<!-- Title: Day shift. Sat Jan 19, 2002 -->
<!-- SciCo: Schlabach -->
<!-- DAQAce: Tonelli/Yamashita -->
<!-- MonAce: Karagoz Unel/Goldstein -->
<!-- CO: Yu -->
<!-- OpManager: Dittman -->
<!-- Notes: -->
<pre>store 914, lum 1.62e30</pre>


<!-- Date: Sat Jan 19 09:01:26 2002 -->
plan is still to dump store at noon; start special CLC run at 10. Run normal physics until then.
<!-- Author: Phil Schlabach -->


<!-- Date: Sat Jan 19 09:03:00 2002 -->
Karen Byrum reported at the 8AM meeting that the hot tower that was present a few days ago (calibrated away by RT) is back. I paged Rick. He will be in soon.
<!-- Author: Phil Schlabach -->


<!-- Date: Sat Jan 19 09:11:33 2002 -->
we inherited the CMX detector on standby in these channels: West 7-10 , 0-3 and 22-23. 
<!-- Author: mon. aces :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137484' target=_top>137484</a>) -->


<!-- Date: Sat Jan 19 09:36:06 2002 -->
we looked at the ymon and trigmon plots. The hot tower, wedge 15, eta 7 is hot on the pmt occupancy. The tower occupancy and the raw adc count look ok. The trigger tower shows nothing anomalous.
<!-- Author: Phil Schlabach -->
<!-- Comment: Sat Jan 19 09;40;58 comment by...ps -->
fter consulting Rick we decided to do nothing. The trigger isn't hot; the data is just crap from this tower.


<!-- Date: Sat Jan 19 09:50:44 2002 -->
paging clc.
<!-- Author: Phil Schlabach -->


<!-- Date: Sat Jan 19 10:09:08 2002 -->
We notify mcr that we are going to screw around with our loss monitors. Jay is trying to provide loss info in the abort gap.
<!-- Author: Phil Schlabach -->


<!-- Date: Sat Jan 19 10:18:49 2002 -->
Jimmy Proudfoot called and asked me to exhort Farrukh to fix the level 3 part of physics_0_03[23,249] which I did to the extent one can exhort an answering machine.
<!-- Author: Phil Schlabach -->
<!-- Comment: Sat Jan 19 10;41;29 comment by...Phil Schlabach -->
Farrukh called back. I was unable to tell him more than what was in the log last night. He said if there was no output that it was probably the trigger table. But he also said that he would read his e-mail and send out the appropriate response. I will try to notify Kirsten or Rob (who just called).
<!-- Date: Sat Jan 19 10:19:18 2002 -->
A couple of antiproton loss alarms. 
CLC people is working at the beam loss counters the alarms are probably due to that.
<!-- Author: Diego/Tomo :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137484' target=_top>137484</a>) -->


<!-- Date: Sat Jan 19 10:42:48 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137484">137484</A>
 Terminated at 2002.01.19 10:42:33
<!-- Author: RunControl -->



<!-- Date: Sat Jan 19 10:44:37 2002 -->
Run ended for start CLC special Run.
<!-- Author: Diego/Tomo :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137484' target=_top>137484</a>) -->


<!-- Date: Sat Jan 19 10:45:07 2002 -->
we finished screwing around with the loss monitors; starting the clc special run (with nominal parameters in clc)
<!-- Author: Phil Schlabach -->

<!-- Date: Sat Jan 19 11:05:58 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137485">137485</A>
 Activated at 2002.01.19 11:05:44
<!-- Author: RunControl -->



<!-- Date: Sat Jan 19 11:06:01 2002 -->
CLC test run beginning.  
CCAL13 not included as asked by R. Tesarek. 
Minimum Bias [5,233] Trigger Table.
<!-- Author: Diego/Tomo :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137485' target=_top>137485</a>) -->


<!-- Date: Sat Jan 19 11:07:25 2002 -->
<pre>Checked beam loss rates with the gate of CDF clock cycles 
(132 nsec), independently of the beam crossing and abort gaps. 
Originally the bunch crossing signal "PW" (396 nsec) is used as  
a gate. 
 
We saw that LOSTPB and B0ALOS both went up higher than LOSTP and 
B0PLOS respectively, and it looks this doesn't make sense. 
 
