BEE-spoke-data/govdocs1-by-extension huggingface dataset:
<!-- Title: 10:57:35  Tue Oct 10 2000 -->
<!-- Crewchief: R. Erbacher -->
<!-- Op1: RJ Tesarek -->
<!-- Op2: &nbsp; -->
<!-- Op3: &nbsp; -->
<!-- Op4: &nbsp; -->
<!-- Op5: &nbsp; -->
<!-- Op6: &nbsp; -->
<!-- Notes: -->
<h2>New Week</h2>


<!-- Date: Tue Oct 10 11:24:15 2000 -->
<h2>Supervised Access Work:</h2> 
<b>b0pcal02:</b><br> 
We tried to get the power supply to trip under the theory that 
"reset_crate" might be causing a glitch.  We reset the crate remotely 10 times but failed to make the supply trip. 
 
<br><b>All crate checkout:</b><br> 
We ran a full ckplug and got the usual suspects: problems in 
pcal00,pcal01,pcal05,pcal11.  We didn't see anything in pcal06 
the second time but did the first, and it's likely that the 
channel is hovering around the tolerance threshold.  See log  
entry for the last access for the particulars on these problems.
<!-- Author: R. Erbacher, RJ Tesarek -->


<!-- Date: Wed Oct 11 06:55:19 2000 -->
<pre><br><b>b0pcal11, EM wedges 20,21:</b> 
<br>Looked into why wedges 20,21 in the PEM do not appear in 
run 101009.  These wedges correspond to the ADMEMs in slots 
16,18.   
<p>Pedestals (passthrough) for all ADMEMs look ok for this crate. 
Tried running a standalone QIE calibration which failed.  Rebooted the crate. I did not think of looking at pedestals in "look-up" mode until after rebooting the crate.  However, these 
pedestals also look normal. 
<br> 
</pre>
<!-- Author: R.J. Tesarek -->


<!-- Date: Wed Oct 11 07:13:14 2000 -->
<pre><br><b>PEM wedge 20,21 problems cont...</b> 
<br>Looking further into the problems observed earlier.  First, 
the numbering convention is different in the plug from the  
central detector (7.5 degree "wedges").  This means that the 
phi slices not seen in the detector really correspond to crates 
8,9.  Looked at pedestals in both passthrough and lookup modes 
for both crates only to find everything looks ok.  Again,  
attempting calibrations fails.  Both crates rebooted. 
<br> 
</pre>
<!-- Author: R.J. Tesarek -->
99.06.15 14:27 B c:\InetPub\wwwroot\controls\Meetings\aug11_98\SVX Overview\CodeFlowChart.JPG --> cdfsga.fnal.gov /cdf/www/internal/upgrades/controls/Meetings/aug11_98/SVX Overview CodeFlowChart.JPG
99.06.15 14:27 B c:\InetPub\wwwroot\controls\Meetings\aug11_98\SVX Overview\PwrSplCntrl.JPG --> cdfsga.fnal.gov /cdf/www/internal/upgrades/controls/Meetings/aug11_98/SVX Overview PwrSplCntrl.JPG
99.06.15 14:27 B c:\InetPub\wwwroot\controls\Meetings\aug11_98\SVX Overview\svx_iiOvrView.htm --> cdfsga.fnal.gov /cdf/www/internal/upgrades/controls/Meetings/aug11_98/SVX Overview svx_iiOvrView.htm
99.06.15 14:27 B c:\InetPub\wwwroot\controls\Meetings\aug11_98\SVX Overview\SVXCntrlIntlk.JPG --> cdfsga.fnal.gov /cdf/www/internal/upgrades/controls/Meetings/aug11_98/SVX Overview SVXCntrlIntlk.JPG
99.06.15 14:27 B c:\InetPub\wwwroot\controls\Meetings\aug11_98\SVX Overview\SVXISLIntrlk.JPG --> cdfsga.fnal.gov /cdf/www/internal/upgrades/controls/Meetings/aug11_98/SVX Overview SVXISLIntrlk.JPG
99.06.15 14:27 B c:\InetPub\wwwroot\controls\Meetings\aug11_98\SVX Overview\SVXStatus.JPG --> cdfsga.fnal.gov /cdf/www/internal/upgrades/controls/Meetings/aug11_98/SVX Overview SVXStatus.JPG
99.06.15 14:27 B c:\InetPub\wwwroot\controls\Meetings\aug11_98\SVX Overview\WWStrcISLCntrl.JPG --> cdfsga.fnal.gov /cdf/www/internal/upgrades/controls/Meetings/aug11_98/SVX Overview WWStrcISLCntrl.JPG
99.06.15 14:27 B c:\InetPub\wwwroot\controls\Meetings\aug11_98\SVX Overview\wwStrcSVXCntrl.JPG --> cdfsga.fnal.gov /cdf/www/internal/upgrades/controls/Meetings/aug11_98/SVX Overview wwStrcSVXCntrl.JPG
99.08.24 14:02 B c:\InetPub\wwwroot\controls\Meetings\aug11_98\SVX Overview\WS_FTP.LOG <-- cdfsga /cdf/www/internal/upgrades/controls/Meetings/aug11_98/SVX Overview WS_FTP.LOG
99.08.24 14:02 B c:\InetPub\wwwroot\controls\Meetings\aug11_98\SVX Overview\WWStrcISLCntrl.JPG <-- cdfsga /cdf/www/internal/upgrades/controls/Meetings/aug11_98/SVX Overview WWStrcISLCntrl.JPG
99.08.24 14:02 B c:\InetPub\wwwroot\controls\Meetings\aug11_98\SVX Overview\wwStrcSVXCntrl.JPG <-- cdfsga /cdf/www/internal/upgrades/controls/Meetings/aug11_98/SVX Overview wwStrcSVXCntrl.JPG
2000.01.20 09:33 B C:\InetPub\wwwroot\myweb\Meetings\aug11_98\SVXOverview\WS_FTP.LOG <-- ncdf03 /samc/InetPub/wwwroot/myweb/Meetings/aug11_98/SVXOverview WS_FTP.LOG
2000.01.20 09:33 B C:\InetPub\wwwroot\myweb\Meetings\aug11_98\SVXOverview\WWStrcISLCntrl.JPG <-- ncdf03 /samc/InetPub/wwwroot/myweb/Meetings/aug11_98/SVXOverview WWStrcISLCntrl.JPG
2000.01.20 09:33 B C:\InetPub\wwwroot\myweb\Meetings\aug11_98\SVXOverview\wwStrcSVXCntrl.JPG <-- ncdf03 /samc/InetPub/wwwroot/myweb/Meetings/aug11_98/SVXOverview wwStrcSVXCntrl.JPG
<!-- Title: 00:02:23  Tue Dec 16 2003 -->
<!-- Crewchief: Jim H. nights -->
<!-- Op1: Dean B. days -->
<!-- Op2: Steve G. nights -->
<!-- Op3: &nbsp; -->
<!-- Op4: &nbsp; -->
<!-- Op5: &nbsp; -->
<!-- Op6: &nbsp; -->
<!-- Notes: -->
Monitoring process systems.


