<!-- Title: 15:04:34  Mon Jan 26 2004 -->
<!-- Crewchief: Rong Shyang -->
<!-- Op1: &nbsp; -->
<!-- Op2: &nbsp; -->
<!-- Op3: &nbsp; -->
<!-- Op4: &nbsp; -->
<!-- Op5: &nbsp; -->
<!-- Op6: &nbsp; -->
<!-- Notes: -->
My first pager carrier week :)


<!-- Date: Mon Jan 26 15:06:10 2004 -->
<center><a href='/elog/notebooks/silicon/Mon_Jan_26_2004_150509.ps'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3371' ></a><a href='/elog/notebooks/silicon/Mon_Jan_26_2004_150534.ps'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3373' ></a><a href='/elog/notebooks/silicon/Mon_Jan_26_2004_150553.ps'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3375' ></a></center>
<!-- Author: David -->
<!-- Comment: Mon Jan 26 15;45;25 comment by...David, Reid, Rong -->
<h3>Light Investigation of SB2W1L2</h3>
Oscilliscope images from left to right: 
<ol>
<li> Bit 6 (top/chan 2) and OBDV (bottom/chan 1) waveforms during RA mode.</li>
<li> Bit 5 (top/chan 2) and OBDV (bottom/chan 1) waveforms during RA mode.</li>
<li> Bit 5 (top/chan 2) and OBDV (bottom/chan 1) waveforms during NN mode.</li>
</ol>
Bit 5 and bit 6 look OK, but OBDV has a strange offset that grows. Also, in NN mode there are frequent missing clock signals as seen in (3).


<!-- Comment: Tue Jan 27 07;54;07 comment by...Gino -->
The missing clock in 3 are expected if they come in between chips.
To recognise if this is the case you should watch BIT 7 (chipID).
</p>
The slowing rising OBDV could be a consequence of the enormous light output (more than 1 mW) for the OBDV. What are the settings on the 2VDOIM and 5VDOIM power lines?

<!-- Comment: Wed Jan 28 15;09;58 comment by...Rong -->
<pre>In the PS Gui - full stats window, it shows the volts setting is
5.39 and 1.64 for Doim. The difference is 3.75V which is indeed a bit high. Some other ladders are around 3.45V.</pre>
<!-- Comment: Fri Jan 30 07;32;26 comment by...Gino -->
The default 3.45 V is a bit higher than the nominal 3.0 V due to the expected voltage drops in the cable between where the voltage is sensed (JC) and the load (DOIM).
</p>
For the wedge here investigated the DOIM voltages were tweaked manually.
</p>
<a href=
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=26&frame=2&anchor=13:52:07'>
here </a> are light output measurements from May 11th 2001.

</p>
<a href=
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=55&frame=2&anchor=12:38:50'>
here </a> are light output measurements from July 25th 2001.

</p>
And <a href=
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=72&frame=2&anchor=18:58:15'>
here </a> is the entry of October 25th 2001 when the Voltage was changed by 300 mV away from nominal as noted above.

</p>


 
<!-- Date: Mon Jan 26 16:24:56 2004 -->
<h3>Swapping madness for SB2W2</h3> 
<a href= 
'/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=181&frame=2&anchor=16:52:58'>Gino's Diagnosis of the problem.</a><br><br> 
We swapped the PS this morning.  Out: S/N #35.  In: S/N #21.  <br> 
Supply looked fine in the loadbox tests.  When we came out, we were 
tripping on 5VDoim OVC.  <br><br> 
Due to a problem with the Tevatron, we had some more time for access.  After some work, I tried to swap in a different supply.  Out: #21.  In: #9.  Again the supply looked good on the loadbox but didn't work when cabled to the wedge.  It was again tripping on 5VDoim.  <br><br> 
Back in the control room we noticed that the 2 and 5 V doim current settings were quite low (about 50% lower than a neighboring wedge).  We reloaded the AVDD/DVDD Currents file and this brought the DOIM current settings back to more conventional values.  With the correct currents in place, the wedge powered okay.
<!-- Author: reid -->
<!-- Comment: Mon Jan 26 17;24;00 comment by...Rong -->
<pre>E224 kept tripping on 'software protection', DVDD current too high thing. DVDD is giving 360mA while the software protection is set on 350mA. 
Rob came in and edit the file $CAENCONTROL_DIR/CAENLVcurrent.dat
and raised the threshold to 375mA.
He tried to load the parameter but the buttom on GUI didn't work, so we restart the PS GUI then it worked.
The IMON current limit should be modified as well. We should ignore the pinky of e224 for one store and see the level and structure to determine the setting on it.</pre>


<!-- Date: Mon Jan 26 23:34:12 2004 -->
<center><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3376&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3376' width='300'></a><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3377&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3377' width='300'></a></center>
<!-- Author: Rong -->
<!-- Comment: Mon Jan 26 23;38;36 comment by...Rong -->
<pre>SB2W2L4 triped on 'DVDD4 overcurrent'. <a href='http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2004&action=view&page=78&frame=2&anchor=23:04:51.647'> shift Elog </a> 
Tried to reset trip but didn't work. But we didn't see any current showing on the plot after trip reset and wonder why it's causing trip.
Anyway, the scraping was finished, so Rainer and I decided to drop this ladder and mark it offline in hardware database.
</pre>


<!-- Comment: Tue Jan 27 03;06;12 comment by...Rainer -->
From PSGUI
<pre>
Mon Jan 26 20:54:31 CST 2004     IFix -  success: Silicon to STANDBY
Mon Jan 26 22:58:53 CST 2004     TRIP: SVX  B2  W2 DVDD4 OverCurrent
Mon Jan 26 23:03:53 CST 2004     success: Resetting trips
Mon Jan 26 23:12:47 CST 2004     success: Channel4 SVX B2 W2 to STANDBY
Mon Jan 26 23:12:54 CST 2004     TRIP: SVX  B2  W2 DVDD4 OverCurrent
Mon Jan 26 23:14:42 CST 2004     success: Resetting trips
</pre>
FROM svxinter <pre>
CaenVme
CaenVme_S.o
InfoToIFixPack.o
InfoToIFixUnpack.o
SponLogPack.o
SponLogUnpack.o
caen_ss.o
caengrps.o
caenlib.o
caentalk.o
psmon.o
value = 0 = 0x0
-> com_int = 0
detector = 0, status = 0
detector = 2, status = 0
com_int = 2
detector = 0, status = 2
detector = 2, status = 2
com_int = 0
detector = 0, status = 0
detector = 2, status = 0
com_int = 2
detector = 0, status = 2
detector = 2, status = 2
SVX B2W2L4 got a trip
SVX B2W2L4 got a trip
SVX B2W2L4 got a trip

</pre>
<!-- Date: Tue Jan 27 00:56:23 2004 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3378&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3378  width='300' align='top'></a></td><td>AVDD2, DVDD1, DVDD2, and DVDD3 of SB2W2 are set to IGNORE due to the offset of new PS current reading is unkonwn. Need to watch this store and feedback from power supply group to determine the value setting.</td></tr></table>
<!-- Author: Rong -->


<!-- Date: Tue Jan 27 01:16:30 2004 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3380&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3381  width='300' align='top'></a></td><td>SVTSPYMon gives empty plot. Ask CO carefully check the SVT.</td></tr></table>
<!-- Author: Rong -->





<!-- Date: Tue Jan 27 03:44:24 2004 -->
<h1> touchy situation in store #3195 </h1> 
<p> 
explained  
<a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2004&action=view&page=79&frame=2&anchor=01:38:13.352'> 
here</a>. The abort turned out to be clean judging from TeV BLMs, but ours did see some small rate as documented  
 
<a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2004&action=view&page=79&frame=2&anchor=01:50:36.873'> 
here</a>. Si-post quench (there was no quench, but still) checkout OK.
<!-- Author: Rainer -->

<!-- Date: Tue Jan 27 03:48:23 2004 -->
<p><font size=+1> D-mode Calibration Job Started </font>
<p><pre>
Job id            : 310
Calibration type  : nonstandard
Run period tag    : Jan_2004
DPS-off run       : 178517
DPS-on run        : 178515
Requested by      : Vadim
Main process id   : 41868245
Comment           : quiet time conditions
</pre>
<!-- Author: CDF silicon D-mode calibration job server v1.4 -->


<!-- Date: Tue Jan 27 03:48:43 2004 -->
<p><font size=+1> D-mode Calibration Job Finished </font>
<p><pre>
This is a computer-generated silicon D-mode calibration status report.
Silicon D-mode calibration job 310 FAILED.

