<!-- Title: Owl shift. Tue Jul 22, 2003 -->
<!-- SciCo: Marjorie Shapiro -->
<!-- DAQAce: Dan Ryan -->
<!-- MonAce: Paola Squillacioti -->
<!-- CO: Anadi Canepai -->
<!-- OpManager: Nathan Eddy -->
<!-- Notes: -->
<pre>Start shift with L=18 and no stack! Run 166479 in progress</pre>


<!-- Date: Tue Jul 22 00:27:18.314 2003 -->
<b>Silicon L1 DTO's</b> (continued from <a href="http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2003&action=copyentry&page=606&time=16:08:39.987">last shift</a>)<p> 
b0dap73.fnal.gov:Thread-11652:12:05:51 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Author: Dan Ryan :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=166479' target=_top>166479</a>) -->
<!-- Comment: Tue Jul 22 01;13;01 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:1:07:18 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Comment: Tue Jul 22 01;22;26 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:1:16:37 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Comment: Tue Jul 22 01;41;02 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:1:30:51 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Comment: Tue Jul 22 02;07;28 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:2:00:25 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Comment: Tue Jul 22 02;14;49 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:2:00:25 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00


<!-- Comment: Tue Jul 22 02;22;40 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:2:14:55 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00


<!-- Comment: Tue Jul 22 02;22;55 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:2:16:41 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00


<!-- Comment: Tue Jul 22 02;32;37 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:2:26:33 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00


<!-- Comment: Tue Jul 22 02;38;58 comment by...dan ryan -->
b0l3gate2:main:2:32:46 AM->Error on L3 node b0l3165 (partition 7) Tue Jul 22 02:32:45 2003 l3_node 
  in refoInt_reformatProc (l3_refevt.c:610)
     ( L3_REFORMAT_ERROR: see Error Handler _reformatter.log file for full message)
b0dap73.fnal.gov:Thread-11652:2:32:13 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Comment: Tue Jul 22 02;59;09 comment by...dan ryan -->
b0l3gate2:main:2:52:03 AM->Error on L3 node b0l3134 (partition 7) Tue Jul 22 02:52:03 2003 l3_node 
  in refoInt_reformatProc (l3_refevt.c:610)
b0dap73.fnal.gov:Thread-11652:2:52:06 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Comment: Tue Jul 22 03;11;29 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:3:05:27 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Comment: Tue Jul 22 03;22;32 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:3:16:16 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Comment: Tue Jul 22 03;41;33 comment by...dan rya -->
b0dap73.fnal.gov:Thread-11652:3:33:06 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Comment: Tue Jul 22 04;06;05 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:3:57:37 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00


<!-- Comment: Tue Jul 22 04;23;47 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:4:17:54 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Comment: Tue Jul 22 04;38;51 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:4:30:26 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Comment: Tue Jul 22 04;39;08 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:4:31:11 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Comment: Tue Jul 22 04;41;20 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:4:35:41 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Comment: Tue Jul 22 04;54;42 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:4:48:37 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Comment: Tue Jul 22 05;00;36 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:4:54:57 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Comment: Tue Jul 22 05;07;24 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:5:01:45 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Comment: Tue Jul 22 05;11;21 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:5:03:23 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00


<!-- Comment: Tue Jul 22 05;19;09 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:5:13:24 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00


<!-- Comment: Tue Jul 22 05;23;34 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:5:17:46 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Comment: Tue Jul 22 05;47;28 comment by...dan ryan -->
b0xft04:Messenger:5:45:02 AM->Event 8474620: Bunchcounters in slot 13 (BC=0) and slot 10 (BC=0) disagree (L2B=0) <br>
b0dap73.fnal.gov:Thread-11652:5:41:23 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Comment: Tue Jul 22 05;50;31 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:5:44:48 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Comment: Tue Jul 22 06;08;45 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:6:02:59 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00


<!-- Comment: Tue Jul 22 06;19;42 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:6:13:37 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00


