<!-- Title: Day shift. Wed Jun  4, 2008 -->
<!-- SciCo: Duncan Carlsmith -->
<!-- DAQAce: Louise Oakes -->
<!-- MonAce: &nbsp; -->
<!-- CO: Silvia Amerio -->
<!-- OpManager: Steve Hahn -->
<!-- Notes: -->
<pre>Start shift with collision just starting and an apparent problem with L3 that is resolved. Store 6191, cdflum232e30, stack=23,rbeam=111,accrat=21. Plan to, gosh, take data!</pre>


<!-- Date: Wed Jun 4 08:17:32.813 2008 -->
cot16 done TO - auto HRR
<!-- Author: Louise :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263057' target=_top>263057</a>) -->


<!-- Date: Wed Jun 4 08:18:16.661 2008 -->
imu01 done TO - auto HRR
<!-- Author: Louise :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263057' target=_top>263057</a>) -->


<!-- Date: Wed Jun 4 08:23:06.553 2008 -->
L2 decision TO - auto HRR
<!-- Author: Louise :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263057' target=_top>263057</a>) -->
<!-- Comment: Wed Jun  4 08;36;54 comment by...Louise -->
again

<!-- Date: Wed Jun 4 08:24:12.407 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263034>263034</A>
general: <tt>checked. Ok.</tt>
<!-- Author: Consumer Operator(Silvia Amerio) -->



<!-- Date: Wed Jun 4 08:26:42.866 2008 -->
Trigger inhibit set from plug HV trip - crate 6. LED for crate 6 was out, so reset following plug HV insetructions
<!-- Author: Louise :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263057' target=_top>263057</a>) -->


<!-- Date: Wed Jun 4 08:33:34.205 2008 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2008&action=view&page=-2723&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-2723  width='300' align='top'></a></td><td>Loss/ ab gaps for start of store 6191 
</td></tr></table>
<!-- Author: Louise -->


<!-- Date: Wed Jun 4 08:34:06.320 2008 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2008&action=view&page=-2724&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-2724  width='300' align='top'></a></td><td>svrad integ for start of store 6191</td></tr></table>
<!-- Author: Louise -->


<!-- Date: Wed Jun 4 08:36:42.891 2008 -->
Front end crate error from svt_tracking_00, auto HRR
<!-- Author: Louise :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263057' target=_top>263057</a>) -->



<!-- Date: Wed Jun 4 08:49:50.280 2008 -->
<table><tr><td><a href=/elog/notebooks/2008/Wed_Jun__4_2008_Day.084927.ps align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-2726  align='top'></a></td><td>Error rate > 0.001 for tower at phi 7, eta 15, east</td></tr></table>
<!-- Author: Silvia -->
<!-- Comment: Wed Jun  4 09;19;39 comment by...Silvia -->
The Expert (Carla) said that this error is due only to a single count difference between data and simulation (the plot has low statistics). If the error is still present at high statistics, we have to call her.
<!-- Date: Wed Jun 4 08:52:18.830 2008 -->
Hot tower in l1 calorimetry. Paged expert.
<!-- Author: Carlsmith :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263057' target=_top>263057</a>) -->

<!-- Date: Wed Jun 4 09:00:37.006 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263057>263057</A>
trigger: <tt>high error rate in a tower in &nbsp;125MeV_DCAS_Occupancy plot. Expert paged.</tt>
<!-- Author: Consumer Operator(Silvia Amerio) -->


<!-- Date: Wed Jun 4 09:00:38.927 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263057>263057</A>
general: <tt>Checlist done. Ok.</tt>
<!-- Author: Consumer Operator(Silvia Amerio) -->



<!-- Date: Wed Jun 4 09:02:33.987 2008 -->
pcal07 done TO - needed shepherding to fix
<!-- Author: Louise :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263057' target=_top>263057</a>) -->


<!-- Date: Wed Jun 4 09:06:00.961 2008 -->
svxmon error - auto HRR
<!-- Author: Louise :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263057' target=_top>263057</a>) -->



<!-- Date: Wed Jun 4 09:36:22.189 2008 -->
Pinkie storm. Greater than 24 pinkies. IMON Java exception out of memory too after I made a few plots. Paged expert.
<!-- Author: Carlsmith :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263057' target=_top>263057</a>) -->
<!-- Comment: Wed Jun  4 09;45;26 comment by...Carlsmith -->
Sasha responds and explains (just a bit defensively, admitting he is the author ;) ) how to kill yellow too many pinkies boxes then close not kill plot history windows (killing introduces the memory leak) then exit the script then restart IMON following instructions next to monitor.