<!-- Date: Tue Dec 16 03:32:46 2003 -->
There might be a small oil leak on LCW pump #1. A month or two ago I wiped clean the platform and it has oil all over it again. The level in the sight glass is fine though.
<!-- Author: Jim H. -->


<!-- Date: Tue Dec 16 07:02:28 2003 -->
Dean on duty, cdf checklist is completed everything is normal at this time
<!-- Author: Dean -->


<!-- Date: Tue Dec 16 13:53:13 2003 -->
I added 10.oz of nalco 7338 to the ecw water sytem
<!-- Author: Dean -->


<!-- Date: Tue Dec 16 17:26:12 2003 -->
added isobutane gas to clc west chamber reading is 22.52 psia
<!-- Author: Dean -->
<!-- Comment: Wed Dec 17 11;42;17 comment by...ming -->
<pre>Thanks !</pre>
<!-- Date: Tue Dec 16 18:18:53 2003 -->
CDF checklist completed. Everything looks okay.
<!-- Author: Steve -->
pact is get-new-atlastrf-AtlasProduction-14.5.1-14.5.1.1-i686-slc4-gcc34-opt
 
 ########################################################### 
 
 The input values you specify for this script to run are :   
 
 Remote gate-keeper is "gridgk03.racf.bnl.gov"
 Remote Gridftp server is "gridgk03.racf.bnl.gov"
 Remote $APP area is "/usatlas/OSG"
 Remote $DATA area is "/usatlas/prodjob/share"
 
 ########################################################### 
 
Thu Jan 22 10:03:11 EST 2009
Sending the install script "ace-install-template-gridgk03.racf.bnl.gov-get-new-atlastrf-AtlasProduction-14.5.1-14.5.1.1-i686-slc4-gcc34-opt-2009-01-22T10-0500-7807" to remote site "gridgk03.racf.bnl.gov". 
globus-url-copy file:\/\/\//direct/usatlas+u/atlaswin/remote-install/scripts\/ace-install-template-gridgk03.racf.bnl.gov-get-new-atlastrf-AtlasProduction-14.5.1-14.5.1.1-i686-slc4-gcc34-opt-2009-01-22T10-0500-7807 gsiftp\:\/\/gridgk03.racf.bnl.gov\//usatlas/OSG\/ace-install-template-gridgk03.racf.bnl.gov-get-new-atlastrf-AtlasProduction-14.5.1-14.5.1.1-i686-slc4-gcc34-opt-2009-01-22T10-0500-7807
Sending the mdsupdate script "fix-MDSConfig-gridgk03.racf.bnl.gov-get-new-atlastrf-AtlasProduction-14.5.1-14.5.1.1-i686-slc4-gcc34-opt-2009-01-22T10-0500-7807" to remote site "gridgk03.racf.bnl.gov". 
Sending the testrun script "KVtest.sh" to remote site "gridgk03.racf.bnl.gov". 
Sending the AGLT2 patch script "copy_libs_ATLAS.sh" to remote site "gridgk03.racf.bnl.gov". 
Sending the skeleton.csc_MergeHIST_trf.py file to remote site "gridgk03.racf.bnl.gov". 
Running the install script in "/usatlas/OSG" on "gridgk03.racf.bnl.gov"
Using http_proxy on "gridgk03.racf.bnl.gov" :  http://squid.sec.bnl.local:3128
hostname is  gridgk03.racf.bnl.gov
PACMAN_CMD is  pacman
Doing a pacman "get-new-atlastrf-AtlasProduction-14.5.1-14.5.1.1-i686-slc4-gcc34-opt" on gridgk03.racf.bnl.gov now... 
Package [AtlasProduction_14_5_1_1_i686_slc4_gcc34_opt] found in [http://cern.ch/atlas-computing/links/kitsDirectory/Production/cache]...
Package [AtlasProduction_14_5_1_1_noarch] found in [http://cern.ch/atlas-computing/links/kitsDirectory/Production/cache]...
Package [AtlasProduction_14_5_1_1_noarch] found in [http://cern.ch/atlas-computing/links/kitsDirectory/Production/cache]...
Downloading [AtlasProduction_14_5_1_1_noarch.tar.gz] from [http://cern.ch/atlas-computing/links/kitsDirectory/Production/cache/../kits]...
Untarring [AtlasProduction_14_5_1_1_noarch.tar.gz]...
Downloading [AtlasProduction_14_5_1_1_i686_slc4_gcc34_opt.tar.gz] from [http://cern.ch/atlas-computing/links/kitsDirectory/Production/cache/../kits]...
Untarring [AtlasProduction_14_5_1_1_i686_slc4_gcc34_opt.tar.gz]...
JobTransforms 14_5_1_1 is installed
gatekeeper is gridgk03.racf.bnl.gov
Cleaning remote install scripts and tmp files...
Installation get-new-atlastrf-AtlasProduction-14.5.1-14.5.1.1-i686-slc4-gcc34-opt is done on site gridgk03.racf.bnl.gov.
--- 16:24:29 3/6/96 top of scan loop ---
1015 MegaBytes free disk space on drive C
samples per pixel=1
bits per sample=8
image startX=0, startY=0, width=2544, length=3392
photometric interpretation=0
ISIS output pipe is: ""
C:\TMP\FPRA\09630012\00000001.JPG 3/6/96 16:24:29 clip=0,255 quant=0,0 SCANJPEG1.04 VT-Compress2.01
jpeg luminance   quantization=0
jpeg chrominance quantization=0 (higher values give greater compression)
scan: 34, convert/display: 0, compress: 11, total: 45 seconds
25.8 compression factor, 335105 bytes file size
bright clip=255, dark clip=0

 --- 16:25:17 3/6/96 top of scan loop ---
1015 MegaBytes free disk space on drive C
samples per pixel=1
bits per sample=8
image startX=0, startY=0, width=2544, length=3392
photometric interpretation=0
ISIS output pipe is: ""
C:\TMP\FPRA\09630012\00000002.JPG 3/6/96 16:25:17 clip=0,255 quant=0,0 SCANJPEG1.04 VT-Compress2.01
jpeg luminance   quantization=0
jpeg chrominance quantization=0 (higher values give greater compression)
scan: 34, convert/display: 0, compress: 12, total: 46 seconds
18.4 compression factor, 469232 bytes file size

 --- 16:26:04 3/6/96 top of scan loop ---
1014 MegaBytes free disk space on drive C
samples per pixel=1
bits per sample=8
image startX=0, startY=0, width=2544, length=3392
photometric interpretation=0
ISIS output pipe is: ""
C:\TMP\FPRA\09630012\00000003.JPG 3/6/96 16:26:04 clip=0,255 quant=0,0 SCANJPEG1.04 VT-Compress2.01
jpeg luminance   quantization=0
jpeg chrominance quantization=0 (higher values give greater compression)
scan: 35, convert/display: 0, compress: 11, total: 46 seconds
18.9 compression factor, 456891 bytes file size

 --- 16:26:57 3/6/96 top of scan loop ---
1014 MegaBytes free disk space on drive C
samples per pixel=1
bits per sample=8
image startX=0, startY=0, width=2544, length=3392
photometric interpretation=0
ISIS output pipe is: ""
C:\TMP\FPRA\09630012\00000004.JPG 3/6/96 16:26:57 clip=0,255 quant=0,0 SCANJPEG1.04 VT-Compress2.01
jpeg luminance   quantization=0
jpeg chrominance quantization=0 (higher values give greater compression)
scan: 35, convert/display: 0, compress: 11, total: 46 seconds
20.3 compression factor, 425491 bytes file size