<!-- Comment: Tue Jul 22 06;23;37 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:6:17:46 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Comment: Tue Jul 22 06;25;08 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:6:19:24 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Comment: Tue Jul 22 06;29;26 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:6:23:37 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00


<!-- Comment: Tue Jul 22 07;48;17 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:7:35:12 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Comment: Tue Jul 22 07;57;48 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:7:50:03 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00


<!-- Comment: Tue Jul 22 08;24;11 comment by...Alberto -->
MLE) b0dap73.fnal.gov:Thread-11652:8:17:18 AM->  Silicon Level 1 Done Timeout Error from SVX, ISL or L00
<!-- Date: Tue Jul 22 00:29:18.617 2003 -->
<b>All other DTO's</b> (continued from <a href="http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2003&action=copyentry&page=606&time=16:18:41.529">last shift</a>)<p> 
b0cot06:Messenger:12:23:26 AM->Event 5381735: Bunchcounters in slot 6 (BC=59) and slot 7 (BC=115) disagree (L2B=2) 
<!-- Author: Dan Ryan :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=166479' target=_top>166479</a>) -->
<!-- Comment: Tue Jul 22 00;58;27 comment by...dan ryan -->
b0xft00:Messenger:12:42:24 AM->Event 5593566: Bunchcounters in slot 13 (BC=0) and slot 9 (BC=0) disagree (L2B=0) <br>
b0xft00:Messenger:12:43:18 AM->Event 5603745: Bunchcounters in slot 13 (BC=0) and slot 5 (BC=0) disagree (L2B=0) <br>
b0xft04:Messenger:12:45:17 AM->Event 5626328: Bunchcounters in slot 13 (BC=0) and slot 10 (BC=0) disagree (L2B=0)<br> b0xft04:Messenger:12:53:56 AM->Event 5723062: Bunchcounters in slot 13 (BC=0) and slot 8 (BC=0) disagree (L2B=0)<br>
b0cot16:Messenger:12:55:08 AM->Event 5736260: Bunchcounters in slot 3 (BC=15) and slot 18 (BC=71) disagree (L2B=2)<br>
<!-- Comment: Tue Jul 22 01;03;27 comment by...dan ryan -->
b0xft00:Messenger:1:00:09 AM->Event 5789246: Bunchcounters in slot 13 (BC=0) and slot 8 (BC=0) disagree (L2B=0) <br>
b0cot10:Messenger:1:01:10 AM->Event 1653503: Bunchcounters in slot 5 (BC=12) and slot 7 (BC=53) disagree (L2B=1) 
<!-- Comment: Tue Jul 22 01;23;41 comment by...dan ryan -->
b0cot09:Messenger:1:21:34 AM->Event 6014997: Bunchcounters in slot 2 (BC=139) and slot 4 (BC=136) disagree (L2B=0) 
<!-- Comment: Tue Jul 22 01;40;28 comment by...dan ryan -->
b0xft02:Messenger:1:25:51 AM->Event 6054105: Bunchcounters in slot 13 (BC=0) and slot 10 (BC=0) disagree (L2B=0) <br>
b0xft00:Messenger:1:27:29 AM->Event 6076387: Bunchcounters in slot 13 (BC=0) and slot 9 (BC=0) disagree (L2B=0)
<!-- Comment: Tue Jul 22 01;48;58 comment by...dan ryan -->
40'19" 1 crate/s: b0cot09(57),  Done TO.[RXPT]
<!-- Comment: Tue Jul 22 02;07;08 comment by...dan ryan -->
