</HTML>
<!-- Author: The Calcard Team -->


<!-- Date: Fri Jun 16 18:13:10 2000 -->
<br><b>b0ccal13:</b> 
<br>Discovered ADMEMs in slots 17-21 had blown F3 (+5V analog) fuses.  Fuses replaced.  On checking this crate, discovered that ADMEM slot=16, chan=0-6 have blown F4 fuses.  <i>These channels have not been replaced.  We should do this first thing Monday morning during checkout.</i> 
<br> 
<br><b>b0ccal11:</b> 
<br> Discovered that ADMEMs in slots 18-21 had blown F3 (+5V analog) fuses.  <i>No action taken.  Save this work for Monday.</i> 
<br>
<!-- Author: R.J. Tesarek, W. Stuermer -->
<!-- Comment: Fri Jun 16 18;15;02 comment by...R.J. Tesarek -->
On further checking in b0ccal11, discovered that only slots 18-20 have F3 problems.  However, channels 0,1 in ADMEM slot=16 have to be replaced due to an F4 problem on the CAFE card.


<!-- Date: Tue Jun 20 13:51:35 2000 -->
<HTML> 
<HEAD> 
   <META NAME="GENERATOR" CONTENT="Mozilla/4.05C-SGI [en] (X11; I; IRIX 6.5 IP32) [Netscape]"> 
</HEAD> 
<BODY> 
 
<UL> 
<LI> 
<B>Se arch, Calcard:</B></LI> 
</UL> 
HAD calibration (b0ccal10, 11 and 13): 
<BR>/mdc1/csl/00/A20000620111215.test 
<BR>Adjusted HV: 
<BR>/mdc1/csl/00/A20000620124606.test 
</BODY> 
</HTML>
<!-- Author: The Calcard Team -->
<!-- Comment: Tue Jun 20 13;55;06 comment by...Ph.Gris -->
This message was put here by error.