<!-- Title: Day shift. Wed Aug 21, 2002 -->
<!-- SciCo: Lina Galtieri -->
<!-- DAQAce: Kathy Cobic/Karen Gibs -->
<!-- MonAce: Max Goncharov/Jason -->
<!-- CO: Andrew Martin -->
<!-- OpManager: Matthew Jones -->
<!-- Notes: -->
<pre>Shot setup expected to start at 6 pm. Finish work by noon. Do calibrations and excercise the system after noon</pre>

<!-- Date: Wed Aug 21 09:27:34 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150215">150215</A>
 Activated at 2002.08.21 09:27:38
<!-- Author: RunControl -->


<!-- Date: Wed Aug 21 09:27:35 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150215">150215</A>
 Activated at 2002.08.21 09:27:38
<!-- Author: RunControl -->


<!-- Date: Wed Aug 21 09:27:35 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150215">150215</A>
 Activated at 2002.08.21 09:27:38
<!-- Author: RunControl -->


<!-- Date: Wed Aug 21 09:27:35 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150215">150215</A>
 Activated at 2002.08.21 09:27:38
<!-- Author: RunControl -->


<!-- Date: Wed Aug 21 09:46:13 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150215">150215</A>
 Terminated at 2002.08.21 09:40:58
<!-- Author: RunControl -->
<!-- Comment: Wed Aug 21 09;58;28 comment by...k copic -->
Also, Sasha took CLC_00 for testing.
<!-- Comment: Wed Aug 21 09;58;50 comment by...k copic -->
This should have been on the entry below.
<!-- Comment: Wed Aug 21 10;57;14 comment by...SCN, Lester, Andrew -->
Silicon studies done, new PROM seems good.  <a href=
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=120&frame=2&anchor=10:55:11'>
Details</a>. 

<!-- Comment: Wed Aug 21 10;58;13 comment by...SCN -->
Odd, my entry went to the wrong place too.  I guess it has to do
with the fact that the timestamps are exactly the same, so 
the elog picks up the first one with the "correct" timestamp.
<!-- Date: Wed Aug 21 09:46:13 2002 -->
<pre>Steve Nahn, Andrew Martin and Lester Miller are working on DAQ problems. Testing at high rates the Silicon system. 
Stephen Miller working on L2. 
</pre>
<!-- Author: Lina -->




<!-- Date: Wed Aug 21 10:18:58 2002 -->
<pre>George Wyatt has taken the alarm bypass key to adjust O2 sensor 
trip levels.</pre>
<!-- Author: M Jones -->
<!-- Comment: Wed Aug 21 10;30;54 comment by...M Jones -->
<pre>The key has been returned.</pre>


<!-- Date: Wed Aug 21 10:39:36 2002 -->
We notice from MCR elog that CDF should ignor fire alarms. 
Then we notice our firus panel was in alarm (although no 
horns  were blaring.).   We call the fire techs and they 
inform us that someone is working on the sprinkler system 
in the assembly hall.  Maybe they called main control room 
instead of cdf control room.   At any rate, we should ignor 
fire alarms at the moment.
<!-- Author: k.byrum -->
<!-- Comment: Wed Aug 21 15;11;55 comment by...Lina -->
<pre>It turns out that they called 2810, i.e., the wrong number. Steve Hahn talked to them and told them to make sure they call the correct number</pre>
<!-- Date: Wed Aug 21 10:42:37 2002 -->
Kevin Lannon just took cmu/cmp/imu crates for calibration tests. 
estimated 15 - 30 min.
<!-- Author: k.byrum -->
<!-- Comment: Wed Aug 21 10;52;38 comment by...hyunsoo -->
<pre>CMX not CMU.</pre>
<!-- Date: Wed Aug 21 10:44:46 2002 -->
dan cyr just took the muon trigger crate and will get imu from 
kevin to look at muon trigger problems and run calibrations. 
estimated 30-60 min.
<!-- Author: k.byrum -->