Job info
--------
Calibration type : nonstandard
DPS-off run      : 178517
DPS-on run       : 178515
Requested by     : Vadim
Submitted        : 2004-01-27 01:48:05
Completed        : 2004-01-27 01:48:25
Status           : FAILURE
Comment          : quiet time conditions

Last processing stage reported : start of the run_calib_analysis.tcl script
Failure diagnostics :
Not enough disk space in the calibration area: only 671688 kB free.
</pre>
<!-- Author: D-mode script run_calib_analysis.tcl -->



<!-- Date: Tue Jan 27 03:54:49 2004 -->
<h1> A night with SB2W2L4 </h1>   aka or it's not that we have ceased to be willing to throw time at a ladder to make it work - see also today's controlled access 
<a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=182&frame=2&anchor=16:24:56'> 
madness.</a>  
 <p> 
not very pleasant night, I suppose. The guy tripped off beginning of this store on SPT. Since there is quiet time now up the whazoo, I bumped up the SPT limit one more time (see  
<a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=182&frame=2&anchor=16:24:56'> 
here. </a>) to 400 mA, the maximum I feel comfortable with (trip is at 450mA). <p>  
 
I made the aces perform a dmod calibration - it turns out the guy trips twice ( 
 
<a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2004&action=view&page=79&frame=2&anchor=03:02:38.989'> 
here  </a>  
 
and  
<a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2004&action=view&page=79&frame=2&anchor=03:26:41.061'> 
here </a>  
 
 on config while preparing for the dmod, which is very reminiscent of  <a href='http://www-cdfonline.fnal.gov/internal/usr/lmiller/cgi-bin/newproblemdb/DBreport.pl?detsys=SVX&wedge=B1W6&layer=L4'> SB1W6L4 </a>. <b> <p> Drawing incessant DVDD current on tripping on coldstart now a new feature of thermocycle/post-shutdown induced ladder hangover ? </b> <p>  
 
It was also observed that SB2W2L4 liked to trip off during dmod calibration, see 
<a href=' http://www-cdfonline.fnal.gov/internal/usr/lmiller/cgi-bin/newproblemdb/DBreport.pl?detsys=SVX&wedge=B2W2&layer=L4'> SB2W2L4 </a> 
Let's see how he fares with cosmics.  
<!-- Author: Rainer -->
<!-- Comment: Tue Jan 27 04;11;59 comment by...Rainer -->
I will leave the ladder in for cosmics running over night - I would like to see how the currents behave and whether there are any bit errors coming up. From the currents, I can reckon what would be a safe new SPT limit - such that we can take some data with this guy. However, there is a certain risk running this ladder in such a high DVDD current mode without clear understanding what is going on. The rest of the wedge is pristine - so SVT would not notice its loss. However, we should not give up ladders too easily, also considering the sweat we invested in this guy.
<!-- Date: Tue Jan 27 03:56:05 2004 -->
<center><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3386&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3387' ></a><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3382&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3383' ></a><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3384&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3385' ></a></center>
<!-- Author: Rainer -->
<!-- Comment: Tue Jan 27 04;08;00 comment by...Rainer -->
SB2W2L4 bitreporter
<a href='http://www-cdfonline.fnal.gov/internal/usr/lmiller/cgi-bin/errmon/makeladder.pl?beginrun=168600&endrun=today&ladder=S224'> overview </a>




<!-- Comment: Tue Jan 27 04;16;09 comment by...Rainer -->
drops in currents (trips) correspond to config transitions, but there were also spontanous trips as well. currently running with SPT limit relaxed to 400mA (was 375 mA set by Rob Napora, earlier 350mA - it would be good to know how much could be attributed to the offsets in the CAENs). Also note that using the GUI button to update trip limits did not work for me as well - PSGui, you are such a beauty ...


<!-- Date: Tue Jan 27 05:07:17 2004 -->
<center><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3388&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3388' width='300'></a></center>
<!-- Author: Rainer -->
<!-- Comment: Tue Jan 27 05;08;57 comment by...Rainer -->
<a href='http://www.quarks.de/dyn/11332.phtml'> digging </a> deeper  in SB1W6L4 history ... lines is Feb 9th, 2003 (alleged end of shutdown Jan 2003).


<!-- Date: Tue Jan 27 09:22:44 2004 -->
<p><font size=+1> D-mode Calibration Job Started </font>
<p><pre>
Job id            : 311
Calibration type  : standard
Run period tag    : Jan_2004
DPS-off run       : 178517
DPS-on run        : 178515
Requested by      : Jason Nielsen
Main process id   : 41844347
</pre>
<!-- Author: CDF silicon D-mode calibration job server v1.4 -->
<!-- Comment: Tue Jan 27 09;44;44 comment by...Jason Nielsen -->
This is a resubmit of the job above.  I am clearing extra space on the calibration disks to delay a repeat of this problem...

<!-- Date: Tue Jan 27 13:25:14 2004 -->
<p><font size=+1> D-mode Calibration Job Finished </font>
<p><pre>
This is a computer-generated silicon D-mode calibration status report.
Silicon D-mode calibration job 311 FAILED.

Job info
--------
Calibration type : standard
DPS-off run      : 178517
DPS-on run       : 178515
Requested by     : Jason Nielsen
Submitted        : 2004-01-27 07:22:30
Completed        : 2004-01-27 11:24:51
Status           : FAILURE

Last processing stage reported : calculating DPS-off pedestals
Failure diagnostics :
child process exited abnormally
</pre>
<!-- Author: D-mode script  -->
<!-- Comment: Wed Jan 28 06;53;59 comment by...rainer -->
<pre>why do the calibrations fail ?</pre>
<!-- Comment: Wed Jan 28 07;45;39 comment by...Jason Nielsen -->
I will check out the bad fraction, but I have submitted these tables to the database.
<!-- Comment: Wed Jan 28 07;46;36 comment by...Jason -->
The failure shown above is because of problems with scp between fcdfsgi2 and pcdaq.lbl.gov.
<!-- Date: Tue Jan 27 14:11:24 2004 -->
Entered in Silicon database: <ul>
<li>Ladder: SB2W2L4 
<li>problem n: 672 
<li>Status: needs_observation
<li>Type: DVDD
<li>Action: Diagnosis_update
<li>entered by: rainer 
<li>description is: DVDD SPT limits updated to 400 mA (was 350mA, but PS changed) - check in next store whether ladder can run. See  <a href= 'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=182&frame=2&anchor=03:56:05'> elog</a>   
<li>See <a href="http://www-cdfonline.fnal.gov/internal/usr/lmiller/cgi-bin/newproblemdb/DBreport.pl?detsys=SVX&wedge=B2W2&layer=L4">report</a></ul>
<!-- Author: Problem DB -->



<!-- Date: Tue Jan 27 15:08:55 2004 -->
<center><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3391&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3392' ></a><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3389&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3390' ></a></center>
<!-- Author: Rong, Rainer -->
<!-- Comment: Tue Jan 27 15;09;51 comment by...Rong, Rainer -->
<pre>guy tripped again - software protection trip  ~ 440 mA. </pre>


<!-- Date: Tue Jan 27 15:31:23 2004 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3393&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3394  width='300' align='top'></a></td><td>Change current setting in IMon for SB2W2L1 AVDD 
norminal 485 -> 515 and corresponding range from (475:490) -> (505:520)</td></tr></table>
<!-- Author: Rong -->


<!-- Date: Tue Jan 27 16:05:57 2004 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3396&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3397  width='300' align='top'></a></td><td>IMon SB2W2L2 AVDD 830 -> 820mA; (820:835)->(810:825)</td></tr></table>
<!-- Author: Rong -->



<!-- Date: Tue Jan 27 16:17:56 2004 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3398&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3399  width='300' align='top'></a></td><td>IMon setting for SB2W2L4 AVDD 1055 -> 1115; (1040:1065)->(1100:1125)</td></tr></table>
<!-- Author: Rong -->


<!-- Date: Tue Jan 27 16:20:15 2004 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3400&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3401  width='300' align='top'></a></td><td>IMon setting 35->74; (24:47)->(63:86)</td></tr></table>
<!-- Author: Rong -->


<!-- Date: Tue Jan 27 16:23:43 2004 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3402&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3403  width='300' align='top'></a></td><td>IMon setting SB2W2L1 DVDD 35->97; (26:54)->(88:116)</td></tr></table>
<!-- Author: Rong -->


<!-- Date: Tue Jan 27 16:26:35 2004 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3404&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3405  width='300' align='top'></a></td><td>IMon setting for SB2W2L2 DVDD 65->134; (45:90)->(114:159)</td></tr></table>
<!-- Author: Rong -->


<!-- Date: Tue Jan 27 16:34:26 2004 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3406&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3407  width='300' align='top'></a></td><td>SB2W2L4 tripped three times in one hour data taking. 
We decide to drop it if it trip again.</td></tr></table>
<!-- Author: Rong -->
<!-- Comment: Tue Jan 27 17;09;37 comment by...Rong -->
<pre>SB2W2L4 DVDD trip again, so I set this ladder offline in hardware database then restart the run.

