<!-- Title: Owl shift. Tue Jan  8, 2002 -->
<!-- SciCo: Vaia Papadimitriou -->
<!-- DAQAce: Abraham Gallas -->
<!-- MonAce: Hung-Chung Fang -->
<!-- CO: Suen Hou -->
<!-- OpManager: Dave Ambrose -->
<!-- Notes: -->
<pre>Goal is to take about 4 hours of data with trigger table [18,244] 
and after that keep running with [20,245]. 
Between these two trigger tables try to take a silicon D-mode 
calibration </pre>

<!-- Date: Tue Jan 8 00:20:02 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=136441">136441</A>
 Activated at 2002.01.08 00:19:29
<!-- Author: RunControl -->



<!-- Date: Tue Jan 8 00:24:19 2002 -->
<pre>We stopped run# 136440 because the reformatter rejected rate  
 was 54%. Mainly those came from FIB_06_channel6. We tried  
first with HRR and did not work. We stopped the run and we  
went through ColdStart. That seemed to cure the problem. </pre>
<!-- Author: Abraham Gallas :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=136441' target=_top>136441</a>) -->


<!-- Date: Tue Jan 8 00:37:31 2002 -->
<pre>BMU HV trip: W 12,24,48,72,108,120.  Reset.</pre>
<!-- Author: H-C -->
<!-- Comment: Tue Jan  8 00;41;58 comment by...H-C -->
<pre>BMU HV trip: W 24.  Reset.</pre>

<!-- Comment: Tue Jan  8 00;50;28 comment by...H-C -->
<pre>BMU HV trip: W 24.  This wedge ON STANDBY.</pre>


<!-- Date: Tue Jan 8 01:31:38 2002 -->
<pre>Persistent TROUBLE alert from FIRUS console from BLUE FRONT END: 
  "The alarm task on the RED front end 
   is down. Consoles and minis are being 
   picked up by the backup front end. 
   CALL FIRUS SOFTWARE PERSONNEL." 
We waked up Dave Ambrose who directed us to the Cryo people. 
Steve Gordon came up and explained to us that the backup FIRUS front end is now in use and MCR needs to reboot the original one.  In other words, we needn't worry about it.</pre>
<!-- Author: H-C -->


<!-- Date: Tue Jan 8 01:59:27 2002 -->
<pre>BMU HV trip: W 276,12,36,48,72,84,96,108,120. 
iFix was slow so Abraham went downstairs and put these wedges on STANDBY.</pre>
<!-- Author: H-C -->


<!-- Date: Tue Jan 8 02:03:58 2002 -->
<pre>Plug HV monitoring communication problems-- PEM/PHA/PSH gray on iFix.  Let's see if it comes back.</pre>
<!-- Author: H-C -->
<!-- Comment: Tue Jan  8 02;59;14 comment by...H-C -->
<pre>Plug HV monitoring back, but PEM in alarm state due to communication problems-- no alarm is listed in the Diagnostic table.</pre>
<!-- Date: Tue Jan 8 02:12:20 2002 -->
<pre>After doing that ColdStart at the beginning of the shift 
 now the reformatter error rate is around 6%. And seems to  
be quite stable around that value. 
</pre>
<!-- Author: Abraham Gallas :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=136441' target=_top>136441</a>) -->



<!-- Date: Tue Jan 8 04:02:25 2002 -->
<pre>B0 integrated luminosities at start of shift: 
 
    Delivered          307.3/nb 
    Acquired           163.3/nb</pre>
<!-- Author: H-C -->

<!-- Date: Tue Jan 8 04:04:23 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=136441">136441</A>
 Terminated at 2002.01.08 04:03:54
<!-- Author: RunControl -->



<!-- Date: Tue Jan 8 04:07:38 2002 -->
<pre>MCR called and tehy say they will drop the store in a few 
minutes. Apparently the schedule changed on Monday afternoon. 
(We knew the store will be dropped at 8 am) 
They will start with machine studies after that which 
will last for 24 hours. 
We ended the current run and are putting on standby everything 
we need to. </pre>
<!-- Author: vaia -->


<!-- Date: Tue Jan 8 04:32:01 2002 -->
<pre>MCR is still doing some 980Gev orbit smoothing before we terminate beam. 
Since we stopped dara taking, Steve Gordon will ramp down  
the magnet.</pre>
<!-- Author: vaia -->


<!-- Date: Tue Jan 8 04:43:09 2002 -->
<pre>The store has been dropped now. 
It took about 40 minutes to put all necessary HV on standby. 
Problems with bringing down ISL and TOF. 
We are still working with exsperts on bringing down TOF HV.</pre>
<!-- Author: vaia -->