b0xft04:Messenger:1:53:43 AM->Event 6350969: Bunchcounters in slot 13 (BC=0) and slot 6 (BC=0) disagree (L2B=0)
<!-- Comment: Tue Jul 22 02;17;24 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:2:11:11 AM->Done Timeout: COT_09 
<!-- Comment: Tue Jul 22 02;42;15 comment by...dan ryan -->
b0cot19:Messenger:2:40:24 AM->Event 6811223: Bunchcounters in slot 5 (BC=3) and slot 8 (BC=74) disagree (L2B=1)
<!-- Comment: Tue Jul 22 02;49;27 comment by...dan ryan -->
b0cot00:Messenger:2:47:14 AM->Event 6878848: Bunchcounters in slot 2 (BC=71) and slot 4 (BC=106) disagree (L2B=1) 
<!-- Comment: Tue Jul 22 03;18;45 comment by...dan ryan -->
b0cot19:Messenger:3:16:04 AM->Event 7139464: Bunchcounters in slot 5 (BC=124) and slot 6-20 (BC=12) disagree (L2B=0)<br>
b0cot19:Messenger:3:16:06 AM->Runtime Error 6, Event 7139464: Bunch counter mismatch, mismatch count = 15
<!-- Comment: Tue Jul 22 03;40;21 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:3:34:22 AM->Done Timeout: COT_09 
<!-- Comment: Tue Jul 22 03;51;06 comment by...dan ryan -->
b0xft04:Messenger:3:43:12 AM->Event 7395832: Bunchcounters in slot 13 (BC=0) and slot 8 (BC=0) disagree (L2B=0) 
<!-- Comment: Tue Jul 22 04;07;58 comment by...dan ryan -->
b0cot11:Messenger:4:06:03 AM->Event 7612458: Bunchcounters in slot 4 (BC=139) and slot 6 (BC=118) disagree (L2B=0) 
<!-- Comment: Tue Jul 22 04;38;32 comment by...dan ryan -->
b0xft02:Messenger:4:29:03 AM->Event 7822532: Bunchcounters in slot 13 (BC=0) and slot 9 (BC=0) disagree (L2B=0) 
<!-- Comment: Tue Jul 22 04;42;16 comment by...dan ryan -->
b0cot10:Messenger:4:40:22 AM->Event 3777088: Bunchcounters in slot 5 (BC=18) and slot 8 (BC=118) disagree (L2B=1) 
<!-- Comment: Tue Jul 22 04;44;41 comment by...dan ryan -->
b0dap73.fnal.gov:Thread-11652:4:38:51 AM->Done Timeout: COT_09 
<!-- Comment: Tue Jul 22 04;55;11 comment by...dan ryan -->
b0xft02:Messenger:4:49:31 AM->Event 8000072: Bunchcounters in slot 13 (BC=0) and slot 6 (BC=0) disagree (L2B=0) 
<!-- Comment: Tue Jul 22 04;55;26 comment by...dan ryan -->
b0cot06:Messenger:4:53:12 AM->Event 8037378: Bunchcounters in slot 6 (BC=12) and slot 7 (BC=127) disagree (L2B=1) 
<!-- Comment: Tue Jul 22 05;01;50 comment by...dan ryan -->
<p>
b0svt03:Messenger:4:59:25 AM->FER_E_SVT, sync error in Wedges  7; HRR required
<br>
<!-- Comment: Tue Jul 22 07;47;29 comment by...dan ryan -->
b0cot19:Messenger:7:45:08 AM->Event 9489090: Bunchcounters in slot 5 (BC=21) and slot 11 (BC=6) disagree (L2B=0) 
<!-- Date: Tue Jul 22 00:36:07.568 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8480&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8480' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8481&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8481' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8482&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8482' width='300'></a></center>
<!-- Author: Paola -->