What I did is :
- ask shifter to 'end' the run (to IDLE state)
- run traceSilicon to set e224 offline (option 4)
- ask ACE to configure
- Write to CAENvolts.dat (option 9)
- load CAENvolts.dat on PS GUI then reset trip.
- after SVX release trigger inhibit, start the run.</pre>


<!-- Date: Tue Jan 27 18:34:07 2004 -->
<table><tr><td><a href=/elog/notebooks/silicon/Tue_Jan_27_2004_183404.ps align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3409  align='top'></a></td><td>dvdd history of SB2W2L4 from midnight 27th jan to ~6pm.</td></tr></table>
<!-- Author: rainer -->





<!-- Date: Wed Jan 28 09:19:32 2004 -->
<h3> L1 SRC <-> Scaler Accounting </h3> 
 
SCN and Marcel have implemented L1  
 
 
<a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=181&frame=2&anchor=21:55:21'> 
accounting </a>  
 
 
 - the # of L1A seen by the SRC is compared to what the scaler crate has counted. The timestamp on the table below corresponds to an HRR sent. There is one incident where information is missing - otherwise no L1A  missing. during this time, about 14 L2 Decision timeout occured. <p> 
Most recent table to be found <a href='http://www-cdfonline.fnal.gov/internal/ops/svxii/finalanal.html'> here  </a> <p> 
 