 --- 16:27:53 3/6/96 top of scan loop ---
1014 MegaBytes free disk space on drive C
samples per pixel=1
bits per sample=8
image startX=0, startY=0, width=2544, length=3392
photometric interpretation=0
ISIS output pipe is: ""
C:\TMP\FPRA\09630012\00000005.JPG 3/6/96 16:27:53 clip=0,255 quant=0,0 SCANJPEG1.04 VT-Compress2.01
User cancelled operation.
  Source file ISIS2RAW.C at line 279
scan: 11, convert/display: 0, compress: 0, total: 11 seconds (CANCELED SCAN)
104.07.13 16:32 B G:\sgpni\1800\180011\18001107\0001u.TIF --> rs5.loc.gov /.mnt.sd.sgp.sgpni/1800/180011/18001107 0001u.TIF
104.07.13 16:32 B G:\sgpni\1800\180011\18001107\0002u.TIF --> rs5.loc.gov /.mnt.sd.sgp.sgpni/1800/180011/18001107 0002u.TIF
104.07.13 16:32 B G:\sgpni\1800\180011\18001107\0003u.TIF --> rs5.loc.gov /.mnt.sd.sgp.sgpni/1800/180011/18001107 0003u.TIF
104.07.13 16:32 B G:\sgpni\1800\180011\18001107\0004u.TIF --> rs5.loc.gov /.mnt.sd.sgp.sgpni/1800/180011/18001107 0004u.TIF
104.08.11 10:47 B G:\sgpni\1800\180011\18001107\0001u.TIF --> rs5.loc.gov /.mnt.ma.sgp.sgpprod/sgpni/1800/180011/18001107 0001u.TIF
104.08.11 10:47 B G:\sgpni\1800\180011\18001107\0002u.TIF --> rs5.loc.gov /.mnt.ma.sgp.sgpprod/sgpni/1800/180011/18001107 0002u.TIF
104.08.11 10:47 B G:\sgpni\1800\180011\18001107\0003u.TIF --> rs5.loc.gov /.mnt.ma.sgp.sgpprod/sgpni/1800/180011/18001107 0003u.TIF
104.08.11 10:48 B G:\sgpni\1800\180011\18001107\0004u.TIF --> rs5.loc.gov /.mnt.ma.sgp.sgpprod/sgpni/1800/180011/18001107 0004u.TIF
104.08.11 10:48 B G:\sgpni\1800\180011\18001107\WS_FTP.LOG --> rs5.loc.gov /.mnt.ma.sgp.sgpprod/sgpni/1800/180011/18001107 WS_FTP.LOG
104.08.27 14:01 B G:\sgpni\1800\180011\18001107\0001u.TIF --> rs5.loc.gov /.mnt.ma.sgp.sgpprod/sgpni/1800/180011/18001107 0001u.TIF
104.08.27 14:01 B G:\sgpni\1800\180011\18001107\0002u.TIF --> rs5.loc.gov /.mnt.ma.sgp.sgpprod/sgpni/1800/180011/18001107 0002u.TIF
104.08.27 14:02 B G:\sgpni\1800\180011\18001107\0003u.TIF --> rs5.loc.gov /.mnt.ma.sgp.sgpprod/sgpni/1800/180011/18001107 0003u.TIF
104.08.27 14:02 B G:\sgpni\1800\180011\18001107\0004u.TIF --> rs5.loc.gov /.mnt.ma.sgp.sgpprod/sgpni/1800/180011/18001107 0004u.TIF
This is TeX, C Version 3.14t3 (format=latex 96.10.4)  17 APR 1998 21:15
**trigger_blt_abs.tex
(trigger_blt_abs.tex
LaTeX2e <1996/06/01>
(/vol/pub/TeX-3.pl141/lib/inputs-latex2e/article.cls
Document Class: article 1996/05/26 v1.3r Standard LaTeX document class
(/vol/pub/TeX-3.pl141/lib/inputs-latex2e/size10.clo
File: size10.clo 1996/05/26 v1.3r Standard LaTeX file (size option)
)
\c@part=\count79
\c@section=\count80
\c@subsection=\count81
\c@subsubsection=\count82
\c@paragraph=\count83
\c@subparagraph=\count84
\c@figure=\count85
\c@table=\count86
\abovecaptionskip=\skip41
\belowcaptionskip=\skip42
\bibindent=\dimen102
) (ieee_tns.sty
(/vol/pub/TeX-3.pl141/lib/inputs-latex2e/ifthen.sty
Package: ifthen 1996/03/22 v1.0l Standard LaTeX ifthen package (DPC)
)
(/vol/pub/TeX-3.pl141/lib/inputs-latex2e/times.sty
Package: times 1995/08/16 5.2 Times PSNFSS2e package
)
Package: ieee_tns 1998/03/12 IEEE Transactions on Nuclear Science Style

Laying out for letter paper
) (/vol/pub/TeX-3.pl141/lib/inputs/psfig.sty
\@unused=\write3
\ps@stream=\read1
\p@intvaluex=\dimen103
\p@intvaluey=\dimen104

psfig/tex 1.10-dvips
) (trigger_blt_abs.aux)
LaTeX Font Info:    Checking defaults for OML/cmm/m/it on input line 5.
LaTeX Font Info:    ... okay on input line 5.
LaTeX Font Info:    Checking defaults for T1/cmr/m/n on input line 5.
LaTeX Font Info:    ... okay on input line 5.
LaTeX Font Info:    Checking defaults for OT1/cmr/m/n on input line 5.
LaTeX Font Info:    ... okay on input line 5.
LaTeX Font Info:    Checking defaults for OMS/cmsy/m/n on input line 5.
LaTeX Font Info:    ... okay on input line 5.
LaTeX Font Info:    Checking defaults for OMX/cmex/m/n on input line 5.
LaTeX Font Info:    ... okay on input line 5.
LaTeX Font Info:    Checking defaults for U/cmr/m/n on input line 5.
LaTeX Font Info:    ... okay on input line 5.
LaTeX Font Info:    Try loading font information for OT1+ptm on input line 5.
 (/vol/pub/TeX-3.pl141/lib/inputs-latex2e/OT1ptm.fd
File: OT1ptm.fd 1996/01/28 Fontinst v1.504 font definitions for OT1/ptm.
)
LaTeX Font Info:    External font `cmex10' loaded for size
(Font)              <7> on input line 35.
LaTeX Font Info:    External font `cmex10' loaded for size
(Font)              <5> on input line 35.

Underfull \hbox (badness 2689) in paragraph at lines 27--42
\OT1/ptm/m/n/10 pro-cess was greatly stream-lined by us-ing state-of-the-art
 []

[1



] (trigger_blt_abs.aux) ) 
Here is how much of TeX's memory you used:
 559 strings out of 3423
 5626 string characters out of 49267
 50366 words of memory out of 262141
 3436 multiletter control sequences out of 9500
 6355 words of font info for 19 fonts, out of 72000 for 255
 14 hyphenation exceptions out of 607
 23i,4n,18p,212b,181s stack positions out of 300i,40n,60p,8000b,4000s

Output written on trigger_blt_abs.dvi (1 page, 2052 bytes).
***********************************************************************
Fatal NI connect error 12547, connecting to:
 (DESCRIPTION=(ADDRESS=(PROTOCOL=TCP)(HOST=209.222.132.141)(PORT=1521))(CONNECT_DATA=(SERVICE_NAME=GRANTS)(CID=(PROGRAM=C:\Program Files\Quest Software\Toad for Oracle\TOAD.exe)(HOST=HQ-NMETROKOTSDT)(USER=upola))))

  VERSION INFORMATION:
	TNS for 32-bit Windows: Version 9.2.0.1.0 - Production
	Windows NT TCP/IP NT Protocol Adapter for 32-bit Windows: Version 9.2.0.1.0 - Production
  Time: 18-JUN-2007 15:45:07
  Tracing not turned on.
  Tns error struct:
    nr err code: 0
    ns main err code: 12547
    TNS-12547: TNS:lost contact
    ns secondary err code: 12560
    nt main err code: 517
    TNS-00517: Lost contact
    nt secondary err code: 54
    nt OS err code: 0
<!-- Title: 00:08:36  Fri Sep 16 2005 -->
<!-- Crewchief: Dean -->
<!-- Op1: Steve -->
<!-- Op2: Jim -->
<!-- Op3: &nbsp; -->
<!-- Op4: &nbsp; -->
<!-- Op5: &nbsp; -->
<!-- Op6: &nbsp; -->
<!-- Notes: -->
isl chiller is running fine