He will mass ignore these pinkies and adjust limits.
<!-- Comment: Wed Jun  4 10;31;23 comment by...Sasha Pronko -->
Small correction.... I was not admitting anything :-), especial the authorship of IMOn. But I'll come back later today to adjust Bias limits. For time being, please put those Bias pinkies into "Ignore". 
<!-- Date: Wed Jun 4 09:44:27.882 2008 -->
Note to shiftcrew: Please power-cycle pcal11 when the current store is dropped and follow with smx calibration.  This is to address the bad channels that were noted in the e-log after calibrations during owl shift.  Thank you.
<!-- Author: Britney Rutherford :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263057' target=_top>263057</a>) -->
<!-- Comment: Wed Jun  4 14;35;47 comment by...Louise -->
I powercycled b0pcal11 
<!-- Date: Wed Jun 4 09:54:14.244 2008 -->
Pinkie flurry (6 pinkies) not storm yet. Paged expert. Previous storm ascribed to prescale change. Advised to check these and ignore if small.
<!-- Author: Carlsmith :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263057' target=_top>263057</a>) -->


<!-- Date: Wed Jun 4 09:56:24.842 2008 -->
Phil informs us the plan is to stack to 350 then shoot. Rbeam is now 302 so that suggests a store termination early afternoon.
<!-- Author: Carlsmith :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263057' target=_top>263057</a>) -->
<!-- Comment: Wed Jun  4 11;50;04 comment by...Carlsmith -->
Correction. I was wondering how they made pbars so fast or had em stashed. Maybe I was looking at an old Acc Division page. RBeam is 140 at 11:49. We have a long way to go.
<!-- Comment: Wed Jun  4 12;15;43 comment by...Phil -->
Runco estimated 8-10 pm at the meeting. There were a couple of hiccups since so maybe it's later.
<!-- Date: Wed Jun 4 10:20:39.999 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263057>263057</A>
 Terminated at 2008.06.04 10:20:14
<!-- Author: RunControl -->


<!-- Date: Wed Jun 4 10:21:02.336 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263057>263057</A>
TERMINATE: stuck in yellow halt state and need to reset L00 trip
<!-- Author: Louise -->
<!-- Comment: Wed Jun  4 10;35;45 comment by...SashaS -->
L00 trip was actually a corrupted readout of L00 B1W0L1 Bias0 voltage. This set L00 trigger inhibit. I hockerized CAEN crate 15 and SAR turned affected L00 and ISL wedges back to ON.
<!-- Date: Wed Jun 4 10:26:12.274 2008 -->
If you again encounter the ProcMon L2Control "Permission denied" failure as reported in the owl shift <a href="http://www-cdfonline.fnal.gov/cgi/internal/elog/elog.pl?nb=2008&action=copyentry&page=466&time=01:17:07.935">here</a>, please disable the L2control check. The failure indicates a problem with $FER_DIR/src/procAction kerberos authentication for "ssh b0felix" and is not related to L2 control/decision processes. 
<!-- Author: Pedro :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263057' target=_top>263057</a>) -->

<!-- Date: Wed Jun 4 10:27:52.902 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263058>263058</A>
 Activated at 2008.06.04 10:27:37
<!-- Author: RunControl -->


<!-- Date: Wed Jun 4 10:27:53.778 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263058>263058</A>
ACTIVATE: RunSet: AAA_CURRENT  Trigger: PHYSICS_5_01 [7,1088,721] 
<!-- Author: Louise -->





<!-- Date: Wed Jun 4 10:39:17.783 2008 -->
Trigger inhibit from L00 HV.  On ifix alarm display PS-CAEN-OFF was in alarm.  Spoke to Sasha who said that this was a readout problem - the voltage was not actually down but the readout was zero, this was causing the trigger inhibit.  To fix this, crate 15 needed Hockerizing which requires RC to be in start or halt state.  I couldn't get it into the halt state because the TI occured at the same time as an SVT crate done TO which had triggered an HRR.  It ended with Halted in yellow, not green and the only way I could sort this out was to end the run and get into the start state so that SAR could work after the Hockerization.
<!-- Author: Louise :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263058' target=_top>263058</a>) -->

