<!-- Title: Owl shift. Mon May  7, 2001 -->
<!-- SciCo: Shin-Shan Yu-->
<!-- DAQAce: Shin-Shan Yu -->
<!-- MonAce: Shin-Shan Yu -->
<!-- CO: Shin-Shan Yu -->
<!-- OpManager: &nbsp; -->
<!-- Notes: -->
<pre>I will do calibrations first and then move on to DAQ test.</pre>


<!-- Date: Mon May 7 00:26:09 2001 -->
<pre>CEM,CHA,WHA, GAM HV lost connection to PC.</pre>
<!-- Author: Eiko -->


<!-- Date: Mon May 7 00:30:03 2001 -->
<pre>When I switched the run type from DAQ test to QIECalibration, 
crates didn't send acknowledgements back for a long time. 
Exited and restart rc, now it's OK.</pre>
<!-- Author: Eiko -->


<!-- Date: Mon May 7 01:30:49 2001 -->
<pre>Calibrations: all in partition 1. 
 
QIE: 113979, in database, no failure 10 
SMQIE: 113980, not in the DB. 
TOF: 113981, in DB.  
  Checked new error -> wcal03 ADMEM=7 fail25: Self Calib Fail 
LED: 113982, in DB 
FlashPed, 113983, not in DB, took another run,still not in the DB 
XEF: 113985, not in DB 
CLC QIE: 113986 done with the configuration file ACE_QIE_NOBEAM, 
         in DB 
COT: 113987, in DB 
Muon: according to elog on May 1st, swap imu00 with imu01.  
      imu01 slot 14,15 18 dropped out, did a bus reset.  
      slot 14 and slot 18 came back, but not slot 15. 
      Removed slot 15 from the crateset but there's a done
      timeout. Did a calibration without imu crates. 
      run 113991 
</pre>
<!-- Author: Eiko -->
<!-- Comment: Mon May  7 01;43;11 comment by...Eiko -->
<pre>Restarted rc and did FlashPed, XEF, SMQIE calibration again.
The calibration data still didn't go into the database.</pre>


<!-- Comment: Mon May  7 02;57;37 comment by...Eiko -->
<pre>DBANA result:
TOF looks OK compared with run 112769.
LED: all channels dead, possible no CEM HV, didn't check when
     taking calibration
QIE: looks OK compared with run 111406
CLC QIE: The peak of SlopeErr/Slope shows up at different 
    channel. Please take a look in DBANA folder.
</pre>
<!-- Date: Mon May 7 02:12:16 2001 -->
<pre>Trying to add back ccal05, ccal07 and pcal04,  
but once I activated the run, ccal05 gave a done  
timeout. The error file says 
 
(MLE) b0ccal05:Messenger:2:02:15 AM->Runtime Error 2, Event 1: 
 SMXR in slot 3 is not responding, header 0x0x3cfc385 
(MLE) b0ccal05:Messenger:2:02:15 AM->Runtime Error 4, Event 1:  
SMXR in slot 4 is not responding, header 0x0x3dfe485 
(RC) Activate -> ACTIVE 
 
 Before I started the run, I did a VISIONdemo mapping of ccal05, 
 the header of slot 3,4,5 were 0xff0011, but after the run,  
 slot 3 became 0x817e0011. ?? 
 
 Removed ccal05 and now run is going well.  
 Note: no l2cal crates, their power is off.</pre>
<!-- Author: Eiko :: (run <a href='/cgi-bin/runSummaryRun.cgi?113998' target=_top>113998</a>) -->
<!-- Comment: Mon May  7 02;23;49 comment by...Eiko -->
L1A rate~92 Hz.
Trigger table:COSMICS_RUN_TABLE_1[3,111]



<!-- Comment: Mon May  7 03;04;19 comment by...Eiko -->
Why is the event size going to CSL zero?


<!-- Comment: Mon May  7 04;58;51 comment by...Eiko -->
<pre>A done timeout from b0cot03 right at the moment when I
opened another run control window and tried to do a
muon calibration with imu01 only. HRR and run is going.</pre>
<!-- Comment: Mon May  7 05;01;50 comment by...Eiko -->
This run is in partition 1.
<!-- Comment: Mon May  7 05;18;45 comment by...Eiko -->
Ended the run to do LED calibration again and include more L3
subfarms.
<!-- Date: Mon May 7 03:12:01 2001 -->
<p> 
I went downstairs and noted that the BLMs are all uncabled, which I assume was done during the opening of the plugs.  Since I believe having cables lying around with 2000 V on them is not such a swell idea, I unplugged the BLM chassis (containing the HV supply) in the control room.</p>
<!-- Author: Andy Hocker -->





<!-- Date: Mon May 7 05:42:58 2001 -->
<pre>1.Did a muon calibration again with only cmx00,01 and cmp00.  
Run # = 114003 
 
2.Did a LED calibration, run#=114004, CEM HV is on, but DBANA 
still shows that all the channels are dead !! 
 