<!-- Date: Wed Aug 21 10:45:40 2002 -->
<pre>Fiodar and Camille are discussing a BSU low voltage monitoring 
problem on the top CCU in the south west rack.  One possible 
explanation is that the monitoring was broken during the access 
last night.  The LV is probably okay but the plan is to try to 
confirm this before we take data.</pre>
<!-- Author: M Jones -->





<!-- Date: Wed Aug 21 11:30:10 2002 -->
Sasha is finished with the CLC.
<!-- Author: k copic -->


<!-- Date: Wed Aug 21 11:30:19 2002 -->
looking at bsu/tsu trigger gate timing	
<!-- Author: dan :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150224' target=_top>150224</a>) -->


<!-- Date: Wed Aug 21 11:45:35 2002 -->
Got a message from RC that Icicle died -- it was, in fact, dead, so I restarted it.
<!-- Author: k copic -->


<!-- Date: Wed Aug 21 11:56:15 2002 -->
<pre>Muon Calibration Consumer: 
-------------------------- 
After much trial and error, I was finally able to reproduce 
the situation where the muon calibration consumer crashes the 
calibration CSL.  The situation that produces this crash is 
running the calibration with only a single muon TDC crate and 
interval0 = 10,000.  By changing interval0 to 500,000 or by  
increasing the number of crates to 3 or more (I didn't try it 
with two crates) allows the calibration to run without crashing 
the CSL.  This supports the theory by experts that this is a 
rate related problem.  I'll take this information to the CSL 
experts and try to find a way to make sure that the crashes  
won't happen when the calibration is run by the shift crew. 
 
I'm now done with the muon TDC crates (cmp00, cmx00, cmx01, 
imu00, and imu01.  However, the latter two were given to Dan 
Cyr about an hour ago).</pre>
<!-- Author: K.Lannon -->


<!-- Date: Wed Aug 21 11:58:03 2002 -->
Stephen and Kevin are finished with their tests.
<!-- Author: k copic -->


<!-- Date: Wed Aug 21 11:59:04 2002 -->
I've used cardEditor to change the QDR settings of COT from 0V 
to -1 V for all the superlayers. This will increase the digital 
width about 10%. I will change the value back at the end of this 
week before the access starts.
<!-- Author: Shin-Shan Yu (Eiko) -->


<!-- Date: Wed Aug 21 12:03:05 2002 -->
<pre>Done with L2 firmware optimization studies. I will need to repeat  
some of these next week without beam before making the optimized 
firmware the default. 
 
Note to shift crew: We replaced one of the interface boards 
which should reduce the rate of L2 decision timeouts. I expect 
there will still be a low level of L2 decision timeouts. If 
the decision timeout can't recover with multiple HRR's, then 
please post the minicom error message in the e-log. Ending the 
run and coldstarting should get things going again.</pre>
<!-- Author: Stephen -->


<!-- Date: Wed Aug 21 12:06:13 2002 -->
<pre>We found a miscabling (cable labels were wrong) of two 
optical fibers which go into the muon matchbox from the TSU 
West TDC's.  The swapped cables were 9<->10.  The labels 
were also replaced.</pre>
<!-- Author: Dan Cyr and Camille Ginsburg -->


<!-- Date: Wed Aug 21 12:28:31 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150256">150256</A>
 Activated at 2002.08.21 12:13:29
<!-- Author: RunControl -->


<!-- Date: Wed Aug 21 12:28:31 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150256">150256</A>
ACTIVATE: Cosmics w/o CCAL07, IMU crates, and MUTRIG

<!-- Author: ace 2080 -->


<!-- Date: Wed Aug 21 12:28:31 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150256">150256</A>
 Terminated at 2002.08.21 12:18:04
<!-- Author: RunControl -->


<!-- Date: Wed Aug 21 12:28:31 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150256">150256</A>
TERMINATE: EVVB problem -- would not activate. All checks OK. (?)
<!-- Author: ace 2080 -->


<!-- Date: Wed Aug 21 12:28:31 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150258">150258</A>
 Activated at 2002.08.21 12:23:01
<!-- Author: RunControl -->