<!-- Date: Wed Jun 4 10:48:01.812 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263057>263057</A>
general: <tt>final checklist done. OK.</tt>
<!-- Author: Consumer Operator(Silvia Amerio) -->



<!-- Date: Wed Jun 4 10:49:04.370 2008 -->
CMX trip - SW channel 15.  Switching the whole CMX on didn't help, switched on that channel from bars which fixed it
<!-- Author: Louise :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263058' target=_top>263058</a>) -->


<!-- Date: Wed Jun 4 10:51:39.126 2008 -->
svt_trackin_00 rxpt error, auto HRR
<!-- Author: Louise :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263058' target=_top>263058</a>) -->
<!-- Comment: Wed Jun  4 11;41;20 comment by...Louise -->
again
<!-- Date: Wed Jun 4 10:56:18.438 2008 -->
Icicle heartbeat alarm - restarted icicle, wait to see if it clears
<!-- Author: Louise :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263058' target=_top>263058</a>) -->
<!-- Comment: Wed Jun  4 11;03;48 comment by...Louise -->
it's cleared

<!-- Comment: Wed Jun  4 11;05;22 comment by...Louise -->
now icilce is red in daqmon - restarted again and it went back to green


<!-- Date: Wed Jun 4 11:09:45.309 2008 -->
DQmon - L00, svxmon found errors in following cells: Silicon/L/B1/W5/L0/C1.  CO is checking this
<!-- Author: Louise :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263058' target=_top>263058</a>) -->
<!-- Comment: Wed Jun  4 11;41;42 comment by...Louise -->
Sasha said to ignore it as long as nothing is yellow in svxmon
<!-- Date: Wed Jun 4 11:26:43.058 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263058>263058</A>
general: <tt>checlist done. ok.</tt>
<!-- Author: Consumer Operator(Silvia Amerio) -->



<!-- Date: Wed Jun 4 11:27:06.592 2008 -->
pcal05 in error - node is not responsive.  Shepherded to recover
<!-- Author: Louise :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263058' target=_top>263058</a>) -->





<!-- Date: Wed Jun 4 12:14:45.271 2008 -->
<a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=operations&action=view&page=193&frame=2&anchor=12:11:03'> 
observations</a> by the dead time police
<!-- Author: Phil -->



<!-- Date: Wed Jun 4 12:33:17.491 2008 -->
An rf station has tripped according to MCR. Abort gap losses jumped. Si alarmed and went into standby. Si paged. Sahsha responds.
<!-- Author: Carlsmith :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263058' target=_top>263058</a>) -->


<!-- Date: Wed Jun 4 12:34:04.850 2008 -->
Phil reports the plan is to abort this store at 4 pm to do Tev work and we will have 1.5 hours access. He is rounding up the troops.
<!-- Author: Carlsmith :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263058' target=_top>263058</a>) -->


<!-- Date: Wed Jun 4 12:34:41.491 2008 -->
<table border=0><td><tr><td bgcolor=#e0e030>RunCo called to loe us know that due to the cryo problems on one of the quads we  may get an 1.5hr access sometime around 4pm.</td></tr></td></table>
<!-- Author: Sebastian -->

<!-- Date: Wed Jun 4 12:40:54.089 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263058>263058</A>
 Terminated at 2008.06.04 12:40:48
<!-- Author: RunControl -->


<!-- Date: Wed Jun 4 12:42:02.586 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263058>263058</A>
TERMINATE: Ended due to loss of RF station - restart without Silicon
<!-- Author: Louise -->



<!-- Date: Wed Jun 4 12:46:11.265 2008 -->
Discussed situation with MCR. Our b0aagc jumped to around 5 khz and is looking rattioer now. MCR watches T:AG1G12 and wasn't interested. MCR says beam is stable and no decision has yet been taken to abort earlier than 4 pm. Sasha makes the call to run without silicon.
<!-- Author: Carlsmith :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263058' target=_top>263058</a>) -->
<!-- Comment: Wed Jun  4 13;12;10 comment by...Louise -->
TevMon automatically took silicon to standby, but L00 tripped on the way down.
<!-- Comment: Wed Jun  4 13;53;48 comment by...Sasha Pronko -->
TevMon sent Silicon to "Standby" after an RF trip. L00 ladder tripped while voltages were rumping down. To prevent Silicon being sent back to ON by SAR while was it recovering L00 ladder, I instructed Luise (the ACE) to turn SAR off (what she eventually did). While I was recovering L00, I instructed the ACE to start a non-silicon run because anti-proton abort gap losses didn't look good (see <a href=
'http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2008&action=view&page=467&frame=2&anchor=12:47:11.006'>here</a>).
During recovery of L00, Sasha Sukhanov, Ricardo and I realized that PS-8 gone bad (PS powering tripped L00 ladder). We hockerized the crate and successfully recovered the offensive ladder after that. Silicon was integrated back to datataking.   
<!-- Date: Wed Jun 4 12:46:39.754 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263059>263059</A>
 Activated at 2008.06.04 12:46:33
<!-- Author: RunControl -->