<head>  
 <title>L1 A Accepts </title>  
 </head> 
<body> 
<table border="1"> 
<tr> <th>SRC Timestamp</th><th>SRC L1A</th><th>Scaler Timestamp</th><th>Run number</th><th>Scaler L1A</th><th>Scaler L1A corrected</th><th>Difference</th></tr> 
<tr> <td>Thu Jan 22 23:06:13 CST 2004</td><td>84635398</td><td>Thu Jan 22 23:06:13 CST 2004</td><td>178310</td><td>86676000</td><td>-1</td><td>0</td></tr> 
<tr> <td>Thu Jan 22 23:10:29 CST 2004</td><td>2464755</td><td>Thu Jan 22 23:10:29 CST 2004</td><td>178310</td><td>89140755</td><td>2464755</td><td>0</td></tr> 
<tr> <td>Thu Jan 22 23:44:15 CST 2004</td><td>20770504</td><td>Thu Jan 22 23:44:15 CST 2004</td><td>178310</td><td>109911259</td><td>20770504</td><td>0</td></tr> 
<tr> <td>Fri Jan 23 03:38:54 CST 2004</td><td>132565754</td><td>Fri Jan 23 03:38:55 CST 2004</td><td>178310</td><td>242477013</td><td>132565754</td><td>0</td></tr> 
<tr> <td>Fri Jan 23 05:57:53 CST 2004</td><td>266297</td><td>Fri Jan 23 05:57:53 CST 2004</td><td>178311</td><td>266297</td><td>266297</td><td>0</td></tr> 
<tr> <td>Fri Jan 23 05:59:32 CST 2004</td><td>223176</td><td>Fri Jan 23 05:59:32 CST 2004</td><td>178311</td><td>489473</td><td>223176</td><td>0</td></tr> 
<tr> <td>Fri Jan 23 06:08:41 CST 2004</td><td>1629827</td><td>Fri Jan 23 06:08:41 CST 2004</td><td>178311</td><td>2119300</td><td>1629827</td><td>0</td></tr> 
<tr> <td>Fri Jan 23 14:01:48 CST 2004</td><td>290095</td><td>Fri Jan 23 14:01:48 CST 2004</td><td>178339</td><td>290095</td><td>290095</td><td>0</td></tr> 
<tr> <td>Fri Jan 23 14:08:48 CST 2004</td><td>219404</td><td>Fri Jan 23 14:08:48 CST 2004</td><td>178339</td><td>509499</td><td>219404</td><td>0</td></tr> 
<tr> <td>Fri Jan 23 14:10:01 CST 2004</td><td>144806</td><td>Fri Jan 23 14:10:01 CST 2004</td><td>178339</td><td>654305</td><td>144806</td><td>0</td></tr> 
<tr> <td>Fri Jan 23 14:11:06 CST 2004</td><td>138735</td><td>Fri Jan 23 14:11:06 CST 2004</td><td>178339</td><td>793040</td><td>138735</td><td>0</td></tr> 
<tr> <td>Fri Jan 23 14:16:30 CST 2004</td><td>1058703</td><td>Fri Jan 23 14:16:30 CST 2004</td><td>178339</td><td>1851743</td><td>1058703</td><td>0</td></tr> 
<tr> <td>Fri Jan 23 16:09:51 CST 2004</td><td>54955247</td><td>Fri Jan 23 16:09:51 CST 2004</td><td>178339</td><td>56806990</td><td>54955247</td><td>0</td></tr> 
<tr> <td>Fri Jan 23 17:27:25 CST 2004</td><td>35131338</td><td>Fri Jan 23 17:27:26 CST 2004</td><td>178339</td><td>91938328</td><td>35131338</td><td>0</td></tr> 
<tr> <td>Fri Jan 23 19:17:36 CST 2004</td><td>46290676</td><td>Fri Jan 23 19:17:37 CST 2004</td><td>178339</td><td>138229004</td><td>46290676</td><td>0</td></tr> 
<tr> <td>Fri Jan 23 19:56:48 CST 2004</td><td>15276773</td><td>Fri Jan 23 19:56:49 CST 2004</td><td>178339</td><td>153505777</td><td>15276773</td><td>0</td></tr> 
<tr> <td>Fri Jan 23 20:44:57 CST 2004</td><td>18084240</td><td>Fri Jan 23 20:44:58 CST 2004</td><td>178339</td><td>171590017</td><td>18084240</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 02:09:28 CST 2004</td><td>17585653</td><td>Sat Jan 24 02:09:28 CST 2004</td><td>178389</td><td>17585653</td><td>17585653</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 02:10:38 CST 2004</td><td>578041</td><td>Sat Jan 24 02:10:38 CST 2004</td><td>178389</td><td>18163694</td><td>578041</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 02:10:54 CST 2004</td><td>16</td><td>Sat Jan 24 02:10:54 CST 2004</td><td>178389</td><td>18163710</td><td>16</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 02:11:51 CST 2004</td><td>17</td><td>Sat Jan 24 02:11:51 CST 2004</td><td>178389</td><td>18163727</td><td>17</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 02:34:30 CST 2004</td><td>15683063</td><td>Sat Jan 24 02:34:30 CST 2004</td><td>178389</td><td>33846790</td><td>15683063</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 03:03:09 CST 2004</td><td>19936378</td><td>Sat Jan 24 03:03:10 CST 2004</td><td>178389</td><td>53783168</td><td>19936378</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 04:43:04 CST 2004</td><td>34646968</td><td>Sat Jan 24 04:43:05 CST 2004</td><td>178390</td><td>34646968</td><td>34646968</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 04:58:30 CST 2004</td><td>8733420</td><td>Sat Jan 24 04:58:31 CST 2004</td><td>178390</td><td>43380388</td><td>8733420</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 05:17:51 CST 2004</td><td>11562711</td><td>Sat Jan 24 05:17:52 CST 2004</td><td>178390</td><td>54943099</td><td>11562711</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 05:22:34 CST 2004</td><td>2588743</td><td>Sat Jan 24 05:22:34 CST 2004</td><td>178390</td><td>57531842</td><td>2588743</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 06:10:08 CST 2004</td><td>30833286</td><td>Sat Jan 24 06:10:09 CST 2004</td><td>178390</td><td>88365128</td><td>30833286</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 06:23:16 CST 2004</td><td>9959950</td><td>Sat Jan 24 06:23:17 CST 2004</td><td>178390</td><td>98325078</td><td>9959950</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 07:11:02 CST 2004</td><td>21449903</td><td>Sat Jan 24 07:11:03 CST 2004</td><td>178390</td><td>119774981</td><td>21449903</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 07:23:57 CST 2004</td><td>10389177</td><td>Sat Jan 24 07:23:58 CST 2004</td><td>178390</td><td>130164158</td><td>10389177</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 08:31:35 CST 2004</td><td>52836173</td><td>Sat Jan 24 08:31:36 CST 2004</td><td>178390</td><td>183000331</td><td>52836173</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 08:39:38 CST 2004</td><td>5974353</td><td>Sat Jan 24 08:39:39 CST 2004</td><td>178390</td><td>188974684</td><td>5974353</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 09:11:44 CST 2004</td><td>23649689</td><td>Sat Jan 24 09:11:45 CST 2004</td><td>178390</td><td>212624373</td><td>23649689</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 09:23:02 CST 2004</td><td>8070409</td><td>Sat Jan 24 09:23:04 CST 2004</td><td>178390</td><td>220694782</td><td>8070409</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 09:32:36 CST 2004</td><td>5802999</td><td>Sat Jan 24 09:32:37 CST 2004</td><td>178390</td><td>226497781</td><td>5802999</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 10:11:48 CST 2004</td><td>27398258</td><td>Sat Jan 24 10:11:50 CST 2004</td><td>178390</td><td>253896039</td><td>27398258</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 10:59:42 CST 2004</td><td>31903601</td><td>Sat Jan 24 10:59:43 CST 2004</td><td>178390</td><td>285799640</td><td>31903601</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 11:44:10 CST 2004</td><td>34267084</td><td>Sat Jan 24 11:44:12 CST 2004</td><td>178390</td><td>320066724</td><td>34267084</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 11:44:36 CST 2004</td><td>881</td><td>Sat Jan 24 11:44:38 CST 2004</td><td>178390</td><td>320067605</td><td>881</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 11:45:00 CST 2004</td><td>545</td><td>Sat Jan 24 11:45:02 CST 2004</td><td>178390</td><td>320068150</td><td>545</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 12:01:05 CST 2004</td><td>6</td><td>Sat Jan 24 12:01:05 CST 2004</td><td>178391</td><td>6</td><td>6</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 12:07:25 CST 2004</td><td>1903464</td><td>Sat Jan 24 12:07:25 CST 2004</td><td>178391</td><td>1903470</td><td>1903464</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 12:11:37 CST 2004</td><td>1510152</td><td>Sat Jan 24 12:11:37 CST 2004</td><td>178391</td><td>3413622</td><td>1510152</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 12:53:48 CST 2004</td><td>33502137</td><td>Sat Jan 24 12:53:48 CST 2004</td><td>178391</td><td>36915759</td><td>33502137</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 13:49:49 CST 2004</td><td>44647185</td><td>Sat Jan 24 13:49:50 CST 2004</td><td>178391</td><td>81562944</td><td>44647185</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 14:18:05 CST 2004</td><td>16852958</td><td>Sat Jan 24 14:18:06 CST 2004</td><td>178391</td><td>98415902</td><td>16852958</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 15:18:36 CST 2004</td><td>45586940</td><td>Sat Jan 24 15:18:37 CST 2004</td><td>178391</td><td>144002842</td><td>45586940</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 16:20:48 CST 2004</td><td>1852705</td><td>Sat Jan 24 16:20:48 CST 2004</td><td>178397</td><td>1852705</td><td>1852705</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 16:26:20 CST 2004</td><td>1641374</td><td>Sat Jan 24 16:26:20 CST 2004</td><td>178397</td><td>3494079</td><td>1641374</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 17:31:17 CST 2004</td><td>43950199</td><td>Sat Jan 24 17:31:17 CST 2004</td><td>178397</td><td>47444278</td><td>43950199</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 17:39:26 CST 2004</td><td>188</td><td>Sat Jan 24 17:39:27 CST 2004</td><td>178397</td><td>47444466</td><td>188</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 17:41:13 CST 2004</td><td>570934</td><td>Sat Jan 24 17:41:14 CST 2004</td><td>178397</td><td>48015400</td><td>570934</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 18:32:10 CST 2004</td><td>32763017</td><td>Sat Jan 24 18:32:11 CST 2004</td><td>178397</td><td>80778417</td><td>32763017</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 18:32:27 CST 2004</td><td>4</td><td>Sat Jan 24 18:32:27 CST 2004</td><td>178397</td><td>80778421</td><td>4</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 18:34:10 CST 2004</td><td>7</td><td>Sat Jan 24 18:34:11 CST 2004</td><td>178397</td><td>80778428</td><td>7</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 18:34:31 CST 2004</td><td>7</td><td>Sat Jan 24 18:34:32 CST 2004</td><td>178397</td><td>80778435</td><td>7</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 20:34:02 CST 2004</td><td>73176378</td><td>Sat Jan 24 20:34:03 CST 2004</td><td>178397</td><td>153954813</td><td>73176378</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 20:36:41 CST 2004</td><td>182</td><td>Sat Jan 24 20:36:42 CST 2004</td><td>178397</td><td>153954995</td><td>182</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 20:36:58 CST 2004</td><td>465</td><td>Sat Jan 24 20:36:59 CST 2004</td><td>178397</td><td>153955460</td><td>465</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 20:37:11 CST 2004</td><td>273</td><td>Sat Jan 24 20:37:12 CST 2004</td><td>178397</td><td>153955733</td><td>273</td><td>0</td></tr> 