Also, we saw some rates corresponding to abort gaps (B0ALOS[1]-[3]).</pre>
<!-- Author: Ming, Jey, Koji -->


<!-- Date: Sat Jan 19 11:13:20 2002 -->
we have started to see frequent big spikes in lostp.
<!-- Author: mon. aces :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137485' target=_top>137485</a>) -->


<!-- Date: Sat Jan 19 11:21:11 2002 -->
<br><b>Hot tower, CEM wedge 15E</b> 
<br>After investigating the hot tower in CEM wedge 15E, I find 
that the pedestal has shifted up from 97 (loaded in the lookup 
table) to 140.  We should re-calibrate and download to correct 
this problem for the immediate future.  A PMT spike run would 
also be useful in helping to identify the source of the problem 
(PMT/electronics). 
<p> 
We need to access the detector to identify the source of the  
problem. 
<br>
<!-- Author: R.J. Tesarek -->


<!-- Date: Sat Jan 19 11:22:40 2002 -->
L3 first farm (c01) is always in 'input' (light blue) state. 
No transition happens there. 
L3 expert paged and he told us to check wheter it stays in that state also in a next Run.
<!-- Author: Diego/Tomo :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137485' target=_top>137485</a>) -->

<!-- Date: Sat Jan 19 11:30:45 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137485">137485</A>
 Terminated at 2002.01.19 11:30:19
<!-- Author: RunControl -->



<!-- Date: Sat Jan 19 11:31:11 2002 -->
CLC special run ended.
<!-- Author: Diego/Tomo :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137485' target=_top>137485</a>) -->


<!-- Date: Sat Jan 19 11:35:51 2002 -->
we are getting set up for the second clc run. This will cause the loss signals and luminosity go to zero - I notified mcr
<!-- Author: Phil Schlabach -->


<!-- Date: Sat Jan 19 11:47:01 2002 -->
Silicon is changed back to nominal (study ladders back to sparse) 
First run to pick up the new settings will be 137486 
 
Aces should take Silicon calibrations during quiet time of shot setup assuming a quench leaves time. 
  
<!-- Author: Lester -->
<!-- Comment: Sat Jan 19 15;44;16 comment by...Doug -->
<pre>The SVT can now unmask the following wedges:
           S-B3W8, S-B5W8

These were read-all for the silicon special runs.  Now that they
are back to sparsified read-out, the SVT can use them.
</pre>
<!-- Date: Sat Jan 19 11:53:01 2002 -->
mcr calls; beam will go away at noon. We are still waiting for the clc to be set up. So we are turning everything else off. If they get set up; we will take data just for the clc.
<!-- Author: Phil Schlabach -->


<!-- Date: Sat Jan 19 11:54:31 2002 -->
end of store. switched  subsytems HV to stby. tof is set to off.  
<!-- Author: mon. aces -->


<!-- Date: Sat Jan 19 12:07:24 2002 -->
beam is gone, we didn't get any clc data. They will put it back to it's normal configuration. Tsarek will do a normal cal_qie run with some expert options enabled.
<!-- Author: Phil Schlabach -->


<!-- Date: Sat Jan 19 12:16:34 2002 -->
TeV didn't quench. (I believe they were going to try to kick the beam out of the abort gap before aborting.) Starting quiet time calibrations. 
 
Farrukh called to say he has a new version of the table that failed last night. We will try physics_0_03[23,250] during shot setup.
<!-- Author: Phil Schlabach -->


<!-- Date: Sat Jan 19 12:20:46 2002 -->
Biased all silicon again to take D-mode calibrations.
<!-- Author: Joel -->


<!-- Date: Sat Jan 19 12:23:53 2002 -->
<center><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-239&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-239' width='300'></a><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-238&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-238' width='300'></a></center>
<!-- Author: mon.aces -->
<!-- Comment: Sat Jan 19 12;24;50 comment by...mon. aces -->
half of summary plots for end of store # 915

<!-- Comment: Sat Jan 19 12;25;41 comment by...muge -->
oops. store 914, not 915!