<!-- Date: Tue Jul 22 01:25:00.340 2003 -->
<table><tr><td><a href=/elog/notebooks/2003/Tue_Jul_22_2003_Owl.012414.ps align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-8484  align='top'></td><td>Left Plots: Et errors per tower. Plots should be empty. Just one entry -> no need to page the expert Right plots: Trigger tower energy-data vs simulation. Few off diagonal elements -> no need to page the expert</a></td></tr></table>
<!-- Author: anadi -->


<!-- Date: Tue Jul 22 01:36:05.527 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8485&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8485' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8486&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8486' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8487&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8487' width='300'></a></center>
<!-- Author: Paola -->
<!-- Comment: Tue Jul 22 01;37;49 comment by...Paola -->
There is a spike in B0PBSM (secon graph, the yellow)





<!-- Date: Tue Jul 22 01:57:25.273 2003 -->
<table><tr><td><a href=/elog/notebooks/2003/Tue_Jul_22_2003_Owl.015713.ps align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-8491  align='top'></td><td>XTRD MONITOR: all plots should be empty. Some errors are OK. Plot n.1: all errors are red; but expert has to be paged only if plot n.3 is full of erros -> no page</a></td></tr></table>
<!-- Author: anadi -->








<!-- Date: Tue Jul 22 02:29:27.859 2003 -->
CSP, BSU, TSU, CSW, CSX, MSK alarms at 2:15. All the CCU and all the pisabox were red. I went downstairs to check the computers, they looked fine and the HV of the ccu were on. When I come back (2:22) in the control room the alarms were cleared.  
<!-- Author: Paola :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=166479' target=_top>166479</a>) -->



<!-- Date: Tue Jul 22 02:33:16.146 2003 -->
<table><tr><td><a href=/elog/notebooks/2003/Tue_Jul_22_2003_Owl.023254.ps align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-8493  align='top'></td><td>SL1 Monitor: L2 accepts + Rejects vs L1 accepts  Off diagonal element </a></td></tr></table>
<!-- Author: anadi -->


<!-- Date: Tue Jul 22 02:35:35.641 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8494&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8494' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8495&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8495' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8496&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8496' width='300'></a></center>
<!-- Author: Paola -->




<!-- Date: Tue Jul 22 02:46:36.723 2003 -->
<table><tr><td><a href=/elog/notebooks/2003/Tue_Jul_22_2003_Owl.024619.ps align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-8498  align='top'></td><td>Object Mon: MUON Monitor charge/pT chi2 fails: the distribution is sensitive to the present triggers used, e.g. spike at 0.25 corresponds to CMUP4 trigger.  The check should not normally fail, the failure would indicate some problems in tracking</a></td></tr></table>
<!-- Author: anadi -->



<!-- Date: Tue Jul 22 02:55:10.341 2003 -->
Trigger Inhibit from TOF.  Paused run to recover but inhibit cleared immediately.  Resumed immediately.
<!-- Author: dan ryan :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=166479' target=_top>166479</a>) -->


<!-- Date: Tue Jul 22 02:58:49.878 2003 -->
TOF alarm, the PC was frozen and I restarted it following the instruction relative the data taking. 
<!-- Author: Paola :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=166479' target=_top>166479</a>) -->



<!-- Date: Tue Jul 22 03:10:46.740 2003 -->
RXPT Errors Loop:<br>  
b0xft04:Messenger:3:03:53 AM->Event 7037173: Bunchcounters in slot 13 (BC=0) and slot 6 (BC=0) disagree (L2B=0) <br> 
b0cot01:Messenger:3:05:35 AM->Runtime Error 3, Event 7054231: TDC in slot 4 not ready for readout<p> 
 
Shepherded b0cot01.
<!-- Author: dan ryan :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=166479' target=_top>166479</a>) -->





<!-- Date: Tue Jul 22 03:22:47.322 2003 -->
PSM alarm, cleared in few seconds
<!-- Author: Paola :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=166479' target=_top>166479</a>) -->


<!-- Date: Tue Jul 22 03:35:36.363 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8499&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8499' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8500&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8500' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8501&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8501' width='300'></a></center>
<!-- Author: Paola -->








<!-- Date: Tue Jul 22 04:24:03.498 2003 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2003&action=view&page=-8503&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-8503  width='300' align='top'></td><td>PSM alarm: 2RR25C_2 was red. This is the plot of the  negative tension, we can note the two spikes! The alarm  cleared by itself in a minute</a></td></tr></table>
<!-- Author: Paola/Dan -->
<!-- Comment: Tue Jul 22 05;05;41 comment by...Paola -->
Another PSM alarm, now the voltage of the channel 1 of 2RR25C_2 was too high (13.6, the maximum is 13)
<!-- Comment: Tue Jul 22 07;21;56 comment by...Paola -->
PSM alarm: the voltage of the channel 0 of 1RR18D_1 was too high (6.5, the maximum is 6). The alarm cleared in few seconds
<!-- Date: Tue Jul 22 04:38:05.983 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8507&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8507' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8508&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8508' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8509&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8509' width='300'></a></center>
<!-- Author: Paola -->
