<!-- Date: Fri Sep 16 04:31:08 2005 -->
power cycle the west alcove rack
<!-- Author: Dean -->


<!-- Date: Fri Sep 16 05:01:48 2005 -->
<table border=0><td><tr><td bgcolor=#e0e030>We need to get CMU bubbler alarm cleared today.</td></tr></td></table>
<!-- Author: Bill Noe -->
<!-- Comment: Fri Sep 16 06;49;06 comment by...KC -->
I removed some alcohol from the cmu bubbler to clear the alarm. I have a repaired bubbler on my dest to replace the cmu bubbler when we can.
<!-- Date: Fri Sep 16 05:50:54 2005 -->
CDF checklist completed. Checked on the silicon chillers; looks good.
<!-- Author: Steve -->



<!-- Date: Fri Sep 16 07:39:35 2005 -->
Discovered that fore pump 70 and turbo pump 70 were off (as well as the cold cathode gauge). Currently pumping down the lines.
<!-- Author: Steve -->
<!-- Comment: Fri Sep 16 08;00;05 comment by...Steve -->
<table border=0><td><tr><td bgcolor=#e0e030>The vacuum is not improving. Suspect something wrong with the forepump or vacuum line. Will turn off the pump for now. An access will be required to investigate further.</td></tr></td></table>


<!-- Date: Fri Sep 16 08:28:09 2005 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=cryo-gas&action=view&page=-586&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=cryo-gas&action=view&page=-587  width='300' align='top'></a></td><td>This is a 2 week plot of the vacuum readouts for the 70 series pumps. It looks like the forepump was dying around Sept. 3rd. There was a collision hall access that day.</td></tr></table>
<!-- Author: Steve -->


<!-- Date: Fri Sep 16 11:14:45 2005 -->
The engine goup was here today. They replaced the bad arm on the #1 side of the north dry engine. There is also some lubricant for the arm to use when the engine is greased. 
<!-- Author: Steve -->


<!-- Date: Fri Sep 16 18:41:56 2005 -->
Cryo and gas readings are normal at this time. The CDF systems checklist and mechanical rooms are also normal at this time. I talked to Stefano a short time ago and we will shutdown the repaired ISL chiller tomorrow. I will monitor it tonight,
<!-- Author: Jim H. -->


<!-- Date: Fri Sep 16 19:21:44 2005 -->
We just received 2200 gallons of LN2.
<!-- Author: Jim H. -->
<!-- Title: 00:22:59  Wed Mar 28 2007 -->
<!-- Crewchief: Dean -->
<!-- Op1: Bruce -->
<!-- Op2: &nbsp; -->
<!-- Op3: &nbsp; -->
<!-- Op4: &nbsp; -->
<!-- Op5: &nbsp; -->
<!-- Op6: &nbsp; -->
<!-- Notes: -->
Zero the gow-mac and started a mix on tank #3


<!-- Date: Wed Mar 28 01:02:55 2007 -->
mix stop on fill in progress
<!-- Author: Dean -->
<!-- Comment: Wed Mar 28 01;11;01 comment by...Dean -->
gow-mac is reading 7750 ppm


<!-- Date: Wed Mar 28 01:24:24 2007 -->
psm alarm 1st floor 1rr21e-2
<!-- Author: Dean -->


<!-- Date: Wed Mar 28 03:34:32 2007 -->
psm alarm 1rr12c-4
<!-- Author: Dean -->


<!-- Date: Wed Mar 28 05:09:28 2007 -->
Chiller 4 & 5 seem to be running well so I asked Dean to shut down chillers 1 & 3. Yesterday Warren found the connector on the back of the control box for chiller 5 was installed on an angle. Looks like that was the problem.
<!-- Author: Bill Noe -->


<!-- Date: Wed Mar 28 05:23:18 2007 -->
Bruce on duty.
<!-- Author: Bruce -->


<!-- Date: Wed Mar 28 05:30:02 2007 -->
PSM alarm 1RR21E_2 reset on it's own.
<!-- Author: Bill Noe -->


<!-- Date: Wed Mar 28 06:06:40 2007 -->
Gas and cryo readings taken all systems are running fine at this time, no problems to report.
<!-- Author: Bruce -->


<!-- Date: Wed Mar 28 07:49:19 2007 -->
PSM Alarm --->1RR21E_2---> cleared itself in less than 2 minutes.
<!-- Author: Jim L -->


<!-- Date: Wed Mar 28 07:51:24 2007 -->
Warren's out in gas shed working on the gow mac and said he was going to leave door open while he was there and to silence the gas alarm.  --> Done
<!-- Author: Jim L -->


<!-- Date: Wed Mar 28 08:16:43 2007 -->
KC and Bruce are installing a new fan blade on Chiller 4. Alarm silenced.
<!-- Author: Bill Noe -->
<!-- Comment: Wed Mar 28 08;27;29 comment by...Bill Noe -->
Job complete alarm back to normal.


<!-- Date: Wed Mar 28 14:48:09 2007 -->
<table border=0><td><tr><td bgcolor=#e0e030>DO NOT mix any gas until tommorow night Warren and Del are working on the gow-mac problem.</td></tr></td></table>
<!-- Author: Bruce -->


<!-- Date: Wed Mar 28 18:17:59 2007 -->
cdf checklist is completed, isl chiller and svx chiller are  set to 6c
<!-- Author: Dean -->


<!-- Date: Wed Mar 28 18:18:56 2007 -->
psm alarm 1rr21e_2
<!-- Author: Dean -->


<!-- Date: Wed Mar 28 18:48:43 2007 -->
I hooked up helium trailer 170 1500 psi
<!-- Author: Dean -->


<!-- Date: Wed Mar 28 19:29:53 2007 -->
flammble gas alarm imu north-west reading trouble reset ok
<!-- Author: Dean -->
/home/mars/rick/pei/src/loc/locsc9 processing 00000003.jpg to /home/mars/rick/images/loc/output7/cdrom/test/t011
image category is text, target size is 60000
decompressing input image...
thresholding with VST options: -t1 30 -t2 12 -tl 40
output size is 123034 with param set 0, attempting retry to reduce below 60000
thresholding with VST options: -t1 38 -t2 15 -tl 40
output size is 116872 with param set 1, attempting retry to reduce below 60000
thresholding with VST options: -t1 47 -t2 18 -tl 40
output size is 108314 with param set 2, attempting retry to reduce below 60000
thresholding with VST options: -t1 57 -t2 22 -tl 40
output size is 100430 with param set 3, attempting retry to reduce below 60000
thresholding with VST options: -t1 68 -t2 27 -tl 40
output size is 94008 with param set 4, unable to reduce below 60000
copying output image...
start time Tue May 13 18:56:32 PDT 1997
  end time Tue May 13 18:59:25 PDT 1997
Feb/16/2004 09:43:17> ako:ako CREATE
Feb/16/2004 09:45:57> ako:ako IMPORT100701 Stations /data/sunny/Taro_Seasoar.data/TOW3/TOW3_ODV.txt (Feb/16/2004 09:26:44)
Feb/16/2004 09:52:32> ako:ako IMPORT100701 Stations /data/sunny/Taro_Seasoar.data/TOW4/TOW4_ODV.txt (Feb/16/2004 09:27:56)
Feb/16/2004 09:54:30> ako:ako IMPORT 53265 Stations /data/sunny/Taro_Seasoar.data/TOW5/TOW5_ODV.txt (Feb/16/2004 09:29:20)
Feb/16/2004 09:57:49> ako:ako DELETE 10653 Stations
Feb/16/2004 09:58:11> ako:ako DELETE 10653 Stations
Feb/16/2004 09:58:27> ako:ako DELETE 10653 Stations
Feb/16/2004 09:58:46> ako:ako DELETE 10653 Stations
Feb/16/2004 09:59:08> ako:ako DELETE 10653 Stations
Feb/16/2004 10:03:07> ako:ako IMPORT 53265 Stations /data/sunny/Taro_Seasoar.data/TOW5/TOW5_ODV.txt (Feb/16/2004 10:01:16)
Feb/16/2004 10:05:42> ako:ako IMPORT100701 Stations /data/sunny/Taro_Seasoar.data/TOW6/TOW6_ODV.txt (Feb/16/2004 09:31:16)
<!-- Title: 05:26:02  Mon Sep 20 2004 -->
<!-- Crewchief: Bill Noe -->
<!-- Op1: &nbsp; -->
<!-- Op2: &nbsp; -->
<!-- Op3: &nbsp; -->
<!-- Op4: &nbsp; -->
<!-- Op5: &nbsp; -->
<!-- Op6: &nbsp; -->
<!-- Notes: -->
Week 5 of Shutdown. Systems look good.