<!-- Date: Sat Jan 19 12:30:50 2002 -->
we start to bring down the solenoid to pump the watt can.
<!-- Author: Phil Schlabach -->


<!-- Date: Sat Jan 19 12:35:47 2002 -->
clc is back in its normal configuration. We got 1 of the 2 runs done. They will be back at an 8 o'clock meeting when they want to do the 2nd.
<!-- Author: Phil Schlabach -->


<!-- Date: Sat Jan 19 12:37:13 2002 -->
SVX D-mode calibration with DPS ON (Extra 0=1)
<!-- Author: Joel :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137489' target=_top>137489</a>) -->


<!-- Date: Sat Jan 19 12:40:54 2002 -->
SVX D-mode calibration with DPS off (Extra 0=0)
<!-- Author: Joel :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137491' target=_top>137491</a>) -->


<!-- Date: Sat Jan 19 12:46:17 2002 -->
clc_qie calib.done 
<!-- Author: muge :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137490' target=_top>137490</a>) -->


<!-- Date: Sat Jan 19 12:46:45 2002 -->
smxqie calib. done
<!-- Author: daq.aces :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137492' target=_top>137492</a>) -->

<!-- Date: Sat Jan 19 12:51:54 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137493">137493</A>
 Activated at 2002.01.19 12:50:42
<!-- Author: RunControl -->



<!-- Date: Sat Jan 19 12:55:09 2002 -->
Silicon switched back to standby
<!-- Author: Joel -->


<!-- Date: Sat Jan 19 12:55:36 2002 -->
COT calib done.
<!-- Author: muge/yamashita :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137494' target=_top>137494</a>) -->

<!-- Date: Sat Jan 19 12:58:20 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137493">137493</A>
 Terminated at 2002.01.19 12:57:49
<!-- Author: RunControl -->



<!-- Date: Sat Jan 19 12:59:42 2002 -->
Rick's run done.the rate is 82 Hz.
<!-- Author: daq aces :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137493' target=_top>137493</a>) -->


<!-- Date: Sat Jan 19 13:04:40 2002 -->
CMX/P/IMU calib. done
<!-- Author: daq aces :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137495' target=_top>137495</a>) -->


<!-- Date: Sat Jan 19 13:14:36 2002 -->
solenoid is coming back up
<!-- Author: Phil Schlabach -->


<!-- Date: Sat Jan 19 13:25:21 2002 -->
tof tac calib done
<!-- Author: daq aces :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137497' target=_top>137497</a>) -->


<!-- Date: Sat Jan 19 13:31:51 2002 -->
tof qie calib done. 2 faulty channels.
<!-- Author: daq aces :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137499' target=_top>137499</a>) -->


<!-- Date: Sat Jan 19 13:40:33 2002 -->
cal_led calib done.
<!-- Author: daq aces :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137501' target=_top>137501</a>) -->


<!-- Date: Sat Jan 19 13:41:57 2002 -->
We did Cal-QIE calibration and found that the LED prep light for 
the CHA/WHA was on (run137487).  Turned off the LED power supply and took a new calibration (run137488).  Downloaded the last run 
to correct problem in CEM wedge 15E.
<!-- Author: R.J. Tesarek :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137487, 137488' target=_top>137487, 137488</a>) -->


<!-- Date: Sat Jan 19 13:42:33 2002 -->
<center><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-245&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-245' width='300'></a><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-246&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-246' width='300'></a></center>
<!-- Author: Diego -->
<!-- Comment: Sat Jan 19 13;54;05 comment by...Diego -->
Store losses plot splitted in two time section trying (..unsuccesfully..) to get a full-time plot of antiproton losses (ACMET bug) 
<!-- Comment: Sat Jan 19 17;30;07 comment by...Jay Dittmann -->
<pre>The abnormalities in the losses between 10am and 11am are 
not real readings.  Koji, Ming, and I were testing a new
cabling configuration to provide information about losses
during the abort gaps.  It isn't easy to do this without
affecting LOSTP and LOSTPB.  We learned a lot but need to
go think some more...
</pre>
<!-- Date: Sat Jan 19 13:47:15 2002 -->
We tried to start plug laser run and set the lasers to standby. But before we start the run, beam injection were started. We turned the lasers off and didn't do plug laser run.
<!-- Author: daq aces -->