<tr> <td>Sat Jan 24 20:48:49 CST 2004</td><td>6203607</td><td>Sat Jan 24 20:48:50 CST 2004</td><td>178397</td><td>160159340</td><td>6203607</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 00:12:06 CST 2004</td><td>111438790</td><td>Sun Jan 25 00:12:08 CST 2004</td><td>178397</td><td>271598130</td><td>111438790</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 00:44:35 CST 2004</td><td>130510</td><td>Sun Jan 25 00:44:35 CST 2004</td><td>178398</td><td>130510</td><td>130510</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 00:47:33 CST 2004</td><td>436102</td><td>Sun Jan 25 00:47:33 CST 2004</td><td>178398</td><td>566612</td><td>436102</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 00:52:32 CST 2004</td><td>1051805</td><td>Sun Jan 25 00:52:32 CST 2004</td><td>178398</td><td>1618417</td><td>1051805</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 00:59:31 CST 2004</td><td>252224</td><td>Sun Jan 25 00:59:31 CST 2004</td><td>178399</td><td>252224</td><td>252224</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:01:08 CST 2004</td><td>191254</td><td>Sun Jan 25 01:01:08 CST 2004</td><td>178399</td><td>443478</td><td>191254</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:09:03 CST 2004</td><td>193688</td><td>Sun Jan 25 01:09:03 CST 2004</td><td>178400</td><td>193688</td><td>193688</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:10:34 CST 2004</td><td>88081</td><td>Sun Jan 25 01:10:34 CST 2004</td><td>178400</td><td>281769</td><td>88081</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:11:32 CST 2004</td><td>101838</td><td>Sun Jan 25 01:11:32 CST 2004</td><td>178400</td><td>383607</td><td>101838</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:12:51 CST 2004</td><td>143204</td><td>Sun Jan 25 01:12:51 CST 2004</td><td>178400</td><td>526811</td><td>143204</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:13:39 CST 2004</td><td>70798</td><td>Sun Jan 25 01:13:39 CST 2004</td><td>178400</td><td>597609</td><td>70798</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:14:28 CST 2004</td><td>21583</td><td>Sun Jan 25 01:14:28 CST 2004</td><td>178400</td><td>619192</td><td>21583</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:15:08 CST 2004</td><td>66440</td><td>Sun Jan 25 01:15:08 CST 2004</td><td>178400</td><td>685632</td><td>66440</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:18:00 CST 2004</td><td>413033</td><td>Sun Jan 25 01:18:00 CST 2004</td><td>178400</td><td>1098665</td><td>413033</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:19:06 CST 2004</td><td>143264</td><td>Sun Jan 25 01:19:06 CST 2004</td><td>178400</td><td>1241929</td><td>143264</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:20:10 CST 2004</td><td>99263</td><td>Sun Jan 25 01:20:10 CST 2004</td><td>178400</td><td>1341192</td><td>99263</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:21:51 CST 2004</td><td>78136</td><td>Sun Jan 25 01:21:51 CST 2004</td><td>178400</td><td>1419328</td><td>78136</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:24:00 CST 2004</td><td>431779</td><td>Sun Jan 25 01:24:00 CST 2004</td><td>178400</td><td>1851107</td><td>431779</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:27:15 CST 2004</td><td>509813</td><td>Sun Jan 25 01:27:15 CST 2004</td><td>178400</td><td>2360920</td><td>509813</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:30:58 CST 2004</td><td>741645</td><td>Sun Jan 25 01:30:58 CST 2004</td><td>178400</td><td>3102565</td><td>741645</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:33:37 CST 2004</td><td>1065341</td><td>Sun Jan 25 01:33:37 CST 2004</td><td>178400</td><td>4167906</td><td>1065341</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:37:12 CST 2004</td><td>1311886</td><td>Sun Jan 25 01:37:12 CST 2004</td><td>178400</td><td>5479792</td><td>1311886</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:38:34 CST 2004</td><td>392242</td><td>Sun Jan 25 01:38:34 CST 2004</td><td>178400</td><td>5872034</td><td>392242</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:40:03 CST 2004</td><td>458157</td><td>Sun Jan 25 01:40:03 CST 2004</td><td>178400</td><td>6330191</td><td>458157</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:40:32 CST 2004</td><td>6</td><td>Sun Jan 25 01:40:32 CST 2004</td><td>178400</td><td>6330197</td><td>6</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:49:01 CST 2004</td><td>3951924</td><td>Sun Jan 25 01:49:01 CST 2004</td><td>178400</td><td>10282121</td><td>3951924</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:50:00 CST 2004</td><td>94966</td><td>Sun Jan 25 01:50:00 CST 2004</td><td>178400</td><td>10377087</td><td>94966</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:50:39 CST 2004</td><td>15736</td><td>Sun Jan 25 01:50:39 CST 2004</td><td>178400</td><td>10392823</td><td>15736</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:53:38 CST 2004</td><td>1180228</td><td>Sun Jan 25 01:53:38 CST 2004</td><td>178400</td><td>11573051</td><td>1180228</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:55:05 CST 2004</td><td>507323</td><td>Sun Jan 25 01:55:05 CST 2004</td><td>178400</td><td>12080374</td><td>507323</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:57:06 CST 2004</td><td>778258</td><td>Sun Jan 25 01:57:06 CST 2004</td><td>178400</td><td>12858632</td><td>778258</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:57:47 CST 2004</td><td>9867</td><td>Sun Jan 25 01:57:47 CST 2004</td><td>178400</td><td>12868499</td><td>9867</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:58:18 CST 2004</td><td>81064</td><td>Sun Jan 25 01:58:18 CST 2004</td><td>178400</td><td>12949563</td><td>81064</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 01:59:32 CST 2004</td><td>3528</td><td>Sun Jan 25 01:59:32 CST 2004</td><td>178400</td><td>12953091</td><td>3528</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:00:20 CST 2004</td><td>22259</td><td>Sun Jan 25 02:00:20 CST 2004</td><td>178400</td><td>12975350</td><td>22259</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:01:35 CST 2004</td><td>3924</td><td>Sun Jan 25 02:01:35 CST 2004</td><td>178400</td><td>12979274</td><td>3924</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:02:02 CST 2004</td><td>2308</td><td>Sun Jan 25 02:02:02 CST 2004</td><td>178400</td><td>12981582</td><td>2308</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:02:27 CST 2004</td><td>15643</td><td>Sun Jan 25 02:02:27 CST 2004</td><td>178400</td><td>12997225</td><td>15643</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:03:39 CST 2004</td><td>2197</td><td>Sun Jan 25 02:03:39 CST 2004</td><td>178400</td><td>12999422</td><td>2197</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:05:26 CST 2004</td><td>608</td><td>Sun Jan 25 02:05:26 CST 2004</td><td>178400</td><td>13000030</td><td>608</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:06:02 CST 2004</td><td>3356</td><td>Sun Jan 25 02:06:02 CST 2004</td><td>178400</td><td>13003386</td><td>3356</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:07:42 CST 2004</td><td>188710</td><td>Sun Jan 25 02:07:42 CST 2004</td><td>178400</td><td>13192096</td><td>188710</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:08:15 CST 2004</td><td>104757</td><td>Sun Jan 25 02:08:15 CST 2004</td><td>178400</td><td>13296853</td><td>104757</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:09:50 CST 2004</td><td>630895</td><td>Sun Jan 25 02:09:50 CST 2004</td><td>178400</td><td>13927748</td><td>630895</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:11:11 CST 2004</td><td>447023</td><td>Sun Jan 25 02:11:11 CST 2004</td><td>178400</td><td>14374771</td><td>447023</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:13:54 CST 2004</td><td>1153034</td><td>Sun Jan 25 02:13:54 CST 2004</td><td>178400</td><td>15527805</td><td>1153034</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:16:34 CST 2004</td><td>1106094</td><td>Sun Jan 25 02:16:34 CST 2004</td><td>178400</td><td>16633899</td><td>1106094</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:17:31 CST 2004</td><td>136608</td><td>Sun Jan 25 02:17:31 CST 2004</td><td>178400</td><td>16770507</td><td>136608</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:18:11 CST 2004</td><td>114145</td><td>Sun Jan 25 02:18:11 CST 2004</td><td>178400</td><td>16884652</td><td>114145</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:18:52 CST 2004</td><td>92468</td><td>Sun Jan 25 02:18:52 CST 2004</td><td>178400</td><td>16977120</td><td>92468</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:19:30 CST 2004</td><td>119330</td><td>Sun Jan 25 02:19:31 CST 2004</td><td>178400</td><td>17096450</td><td>119330</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:20:05 CST 2004</td><td>47663</td><td>Sun Jan 25 02:20:05 CST 2004</td><td>178400</td><td>17144113</td><td>47663</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:20:33 CST 2004</td><td>554</td><td>Sun Jan 25 02:20:33 CST 2004</td><td>178400</td><td>17144667</td><td>554</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:21:18 CST 2004</td><td>362</td><td>Sun Jan 25 02:21:18 CST 2004</td><td>178400</td><td>17145029</td><td>362</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:24:25 CST 2004</td><td>151</td><td>Sun Jan 25 02:24:25 CST 2004</td><td>178400</td><td>17145180</td><td>151</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:26:13 CST 2004</td><td>312</td><td>Sun Jan 25 02:26:13 CST 2004</td><td>178400</td><td>17145492</td><td>312</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:35:55 CST 2004</td><td>356</td><td>Sun Jan 25 02:35:55 CST 2004</td><td>178401</td><td>356</td><td>356</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:36:17 CST 2004</td><td>174</td><td>Sun Jan 25 02:36:17 CST 2004</td><td>178401</td><td>530</td><td>174</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:36:39 CST 2004</td><td>222</td><td>Sun Jan 25 02:36:40 CST 2004</td><td>178401</td><td>752</td><td>222</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:37:17 CST 2004</td><td>201</td><td>Sun Jan 25 02:37:17 CST 2004</td><td>178401</td><td>953</td><td>201</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:39:27 CST 2004</td><td>120</td><td>Sun Jan 25 02:39:27 CST 2004</td><td>178401</td><td>1073</td><td>120</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:48:33 CST 2004</td><td>307</td><td>Sun Jan 25 02:48:33 CST 2004</td><td>178401</td><td>1380</td><td>307</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:49:27 CST 2004</td><td>113</td><td>Sun Jan 25 02:49:27 CST 2004</td><td>178401</td><td>1493</td><td>113</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 02:51:18 CST 2004</td><td>212</td><td>Sun Jan 25 02:51:18 CST 2004</td><td>178401</td><td>1705</td><td>212</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 03:19:22 CST 2004</td><td>280</td><td>Sun Jan 25 03:19:22 CST 2004</td><td>178402</td><td>280</td><td>280</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 03:20:42 CST 2004</td><td>146</td><td>Sun Jan 25 03:20:42 CST 2004</td><td>178402</td><td>426</td><td>146</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 03:24:16 CST 2004</td><td>211082</td><td>Sun Jan 25 03:24:16 CST 2004</td><td>178402</td><td>211508</td><td>211082</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 03:25:48 CST 2004</td><td>148759</td><td>Sun Jan 25 03:25:48 CST 2004</td><td>178402</td><td>360267</td><td>148759</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 03:27:15 CST 2004</td><td>239253</td><td>Sun Jan 25 03:27:15 CST 2004</td><td>178402</td><td>599520</td><td>239253</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 03:29:12 CST 2004</td><td>330413</td><td>Sun Jan 25 03:29:12 CST 2004</td><td>178402</td><td>929933</td><td>330413</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 03:32:26 CST 2004</td><td>590909</td><td>Sun Jan 25 03:32:26 CST 2004</td><td>178402</td><td>1520842</td><td>590909</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 05:02:06 CST 2004</td><td>202233</td><td>Sun Jan 25 05:02:06 CST 2004</td><td>178405</td><td>202233</td><td>202233</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 05:03:10 CST 2004</td><td>85813</td><td>Sun Jan 25 05:03:10 CST 2004</td><td>178405</td><td>288046</td><td>85813</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 05:06:36 CST 2004</td><td>457892</td><td>Sun Jan 25 05:06:36 CST 2004</td><td>178405</td><td>745938</td><td>457892</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 05:08:53 CST 2004</td><td>392583</td><td>Sun Jan 25 05:08:54 CST 2004</td><td>178405</td><td>1138521</td><td>392583</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 05:10:54 CST 2004</td><td>328733</td><td>Sun Jan 25 05:10:54 CST 2004</td><td>178405</td><td>1467254</td><td>328733</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 05:13:04 CST 2004</td><td>359024</td><td>Sun Jan 25 05:13:04 CST 2004</td><td>178405</td><td>1826278</td><td>359024</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 05:15:45 CST 2004</td><td>469789</td><td>Sun Jan 25 05:15:45 CST 2004</td><td>178405</td><td>2296067</td><td>469789</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 07:49:14 CST 2004</td><td>63563637</td><td>Sun Jan 25 07:49:14 CST 2004</td><td>178405</td><td>65859704</td><td>63563637</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 08:25:51 CST 2004</td><td>473334</td><td>Sun Jan 25 08:25:51 CST 2004</td><td>178407</td><td>473334</td><td>473334</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 08:28:08 