<!-- Date: Mon Sep 20 05:26:36 2004 -->
I switched HVAC back to low and disabled the RFs.
<!-- Author: Bill Noe -->


<!-- Date: Mon Sep 20 06:30:23 2004 -->
I switched the O2 analyzer to Manual Zero.  Del and I will try to figure out why it is reading about 100 ppm high.
<!-- Author: WB -->
<!-- Comment: Mon Sep 20 07;40;56 comment by...WB -->
I'm removing the analyzer from the rack to set it up for a bench test.


<!-- Date: Mon Sep 20 08:02:54 2004 -->
We are crashing the cryo plant to pull the wet exhaust u-tube in order to change the valves. 
<!-- Author: Steve & Jim L.  -->


<!-- Date: Mon Sep 20 09:50:57 2004 -->
Running cryo on the JT. Valve setting is at 36% and finally it's making an increase in storage dewar level. Jim and Bruce changed the wet engine valves. We are warming up the exhaust u-tube to get the moisture out of it. 
<!-- Author: Steve -->


<!-- Date: Mon Sep 20 10:56:53 2004 -->
The wet is back online. 22.8% in storage dewar. Plant recovery in progress. 
<!-- Author: Steve -->


<!-- Date: Mon Sep 20 13:33:10 2004 -->
Established normal magnet flow and a nice steady liquid makeup rate into the dewar. The wet engine perfomance is greatly improved (thanks, Jim!). 
<!-- Author: Steve -->


<!-- Date: Mon Sep 20 14:00:34 2004 -->
Hooked up helium tube trailer #172, east trailer, with 1420 psig (1500 on the analog gauge at the trailer). The west trailer, #171, is valved out but still connected. 
<!-- Author: Steve -->


<!-- Date: Mon Sep 20 14:16:10 2004 -->
Slowed the make rate down a little so tht the dewar will likely be near 60% when we show up in the morning. Magnet flow will vary, depending on PVBP and engine performance. 
<!-- Author: Steve -->


<!-- Date: Mon Sep 20 14:17:55 2004 -->
<pre>Del established O2 sampling of the chambers. There are 5 left. The alarm is silenced and the autodialer is unplugged. No one here to answer the phone.</pre>
<!-- Author: Steve -->


<!-- Date: Mon Sep 20 19:49:17 2004 -->
Steve Hahn called about a cryo alarm <dewar low level>. I checked the system and discovered high pressure in the dewar. Suspected wet engine valve blow-by so I opened the bypass and slowed the wet engine speed. 
<!-- Author: Bill Noe -->
<!-- Title: 00:00:37  Fri Jun 27 2008 -->
<!-- Crewchief: KC nights -->
<!-- Op1: Jim H. days -->
<!-- Op2: Bruce nights -->
<!-- Op3: &nbsp; -->
<!-- Op4: &nbsp; -->
<!-- Op5: &nbsp; -->
<!-- Op6: &nbsp; -->
<!-- Notes: -->
CDF is running very smooth tonight, all green.


<!-- Date: Fri Jun 27 06:18:49 2008 -->
The CDF systems checklist and mechanical rooms are normal at this time.
<!-- Author: Jim H. -->


<!-- Date: Fri Jun 27 08:16:53 2008 -->
Increased CLC west pressure from 22.05 to 22.57 psia.
<!-- Author: Jim H. -->


<!-- Date: Fri Jun 27 08:59:14 2008 -->
Steve Hahn is replacing the CEM SE arch power supply. I've silenced the alarm for now.
<!-- Author: Jim H. -->
<!-- Comment: Fri Jun 27 10;21;13 comment by...Jim H. -->
Steve has completed the replacement.


<!-- Date: Fri Jun 27 12:03:15 2008 -->
Cryo alarm. The east dry engine stopped. In the controller the small glass fuse was blown. We replaced it and so far so good.
<!-- Author: Jim H. & Warren B. -->


<!-- Date: Fri Jun 27 12:33:41 2008 -->
I started an A/E mix into tank #3.
<!-- Author: Jim H. -->
<!-- Comment: Fri Jun 27 16;53;02 comment by...Jim H. -->
The mix completed ok.


<!-- Date: Fri Jun 27 17:52:33 2008 -->
Bruce on duty,gas and cryo readings taken all systems are running fine at this time.
<!-- Author: Bruce -->


<!-- Date: Fri Jun 27 18:36:42 2008 -->
On the rack page the 2R23C rack is hilighted in yelow and next to it there is a FLT in red.
<!-- Author: Bruce -->
<!-- Comment: Fri Jun 27 19;46;13 comment by...Bruce -->
I called the 2nd floor they called the ops manager he said to override the fan failure ,it is done they will add it to the list of fans that have to be repaired he said.

