<!-- Title: Day shift. Wed Aug 17, 2005 -->
<!-- SciCo: Soo-Bong Kim -->
<!-- DAQAce: Collin Wolfe -->
<!-- MonAce: Sasha Baroiant -->
<!-- CO: Ben Brau -->
<!-- OpManager: Bob Wagner -->
<!-- Notes: -->
<pre>Store 4336 (Run 202973) in progress; L=35e30</pre>


<!-- Date: Wed Aug 17 08:11:03.989 2005 -->
Flying wire scan.
<!-- Author: collin :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=202973' target=_top>202973</a>) -->
<!-- Comment: Wed Aug 17 09;10;42 comment by...collin -->
again.
<!-- Comment: Wed Aug 17 10;10;41 comment by...collin -->
again.
<!-- Comment: Wed Aug 17 11;10;48 comment by...collin -->
again.
<!-- Comment: Wed Aug 17 12;10;45 comment by...collin -->
again.
<!-- Comment: Wed Aug 17 13;10;38 comment by...collin -->
again.
<!-- Comment: Wed Aug 17 14;10;41 comment by...collin -->
again.
<!-- Comment: Wed Aug 17 15;10;46 comment by...collin -->
again.
<!-- Date: Wed Aug 17 08:21:19.680 2005 -->
<pre>18'7" b0l3pcom1.fnal.gov:main:Aug 17, 2005 8:18:07 AM-&gt;Host b0eb16.fnal.gov, task tRec_0 
SCPU-P0-E-TracerEventId: Event 7978848, crate 44, channel 6 has either bad Tracer ID or bad markers around Tracer word. 
 Tracer word event ID is 7, should be 0. 
Tracer word = 0x0007002c (I think). 
Channel data = 0x000000c7 0x0007002c 0x50454d44 0x000c000a 0x00000000 0x00000051 0x00000001 0x0ad40031 ... 
... 0x019c511e 0x01a9d31e 0x019f751e 0x00000000 0xaaaaaaaa 0x55555555 0x0007002c 0x80000000 --&gt;   
 Additional Information:  
 
 Attention!!!. SCPU_TRACER_EVENT_ID Error !!!  
 Hardware EVB has detected a problem with data quality in  
 SCPU b0eb16 (forwarded by FER crate PCAL_09).  
 Datataking is paused: events are not being processed.  
  
 AUTO HRR will be issued.   
 N.B. an automatic HRR for HEVB errors will be re-issued (if still needed)   
 in 30 secs. [MLE] 
</pre>
<!-- Author: collin :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=202973' target=_top>202973</a>) -->


<!-- Date: Wed Aug 17 08:45:13.252 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14478&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14478' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14479&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14479' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14480&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14480' width='300'></a></center>
<!-- Author: sasha -->
<!-- Comment: Wed Aug 17 08;47;38 comment by...sasha -->
Plots for 7:30 to 8:30


<!-- Date: Wed Aug 17 09:03:15.241 2005 -->
Retrospect Client on VNode2 warns us that we have not been backed up for a week. 
<!-- Author: sasha -->



<!-- Date: Wed Aug 17 09:15:37.568 2005 -->
Done TO for b0pcal00. Wouldn't transition to Halt on auto HRR, so shepherded.
<!-- Author: collin :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=202973' target=_top>202973</a>) -->
<!-- Comment: Wed Aug 17 15;26;53 comment by...collin -->
again.
<!-- Comment: Wed Aug 17 15;30;46 comment by...collin -->
Again. This time very quickly after the second done TO...
<!-- Date: Wed Aug 17 09:45:27.368 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14481&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14481' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14482&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14482' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14483&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14483' width='300'></a></center>
<!-- Author: sasha -->


<!-- Date: Wed Aug 17 10:07:33.119 2005 -->
ICICLE HEartbeat Alarm. Restarted ICICLE.
<!-- Author: sasha :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=202973' target=_top>202973</a>) -->



<!-- Date: Wed Aug 17 10:33:26.908 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14484&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14484' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14485&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14485' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14486&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14486' width='300'></a></center>
<!-- Author: sasha -->


<!-- Date: Wed Aug 17 10:50:30.717 2005 -->
Done TO for b0pcal05. Wouldn't transition to Halt on auto HRR, so shepherded.
<!-- Author: collin :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=202973' target=_top>202973</a>) -->


<!-- Date: Wed Aug 17 11:00:22.457 2005 -->
Got a strange cluster of six warnings on VNode2: 
-All had the following error code: 
 Error Number 1750 (000006d6) 
-the descriptions are 
CDF_S3R_SOUND_ALARM.F_CV: Tag name is not defined. 
CDF_S3R_NO_ALARM.F_CV: Tag name is not defined. 
CDF_S3R_VISUAL_ALARM.F_CV: Tag name is not defined. 
CDF_S3R_SUM_SILF_CV: Tag name is not defined. 
CDF_S3SVX_ISL_SUM_SILF_CV: Tag name is not defined. 
CDF_S3.S3_HRTBT.F_CV: Tag name is not defined. 
 