CST 2004</td><td>342058</td><td>Sun Jan 25 08:28:08 CST 2004</td><td>178407</td><td>815392</td><td>342058</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 08:31:22 CST 2004</td><td>511698</td><td>Sun Jan 25 08:31:22 CST 2004</td><td>178407</td><td>1327090</td><td>511698</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 08:33:32 CST 2004</td><td>320321</td><td>Sun Jan 25 08:33:32 CST 2004</td><td>178407</td><td>1647411</td><td>320321</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 08:35:24 CST 2004</td><td>289455</td><td>Sun Jan 25 08:35:24 CST 2004</td><td>178407</td><td>1936866</td><td>289455</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 08:39:51 CST 2004</td><td>1363889</td><td>Sun Jan 25 08:39:51 CST 2004</td><td>178407</td><td>3300755</td><td>1363889</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 09:27:26 CST 2004</td><td>131804</td><td>Sun Jan 25 09:27:25 CST 2004</td><td>178408</td><td>131804</td><td>131804</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 09:29:26 CST 2004</td><td>188220</td><td>Sun Jan 25 09:29:25 CST 2004</td><td>178408</td><td>320024</td><td>188220</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 09:30:07 CST 2004</td><td>42748</td><td>Sun Jan 25 09:30:06 CST 2004</td><td>178408</td><td>362772</td><td>42748</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 09:33:21 CST 2004</td><td>426763</td><td>Sun Jan 25 09:33:20 CST 2004</td><td>178408</td><td>789535</td><td>426763</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 09:36:11 CST 2004</td><td>438539</td><td>Sun Jan 25 09:36:10 CST 2004</td><td>178408</td><td>1228074</td><td>438539</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 09:37:15 CST 2004</td><td>138260</td><td>Sun Jan 25 09:37:15 CST 2004</td><td>178408</td><td>1366334</td><td>138260</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 09:38:36 CST 2004</td><td>175101</td><td>Sun Jan 25 09:38:35 CST 2004</td><td>178408</td><td>1541435</td><td>175101</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 09:41:44 CST 2004</td><td>83071</td><td>Sun Jan 25 09:41:43 CST 2004</td><td>178408</td><td>1624506</td><td>83071</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 09:50:44 CST 2004</td><td>136343</td><td>Sun Jan 25 09:50:45 CST 2004</td><td>178409</td><td>136343</td><td>136343</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 09:52:05 CST 2004</td><td>111900</td><td>Sun Jan 25 09:52:05 CST 2004</td><td>178409</td><td>248243</td><td>111900</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 09:53:02 CST 2004</td><td>80646</td><td>Sun Jan 25 09:53:02 CST 2004</td><td>178409</td><td>328889</td><td>80646</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 09:53:25 CST 2004</td><td>8</td><td>Sun Jan 25 09:53:26 CST 2004</td><td>178409</td><td>328897</td><td>8</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 09:54:47 CST 2004</td><td>121123</td><td>Sun Jan 25 09:54:47 CST 2004</td><td>178409</td><td>450020</td><td>121123</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 09:57:04 CST 2004</td><td>260385</td><td>Sun Jan 25 09:57:04 CST 2004</td><td>178409</td><td>710405</td><td>260385</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 09:58:01 CST 2004</td><td>119826</td><td>Sun Jan 25 09:58:01 CST 2004</td><td>178409</td><td>830231</td><td>119826</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 09:59:13 CST 2004</td><td>153297</td><td>Sun Jan 25 09:59:13 CST 2004</td><td>178409</td><td>983528</td><td>153297</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 10:01:07 CST 2004</td><td>259385</td><td>Sun Jan 25 10:01:07 CST 2004</td><td>178409</td><td>1242913</td><td>259385</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 10:02:27 CST 2004</td><td>169717</td><td>Sun Jan 25 10:02:27 CST 2004</td><td>178409</td><td>1412630</td><td>169717</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 10:04:28 CST 2004</td><td>272755</td><td>Sun Jan 25 10:04:28 CST 2004</td><td>178409</td><td>1685385</td><td>272755</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 10:11:45 CST 2004</td><td>1906842</td><td>Sun Jan 25 10:11:45 CST 2004</td><td>178409</td><td>3592227</td><td>1906842</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 10:43:32 CST 2004</td><td>11556657</td><td>Sun Jan 25 10:43:32 CST 2004</td><td>178409</td><td>15148884</td><td>11556657</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 11:47:16 CST 2004</td><td>22740852</td><td>Sun Jan 25 11:47:16 CST 2004</td><td>178409</td><td>37889736</td><td>22740852</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 15:47:49 CST 2004</td><td>13830557</td><td>Sun Jan 25 15:47:48 CST 2004</td><td>178432</td><td>13830557</td><td>13830557</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 15:50:04 CST 2004</td><td>1260066</td><td>Sun Jan 25 15:50:03 CST 2004</td><td>178432</td><td>15090623</td><td>1260066</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 15:50:33 CST 2004</td><td>336</td><td>Sun Jan 25 15:50:32 CST 2004</td><td>178432</td><td>15090959</td><td>336</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 16:32:26 CST 2004</td><td>20472720</td><td>Sun Jan 25 16:32:26 CST 2004</td><td>178433</td><td>20472720</td><td>20472720</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 16:33:12 CST 2004</td><td>308022</td><td>Sun Jan 25 16:33:12 CST 2004</td><td>178433</td><td>20780742</td><td>308022</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 16:40:04 CST 2004</td><td>4394189</td><td>Sun Jan 25 16:40:04 CST 2004</td><td>178433</td><td>25174931</td><td>4394189</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 16:40:27 CST 2004</td><td>131618</td><td>Sun Jan 25 16:40:28 CST 2004</td><td>178433</td><td>25306549</td><td>131618</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 16:42:05 CST 2004</td><td>263</td><td>Sun Jan 25 16:42:05 CST 2004</td><td>178433</td><td>25306812</td><td>263</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 16:42:20 CST 2004</td><td>114</td><td>Sun Jan 25 16:42:21 CST 2004</td><td>178433</td><td>25306926</td><td>114</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 16:44:04 CST 2004</td><td>137</td><td>Sun Jan 25 16:44:04 CST 2004</td><td>178433</td><td>25307063</td><td>137</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 16:44:55 CST 2004</td><td>216</td><td>Sun Jan 25 16:44:55 CST 2004</td><td>178433</td><td>25307279</td><td>216</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 16:52:19 CST 2004</td><td>97</td><td>Sun Jan 25 16:52:20 CST 2004</td><td>178434</td><td>97</td><td>97</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 16:53:44 CST 2004</td><td>171</td><td>Sun Jan 25 16:53:44 CST 2004</td><td>178434</td><td>268</td><td>171</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 16:57:05 CST 2004</td><td>112</td><td>Sun Jan 25 16:57:05 CST 2004</td><td>178434</td><td>380</td><td>112</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 16:59:01 CST 2004</td><td>101</td><td>Sun Jan 25 16:59:01 CST 2004</td><td>178434</td><td>481</td><td>101</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 17:23:46 CST 2004</td><td>11967668</td><td>Sun Jan 25 17:23:46 CST 2004</td><td>178434</td><td>11968149</td><td>11967668</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 18:53:08 CST 2004</td><td>34174516</td><td>Sun Jan 25 18:53:08 CST 2004</td><td>178435</td><td>34174516</td><td>34174516</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 19:17:55 CST 2004</td><td>110310</td><td>Sun Jan 25 19:17:55 CST 2004</td><td>178437</td><td>110310</td><td>110310</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 20:00:33 CST 2004</td><td>29999893</td><td>Sun Jan 25 20:00:33 CST 2004</td><td>178437</td><td>30110203</td><td>29999893</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 20:54:50 CST 2004</td><td>41740379</td><td>Sun Jan 25 20:54:49 CST 2004</td><td>178437</td><td>71850582</td><td>41740379</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 20:54:50 CST 2004</td><td>41740379</td><td>Sun Jan 25 20:54:51 CST 2004</td><td>178437</td><td>71850582</td><td>-1</td><td>Warning</td></tr> 
<tr> <td>Sun Jan 25 20:55:20 CST 2004</td><td>1015</td><td>Sun Jan 25 20:55:21 CST 2004</td><td>178437</td><td>71851597</td><td>1015</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 22:11:14 CST 2004</td><td>40045048</td><td>Sun Jan 25 22:11:14 CST 2004</td><td>178438</td><td>40045048</td><td>40045048</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 22:57:59 CST 2004</td><td>36937290</td><td>Sun Jan 25 22:57:59 CST 2004</td><td>178438</td><td>76982338</td><td>36937290</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 23:14:23 CST 2004</td><td>14236361</td><td>Sun Jan 25 23:14:24 CST 2004</td><td>178438</td><td>91218699</td><td>14236361</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 23:14:39 CST 2004</td><td>4</td><td>Sun Jan 25 23:14:39 CST 2004</td><td>178438</td><td>91218703</td><td>4</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 23:15:37 CST 2004</td><td>13</td><td>Sun Jan 25 23:15:38 CST 2004</td><td>178438</td><td>91218716</td><td>13</td><td>0</td></tr> 
<tr> <td>Sun Jan 25 23:56:28 CST 2004</td><td>34250709</td><td>Sun Jan 25 23:56:28 CST 2004</td><td>178438</td><td>125469425</td><td>34250709</td><td>0</td></tr> 
<tr> <td>Mon Jan 26 00:00:47 CST 2004</td><td>3372314</td><td>Mon Jan 26 00:00:47 CST 2004</td><td>178438</td><td>128841739</td><td>3372314</td><td>0</td></tr> 
<tr> <td>Mon Jan 26 00:41:18 CST 2004</td><td>3159287</td><td>Mon Jan 26 00:41:18 CST 2004</td><td>178440</td><td>3159287</td><td>3159287</td><td>0</td></tr> 
<tr> <td>Mon Jan 26 00:41:34 CST 2004</td><td>6</td><td>Mon Jan 26 00:41:34 CST 2004</td><td>178440</td><td>3159293</td><td>6</td><td>0</td></tr> 
<tr> <td>Mon Jan 26 02:46:02 CST 2004</td><td>93609588</td><td>Mon Jan 26 02:46:03 CST 2004</td><td>178440</td><td>96768881</td><td>93609588</td><td>0</td></tr> 
<tr> <td>Mon Jan 26 04:14:45 CST 2004</td><td>437249</td><td>Mon Jan 26 04:14:45 CST 2004</td><td>178446</td><td>437249</td><td>437249</td><td>0</td></tr> 
<tr> <td>Mon Jan 26 04:45:19 CST 2004</td><td>252743</td><td>Mon Jan 26 04:45:20 CST 2004</td><td>178448</td><td>252743</td><td>252743</td><td>0</td></tr> 
<tr> <td>Mon Jan 26 04:47:13 CST 2004</td><td>292449</td><td>Mon Jan 26 04:47:13 CST 2004</td><td>178448</td><td>545192</td><td>292449</td><td>0</td></tr> 
<tr> <td>Mon Jan 26 04:53:32 CST 2004</td><td>1581658</td><td>Mon Jan 26 04:53:32 CST 2004</td><td>178448</td><td>2126850</td><td>1581658</td><td>0</td></tr> 
<tr> <td>Mon Jan 26 06:05:05 CST 2004</td><td>42587810</td><td>Mon Jan 26 06:05:06 CST 2004</td><td>178448</td><td>44714660</td><td>42587810</td><td>0</td></tr> 
<tr> <td>Mon Jan 26 07:08:36 CST 2004</td><td>37368755</td><td>Mon Jan 26 07:08:37 CST 2004</td><td>178448</td><td>82083415</td><td>37368755</td><td>0</td></tr> 
<tr> <td>Mon Jan 26 07:09:57 CST 2004</td><td>350</td><td>Mon Jan 26 07:09:58 CST 2004</td><td>178448</td><td>82083765</td><td>350</td><td>0</td></tr> 
<tr> <td>Mon Jan 26 07:14:53 CST 2004</td><td>206</td><td>Mon Jan 26 07:14:54 CST 2004</td><td>178448</td><td>82083971</td><td>206</td><td>0</td></tr> 
<tr> <td>Mon Jan 26 07:34:42 CST 2004</td><td>252</td><td>Mon Jan 26 07:34:43 CST 2004</td><td>178448</td><td>82084223</td><td>252</td><td>0</td></tr> 
<tr> <td>Mon Jan 26 07:59:51 CST 2004</td><td>475279</td><td>Mon Jan 26 07:59:51 CST 2004</td><td>178450</td><td>475279</td><td>475279</td><td>0</td></tr> 
<tr> <td>Mon Jan 26 08:06:52 CST 2004</td><td>1527046</td><td>Mon Jan 26 08:06:52 CST 2004</td><td>178450</td><td>2002325</td><td>1527046</td><td>0</td></tr> 
<tr> <td>Mon Jan 26 08:11:19 CST 2004</td><td>1470809</td><td>Mon Jan 26 08:11:19 CST 2004</td><td>178450</td><td>3473134</td><td>1470809</td><td>0</td></tr> 
<tr> <td>Mon Jan 26 08:31:06 CST 2004</td><td>10403506</td><td>Mon Jan 26 08:31:07 CST 2004</td><td>178450</td><td>13876640</td><td>10403506</td><td>0</td></tr> 
</table> 
 Written by analyseL1A.pl  
<!-- Author: rsw for Marcel, SCN. -->