<!-- Comment: Fri Jun 27 20;04;50 comment by...Bruce -->
Sent e mail to bill and dervin about the fan failure on 2R23C bottom fan failure.
sources_cache = http://cmsdoc.cern.ch/cms/cpt/Software/download/cms
apt_server = http://cmsdoc.cern.ch/cms/cpt/Software/download/cms
Preparing spec for xsl-virtual...
Preparing spec for gcc...
Preparing sources for xsl-virtual...
Preparing sources for gcc...
File already downloaded.
slc4_ia32_gcc345
/usr/bin/apt-get
Using apt to fetch available binary packages
Get:1 http://cmsdoc.cern.ch cms/cpt/Software/download/cms/apt/slc4_ia32_gcc345 release [1085B]
Fetched 1085B in 0s (4170B/s)
Hit http://cmsdoc.cern.ch cms/cpt/Software/download/cms/apt/slc4_ia32_gcc345/lcg pkglist
Hit http://cmsdoc.cern.ch cms/cpt/Software/download/cms/apt/slc4_ia32_gcc345/lcg release
Hit http://cmsdoc.cern.ch cms/cpt/Software/download/cms/apt/slc4_ia32_gcc345/cms pkglist
Hit http://cmsdoc.cern.ch cms/cpt/Software/download/cms/apt/slc4_ia32_gcc345/cms release
Hit http://cmsdoc.cern.ch cms/cpt/Software/download/cms/apt/slc4_ia32_gcc345/external pkglist
Hit http://cmsdoc.cern.ch cms/cpt/Software/download/cms/apt/slc4_ia32_gcc345/external release
Hit http://cmsdoc.cern.ch cms/cpt/Software/download/cms/apt/slc4_ia32_gcc345/lcg srclist
Hit http://cmsdoc.cern.ch cms/cpt/Software/download/cms/apt/slc4_ia32_gcc345/cms srclist
Hit http://cmsdoc.cern.ch cms/cpt/Software/download/cms/apt/slc4_ia32_gcc345/external srclist
Reading Package Lists...
Building Dependency Tree...
rpminit done
virtual+xsl-virtual+1.0.33-1-1001MD5natasha1158608323 not found in RPM DB, trying installation with apt.
Reading Package Lists...
Building Dependency Tree...
E: Couldn't find package virtual+xsl-virtual+1.0.33
Required RPMs for xsl-virtual: gcc  
slc4_ia32_gcc345
/usr/bin/apt-get
Using apt to fetch available binary packages
Get:1 http://cmsdoc.cern.ch cms/cpt/Software/download/cms/apt/slc4_ia32_gcc345 release [1085B]
Fetched 1085B in 0s (4183B/s)
Hit http://cmsdoc.cern.ch cms/cpt/Software/download/cms/apt/slc4_ia32_gcc345/lcg pkglist
Hit http://cmsdoc.cern.ch cms/cpt/Software/download/cms/apt/slc4_ia32_gcc345/lcg release
Hit http://cmsdoc.cern.ch cms/cpt/Software/download/cms/apt/slc4_ia32_gcc345/cms pkglist
Hit http://cmsdoc.cern.ch cms/cpt/Software/download/cms/apt/slc4_ia32_gcc345/cms release
Hit http://cmsdoc.cern.ch cms/cpt/Software/download/cms/apt/slc4_ia32_gcc345/external pkglist
Hit http://cmsdoc.cern.ch cms/cpt/Software/download/cms/apt/slc4_ia32_gcc345/external release
Hit http://cmsdoc.cern.ch cms/cpt/Software/download/cms/apt/slc4_ia32_gcc345/lcg srclist
Hit http://cmsdoc.cern.ch cms/cpt/Software/download/cms/apt/slc4_ia32_gcc345/cms srclist
Hit http://cmsdoc.cern.ch cms/cpt/Software/download/cms/apt/slc4_ia32_gcc345/external srclist
Reading Package Lists...
Building Dependency Tree...
rpminit done
external+gcc+3.4.5-1-1010MD5natasha1158608322 found in RPM DB@
RPM for gcc already exists
Creating RPM for xsl-virtual...\nTesting installation of the rpm
Preparing...                ##################################################
virtual+xsl-virtual+1.0.33  ##################################################
Relocating to /scratch2/SL4/captor/tryeach
 ok
0  in cpl ===  8  in global
 MPH_local_proc_id()= 0
 MPH_global_proc_id()= 8
  sum of local ranks in cpl is  0
  sum of global ranks in cpl is  8
  sum of global ranks in world is  36
 I am cpl 0 (g=8), recv from ccm 1 (g=7)
 I am cpl 0, expect a2c(1)=100
 cpl 0 gets a2c(1)= 100
 I am cpl 0 (g=8), send to pop 5 (g=5)
 cpl 0 sent to pop 5 successfully
 myProc_joined= 0
  sum of joined ranks in cpl_pop is  21
 I am proc  8  of global,                   proc  0  of coupler         
 ==============================================
GAMIT PROCESSING FOR 063, 98 REGION ties.
 
    RINEX FILES DOWNLOADED
Archive: sio  File: cit10630.98o.Z  Mine: 630787  Theirs: 630787

    SITE STATUS - PROCESSING LIST
Sites being processed:  cit1 nmd1 nmud
Missing Regional Sites:

KWH mod. : TFILE tusgs8.063 CREATED
STATUS :980401:1312: 4.0 FIXDRV/fixdrv: Normal end

   STARTING GAMIT SOLUTION Wed Apr 1 13:12:04 PST 1998
   FINISHED GAMIT SOLUTION Wed Apr 1 13:13:00 PST 1998

    SOLUTION RMS
    Postfit RMS 0.16664E+00
    Postfit RMS 0.16990E+00
    Postfit RMS 0.16733E+00
    Postfit RMS 0.16989E+00

BIAS SUMMARY FOR 063 ties
CIT1-NMD1 26 0
NMD1-NMUD 1 0
NMD1-NMUD 26 0

    PERFORMING BACKUP 
POST CTOX: X-file/C-file sizes and ratios
cit1 717822 2127493 33
nmd1 743445 2211589 33
nmud 738765 2196229 33
/home/dgga/scign/ties/daily/063_es backed up and removed.
<!-- Title: 00:01:20  Wed Oct 25 2006 -->
<!-- Crewchief: Jim H. night -->
<!-- Op1: Steve G. day -->
<!-- Op2: &nbsp; -->
<!-- Op3: &nbsp; -->
<!-- Op4: &nbsp; -->
<!-- Op5: &nbsp; -->
<!-- Op6: &nbsp; -->
<!-- Notes: -->
Monitoring process systems.


<!-- Date: Wed Oct 25 05:09:10 2006 -->
The diesel generator started at 4:57 AM.
<!-- Author: Bill Noe -->


<!-- Date: Wed Oct 25 05:56:37 2006 -->
CDF systems checklist completed. No problems at this time.
<!-- Author: Steve -->


<!-- Date: Wed Oct 25 07:06:18 2006 -->
Steve Shirley from FESS just called to let us know they are planning to pull VP-1 (CPW pumps)to find out what is causing it's troubles.
<!-- Author: Bill Noe -->
<!-- Comment: Wed Oct 25 09;04;41 comment by...Steve -->
Tried starting VP-1 for them, but once again there was no pressure and the system switched to VP-2.


<!-- Comment: Wed Oct 25 11;43;10 comment by...Bill Noe -->
Muskrats walk on four legs like most mammals do. The muskrat has little hairs that are waterproof, and the hair that is called protective guard hair is long and brown. The muskrat has webbed feet for swimming. However, the muskrat has difficulty swimming through a pump. This was the problem with Pond Water pump VP-1. FESS should have it back online today… minus one muskrat.
<!-- Date: Wed Oct 25 10:47:52 2006 -->
There are 29 bottles of alcohol in the cabinet.
<!-- Author: Steve -->



<!-- Date: Wed Oct 25 14:05:15 2006 -->
We are back to running VP-1 with a reading of 72 psig after FESS administered the fix for muskratitis. 
<!-- Author: Steve -->


<!-- Date: Wed Oct 25 15:58:24 2006 -->
Zero'd the gow-mac and started an A/E mix to tank 1.
<!-- Author: Steve -->
<!-- Comment: Wed Oct 25 21;18;49 comment by...Jim H. -->
The mix completed ok.
<!-- Date: Wed Oct 25 18:28:47 2006 -->
The CDF systems checklist and mechanical rooms are normal at this time.
<!-- Author: Jim H. -->



<!-- Date: Wed Oct 25 21:49:39 2006 -->
I put the HVAC on low for an access.
<!-- Author: Jim H. -->
<!-- Comment: Wed Oct 25 22;05;17 comment by...Jim H. -->
The access is over. The HVAC is back on full.
2004.08.30 19:45 B Q:\GSDD\NCD\newsroom\testimony\04testimony.htm --> web2.access.gpo.gov /diska/netsite-docs/ncd/newsroom/testimony 04testimony.htm

2004.10.26 14:05 B Q:\GSDD\NCD\newsroom\testimony\04testimony.htm --> web2bck.access.gpo.gov /diska/netsite-docs/ncd/newsroom/testimony 04testimony.htm

2004.10.26 14:05 B Q:\GSDD\NCD\newsroom\testimony\04testimony.htm --> web2.access.gpo.gov /diska/netsite-docs/ncd/newsroom/testimony 04testimony.htm

2004.10.28 10:41 B Q:\GSDD\NCD\newsroom\testimony\04testimony.htm --> web2.access.gpo.gov /diska/netsite-docs/ncd/newsroom/testimony 04testimony.htm

2004.10.28 10:41 B Q:\GSDD\NCD\newsroom\testimony\04testimony.htm --> web2bck.access.gpo.gov /diska/netsite-docs/ncd/newsroom/testimony 04testimony.htm