<!-- Date: Tue Jul 22 05:13:19.432 2003 -->
High Level 1 Deadtime: 
b0dap57.fnal.gov:resmgr.ReplyMonitor 5:5:07:03 AM-> Dead Time is High  -->  
Not sure if this is related to silicon DTO's.  CO instructed to look for hot triggers.
<!-- Author: Dan Ryan :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=166479' target=_top>166479</a>) -->






<!-- Date: Tue Jul 22 05:36:39.825 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8513&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8513' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8514&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8514' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8515&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8515' width='300'></a></center>
<!-- Author: Paola -->









<!-- Date: Tue Jul 22 06:34:47.212 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8516&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8516' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8517&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8517' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8518&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8518' width='300'></a></center>
<!-- Author: Paola -->



<!-- Date: Tue Jul 22 07:34:40.956 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8519&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8519' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8520&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8520' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8521&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8521' width='300'></a></center>
<!-- Author: Paola -->
<!-- Comment: Tue Jul 22 07;35;52 comment by...Paola -->
There is a spike in B0PBSM (second graph, the yellow one)



<!-- Date: Tue Jul 22 07:47:37.841 2003 -->
<table><tr><td><a href=/elog/notebooks/2003/Tue_Jul_22_2003_Owl.074713.ps align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-8524  align='top'></td><td>CEM Tower: average raw ADC count per hit TO BE INVESTIGATED -> expert comes at 8 AM</a></td></tr></table>
<!-- Author: anadi -->
<!-- Comment: Tue Jul 22 07;52;05 comment by...Larry Nodulman -->
Lokks fine to me!
<!-- Comment: Tue Jul 22 08;04;10 comment by...Larry Nodulman -->
rate is high - triggered data not minbias?
<!-- Comment: Tue Jul 22 08;05;42 comment by...Larry Nodulman -->
oops rate ok for 60 counts th.
<!-- Date: Tue Jul 22 07:47:43.717 2003 -->
<center><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8522&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8522' width='300'></a><a href='/cgi/elog/elog.pl?nb=2003&action=view&page=-8525&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2003&action=view&page=-8525' width='300'></a></center>
<!-- Author: Paola -->



<!-- Date: Tue Jul 22 07:48:37.836 2003 -->
<TABLE bgcolor=#cdc9a5> 
<TR><TH>Date</th><th>Time</th><th>BLM</th><th colspan=2>Dose</th></tr> 
<tr><td>2003.07.22</td><td>07:39:52</td><td>W Inner BLM</td><td>512.06</td><td>RADS</td></tr> 
<tr><td>2003.07.22</td><td>07:39:52</td><td>W Outer BLM</td><td>0.00</td><td>RADS</td></tr> 
<tr><td>2003.07.22</td><td>07:39:52</td><td>E Inner BLM</td><td>0.00</td><td>RADS</td></tr> 
<tr><td>2003.07.22</td><td>07:39:52</td><td>E Outer BLM</td><td>232.66</td><td>RADS</td></tr> 
</TABLE>
<!-- Author: Paola :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=166479' target=_top>166479</a>) -->


<!-- Date: Tue Jul 22 07:49:53.921 2003 -->
<TABLE bgcolor=#cdc9a5> 
<tr><th colspan=2>Totals</th></tr> 
<tr><td>Date:</td><td>2003.07.22</td> 
<tr><td>Shift:</td><td>owl</td> 
<tr><td>Delivered luminosity:</td><td>431.02 nb<sup>-1</sup></td></tr> 
<tr><td>Acquired luminosity:</td><td>406.06 nb<sup>-1</sup></td></tr> 
<tr><td>Efficiency:</td><td>94.21</td></tr> 
</TABLE>  
<!-- Author: Paola :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=166479' target=_top>166479</a>) -->