<!-- Date: Wed Aug 21 12:28:32 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150258">150258</A>
ACTIVATE: Trying again - no IMU, MUTRIG, CCAL07
<!-- Author: ace 2080 -->


<!-- Date: Wed Aug 21 12:29:03 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150258">150258</A>
 Terminated at 2002.08.21 12:29:02
<!-- Author: RunControl -->


<!-- Date: Wed Aug 21 12:40:58 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150260">150260</A>
 Activated at 2002.08.21 12:40:46
<!-- Author: RunControl -->



<!-- Date: Wed Aug 21 12:45:24 2002 -->
This is what happened after aces switched from partition 2 to partition 0  
 
(RC)   12:21:47 Config -> READY 
(MLE) b0l3gate1.fnal.gov:main:12:23:04 PM->Host b0eb16.fnal.gov, task tCtrl_0 
SCPU-P0-E-WhereIsL3: Partition has no L3 subfarms registered. 
(MLE) b0l3gate1.fnal.gov:main:12:23:04 PM->Host b0eb13.fnal.gov, task tCtrl_0 
SCPU-P0-E-WhereIsL3: Partition has no L3 subfarms registered. 
(MLE) b0l3gate1.fnal.gov:main:12:23:04 PM->Host b0eb14.fnal.gov, task tCtrl_0 
SCPU-P0-E-WhereIsL3: Partition has no L3 subfarms registered. 
 
 
t least one SCPU rejected Define Partition. 
(MLE) b0l3gate1.fnal.gov:main:12:23:34 PM->Host b0eb10.fnal.gov, task tsmHot 
SM-P0-E-ScpuUnresponsive: One or more SCPUs failed to respond to Define Partition. Check SCRAMNet. 
(MLE) b0l3gate1.fnal.gov:main:12:23:39 PM->Host b0eb10.fnal.gov, task tsmHot 
SM-P0-E-ScpuOperationFailed: SCPU b0eb25 sent error code 0x000001ff in reply to Delete Partition. 
(MLE) b0l3gate1.fnal.gov:main:12:23:39 PM->Host b0eb10.fnal.gov, task tsmHot 
SM-P0-E-ScpuOperationFailed: SCPU b0eb24 sent error code 0x000001ff in reply to Delete Partition. 
(MLE) b0l3gate1.fnal.gov:main:12:23:39 PM->Host b0eb10.fnal.gov, task tsmHot 
 
(MLE) b0l3gate1.fnal.gov:main:12:23:39 PM->Host b0eb10.fnal.gov, task tsmHot 
SM-P0-E-ScpuOperationFailed: SCPU b0eb25 sent error code 0x000001ff in reply to Delete Partition. 
(MLE) b0l3gate1.fnal.gov:main:12:23:39 PM->Host b0eb10.fnal.gov, task tsmHot 
 
 
The problem was i think that the partition was not cleanly closed  
so the EVB and Level3 thought that they belong to different partitions  
 
After i cleaned EVB and Level3  
the problem went away 
  
<!-- Author: Arkadiy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150258' target=_top>150258</a>) -->


<!-- Date: Wed Aug 21 12:53:44 2002 -->
<pre>The TSU-SW data look ok, and do not suggest any problem with 
the newly-installed CCU03, which has a LV monitor communication 
problem.</pre>
<!-- Author: Dan Cyr and Camille Ginsburg -->

<!-- Date: Wed Aug 21 12:59:17 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150260">150260</A>
 Terminated at 2002.08.21 12:59:15
<!-- Author: RunControl -->


<!-- Date: Wed Aug 21 13:03:55 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150261">150261</A>
ACTIVATE: CLC QIE calibration
<!-- Author: ace 2080 -->
<!-- Comment: Wed Aug 21 13;17;51 comment by...k copic -->
Several errors in errlog, otherwise OK.
<!-- Date: Wed Aug 21 13:06:31 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150262">150262</A>
ACTIVATE: CAL QIE calibration
<!-- Author: ace 2080 -->
<!-- Comment: Wed Aug 21 13;19;35 comment by...k copic -->
Lots of errors in errlog, otherwise OK.
<!-- Comment: Wed Aug 21 13;20;32 comment by...Andrew Martin -->
<pre>This is the bad channel list from DBAna for the Cal QIE
run.