<!-- Date: Wed Jun 4 12:46:50.824 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263059>263059</A>
ACTIVATE: RunSet: AAA_NOSILICON  Trigger: PHYSICS_5_01 [7,1088,721] 
<!-- Author: Louise -->



<!-- Date: Wed Jun 4 12:47:11.006 2008 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2008&action=view&page=-2727&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-2727  width='300' align='top'></a></td><td>Antiproton abort gap losses after RF station trip</td></tr></table>
<!-- Author: Louise -->

<!-- Date: Wed Jun 4 12:56:21.734 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263059>263059</A>
 Terminated at 2008.06.04 12:56:04
<!-- Author: RunControl -->


<!-- Date: Wed Jun 4 12:57:33.309 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263059>263059</A>
TERMINATE: without silicon while waiting for beam losses to settle
<!-- Author: Louise -->


<!-- Date: Wed Jun 4 13:02:00.741 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263060>263060</A>
TERMINATE: failed config
<!-- Author: Louise -->


<!-- Date: Wed Jun 4 13:05:40.821 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263058>263058</A>
general: <tt>final checklist done. Ok.</tt>
<!-- Author: Consumer Operator(Silvia Amerio) -->


<!-- Date: Wed Jun 4 13:07:46.047 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263061>263061</A>
 Activated at 2008.06.04 13:07:36
<!-- Author: RunControl -->


<!-- Date: Wed Jun 4 13:08:44.502 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263061>263061</A>
ACTIVATE: RunSet: AAA_CURRENT  Trigger: PHYSICS_5_01 [7,1088,721] 
<!-- Author: Louise -->




<!-- Date: Wed Jun 4 13:15:58.250 2008 -->
Back online running with silicon and somehwat elevated antiproton abort gap losses.  
 
MCR gives us 15 minute warning for end of store. Expects at tiddy bit (5 minutes) of tev studies, then "several" hours of TeV access.
<!-- Author: Carlsmith :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263061' target=_top>263061</a>) -->


<!-- Date: Wed Jun 4 13:16:33.584 2008 -->
pcal04 in error - needed shepherding
<!-- Author: Louise :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263061' target=_top>263061</a>) -->

<!-- Date: Wed Jun 4 13:19:28.995 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263061>263061</A>
 Terminated at 2008.06.04 13:19:02
<!-- Author: RunControl -->


<!-- Date: Wed Jun 4 13:19:29.100 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263061>263061</A>
TERMINATE: End of store
<!-- Author: Louise -->



<!-- Date: Wed Jun 4 13:20:31.475 2008 -->
Nother random pointless "Cryo alarm" with nothing red. Steve (process tech) didn't understand it before and still doesn't. 
Ops man Steve says access time translates more precisely to "several hours"== 1.3 to 3 or more
<!-- Author: Carlsmith :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263061' target=_top>263061</a>) -->


<!-- Date: Wed Jun 4 13:20:46.045 2008 -->
<p>Just talked to Cons.  They have put us down for a minimum 
1 1/2 hours in the collision hall.  Depending on what they find 
regarding the water leak, this could go to 3 hour or more (or 
it could remain just 1 1/2 hours).</p>
<!-- Author: Steve Hahn :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263061' target=_top>263061</a>) -->

<!-- Date: Wed Jun 4 13:46:06.111 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263062>263062</A>
 Activated at 2008.06.04 13:45:54
<!-- Author: RunControl -->


<!-- Date: Wed Jun 4 13:46:21.385 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263062>263062</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,1019,706] 
<!-- Author: Louise -->



<!-- Date: Wed Jun 4 13:55:07.908 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263061>263061</A>
clc: <tt>low statistics run. in cotcanvas1 mean Z is in the good range but there are only 416 entries.</tt>
<!-- Author: Consumer Operator(Silvia Amerio) -->