<!-- Date: Tue Jan 8 04:57:23 2002 -->
<pre>Tev store aborted, BA and BB were quenched. Main Tev bus was dumped after the $5E before it quenched. 
Called MCR. They said for an hour it will be quiet time. 
We decide to do quiet time calibrations.</pre>
<!-- Author: vaia -->


<!-- Date: Tue Jan 8 05:00:23 2002 -->
<pre>We had problems turning the silicon to standby.  Andy Foland was paged.  The problem was that B2 of the ISL would not go to standby.  Andy suspected it was because CAENvolts.dat and powerset.dat were not loaded.  We loaded the two files and clicked on the ISL standby button on the PS GUI.  This still failed to put B2 on standby.  Only when L0,1,2 were individually put to standby did the voltages go down.  This gave an alarm in the ISL which went away after acknowledging it.  Strangely enough, the ISL global power button is still ON, although no wedges are powered. 
 
A software protection trip in SB4W1L3 also occured when the silicon was put on standby.  This is the wedge that was entirely (including the port card) powered off.  We set it to standby and the alarm went away.  The SVX power button is still in the ON state, much like the ISL button mentioned above, for some unknown reason.  Despite this, the detector is on standby now.</pre>
<!-- Author: H-C -->


<!-- Date: Tue Jan 8 05:01:17 2002 -->
<pre>We took Cal QIE calibration. run # 136442 . We look at  
the results later. We proceed with the other quiet time calibrations.  
</pre>
<!-- Author: Abraham Gallas :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=136442' target=_top>136442</a>) -->


<!-- Date: Tue Jan 8 05:07:52 2002 -->
<pre>CLC QIE Calibration done. Run number #136443.
     Warning during the calibration:
     Crate CLC_00 is missing card ADMEN_BSC_00 in slot 16
     Crate CLC_00 is missing card ADMEN_BSC_01 in slot 17</pre>
<!-- Author: Abraham Gallas :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=136443' target=_top>136443</a>) -->


<!-- Date: Tue Jan 8 05:13:55 2002 -->
<pre>We could not power off the TOF from the control room.  IFIX was not communicating with the TOF PC.  We went downstairs and saw that the PC was frozen.  We paged Matt Jones.  He instructed us to turn the interlock switches on the 1st floor TOF PS crates from the "Solenoid On" position to off.  After doing this, we realized that the crates were already off in the first place, although the HV bars in the iFix HV summary page were still at full height.   
 
That the HV bars falsely suggested the TOF was on is no mystery, of course, because iFix was not communicating with the TOF monitoring at that time.  The real question is when the TOF went off.  One possibility (among many) is that the TOF went off when the store was dropped and Cryo ramped down the solenoid (this happened unbeknownst to me--I was on the phone with Matt), since the TOF is interlocked with the solenoid.  Then again, it might have occured earlier.</pre>
<!-- Author: H-C -->


<!-- Date: Tue Jan 8 05:16:02 2002 -->
<pre>SMX QIE Calibration has been done. Run # 136444.</pre>
<!-- Author: Abraham Gallas :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=136444' target=_top>136444</a>) -->


<!-- Date: Tue Jan 8 05:17:13 2002 -->
<pre>Luciana just e-mailed me about stopping and restarting  
the monitor program for plug HV with a larger timeout. 
Told her to go ahead.</pre>
<!-- Author: vaia -->


<!-- Date: Tue Jan 8 05:29:10 2002 -->
<pre>By looking at QIE calibration just taken we see that again  
CHA LEDs affect the CHA pedetsal RMS. I switched of the  
power supply and we took another QIE calibration run. 
(136444) 
After that we switch the power supply ON again.</pre>
<!-- Author: vaia :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=136444' target=_top>136444</a>) -->


<!-- Date: Tue Jan 8 05:29:30 2002 -->
<pre>We took another Cal QIE calibration, since the previous one was not good after Suen Hou analized the results.  
</pre>
<!-- Author: Abraham Gallas :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=136447' target=_top>136447</a>) -->


<!-- Date: Tue Jan 8 05:51:56 2002 -->
<pre>  We took a TOF QIE Calibration. Run number 136449. 
</pre>
<!-- Author: Abraham Gallas :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=136449' target=_top>136449</a>) -->


<!-- Date: Tue Jan 8 06:22:58 2002 -->
<pre>We took Cal LED Calibration, 136451 Run number.  
</pre>
<!-- Author: Abraham Gallas :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=136451' target=_top>136451</a>) -->


<!-- Date: Tue Jan 8 06:25:24 2002 -->
<pre>We took a COT Calibration Run number # 136448. 
          muon Calibration Run Number # 136446 
          Plug Laser Calibration 13650 
          </pre>
<!-- Author: Abraham Gallas -->