Run, Wedge, Channel, Cap, Detector, Mean(ref), Mean, RMS(ref), Sigma
150262   2 W   4  1  PHA  208#pm 5  203  0&lt;rms&lt;4   1.47 
150262   3 W  13  2  PHA  186#pm 5  181  0&lt;rms&lt;4   1.30 
150262   3 W  16  0  PHA  186#pm 5  181  0&lt;rms&lt;4   1.27 
150262   3 W  16  2  PHA  185#pm 5  180  0&lt;rms&lt;4   1.29 
150262   5 W   2  1  PHA  192#pm 5  187  0&lt;rms&lt;4   1.42 
150262   5 W   2  2  PHA  197#pm 5  192  0&lt;rms&lt;4   1.28 
150262   5 W  16  0  PHA  215#pm 5  210  0&lt;rms&lt;4   1.36 
150262   6 W  10  0  PHA  188#pm 5  183  0&lt;rms&lt;4   1.41 
150262  19 W   7  2  PHA  179#pm 5  174  0&lt;rms&lt;4   1.33 
150262  22 W  17  3  PHA  200#pm 5  195  0&lt;rms&lt;4   1.38 
150262  14 W  10  3  WHA  179#pm 5  174  0&lt;rms&lt;4   2.91 
150262   2 W   2  0  PEM  180#pm 5  175  0&lt;rms&lt;4   1.49 
150262   2 W   2  1  PEM  181#pm 5  176  0&lt;rms&lt;4   1.46 
150262   2 W  10  0  PEM  191#pm 5  186  0&lt;rms&lt;4   1.47 
150262   2 W  10  1  PEM  189#pm 5  184  0&lt;rms&lt;4   1.49 
150262   2 W  10  2  PEM  191#pm 5  185  0&lt;rms&lt;4   1.45 
150262   2 W  10  3  PEM  197#pm 5  192  0&lt;rms&lt;4   1.37 
150262   3 W   5  3  PEM  188#pm 5  183  0&lt;rms&lt;4   1.39 
150262   6 W   2  2  PEM  189#pm 5  184  0&lt;rms&lt;4   1.20 
150262   6 W   4  0  PEM  178#pm 5  172  0&lt;rms&lt;4   1.49 
150262   6 W   5  2  PEM  174#pm 5  169  0&lt;rms&lt;4   1.49 
150262   6 W   5  3  PEM  178#pm 5  173  0&lt;rms&lt;4   1.50 
150262  19 W   6  2  PEM  193#pm 5  188  0&lt;rms&lt;4   1.34 
150262  19 W  15  2  PEM  174#pm 5  169  0&lt;rms&lt;4   1.42 
150262  22 W  10  2  PEM  178#pm 5  173  0&lt;rms&lt;4   1.42 
150262  23 E  10  0  PEM  183#pm 5  178  0&lt;rms&lt;4   1.39 
150262  12 W   3  1  CEM  176#pm 5  171  0&lt;rms&lt;4   1.60 
Run, Wedge, Channel, Cap, Detector, Mean(ref), Mean, RMS(ref), Sigma
150262  12 W   3  3  CEM  182#pm 5  177  0&lt;rms&lt;4   1.48 
150262   1 E  13  3  CEM  184#pm 5  179  0&lt;rms&lt;4   1.26 
150262   1 E  15  3  CEM  193#pm 5  187  0&lt;rms&lt;4   1.26 
150262   1 E  16  3  CEM  176#pm 5  171  0&lt;rms&lt;4   1.51 
150262   1 E  19  3  CEM  183#pm 5  178  0&lt;rms&lt;4   1.34 
150262   4 E  15  2  CEM  185#pm 5  180  0&lt;rms&lt;4   1.37 
150262  18 E   0  0  CEM  182#pm 5  190  0&lt;rms&lt;4   1.32 
150262  18 E   0  1  CEM  178#pm 5  190  0&lt;rms&lt;4   1.50 
150262  18 E   0  2  CEM  181#pm 5  193  0&lt;rms&lt;4   1.32 
150262  18 E   0  3  CEM  183#pm 5  197  0&lt;rms&lt;4   1.30 
150262  18 E   3  0  CEM  187#pm 5  182  0&lt;rms&lt;4   1.37 
</pre>
<!-- Date: Wed Aug 21 13:08:43 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150263">150263</A>
ACTIVATE: CAL LED calibration
<!-- Author: ace 2080 -->
<!-- Comment: Wed Aug 21 13;20;34 comment by...k copic -->
No errors.
<!-- Date: Wed Aug 21 13:08:43 2002 -->
Returned ccal07 to shift crew (I did some timing measurements for 
showermax). <P> 
Since it wasn't mentioned in the e-log: earlier I was called 
because of rc problems; it turned out that smartsockets licenses 
were exceeded. I "restarted" the rtserver to clean things up.  
<!-- Author: Arnd -->