2004.10.28 10:43 B Q:\GSDD\NCD\newsroom\testimony\04testimony.htm --> web2bck.access.gpo.gov /diska/netsite-docs/ncd/newsroom/testimony 04testimony.htm

2004.10.28 10:44 B Q:\GSDD\NCD\newsroom\testimony\04testimony.htm --> web2.access.gpo.gov /diska/netsite-docs/ncd/newsroom/testimony 04testimony.htm

2004.11.19 09:51 B Q:\GSDD\NCD\newsroom\testimony\04testimony.htm --> web2.access.gpo.gov /diska/netsite-docs/ncd/newsroom/testimony 04testimony.htm

2004.11.19 09:55 B Q:\GSDD\NCD\newsroom\testimony\04testimony.htm --> web2bck.access.gpo.gov /diska/netsite-docs/ncd/newsroom/testimony 04testimony.htm
<!-- Title: 05:14:58  Mon Oct 29 2001 -->
<!-- Crewchief: Cryo -->
<!-- Op1: Gas -->
<!-- Op2: &nbsp; -->
<!-- Op3: &nbsp; -->
<!-- Op4: &nbsp; -->
<!-- Op5: &nbsp; -->
<!-- Op6: &nbsp; -->
<!-- Notes: -->
Preparing for planned controls outage.


<!-- Date: Mon Oct 29 05:45:02 2001 -->
Working on turning off the cryo system. Rob notified MCR and dispatch to expect alarms.
<!-- Author: Steve -->


<!-- Date: Mon Oct 29 09:16:43 2001 -->
Reduced the N2 baggy flow to 100 scfh per side from 230 per side so the baggy can be slit fo ISL laser work. (per Rich Stanek)
<!-- Author: Steve -->


<!-- Date: Mon Oct 29 09:23:22 2001 -->
During shutdown, On racks control, CDF3:  moved APACS RTM modules from R8S3 to R11S9, and from R8S4 to R11S9.  Moved ACMs from R8S1 and R8S2 to R8S3 and R8S4 
<!-- Author: Sanders -->


<!-- Date: Mon Oct 29 11:27:04 2001 -->
<pre>Reset N2 purge on baggies to 230 scfh per side (what it was set to this morning).  Set up flow to SVX and ISL East at normal flow rates.  Temp setting is 6.0 C for both systems.</pre>
<!-- Author: R Stanek -->


<!-- Date: Mon Oct 29 12:56:53 2001 -->
Recovering cryo system.  <BR><BR> Had problems with FIMF not reading. <BR><BR> The adsorber LN2 valve was closed and Sanders got it open. Then it stuck open. Removed the cover and cleaned out the spiders. <BR><BR> 
Dan Freeman came out and worked on the engines. He put new valves in the east dry and replaced the motor with a spare. Needs a fan mounted on it. He also tightened the belts on the wet and north dry engines. <BR><BR> 
The default settings for the loops came back with different numbers, not reflecting any recent changes. Some loops like the lead flow standby settings had to be re-entered  so it would be recognized.  
<BR><BR> 
The wet engine does not seem to be working as well as last week. The setpoint is 40%.   
<!-- Author: Steve -->


<!-- Date: Mon Oct 29 16:43:48 2001 -->
CLC West lines pressurized to 23.83 psia.
<!-- Author: M. Wong -->


<!-- Date: Tue Oct 30 07:33:59 2001 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=cryo-gas&action=view&page=-330&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=cryo-gas&action=view&page=-331  width='300' align='top'></td><td>The plant came back bumpy. I suspect that the loops that have been tuned have lost their parameters such as gains and pg's, etc. We need to smooth this out before the solenoid is powered. </a></td></tr></table>
<!-- Author: Steve -->


<!-- Date: Tue Oct 30 09:08:50 2001 -->
On EVBP loop controlling dewar changed adtaptive gain variable from 25 to 5.  This should help PVMF from oscillatting (yes, I know its not spelled correctly).
<!-- Author: Sanders -->


<!-- Date: Tue Oct 30 17:40:23 2001 -->
Dean and I found and repaired several leaks on clc west today. The flow meter on the front panel and we glued most of the connections on the rear on 730 platform. We think that the pressure releaf valve may be leaking to the exhaust. We left it pressurized at 24.96 psi at 5:10. We included the lines in the test which we think doesnot leak.
<!-- Author: KC -->
--------------------------------------------------------
Running Yohkoh SW Version: 950729.04

IDL> show_pix 	        (NEW_DATA, Transit, Eclipse, Notable Images)
IDL> first_light,/summ  (Display collage of latest Yohkoh, GBO, and GOES) 
UNIX%                   rmosaic [today sxt trace soho yag sdac ydac selsis news]
 
PROBLEMS? - e-mail TO: software@isass0.solar.isas.ac.jp
--------------------------------------------------------
IDL. Version 3.5.1 (ultrix mipsel).
Copyright 1989-1993, Research Systems, Inc.
All rights reserved.  Unauthorized reproduction prohibited.
Installation number: 994.
Licensed for use by: isass2  --  DECstation 5000/200 PXG

Executing Yohkoh IDL_STARTUP...
Executing Site IDL_STARTUP
% Compiled module: $MAIN$.
Tue Oct 31 18:07:45 1995
junk took     0.0081333339 minutes to run
Importing Model from XMI
***** Root Package: Logical Model *****
Add Classifier Role: java
Add Classifier Role: ValueDomain
Add Classifier Role: org
***** Append Package: java *****
Add Classifier Role: lang
***** Append Package: lang *****
Add Classifier: Integer
Add Classifier: Boolean
Add Classifier: String
Add Classifier: Float
***** Append Package: ValueDomain *****
Add Classifier: GeneNeighborsDistanceMetricType
Add Classifier: GeneNeighborsSimilarOrDifferentIndicator
***** Append Package: org *****
Add Classifier Role: genepattern
***** Append Package: genepattern *****
Add Classifier Role: cagrid
***** Append Package: cagrid *****
Add Classifier: GeneNeighborsParameterSet
***** Resolving Classifiers *****
***** Resolving Owners *****
***** Resolve Package Links *****
***** Add Extensions ******
***** Add Diagrams *****
Adding Diagram : Domain Model
***** Resolving DiagramIDs *****
***** Resolving Association Classes *****
***** Resolving Parameter Types *****
***** Resolving Attribute Types *****
***** Resolving Note links *****
***** Fixup external references *****
Import Complete!
pact is get-new-JobTransforms-12.0.6-12.0.6.5
 
 ########################################################### 
 
 The input values you specify for this script to run are :   
 
 Remote gate-keeper is "hep.westgrid.ca"
 Remote Gridftp server is "hep.westgrid.ca"
 Remote $APP area is "/export/LHC/software/Panda"
 Remote $DATA area is "/export/LHC/software/Panda"
 