<!-- Date: Wed Jun 4 13:55:08.637 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263061>263061</A>
general: <tt>low stat run. Checklist done. ok.</tt>
<!-- Author: Consumer Operator(Silvia Amerio) -->



<!-- Date: Wed Jun 4 14:01:33.763 2008 -->
MCR says collision hall is ready for access. Steve and Sebastian are going in to work on a COT TDC crate 1. 
Page Shin-Shan Eiko Yu to do post access TDC calibration.
<!-- Author: Carlsmith :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263062' target=_top>263062</a>) -->
<!-- Comment: Wed Jun  4 14;28;11 comment by...Phil -->
They are swapping a transition card for the TDC. This is the next step in tracking down the "XTC" problem now known not to be the XTC.
<!-- Comment: Wed Jun  4 14;37;29 comment by...Carlsmith -->
Unable to reach Shin-Shan (pager, cell, office).
<!-- Comment: Wed Jun  4 14;53;27 comment by...Carlsmith -->
Here she is!
<!-- Comment: Wed Jun  4 14;58;50 comment by...Eiko -->
sorry, was in the User's meeting.
<!-- Date: Wed Jun 4 14:05:46.976 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263062>263062</A>
 Terminated at 2008.06.04 14:05:33
<!-- Author: RunControl -->


<!-- Date: Wed Jun 4 14:07:36.415 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263063>263063</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Louise -->


<!-- Date: Wed Jun 4 14:09:20.613 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263063>263063</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Wed Jun  4 14;23;10 comment by...Silvia -->
Checked. Everything ok, apart the <a href=http://www-cdfonline.fnal.gov/cgi/internal/elog/elog.pl?nb=2008&action=copyentry&page=466&time=01:47:40.599> known problem </a> on WHA 7E channel 1 


<!-- Date: Wed Jun 4 14:10:15.411 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263064>263064</A>
ACTIVATE: RunSet: CALIB_CLC_QIE 
<!-- Author: Louise -->


<!-- Date: Wed Jun 4 14:11:23.423 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263064>263064</A>
CALIBRATION: <U><B>clcqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#clcqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Wed Jun  4 14;26;05 comment by...Silvia -->
Checked. ok. 
<!-- Date: Wed Jun 4 14:19:14.837 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263065>263065</A>
CALIBRATION: <U><B>bscqie</B></U> calibration FAILED, please investigate
<!-- Author: CDFCALIB -->



<!-- Date: Wed Jun 4 14:22:03.403 2008 -->
Farrukh and Bill are running a 15 minute IO test with cosmics. 
Jonathan Lewis has a shopping list of things to fix during the access to discuss with Steve when Steve returns from the hall.
<!-- Author: Carlsmith :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263066' target=_top>263066</a>) -->



<!-- Date: Wed Jun 4 14:26:22.966 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263068>263068</A>
 Activated at 2008.06.04 14:26:09
<!-- Author: RunControl -->


<!-- Date: Wed Jun 4 14:26:23.570 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263068>263068</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,1019,707] 
<!-- Author: Louise -->
<!-- Comment: Wed Jun  4 14;27;15 comment by...badgett -->
This run has CAL_PULSER_01 crate, which contains the 
new TFAD bank;  this bank was mysteriously being 
dropped in our previous test with AAA_SHOTSETUP;  
we're trying a NULL L3 exe now.


<!-- Date: Wed Jun 4 14:28:44.638 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263068>263068</A>
 Terminated at 2008.06.04 14:28:28
<!-- Author: RunControl -->




<!-- Date: Wed Jun 4 14:40:43.530 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263070>263070</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_5_01 [7,1088,721] 
<!-- Author: Louise -->


<!-- Date: Wed Jun 4 14:44:39.995 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263071>263071</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Louise -->


<!-- Date: Wed Jun 4 14:45:49.309 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263071>263071</A>
TERMINATE: SMX calibration as requested after powercycling b0pcal11
<!-- Author: Louise -->


<!-- Date: Wed Jun 4 14:47:30.956 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263072>263072</A>
ACTIVATE: RunSet: CALIB_QIE_FWD 
<!-- Author: Louise -->