<!-- Date: Wed Aug 21 13:12:01 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150264">150264</A>
ACTIVATE: BSC calibration
<!-- Author: ace 2080 -->
<!-- Comment: Wed Aug 21 13;21;03 comment by...k copic -->
Some errors in errlog, otherwise OK.
<!-- Date: Wed Aug 21 13:13:58 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150265">150265</A>
ACTIVATE: SMX calibration - CES/CPR/CCR off
<!-- Author: ace 2080 -->
<!-- Comment: Wed Aug 21 13;49;31 comment by...k copic -->
CES/CPR/CCR were tuned back to standby shortly afterwards.
<!-- Date: Wed Aug 21 13:18:40 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150266">150266</A>
ACTIVATE: COT calibration -- beam in studies.
<!-- Author: ace 2080 -->



<!-- Date: Wed Aug 21 13:20:27 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150267">150267</A>
ACTIVATE: Muon calibration - beam in studies.
<!-- Author: ace 2080 -->






<!-- Date: Wed Aug 21 13:25:33 2002 -->
<pre>I'm turning off the TOF high voltage so that we can do 
calibrations and get the detector into the correct configuration 
for shot setup.</pre>
<!-- Author: M Jones -->

<!-- Date: Wed Aug 21 13:31:20 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150268">150268</A>
ACTIVATE: CAL XEF calibration - beam in studies
<!-- Author: ace 2080 -->


<!-- Date: Wed Aug 21 13:36:30 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150269">150269</A>
ACTIVATE: TOF TAC calibration - beam in studies.
<!-- Author: ace 2080 -->


<!-- Date: Wed Aug 21 13:40:21 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150270">150270</A>
ACTIVATE: TOF QIE calibration - beam in studies.
<!-- Author: ace 2080 -->



<!-- Date: Wed Aug 21 13:49:07 2002 -->
<table border=0><td><tr><td bgcolor=#e0e030>I fixed the bug that lately was causing Error Handler (EH) to hang. I've rebuild v1_06_3 and declared it current. When convenient, please close the current EH window and restart EH from the RC Partitioning Menu. As usual, please report any problems in the e-log. Thanks.</td></tr></td></table>
<!-- Author: Donatella T. -->


<!-- Date: Wed Aug 21 13:50:56 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150271">150271</A>
 Activated at 2002.08.21 13:50:44
<!-- Author: RunControl -->


<!-- Date: Wed Aug 21 13:52:15 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150271">150271</A>
ACTIVATE: Rick's Run. No beam in the machine, but not official "quiet time." Plug HV is on standby.
<!-- Author: ace 2080 -->
<!-- Comment: Wed Aug 21 13;55;53 comment by...k copic -->
<pre>L1_SUMET5 &lt;rate&gt; at 111 Hz, MBXING at 1.7 MHz.
Instantaneous SUMET fluctuates sometimes to 200 Hz.</pre>