</pre>
<!-- Author: Eiko -->


<!-- Date: Mon May 7 05:55:52 2001 -->
<pre>Include all the L3 subfarms except 15. 
Run configuration same as 113998. 
L1A rate=93 Hz, mainly 2_JET_1GEV after prescale.</pre>
<!-- Author: Eiko :: (run <a href='/cgi-bin/runSummaryRun.cgi?114006' target=_top>114006</a>) -->
<!-- Comment: Mon May  7 06;25;38 comment by...Eiko -->
Around 190K L1As, states of L3 converter b0l3c09, processor b0l3067-74 all became OLD. Not recovered.


<!-- Date: Mon May 7 07:28:22 2001 -->
<pre>Stopped last DAQ run and wanted to do a CLC calibration with 
consumer CLCAMPLCALIB2 but failed. 
 
Started a new DAQ run with all L3 initially, however, L3 
display showed that as RC state proceeded, L3 nodes were 
not cleaned up. The display also didn't reflect a change 
when I cleaned up L3 subfarms. Started a new DAQ run with 
only 4 L3 subfarms. b0l3104 was not initialized. 
</pre>
<!-- Author: Eiko :: (run <a href='/cgi-bin/runSummaryRun.cgi?114009' target=_top>114009</a>) -->
<!-- Comment: Mon May  7 07;47;22 comment by...Eiko -->
<pre>Ended this run. Busy timeout from ccal06,07,13,14,15.
HRR didn't let the run keep moving. 
The error logger says,

 b0l3gate1.fnal.gov:main:7:37:58 AM->Host b0eb15.fnal.gov, task tRec_0
SCPU-P0-E-TracerEventId: Event 100122, crate 16, channel 5 has either bad Tracer ID or bad markers around Tracer word.
 Tracer word event ID is 0, should be 10.
Tracer word = 0x00000010 (I think).
Channel data = 0x000000af 0xe60a0016 0x57484144 0x00080006 0x00000000 0x00000049 0x00000001 0x098c0030 ...
... 0x00000030 0x00000000 0x00000034 0x00000000 0x00020020 0x00010011 0x00000010 0x00000000
(MLE) b0l3gate1.fnal.gov:main:7:37:58 AM->Host b0eb15.fnal.gov, task tRec_0
SCPU-P0-E-TracerEventId: Event 100122, crate 23, channel 6 has either bad Tracer ID or bad markers around Tracer word.
 Tracer word event ID is 10, should be 10.
Tracer word = 0x000a0017 (I think).
Channel data = 0x00000020 0x00080011 0x00080000 0x00080004 0x00000000 0x00000000 0x00000000 0x000c0030 ...
... 0x00000020 0x00000000 0x00000001 0x00000000 0x00280022 0x00150005 0x000a0017 0x00000000
(MLE) b0l3gate1.fnal.gov:main:7:37:58 AM->Host b0eb15.fnal.gov, task tRec_0
SCPU-P0-E-TracerEventId: Event 100122, crate 16, channel 7 has either bad Tracer ID or bad markers around Tracer word.
 Tracer word event ID is 10, should be 10.
Tracer word = 0x000a0010 (I think).
Channel data = 0x00000025 0x00020018 0x00400000 0x00000008 0x00000000 0x00000000 0x00000000 0x00080033 ...
... 0x00000030 0x00000000 0x00000031 0x00000000 0x00280020 0x01050011 0x000a0010 0x00000000
(MLE) b0l3gate1.fnal.gov:main:7:37:58 AM->Host b0eb15.fnal.gov, task tRec_0
SCPU-P0-E-VrbHeader: Dump of header words for event 100122 from VRB in slot 14:
0x00000000 0x000019c8 0x00190019 0x02bc02bc 0x02bc02bc 0x02bc02bc 0x02bc02bc 0x03a40000
37'58" 5 crate/s: b0ccal06(6), b0ccal07(7), b0ccal13(13), b0ccal14(14), b0ccal15(15),  busy.[RXPT]</pre>


<!-- Date: Mon May 7 07:53:26 2001 -->
<br><b>ACE_CALORIMETER_MINIMAL Configuration</b> 
<br>Modified the above DAQ configuration to include <b>ONLY</b> 
the calorimeter crates without L2 calorimeter trigger.  This is 
a temporary measure until the L2 calorimeter trigger modules 
(DCAS) return and are confirmed to be working correctly. 
<br>
<!-- Author: R.J. Tesarek -->


<!-- Date: Mon May 7 07:59:59 2001 -->
<font color=red>Shift Summary: </font><pre>1. all the calibrations done.  
   LED: all the channels dead 
   XEF, FlashPed, SMQIE didn't write to the database. 
   
2. Did several DAQ tests, ccal05 was not included in the run 
   because it failed to recognize header words of ShowerMax 
 
3. Muon calibration reveled the problem of IMU01, 
   TDC in slot15 didn't respond. </pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   113979-114010
Delivered Luminosity   0  pb-1
Acquired Luminosity    0  pb-1
Efficiency             200
   
</td></tr></table></pre>
<!-- Author: Eiko -->