<!-- Date: Wed Jun 4 14:48:11.135 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263072>263072</A>
CALIBRATION: <U><B>bscqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#bscqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Wed Jun  4 15;10;36 comment by...Silvia -->
Checked. OK.
<!-- Comment: Wed Jun  4 15;13;27 comment by...Silvia -->
Checked. OK.
<!-- Date: Wed Jun 4 14:48:40.324 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263071>263071</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Wed Jun  4 15;13;56 comment by...Silvia -->
Checked. OK.
<!-- Date: Wed Jun 4 14:52:31.069 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263073>263073</A>
ACTIVATE: RunSet: ACE_TOF_QIE_NEW 
<!-- Author: Louise -->


<!-- Date: Wed Jun 4 14:52:43.225 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263073>263073</A>
CALIBRATION: <U><B>tofqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#tofqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Wed Jun  4 15;15;03 comment by...Silvia -->
Checked. OK.
<!-- Date: Wed Jun 4 14:58:03.923 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263074>263074</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_5_01 [7,1088,721] 
<!-- Author: Louise -->
<!-- Comment: Wed Jun  4 15;20;38 comment by...badgett -->
<P>
This run was a success in reading out the TFAD_StorableBank 
without any problems.   Checked all streams in the lookarea, 
and they all had at least one instance.   Stream G had 
the bank on every event, as expected on readoutlist 7.
<P>
The next step will be to request an end-of-store test.
<P>

<!-- Date: Wed Jun 4 15:02:32.298 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263075>263075</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL 
<!-- Author: Louise -->


<!-- Date: Wed Jun 4 15:05:01.921 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263076>263076</A>
ACTIVATE: RunSet: COT_ACE after work on b0cot11
<!-- Author: Eiko X 2080 -->


<!-- Date: Wed Jun 4 15:06:43.736 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263077>263077</A>
ACTIVATE: RunSet: TDCTEST_B0DAIG01 
<!-- Author: Eiko X 2080 -->



<!-- Date: Wed Jun 4 15:11:27.353 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263078>263078</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: Louise -->



<!-- Date: Wed Jun 4 15:13:56.544 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263078>263078</A>
CALIBRATION: <U><B>pluglaser</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#pluglaser target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Wed Jun  4 15;16;33 comment by...Silvia -->
Checked. OK.


<!-- Date: Wed Jun 4 15:17:07.933 2008 -->
Water alarm. Process tech investigating a compressor 4 failure.
<!-- Author: Carlsmith :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263078' target=_top>263078</a>) -->



<!-- Date: Wed Jun 4 15:25:58.201 2008 -->
Shalhout requested that we powercycle crate wcal05 then redo CAL QIE calibration
<!-- Author: Louise -->


<!-- Date: Wed Jun 4 15:26:11.519 2008 -->
<pre>WCAL05 slot 17 channel 1 is showing zero slopes in recent calqie calibration. 
 
A pedestal run shows a low mean value for slot 17 channel 1 : 
 
   17    0    3    1000   0.0      170.7   2.0 
   17    1    0    1000   0.0       47.0   0.0 
   17    1    1    1000   0.0      196.1   1.6 
   17    1    2    1000   0.0      193.2   1.6 
 
I have asked for a powercycle of the crate and a repeat of the calibration.</pre>
<!-- Author: Shalhout -->

<!-- Date: Wed Jun 4 15:27:11.054 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263079>263079</A>
 Activated at 2008.06.04 15:26:59
<!-- Author: RunControl -->


<!-- Date: Wed Jun 4 15:27:12.194 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263079>263079</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,1019,706] 
<!-- Author: Louise -->


<!-- Date: Wed Jun 4 15:29:42.459 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263079>263079</A>
 Terminated at 2008.06.04 15:28:46
<!-- Author: RunControl -->


<!-- Date: Wed Jun 4 15:31:07.081 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263080>263080</A>
ACTIVATE: RunSet: BOTHPLUGS_MM 
<!-- Author: Louise -->


<!-- Date: Wed Jun 4 15:35:55.027 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263081>263081</A>
 Activated at 2008.06.04 15:35:42
<!-- Author: RunControl -->


<!-- Date: Wed Jun 4 15:35:56.724 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263081>263081</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [7,1023,706] 
<!-- Author: Louise -->



<!-- Date: Wed Jun 4 15:42:49.391 2008 -->
Yuri will update the CLC acceptances, rebooting the PC
<!-- Author: Carlsmith :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263081' target=_top>263081</a>) -->