<!-- Date: Tue Jan 8 06:32:18 2002 -->
<pre>New version of the Plug-HV monitor/control program has been installed. 
The timeout range, i.e. the number of times to repeat the request of 
access to a crate is changed from 150 to 350.</pre>
<!-- Author: Luciana -->


<!-- Date: Tue Jan 8 06:43:51 2002 -->
<pre>We made a CEM XEF Calibration. Run number # 136452. </pre>
<!-- Author: Abraham Gallas :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=136452' target=_top>136452</a>) -->


<!-- Date: Tue Jan 8 07:53:18 2002 -->
<pre>We start a new eun, ACE_COSMIC_NOTRACKS cosmics_notracks[3,233]. 
Run number # 136460</pre>
<!-- Author: Abraham Gallas :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=136460' target=_top>136460</a>) -->


<!-- Date: Tue Jan 8 07:54:41 2002 -->
<font color=red>Shift Summary: </font><pre>1) Took about 4 hours of data with Physics_0_03[18,244] 
with 330K L3 accepts and compared L2 rates. 
There are some differences in the rates. Informed YKK. 
There were some spikes in losses. BMU, CMX, CMP standby 
 
2)At about 4 am MCR informed us that they will dump the store 
 
3) Had problems turning the silicon to standby.   
The problem was that B2 of the ISL would not go to standby.   
Andy suspected it was because CAENvolts.dat and powerset.dat 
were not loaded.  We loaded the two files but 
still failed to put B2 on standby.  Only when L0,1,2 were individually put  
to standby did the voltages go down.  This gave an alarm in the ISL which  
went away after acknowledging it.  Strangely 
enough, the ISL global power button is still ON, although no wedges are  
powered. 
 
A software protection trip in SB4W1L3 also occured when the silicon was put  
on standby.  This is the wedge that was entirely powered off.   
We set it to standby and the alarm 
went away.  The SVX power button is still in the ON state, much like the ISL  
button mentioned above, 
for some unknown reason.  Despite this, the detector is on standby now. 
 
We could not power off TOF from control room. PC was frozen. 
One possibility (among many) is that the TOF went off when the store was  
dropped and Cryo ramped 
down the solenoid since the TOF is 
interlocked with the solenoid. 
 
Took about 45 minutes to power everything down succesfully 
  
4) Took all calibrations including plug laser calibration. 
(We had about one hour of quiet time ) 
 
5) New version of plug-HV program installed 
Does not seem to work well. 
 
6) Started cosmics_notrack</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   136440-136460
Delivered Luminosity   29  
Acquired Luminosity    25.6  
Efficiency             0.88
   
</td></tr></table></pre>
<!-- Author: vaia -->


<!-- Date: Tue Jan 8 07:54:57 2002 -->
<center><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-107&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-107' width='300'></a><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-108&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-108' width='300'></a></center>
<!-- Author: H-C -->
<!-- Comment: Tue Jan  8 07;56;23 comment by...H-C -->
<pre>Luminosity and losses for shift.

SVRAD sums at end of shift:
     SVRAD0      0
     SVRAD1      0
     SVRAD2      0
     SVRAD3      0</pre>
<!-- Date: Tue Jan 8 07:55:15 2002 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=136440>
136440</A> 
x214F8</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_0_03 [18,244]</TD>                                          
<TD ALIGN=CENTER>23:58:40</TD>                                          
<TD ALIGN=CENTER>00:14:25</TD>                                          
<TD ALIGN=CENTER>00:14:43</TD>                                          
<TD ALIGN=RIGHT>        192,748</TD>                                    
<TD ALIGN=RIGHT>        98,592</TD>                                     
<TD ALIGN=RIGHT>      11,811</TD>                                       
<TD ALIGN=RIGHT>    1.874</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=136441>
136441</A> 
x214F9</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_0_03 [18,244]</TD>                                          
<TD ALIGN=CENTER>00:19:29</TD>                                          
<TD ALIGN=CENTER>04:03:54</TD>                                          
<TD ALIGN=CENTER>03:22:41</TD>                                          
<TD ALIGN=RIGHT>      2,771,007</TD>                                    
<TD ALIGN=RIGHT>     1,558,556</TD>                                     
<TD ALIGN=RIGHT>     333,445</TD>                                       
<TD ALIGN=RIGHT>   23.873</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>07:55:01</TD>                                                       
<TD ALIGN=CENTER>03:37:24</TD>                                          
<TD ALIGN=RIGHT>       2,963,755</TD>                                   
<TD ALIGN=RIGHT>      1,657,148</TD>                                    
<TD ALIGN=RIGHT>       345,256</TD>                                     
<TD ALIGN=RIGHT>   25.747</TD>                                          
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->