 ########################################################### 
 
Tue Aug  7 12:57:54 PDT 2007
Sending the install script "ace-install-template-hep.westgrid.ca-get-new-JobTransforms-12.0.6-12.0.6.5-2007-08-07T15-0400-5069" to remote site "hep.westgrid.ca". 
globus-url-copy file:\/\/\//direct/usatlas+u/atlaswin/remote-install/scripts\/ace-install-template-hep.westgrid.ca-get-new-JobTransforms-12.0.6-12.0.6.5-2007-08-07T15-0400-5069 gsiftp\:\/\/hep.westgrid.ca\//export/LHC/software/Panda\/ace-install-template-hep.westgrid.ca-get-new-JobTransforms-12.0.6-12.0.6.5-2007-08-07T15-0400-5069
Sending the mdsupdate script "fix-MDSConfig-hep.westgrid.ca-get-new-JobTransforms-12.0.6-12.0.6.5-2007-08-07T15-0400-5069" to remote site "hep.westgrid.ca". 
Sending the testrun script "KVtest.sh" to remote site "hep.westgrid.ca". 
Sending the AGLT2 patch script "copy_libs_ATLAS.sh" to remote site "hep.westgrid.ca". 
Running the install script in "/export/LHC/software/Panda" on "hep.westgrid.ca"
hostname is  hep.westgrid.ca
PACMAN_CMD is  pacman
installing pacman pacman-3.20.tar.gz in "/export/LHC/software/Panda" ...
Doing a pacman "get-new-JobTransforms-12.0.6-12.0.6.5" on hep.westgrid.ca now... 
Package [AtlasProduction_12_0_6_5_i686_slc3_gcc323_opt] found in [http://cern.ch/atlas-computing/links/kitsDirectory/Production/cache]...
Package [AtlasProduction_12_0_6_5_noarch] found in [http://cern.ch/atlas-computing/links/kitsDirectory/Production/cache]...
Package [AtlasProduction_12_0_6_5_noarch] found in [http://cern.ch/atlas-computing/links/kitsDirectory/Production/cache]...
Downloading [AtlasProduction_12_0_6_5_i686_slc3_gcc323_opt.tar.gz] from [http://cern.ch/atlas-computing/links/kitsDirectory/Production/cache/../kits]...
Untarring [AtlasProduction_12_0_6_5_i686_slc3_gcc323_opt.tar.gz]...
Downloading [AtlasProduction_12_0_6_5_noarch.tar.gz] from [http://cern.ch/atlas-computing/links/kitsDirectory/Production/cache/../kits]...
Untarring [AtlasProduction_12_0_6_5_noarch.tar.gz]...
Generating cmt setup scripts...
Python JobTransforms 12_0_6_5 is installed
gatekeeper is hep.westgrid.ca
Cleaning remote install scripts and tmp files...
Installation get-new-JobTransforms-12.0.6-12.0.6.5 is done on site hep.westgrid.ca.
100.02.15 15:23 B C:\TEMP\vickitemp\of00_027\data\plot_files\postscript\em\33k.ps.gz <-- musette.cr.usgs.gov /tmp_mnt/hime/d3/vrystrom/of00_027/data/plot_files/postscript/em 33k.ps.gz
100.02.15 15:23 B C:\TEMP\vickitemp\of00_027\data\plot_files\postscript\em\4341.ps.gz <-- musette.cr.usgs.gov /tmp_mnt/hime/d3/vrystrom/of00_027/data/plot_files/postscript/em 4341.ps.gz
100.02.15 15:23 B C:\TEMP\vickitemp\of00_027\data\plot_files\postscript\em\4737.ps.gz <-- musette.cr.usgs.gov /tmp_mnt/hime/d3/vrystrom/of00_027/data/plot_files/postscript/em 4737.ps.gz
100.02.15 15:23 B C:\TEMP\vickitemp\of00_027\data\plot_files\postscript\em\837.ps.gz <-- musette.cr.usgs.gov /tmp_mnt/hime/d3/vrystrom/of00_027/data/plot_files/postscript/em 837.ps.gz
100.02.15 15:30 B C:\TEMP\vickitemp\of00_027\data\plot_files\postscript\em\33k.ps.gz --> ctr6.cr.usgs.gov /incoming/Rystrom/of00_027/data/plot_files/postscript/em 33k.ps.gz
100.02.15 15:30 B C:\TEMP\vickitemp\of00_027\data\plot_files\postscript\em\4341.ps.gz --> ctr6.cr.usgs.gov /incoming/Rystrom/of00_027/data/plot_files/postscript/em 4341.ps.gz
100.02.15 15:30 B C:\TEMP\vickitemp\of00_027\data\plot_files\postscript\em\4737.ps.gz --> ctr6.cr.usgs.gov /incoming/Rystrom/of00_027/data/plot_files/postscript/em 4737.ps.gz
100.02.15 15:30 B C:\TEMP\vickitemp\of00_027\data\plot_files\postscript\em\837.ps.gz --> ctr6.cr.usgs.gov /incoming/Rystrom/of00_027/data/plot_files/postscript/em 837.ps.gz
<!-- Title: 00:00:23  Mon Aug  9 2004 -->
<!-- Crewchief: KC... nights -->
<!-- Op1: Jim...days -->
<!-- Op2: &nbsp; -->
<!-- Op3: &nbsp; -->
<!-- Op4: &nbsp; -->
<!-- Op5: &nbsp; -->
<!-- Op6: &nbsp; -->
<!-- Notes: -->
Goodmorning, all systems are running fine. 


<!-- Date: Mon Aug 9 00:42:03 2004 -->
I put the MKS cold cathode ionization guage transducer back together. I cleaned these parts last week in the ultrasonic cleaner using alconox. Bob and I or Steve may test it this week. I expect very good results, maybe like new. Bob ordered a rebuild kit that should be in soon.
<!-- Author: KC -->


<!-- Date: Mon Aug 9 05:58:25 2004 -->
Cryo and gas readings are normal at this time. The CDF systems checklist and mechanical rooms are also normal at this time.
<!-- Author: Jim H. -->


<!-- Date: Mon Aug 9 07:31:15 2004 -->
Started an A/E mix into tank #3.
<!-- Author: Jim H. -->
<!-- Comment: Mon Aug  9 10;46;25 comment by...Jim H. -->
The mix stopped on system watch.
<!-- Comment: Mon Aug  9 11;38;06 comment by...Jim H. -->
Zeroed the gowmac and restarted the mix.
<!-- Comment: Mon Aug  9 13;03;51 comment by...Jim H. -->
The mix stopped again on system watch. Couldn't find anything wrong. Tank pressure is 266psig so I started recirculation.
<!-- Comment: Mon Aug  9 13;23;07 comment by...Jim H. -->
Del is going to look in 4-mation to find the problem.
<!-- Comment: Mon Aug  9 13;37;11 comment by...Jim H. -->
Recirculation is complete.
<!-- Comment: Mon Aug  9 14;30;08 comment by...D. Allspach -->
<table border=0><td><tr><td bgcolor=#e0e030>The Liquid Argon Dewar is going dry. Mix tripped on low argon supply pressure. The dewar must be filled before we perform another mix!</td></tr></td></table>
<!-- Date: Mon Aug 9 09:36:42 2004 -->
The COT Baggy Ethane Level Warning Limit has been changed to 25%. Change made in Structured Text "GAS_DET_COTBOX" on sheet CDF4.GAS_CH2.
<!-- Author: D. Allspach -->


<!-- Date: Mon Aug 9 09:39:06 2004 -->
A/E/O2 cylinder pressure for the COT is at 120 psig.
<!-- Author: WB -->








<!-- Date: Mon Aug 9 15:52:16 2004 -->
<table border=0><td><tr><td bgcolor=#e0e030>Called BOC air gas about an argon fill. They are checking on it and  will call back. By the way the new phone number is 630/876/2559.</td></tr></td></table>
<!-- Author: Jim H. -->


<!-- Date: Mon Aug 9 20:06:57 2004 -->
I did the CDF system checklist and all systems are doing fine.
<!-- Author: KC -->


<!-- Date: Mon Aug 9 23:54:26 2004 -->
I tripped the 400 hz on RR# SVNWT on Detector Racks West by mistake for Wallny #10750B. Then I reset it. I turned off RR# SVNWB and now waiting the 15 min to reset.
<!-- Author: KC -->
<!-- Comment: Mon Aug  9 23;58;09 comment by...wallny 10750V -->
<pre>the 400 Hz does not do anything on fib crates so no damage was done. in order to powercycle fib crates, one needs to switch OFF the whole crate (my mistake)</pre>