<!-- Date: Sat Jan 19 13:47:17 2002 -->
TOF HV alarm for a few seconds, then cleared itself.
<!-- Author: Joel -->


<!-- Date: Sat Jan 19 13:50:36 2002 -->
cal_xenon calib done.
<!-- Author: daq aces :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137502' target=_top>137502</a>) -->


<!-- Date: Sat Jan 19 13:51:33 2002 -->
<center><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-248&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-248' width='300'></a><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-247&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-247' width='300'></a></center>
<!-- Author: Diego -->
<!-- Comment: Sat Jan 19 13;55;35 comment by...Diego -->
Same as previous plot comment but for B0PLUM.
<!-- Comment: Sat Jan 19 13;56;47 comment by...Diego -->
..oopss B0TLUM...
<!-- Date: Sat Jan 19 13:53:30 2002 -->
solenoid is at full strength
<!-- Author: Phil Schlabach -->





<!-- Date: Sat Jan 19 14:07:07 2002 -->
it looks like they're putting in real proton bunches
<!-- Author: Phil Schlabach -->
<!-- Comment: Sat Jan 19 14;11;31 comment by...ps -->
yup


<!-- Date: Sat Jan 19 14:12:07 2002 -->
calibrations are finally finished, test new trigger table.
<!-- Author: Phil Schlabach -->


<!-- Date: Sat Jan 19 14:13:37 2002 -->
plug laser run done. there were a few bunches in the machine
<!-- Author: daq aces :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137504' target=_top>137504</a>) -->


<!-- Date: Sat Jan 19 14:16:10 2002 -->
Final p are loaded, and this are the SVRAD values: 
<pre> 
W Inner BLM .5171166 
W Outer BLM .021525 
E Inner BLM 1.055536 
E Outer BLM 1.206742 
</pre>
<!-- Author: Diego -->

<!-- Date: Sat Jan 19 14:22:52 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137505">137505</A>
 Activated at 2002.01.19 14:22:38
<!-- Author: RunControl -->



<!-- Date: Sat Jan 19 14:23:05 2002 -->
pbars going in
<!-- Author: Phil Schlabach -->


<!-- Date: Sat Jan 19 14:28:58 2002 -->
BSC 2-3-4 alarm, it cleared itself.
<!-- Author: Diego -->


<!-- Date: Sat Jan 19 14:30:29 2002 -->
physics_0_03[23,250] is running w.o. beam. There are no obvious problems
<!-- Author: Phil Schlabach -->

<!-- Date: Sat Jan 19 14:34:45 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137505">137505</A>
 Terminated at 2002.01.19 14:34:28
<!-- Author: RunControl -->



<!-- Date: Sat Jan 19 14:35:58 2002 -->
we'll stop the run with the new table and start a run with the default to get ready for beam. We ran about 4000 events through the new table.
<!-- Author: Phil Schlabach -->


<!-- Date: Sat Jan 19 14:36:23 2002 -->
pbars are all in.
<!-- Author: Diego -->


<!-- Date: Sat Jan 19 14:36:28 2002 -->
TeV@flattop
<!-- Author: Phil Schlabach -->

<!-- Date: Sat Jan 19 14:45:22 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137506">137506</A>
 Activated at 2002.01.19 14:44:46
<!-- Author: RunControl -->


<!-- Date: Sat Jan 19 14:47:16 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137506">137506</A>
 Terminated at 2002.01.19 14:46:50
<!-- Author: RunControl -->



<!-- Date: Sat Jan 19 14:47:24 2002 -->
BSC 2-3-4 alarm, it cleared itself.
<!-- Author: Diego -->


<!-- Date: Sat Jan 19 14:49:43 2002 -->
scraping is over; L=5.5e30
<!-- Author: Phil Schlabach -->


<!-- Date: Sat Jan 19 14:55:57 2002 -->
<pre>MCR called, scraping finished.  
We raise the high voltages. 
Store #: 916 
Initial instantaneous luminosity : 5.47E30 
p current: 4.239E09 
pbar current 170E09 
</pre>
<!-- Author: Diego -->


<!-- Date: Sat Jan 19 14:57:10 2002 -->
detectors are on.
<!-- Author: Phil Schlabach -->