Is there supposed to be some kind of warning regarding silicon, but the program does not know what kind of warning sound/picture to make?  
 As far as I can tell, everything looks OK on IMON (and it is updating). 
 TevMon is way green. 
 I'm checking out slow control status for Si(though things look fine so far)
<!-- Author: sasha :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=202973' target=_top>202973</a>) -->
<!-- Comment: Wed Aug 17 11;01;21 comment by...sasha -->
I meant to submit as plain text. HAve posted identical but easier to read message below.
<!-- Date: Wed Aug 17 11:00:38.895 2005 -->
<pre>Got a strange cluster of six warnings on VNode2: 
-All had the following error code: 
 Error Number 1750 (000006d6) 
-the descriptions are 
CDF_S3R_SOUND_ALARM.F_CV: Tag name is not defined. 
CDF_S3R_NO_ALARM.F_CV: Tag name is not defined. 
CDF_S3R_VISUAL_ALARM.F_CV: Tag name is not defined. 
CDF_S3R_SUM_SILF_CV: Tag name is not defined. 
CDF_S3SVX_ISL_SUM_SILF_CV: Tag name is not defined. 
CDF_S3.S3_HRTBT.F_CV: Tag name is not defined. 
 
Is there supposed to be some kind of warning regarding silicon, but the program does not know what kind of warning sound/picture to make?  
 As far as I can tell, everything looks OK on IMON (and it is updating). 
 TevMon is way green. 
 I'm checking out slow control status for Si(though things look fine so far)</pre>
<!-- Author: sasha :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=202973' target=_top>202973</a>) -->




<!-- Date: Wed Aug 17 11:11:54.595 2005 -->
<pre>When chacking global alarm page for ISL, I get more of those little warnings (Error Number 1750 ...) 
 
CDF_S3ISL_TE_CHSUP.F_CV.Tag is not defined. 
CDF_S3ISL_C1_TE_HLF_CV.Tag is not defined. 
there are about 9 such warnings.  
 
The only red line I see for ISL is a west chiller outlet temp(ccw2-1out) at -200 C, and this has the 'ignored' status as 'YES'. 
BTE, the 'IN' value for this chiller is 8.9C. 
SVX and L00 'D' buttons also bring such warnings. But nothing seems wrong! 
 I will check elog for previous occurences.</pre>
<!-- Author: sasha -->
<!-- Comment: Wed Aug 17 11;20;19 comment by...sasha -->
<pre>I havn't found too many entries of this type in the elog. Once there were these errors when the cryo tech was updating a database.
Another time the warning was acknowledged by ACE w/o any further explanation. 
</pre>


<!-- Comment: Wed Aug 17 12;08;27 comment by...Ulrich H. -->
The ops manager found out that the cryo people are working on the iFix server CDF_S3, therefore the warnings. The system itself is not endangered, but some iFix readings are missing. The -200C reading is a broken temperature sensor, and should be ignored.
<!-- Comment: Wed Aug 17 12;31;19 comment by...sasha -->
Checked ISL 'D' page on iFix. All readings are there now, and there are no warnings.
<!-- Date: Wed Aug 17 11:22:15.833 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14487&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14487' width='300'></a><a href='/elog/notebooks/2005/Wed_Aug_17_2005_Day.112130.ps'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14490' ></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14488&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14488' width='300'></a><a href='/elog/notebooks/2005/Wed_Aug_17_2005_Day.112142.ps'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14492' ></a></center>
<!-- Author: bbrau -->
<!-- Comment: Wed Aug 17 11;23;14 comment by...Ben Brau -->
L1_GAP_EAST_&_JET5_&_RP_v2 GIF (left) and EPS (right).

Rates seem to be quite high for the current store (brown upside-down triangles).



<!-- Comment: Wed Aug 17 15;18;08 comment by...convery -->
All the RP triggers appear to be hot.  Pots are in the same place as always, as is the a48 collimator.  Maybe more beam halo.
<!-- Date: Wed Aug 17 11:38:02.720 2005 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2005&action=view&page=-14493&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2005&action=view&page=-14493  width='300' align='top'></a></td><td>SVX B3W7L2 FE current dropped, back up after HRR. </td></tr></table>
<!-- Author: sasha -->


<!-- Date: Wed Aug 17 11:43:47.031 2005 -->
Got a chirping alarm saying 'Lost Communication With Node'. Everything looked fine on global alarm page.  
Could also be related w/database work downstairs.
<!-- Author: sasha :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=202973' target=_top>202973</a>) -->


<!-- Date: Wed Aug 17 11:52:20.230 2005 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2005&action=view&page=-14496&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2005&action=view&page=-14496  width='300' align='top'></a></td><td>10:30-11:30 ace #3</td></tr></table>
<!-- Author: sasha -->