<!-- Date: Wed Jan 28 10:01:49 2004 -->
<h3> Further investigation in SVXMon Missing HDI issue </h3> 
<p> 
We looked into single events where a single HDI (i.e. e1a0) was incriminated to be missing by SVXMon,  <pre> 
errlog 13 {Missing HDI} SIXDMon Silicon/S/B1/W10/L0 79074 1074592807 {HDI with fib id0 225 fib id1 160 not found in the data 
stream for trigger 79074} 
errlog 13 {Wrong HDI Count} SIXDMon Silicon/DAQ 79074 1074592807 {The number of HDIs in the data stream has changed from 514 
to 513} 
</pre> 
 
Contrary to what we <a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=181&frame=2&anchor=14:06:36'> 
found </a> in failures where 200 HDIs are missing, this does seem to be real - 
<pre>  
     run 178165 evt 79094 
 
     1600: 3003a120 a0207f05 c1c10000 e19441fc  
     1604: 8d2000f3 8c203b04 3c1e3d20 3e088b20  
     1608: 49074a2f 4b364c05 69046a16 6b276c2f  
     1612: 6d068a20 89208820 8720a620 a5202604  
     1616: 270f2823 290b2a02 a4203307 3423353f  
     1620: 360a37f3 53045409 55445620 5704a320  
     1624: a22050f3 511352f3 a120a020 7ff3c1c1  
     1628: <b>e1a041fc </b> e1a141fc 852000f3 7307740b  
</pre> 
The HDI is there, but the data is not terminated correctly - SCN says this is the re-formatter 'fixing' garbage data and cutting the HDI out, not rejecting the event. 
<p> 
We found also incidents like  
<pre> 
     run 178165 evt 1204546 
 
     2148: e1947ffc 8d1f00f3 8c1f8b1f 8a1f891f  
     2152: 881f871f a61fa51f a41fa31f a21f50f3  
     2156: 510f52f3 a11fa01f 7ff3c1c1 <b>e1a07ffc   
     2160: f0d0d38f 100c3108 32082307 00d0d28f  
     2164: 10d0d1af 1263f368 19652c66 226000d0  
     2168: dfaf1371 f1c4c4c4 </b> e1a17ffc 851f0001  
</pre> 
where garbage is visible - it has been spotted already in the  
<a href='http://www-cdfonline.fnal.gov/internal/usr/lmiller/cgi-bin/newproblemdb/DBreport.pl?detsys=SVX&wedge=B1W10&layer=L0'> PDB </a> and DSEM reports garbage data at the level of <1%, sometimes flaring up to 20-30%, see <a href='http://nucsb03.fnal.gov/cgi-bin/errorMon?HDI=e1a0&RUN=all'> here </a> 
<!-- Author: Rong, Rainer -->