<!-- Date: Wed Aug 21 13:56:19 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150271">150271</A>
 Terminated at 2002.08.21 13:56:07
<!-- Author: RunControl -->


<!-- Date: Wed Aug 21 13:56:19 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150271">150271</A>
TERMINATE: 35k events.
<!-- Author: ace 2080 -->


<!-- Date: Wed Aug 21 14:06:45 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150272">150272</A>
TERMINATE: Starting new error handler.
<!-- Author: ace 2080 -->


<!-- Date: Wed Aug 21 14:13:41 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150273">150273</A>
 Activated at 2002.08.21 14:13:35
<!-- Author: RunControl -->


<!-- Date: Wed Aug 21 14:14:23 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150273">150273</A>
ACTIVATE: Cosmics with all crates but silicon.
<!-- Author: ace 2080 -->


<!-- Date: Wed Aug 21 14:29:37 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150273">150273</A>
 Terminated at 2002.08.21 14:28:51
<!-- Author: RunControl -->


<!-- Date: Wed Aug 21 14:29:37 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150273">150273</A>
TERMINATE: Ending to test trigger tables.
<!-- Author: ace 2080 -->



<!-- Date: Wed Aug 21 14:31:10 2002 -->
<table border=0><td><tr><td bgcolor=#e0e030>Power supply BSU_SW_1 generates PSM alarms. The monitored voltages on channels #0 and #6 are 25V instead of 14.5V and 11V. The power supply was recently replaced and after the replacement problems appeared. This is probably due to the problems with monitoring box (swapped cables, ground on wrong cable etc). The power supply is masked for the time being</td></tr></td></table>
<!-- Author: M. Wolter -->
<!-- Comment: Wed Aug 21 14;37;43 comment by...Fiodar -->
Check of the Occupancy plots for channels controlled by replaced CCU shows, that counters somehow working (data is present). We need more statistics to make shure that everything is OK.

<!-- Date: Wed Aug 21 14:39:14 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150274">150274</A>
ACTIVATE: Testing Physics_1_02[3,136,313]. no beam.
<!-- Author: ace 2080 -->


<!-- Date: Wed Aug 21 14:51:24 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150274">150274</A>
TERMINATE: Seems OK.
<!-- Author: ace 2080 -->



<!-- Date: Wed Aug 21 15:17:07 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150276">150276</A>
ACTIVATE: Testing COSMICS_NOTRACKS[4,138,292]. no beam.
<!-- Author: ace 2080 -->
<!-- Comment: Wed Aug 21 15;18;54 comment by...k copic -->
There is now beam in the machine. Trigger Table seems fine.
]


<!-- Comment: Wed Aug 21 15;54;03 comment by...k copic -->
<pre>Using the default COSMICS_NOTRACKS configuration, I discovered that the FIB and VRB crates were included in the default configuration. I took them out before starting the run. 

After talking to Gino and Arnd, I changed the default configuration to start up without FIBs or VRBs.</pre>
<!-- Date: Wed Aug 21 15:27:37 2002 -->
Fire techs called to tell us that they are finished with their work. They have a pager and we should use it when they do not inform us and we find out second hand. 
<!-- Author: Lina -->


<!-- Date: Wed Aug 21 15:37:41 2002 -->
MCR called. they will start shot setup at 8 pm. testing the BLT now
<!-- Author: Lina -->


<!-- Date: Wed Aug 21 15:39:34 2002 -->
Run 150277 was a very short run to see if the new MinBias trigger goes through all the transitions. It was successful, but we did not accumulate many events. Trigger= MINIMUM_BIAS_1[7,139,292]  
 
For some reason, it did not show up in the e-log. (??)
<!-- Author: k copic :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=150277' target=_top>150277</a>) -->


<!-- Date: Wed Aug 21 15:55:08 2002 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>15:55:01</TD>                                                       
<TD ALIGN=CENTER>::</TD>                                                
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->