<!-- Date: Wed Jun 4 15:43:43.980 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263081>263081</A>
 Terminated at 2008.06.04 15:43:34
<!-- Author: RunControl -->



<!-- Date: Wed Jun 4 15:44:15.133 2008 -->
Mike Lindgren found power cylcing PSM crate and running shower max calib still showed 20 odd bad channels. He is stepping out for 1/2 hour and will check back if/after the 400 Hz is power cycled.
<!-- Author: Carlsmith :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263081' target=_top>263081</a>) -->

<!-- Date: Wed Jun 4 15:44:40.159 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263081>263081</A>
TERMINATE: L1 accept rate 64.2Hz
<!-- Author: Louise -->



<!-- Date: Wed Jun 4 15:45:41.413 2008 -->
<pre>Updated the acceptances for CLC. It will result in instant luminosity increase of about 5%</pre>
<!-- Author: Yuri -->
<!-- Comment: Wed Jun  4 15;49;48 comment by...Y -->
refilling CLC west now...
<!-- Date: Wed Jun 4 15:46:34.443 2008 -->
I changed a XTRP board (#11) for a spare one. This board was giving  us some problem. We will take it to Urbana to check. 
<!-- Author: Olga Norniella :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263081' target=_top>263081</a>) -->

<!-- Date: Wed Jun 4 15:47:17.282 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263083>263083</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: Louise -->


<!-- Date: Wed Jun 4 15:48:30.298 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263083>263083</A>
CALIBRATION: <U><B>cemled</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cemled target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Wed Jun  4 15;57;21 comment by...Silvia -->
Checked. ok.
<!-- Date: Wed Jun 4 15:50:40.384 2008 -->
CLC gas will be "refueled." This will take 15 minutes.
<!-- Author: Carlsmith :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263083' target=_top>263083</a>) -->


<!-- Date: Wed Jun 4 15:54:03.945 2008 -->
There was a stuck bit in b0xft06 slot 14.<br> 
Heather + Nils suggested to swap the board with its neighbour (slot 13).<br> 
Swapped boards! <br> 
Heather has TrigMon plots to check this when beam comes back.<br>
<!-- Author: Alison -->
<!-- Comment: Wed Jun  4 17;54;48 comment by...Heather -->
An example of the original problem is <a href= 'http://www-cdfonline.fnal.gov/cgi/internal/elog/elog.pl?nb=2008&action=copyentry&page=434&time=12:03:26.094'> here </a>.  I tried to "fix" it by masking on the offending channel <a href= 'http://www-cdfonline.fnal.gov/cgi/internal/elog/elog.pl?nb=2008&action=copyentry&page=446&time=11:34:16.111'> here </a>.  It didn't <a href = 'http://www-cdfonline.fnal.gov/cgi/internal/elog/elog.pl?nb=2008&action=copyentry&page=446&time=15:58:28.642'>  help </a>), and looked like a but stuck low in the finder for the particular channel.  So Alison swapped the finder board with its neighbor to see if the error moved.  In cosmic run 263086, the error has cleared (1.5.5 top left plot).  I suspect we could have fixed the stuck bit by simply power cycling the crate.  I have unmasked SL 5, cell 42, wire 2.
<!-- Comment: Wed Jun  4 18;04;25 comment by...Heather -->
I did not mean to click "Add comment to log".  I will try again:

<p> An example of the original problem is <a href= 'http://www-cdfonline.fnal.gov/cgi/internal/elog/elog.pl?nb=2008&action=copyentry&page=434&time=12:03:26.094'> here </a>.  
<br>I tried to "fix" it by masking on the offending channel <a href= 'http://www-cdfonline.fnal.gov/cgi/internal/elog/elog.pl?nb=2008&action=copyentry&page=446&time=11:34:16.111'> here </a>.  
<br> It did not <a href = 'http://www-cdfonline.fnal.gov/cgi/internal/elog/elog.pl?nb=2008&action=copyentry&page=446&time=15:58:28.642'>  help </a>, and appeared to be a bit stuck low in the finder for the particular channel, downstream of the masking.  
<br> Alison swapped the finder board with its neighbor to see if the error moved.  
<br> In cosmic run 263086, the error has cleared (1.5.5 top left plot).  
<br>I suspect simply power cycling the crate would have "fixed" the stuck bit since it is gone and did not move with the finder swap.  
<br>I have unmasked SL 5, cell 42, wire 2.
<!-- Date: Wed Jun 4 15:55:31.627 2008 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>PhysicsTable</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L3Accepts</TH>
<TH>DelivLumi nb<SUP>-1</SUP></TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN=263057 target=_runSummary>263057</A> x40391</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_5_01 [7,1088,721]</FONT></TD>
<TD ALIGN=CENTER>07:54:54</TD>
<TD ALIGN=CENTER>10:20:14</TD>
<TD ALIGN=CENTER>02:00:50</TD>
<TD ALIGN=RIGHT>   1,031,906</TD>   <TD ALIGN=RIGHT>  1,754.70</TD>
<TD ALIGN=RIGHT>  1,450.74</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN=263058 target=_runSummary>263058</A> x40392</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_5_01 [7,1088,721]</FONT></TD>
<TD ALIGN=CENTER>10:27:37</TD>
<TD ALIGN=CENTER>12:40:48</TD>
<TD ALIGN=CENTER>01:47:06</TD>
<TD ALIGN=RIGHT>     865,605</TD>   <TD ALIGN=RIGHT>  1,163.19</TD>
<TD ALIGN=RIGHT>    938.74</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN=263059 target=_runSummary>263059</A> x40393</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_5_01 [7,1088,721]</FONT></TD>
<TD ALIGN=CENTER>12:46:33</TD>
<TD ALIGN=CENTER>12:56:04</TD>
<TD ALIGN=CENTER>00:09:22</TD>
<TD ALIGN=RIGHT>      38,701</TD>   <TD ALIGN=RIGHT>     73.26</TD>
<TD ALIGN=RIGHT>     69.01</TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN=263061 target=_runSummary>263061</A> x40395</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_5_01 [7,1088,721]</FONT></TD>
<TD ALIGN=CENTER>13:07:36</TD>
<TD ALIGN=CENTER>13:19:02</TD>
<TD ALIGN=CENTER>00:10:15</TD>
<TD ALIGN=RIGHT>      43,649</TD>   <TD ALIGN=RIGHT>     86.05</TD>
<TD ALIGN=RIGHT>     74.02</TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>15:55:01</TD>
<TD ALIGN=CENTER>04:07:35</TD>
<TD ALIGN=RIGHT>     1,979,861</TD> <TD ALIGN=RIGHT>  3,077.20</TD>
<TD ALIGN=RIGHT>  2,532.51</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->


<!-- Date: Wed Jun 4 15:56:31.148 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263059>263059</A>
<HTML>Run <A HREF=http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN=263059 target=_runSummary>263059</A> has not yet been checked for goodrun status; please do so.</HTML>
<!-- Author: RunControl -->



<!-- Date: Wed Jun 4 15:59:29.292 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263084>263084</A>
 Activated at 2008.06.04 15:59:18
<!-- Author: RunControl -->


<!-- Date: Wed Jun 4 15:59:30.884 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263084>263084</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,1019,706] 
<!-- Author: Louise -->



<!-- Date: Wed Jun 4 16:05:11.528 2008 -->
<font color=red>Shift Summary: </font><pre>Shift summary: 
 
Started shift with store 6191 collisions just beginning. 4 hours of good data 
At 12:33, Tev lost an rf station and our favorite abort gap losses monitor surged. Tevmon took Si to standby and L00 tripped on the way down. By the time we recovered, MCR decided to drop the store. 
Access started. Steve Hahn swapped a COT transition card and it seemed to be verified by Aiko but it had no front panel lights so Sebastian had another shot at it and Alison verified.  Pcall11 was power cycled for Britany but this didn't  fix the problem so the proposal is to powercyle the whole east PEM (400 Hz) then run shower max calibration again then Mike Lindgren will recheck. 
 
Alison is swaped an xtrp (?) card. Olga swapped a different xtrp card. 
Process tech working on compressor that failed. 
 
MCR expects Tev access will soon end and they will be waiting on us.</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   263057-263084
Delivered Luminosity   3179  
Acquired Luminosity    2465  
Efficiency             77.6
   
</td></tr></table></pre>
<!-- Author: Carlsmith -->

<!-- Date: Wed Jun 4 16:08:46.465 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263084>263084</A>
 Terminated at 2008.06.04 16:08:23
<!-- Author: RunControl -->


<!-- Date: Wed Jun 4 16:08:47.381 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=263084>263084</A>
TERMINATE: restarting new cosmic run
<!-- Author: Chang-Seong -->