<!-- Date: Wed Aug 17 11:53:02.788 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14494&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14494' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14495&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14495' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14496&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14496' width='300'></a></center>
<!-- Author: sasha -->
<!-- Comment: Wed Aug 17 11;53;35 comment by...sasha -->
Hourly Plots. 10:30 to 11:30


<!-- Date: Wed Aug 17 12:03:44.370 2005 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2005&action=view&page=-14497&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2005&action=view&page=-14497  width='300' align='top'></a></td><td>Saw two spikes in Proton Loss. Nothing tripped. Called MCR anyway.</td></tr></table>
<!-- Author: sasha -->





<!-- Date: Wed Aug 17 12:36:01.947 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14498&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14498' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14499&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14499' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14500&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14500' width='300'></a></center>
<!-- Author: sasha -->


<!-- Date: Wed Aug 17 13:01:19.872 2005 -->
L2 Decision TO.
<!-- Author: collin :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=202973' target=_top>202973</a>) -->



<!-- Date: Wed Aug 17 13:33:25.061 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14501&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14501' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14502&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14502' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14503&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14503' width='300'></a></center>
<!-- Author: sasha -->



<!-- Date: Wed Aug 17 14:55:13.434 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14504&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14504' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14505&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14505' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14506&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14506' width='300'></a></center>
<!-- Author: sasha -->
<!-- Comment: Wed Aug 17 14;56;01 comment by...sasha -->
Plots for 13:30 to 14:35. Note Abort Gap Activity arount 2:30. 



<!-- Date: Wed Aug 17 14:58:08.197 2005 -->
Abort Gap Count up to 85kHz, TevMon warnings. We put Si in Standby, waited for MCR to clean up gap.   
<!-- Author: sasha :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=202973' target=_top>202973</a>) -->
<!-- Comment: Wed Aug 17 14;59;18 comment by...sasha -->
This happened at 14:28. See hourly plots above. 



<!-- Date: Wed Aug 17 15:15:52.425 2005 -->
pinkie: Bias Current for ISL B2W0L3 is stable just past 100muA. Stays pink after unmarking/HRR.. Will paste plot next.
<!-- Author: sasha :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=202973' target=_top>202973</a>) -->


<!-- Date: Wed Aug 17 15:17:59.239 2005 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2005&action=view&page=-14507&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2005&action=view&page=-14507  width='300' align='top'></a></td><td>ISL Bias pinkie: 101muA and perhaps growing slowly.</td></tr></table>
<!-- Author: sasha -->
<!-- Comment: Wed Aug 17 15;31;40 comment by...Ulrich H. -->
The bias current for ISL B2W0L3 seems to be stable now, therefore we bumped up limit on bias current to 105 &mu;A.
<!-- Date: Wed Aug 17 15:22:57.050 2005 -->
Done TO for crate b0clc00.
<!-- Author: collin :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=202973' target=_top>202973</a>) -->





<!-- Date: Wed Aug 17 15:35:49.744 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14508&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14508' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14509&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14509' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-14512&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-14512' width='300'></a></center>
<!-- Author: sasha -->
<!-- Comment: Wed Aug 17 15;36;42 comment by...sasha -->
Plots for 14:30 to 15:30

<!-- Date: Wed Aug 17 15:55:27.681 2005 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>Physics able</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L1Accepts</TH>
<TH>L2Accepts</TH>
<TH>L3Accepts</TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=202973 target=_blank>202973</A> x318DD</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_3_07 [1,691,588]</FONT></TD>
<TD ALIGN=CENTER>23:40:16</TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>15:07:58</TD>
<TD ALIGN=RIGHT>  1,171,013,877</TD>
<TD ALIGN=RIGHT>    13,765,640</TD>
<TD ALIGN=RIGHT>   3,906,888</TD>
<TD ALIGN=RIGHT> 2138.753</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>15:55:01</TD>
<TD ALIGN=CENTER>15:07:58</TD>
<TD ALIGN=RIGHT>   1,171,013,877</TD>
<TD ALIGN=RIGHT>     13,765,640</TD>
<TD ALIGN=RIGHT>     3,906,888</TD>
<TD ALIGN=RIGHT> 2138.753</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Wed Aug 17 15:59:55.382 2005 -->
<font color=red>Shift Summary: </font><pre>Quiet and efficient data-taking 
 
8:00 Store 4336 in progress; L=35e30 
12:00 Saw two spikes in proton loss. Called MCR, but they cannot find any reason for it. 
14:25 ACTNET shows abort gap counting is rather high; called to MCR 
</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   202973
Delivered Luminosity    794.9 nb-1  
Acquired Luminosity     743.4 nb-1  
Efficiency             	 93.5%
   
</td></tr></table></pre>
<!-- Author: Soo-Bong Kim -->


<!-- Date: Wed Aug 17 16:10:52.750 2005 -->
Collider Flying Wire Scan
<!-- Author: Shawn K :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=202973' target=_top>202973</a>) -->