<!-- Date: Tue Jul 22 07:52:53.180 2003 -->
<h3>Belated entry about L1 Done timeouts</h3> 
So, with the advent of 2 SRC running, we get a L1 DONE timeout 
every 10-60 minutes.  I did some hunting around with Logic Analyzers to try and find the cause, and can come up with two theories<ol> 
<li> It is clear that if there is a L2 Decision timeout, there will be a L1 DONE timeout (L1TO) as well.  This suggests that the L1TO period starts with the L1A, and perhaps sometimes the L2 decision is so long in coming that the time expires before the SRC can receive the L2A and send out the L1D signal (It takes 
10 microsecs to turn a "special L2A" into a L1D).  We would like to check with the expert <ol> 
<li> How long is the L1D timeout period 
<li> Does it indeed start counting from L1A 
</ol> 
<li> With no Silicon involved, the L1D should come 80*132 clock 
ticks after the "special L2A" or ~ 1-2 uS after L2R.  So, I set up the logic analyzer to look for times when it hasn't appeared for 20 uS after one or the other, and 
got a few  trigger, one of which is in the <a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=tsi&action=view&page=11&frame=2&anchor=23:58:26'>TSI log</a>.  Here there was a "special L2A" (TAXI word 0x01) with very little activity around-no normal L1 or L2 decisions within the 60 uS window, and the TS is sending 0x08, which is L1 reject with the GLIVE down, meaning we are not live, i.e. all 4 buffers are occupied.  This suggests that something is taking a long time with this L2 decision.  
The second SRC responds to the special L2A, but ?instead? of sending the L1DONE it sends a "Keep Alive" command 
which is something it is suppossed to do if it hasn't seen a L1A in something like 287 uS.  So perhaps the 2nd SRC gets confused when it is a very long time between L1A and the L2 decision, and  
swaps the L1D command with a Keep Alive?  We will pursue this 
with the SRC people. 
</ol> 
This needs to be fixed, although it is causing roughly as much 
downtime as the BC mismatch errors, which we've been living with for a few months now I guess.  Before we back out the 2nd SRC change, it may be good to actually use this everlasting store with low lumi to actually run L00 and ISL on the second SRC to 
make sure there is nothing else we need to fix. In addition, I can probably make some refinements on the diagnostics done last night now that I've thought about it some more.  
<!-- Author: SCN -->


<!-- Date: Tue Jul 22 07:54:09.207 2003 -->
b0cot09:Messenger:7:49:38 AM->Runtime Error 16, Event 9524527: TDC in slot 4 not ready for readout 
 
Shepherded b0cot09.
<!-- Author: dan ryan :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=166479' target=_top>166479</a>) -->

<!-- Date: Tue Jul 22 07:56:59.118 2003 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=166479>
166479</A> 
x28A4F</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_1_05 [5,319,391]</TD>                                       
<TD ALIGN=CENTER>17:10:57</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>13:44:09</TD>                                          
<TD ALIGN=RIGHT>    572,674,312</TD>                                    
<TD ALIGN=RIGHT>     9,545,173</TD>                                     
<TD ALIGN=RIGHT>   2,143,295</TD>                                       
<TD ALIGN=RIGHT>  910.298</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>07:55:02</TD>                                                       
<TD ALIGN=CENTER>13:44:09</TD>                                          
<TD ALIGN=RIGHT>     572,674,312</TD>                                   
<TD ALIGN=RIGHT>      9,545,173</TD>                                    
<TD ALIGN=RIGHT>     2,143,295</TD>                                     
<TD ALIGN=RIGHT>  910.298</TD>                                          
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->




<!-- Date: Tue Jul 22 08:01:23.337 2003 -->
<font color=red>Shift Summary: </font><pre>Quiet shift:  low luminosity but smooth data taking</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   166479
Delivered Luminosity   435.72  
Acquired Luminosity    409.39  
Efficiency             93.96
   
</td></tr></table></pre>
<!-- Author: Marjorie Shapiro -->