<!-- Date: Sat Jan 19 14:59:36 2002 -->
<pre> 
B0PLOS: around 10 KHz 
B0ALOS: around 0.8 KHz 
</pre>
<!-- Author: Diego -->


<!-- Date: Sat Jan 19 15:21:30 2002 -->
we're having trouble getting a run going (l3 won't start; one of the output nodes refuses to go.) Expert called, no est.
<!-- Author: Phil Schlabach -->


<!-- Date: Sat Jan 19 15:24:29 2002 -->
<center><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-249&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-249' width='300'></a><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-250&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-250' width='300'></a></center>
<!-- Author: Diego -->
<!-- Comment: Sat Jan 19 15;25;37 comment by...Diego -->
End of shot plots: SVX radiation and p/pbar losses.


<!-- Date: Sat Jan 19 15:31:58 2002 -->
expert arrived
<!-- Author: Phil Schlabach -->


<!-- Date: Sat Jan 19 15:55:21 2002 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137484>137484</A> 
x2190C</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_0_03 [21,246]</TD>                                          
<TD ALIGN=CENTER>04:56:38</TD>                                          
<TD ALIGN=CENTER>10:42:33</TD>                                          
<TD ALIGN=CENTER>05:38:07</TD>                                          
<TD ALIGN=RIGHT>      2,401,872</TD>                                    
<TD ALIGN=RIGHT>       592,525</TD>                                     
<TD ALIGN=RIGHT>     247,586</TD>                                       
<TD ALIGN=RIGHT>   34.134</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=137485>137485</A> 
x2190D</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>MINIMUM_BIAS_1 [5,233]</TD>                                         
<TD ALIGN=CENTER>11:05:44</TD>                                          
<TD ALIGN=CENTER>11:30:19</TD>                                          
<TD ALIGN=CENTER>00:23:49</TD>                                          
<TD ALIGN=RIGHT>        157,466</TD>                                    
<TD ALIGN=RIGHT>       157,466</TD>                                     
<TD ALIGN=RIGHT>      99,811</TD>                                       
<TD ALIGN=RIGHT>    1.917</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>15:55:01</TD>                                                       
<TD ALIGN=CENTER>06:01:56</TD>                                          
<TD ALIGN=RIGHT>       2,559,338</TD>                                   
<TD ALIGN=RIGHT>        749,991</TD>                                    
<TD ALIGN=RIGHT>       347,397</TD>                                     
<TD ALIGN=RIGHT>   36.051</TD>                                          
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Sat Jan 19 15:55:21 2002 -->
during cold start l3 did not respond for a long time> 6 mins or so. we aborted and reset and we tried to clean up L# farm, it did not respnd and gave errors for the 1st subfarm. we stopped L# proxy and clean up L3 again. we could not partition.we paged the L3 expert. he is now here trying to solve the problem.
<!-- Author: daq aces -->


<!-- Date: Sat Jan 19 16:16:52 2002 -->
<center><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-251&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-251' width='300'></a><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-252&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-252' width='300'></a><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-253&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-253' width='300'></a><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-254&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-254' width='300'></a></center> <font color=red><cite>(entry outside this shift's time range )</cite></font>
<!-- Author: Diego -->
<!-- Comment: Sat Jan 19 16;17;44 comment by...Diego -->
End of shift plots.


<!-- Date: Sat Jan 19 16:20:30 2002 -->
<font color=red>Shift Summary: </font><pre>we inherited a store and ran until noon. In the last 2 hours we tried a test changing the loss monitoring gates to include the abort gap. We also did a minimum bias run for clc with clc in normal mode. We were set up to do a minimum bias run with clc in a different mode, but we ran out of beam. The beam did not quench the TeV when it was aborted! We did the usual calibrations. Rick Tesarek was here and tried to do something to fix the hot tower in CEM (wedge 15, eta 7). Next shift needs to check if it worked. Store 916 at a luminosity of 5.5e30 was in at 14:57. L3 did not start. L3 is still hosed. L=4.7e30</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   137484, 137485
Delivered Luminosity   40  
Acquired Luminosity    15  
Efficiency             .4
   
</td></tr></table></pre>
<!-- Author: Phil Schlabach -->