<!-- Date: Wed Jan 28 11:01:24 2004 -->
Entered in Silicon database: <ul>
<li>Ladder: SB2W2L4 
<li>problem n: 672 
<li>Status: needs_quiet_time
<li>Type: DVDD
<li>Action: Diagnosis
<li>entered by: rong 
<li>description is: e224 could be powered up during quiet time and had run through cosmic data over night, but it started to trip before collision and had 3 times of trip during first hour data taking. It was removed from hardware database and need quiet time to study it by Svxdaq. 
<li>See <a href="http://www-cdfonline.fnal.gov/internal/usr/lmiller/cgi-bin/newproblemdb/DBreport.pl?detsys=SVX&wedge=B2W2&layer=L4">report</a></ul>
<!-- Author: Problem DB -->



<!-- Date: Wed Jan 28 12:18:46 2004 -->
<center><a href='/elog/notebooks/silicon/Wed_Jan_28_2004_121817.ps'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3411' ></a><a href='/elog/notebooks/silicon/Wed_Jan_28_2004_121826.ps'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3413' ></a></center>
<!-- Author: SCN -->
<!-- Comment: Wed Jan 28 12;34;40 comment by...SCN, Taka -->
<h3>4 Channel Estring Testing</h3>
Taka and I met and did some testing of the new GB firmware to handle the Readout Return from 4 different FFOs.  We found <ul>
<li> Unused channels pull error immediately.  This is because
the SRCTM-Mezzanine card pulls the input to the ghostbuster high
if the channel is not present.  Two options to fix this <ol>
<li> Ken modifies the Mezzanine card to make the lines default low
<li> Taka adds a register to selectively disable particular channels in the GB.
</ol>
Option 2 is the safer way to go, since if the connection is broken or a Fib crate dies, we will immediately pull ERROR, and then a human can disable this channel by VME via the HDWDB, so I asked Taka to implement it.  He is going to do it, should take a day or so.
<li> (Left plot) The condition for pulling Error is <i>(# kicks >= threshold)</i> which is shown for Channel 0 above.  I was
confused that the GB was pulling Error (BRAKE0 and GBBOUT 0 above) "every time" but in fact we triggered on it and started a run and saw that it is every time after the threshold.
<li> (Right plot) There is a considerable delay in processing, something like 140 microSecs between when the Mezzanine board sends the signal to the GB (GBIN 0) and the GB starts processing
(CABLIN 0).  This is not so good, because it means you only 
pull the error 140 microSecs <i>after</i> the condition happened, so
what you think was 10 kicks could in fact be 10 + (140 uS/kick period) so at most ~20 kicks.  Taka will try to understand this delay and see if it cannot be increased, but I'd expect that
the kicktime for dangerous cases would be something like 50 uS or more, which only means something like <= 3 extra kicks.
</ul>
More testing later when new firmware arrives in the next few days.



<!-- Date: Wed Jan 28 15:20:27 2004 -->
<center><a href='/elog/notebooks/silicon/Wed_Jan_28_2004_151942.ps'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3417' ></a><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3418&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3419' ></a><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3420&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3421' ></a></center>
<!-- Author: David -->
<!-- Comment: Wed Jan 28 16;05;16 comment by...David -->
<h3>SB2W2L4</h3>
Current plots for AVDD (black) and DVDD (red) for important transitional dates:
<ol>
<li>10/05/03 (day 1007): first sign of DVDD SPT trips after <a href=
'/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=171&frame=2&anchor=09:55:39'>
most of SVXII tripped off at 9:48am</a> from SRC bug (no Keep Alives sent).</li>
<li>11/24/03-12/6/03 (1057-1069): first physics runs after shutdown on 11/24. On 11/28 (1061), SB2W2L4 was taken out of running because of high readout times due to garbage data coming from z side (chips a2, a1 and a0). On 12/1/03 (1064) at ~4pm - after the big quench -  the offending chips were chopped, the RDriver for chip 89 increased to 7, and the ladder reintegrated. When data taking resumed on the 6th (1069) this ladder had to be removed again because of persistent DVDD tripping.</li>
<li>1/21/04-1/28/04 (1115-1122): reintegrated for data taking on 1/10/04 and taken out on 1/22 due to DVDD SPT. Swapped power supply on 1/26 (1120) and reintegrated. SPTs still there, but not the same as before - only see DVDD going over 350mA (now 400mA) limit, don't see DVDD rising to AVDD current level (the one DVDD jump to near AVDD levels at 1120.6 is at transition to standby and does not cause trip).</li>
</ol>
The trips since the PS swap have been due to the currents jumping just beyond the SPT limit. No trips have been seen of the kind that jump to the AVDD current level. Perhaps more time is required to determine if this feature is gone thanks to the PS swap, but this means we have to run it with cosmic, calib, and data taking. Will run it with svxdaq in next quiet time to determine current draw and reintegrate it near the end of the next store.

<!-- Date: Wed Jan 28 16:20:43 2004 -->
Entered in Silicon database: <ul>
<li>Ladder: SB2W2L4 
<li>problem n: 672 
<li>Status: needs_quiet_time
<li>Type: DVDD
<li>Action: Diagnosis
<li>entered by: dkclark 
<li>description is: <a href= 'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=182&frame=2&anchor=15:20:27'> Traced the DVDD SP trips</a> back to SVX-wide trip on Oct. 5th due to SRC bug. PS swap on 1/26 has not eliminated all trips, still requires study with svxdaq before re-integrating. 
<li>See <a href="http://www-cdfonline.fnal.gov/internal/usr/lmiller/cgi-bin/newproblemdb/DBreport.pl?detsys=SVX&wedge=B2W2&layer=L4">report</a></ul>
<!-- Author: Problem DB -->
<!-- Comment: Wed Jan 28 17;27;04 comment by...rainer -->
<pre>oups ...</pre>


<!-- Date: Thu Jan 29 00:01:43 2004 -->
<center><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3426&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3427' ></a><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3428&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3429' ></a></center>
<!-- Author: Rong -->
<!-- Comment: Thu Jan 29 00;16;56 comment by...David, Rong -->
<p>Quiet time study of SB2W2L4 with SVXDAQ: <p>
<ol>
<li>Ran with all chips with normal RDriver settings. Had garbage data from z side as seen in the past.</li>
<li>Ran without last 3 chips (see step down in AVDD current for time stamp). No errors except for very low rate c2c2 in first few events.</li>
<li>Ran with several higher Rdriver settings for chip 89 as it was suspected of dropping readout. Currents remained the same. c2c2 errors remained - suspect chip 88 may need RDriver changes, but need more time to determine proper setting.</li>
<li>Ran with above settings (same as in HDWDB) in NN mode. Currents remained the same.</li>
<li>Performed a COLDSTART - currents still stable.</li>
</ol>
I would like to leave this ladder on standby until Shotsetup Run is started to see how the currents behave on their own. Remove before AAA_CURRENT run is config'd (don't forget to update and load CAENvolts file). Near the end of the store, at low luminosities so that downtime incurred is not so costly, integrate this ladder to see how it behaves with the other kids.

<!-- Comment: Thu Jan 29 00;33;30 comment by...Rong, David -->
<pre>Rough time information:
     power up the ladder @ 22:58 (rapidly increaing of current)
1. ~ 23:05 start to run Svxdaq with all chips (drop on DVDD)
2. ~ 23:21 chop 3 chips (step down on AVDD)
3. ~ 23:27 change R-driver setting...
4. ~ 23:35 turn on the pedestal subtraction (current remain)
5. ~ 23:53 cold start (see 50mA drop on AVDD and 5mA drop on DVDD)
     then we leave the ladder at 'standby' state. The current is stable.</pre>


<!-- Date: Thu Jan 29 09:15:40 2004 -->
<center><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3434&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3435' ></a><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3436&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3437' ></a></center>
<!-- Author: Rong -->
<!-- Comment: Thu Jan 29 09;24;07 comment by...Rong -->
<pre>- AVDD and DVDD are both stable overnight and had no trip.
- 08:05  The current of AVDD and DVDD of SB2W2L4.
Both current jump up around 08:05 while Run Control started 
to take Cosmic data without Silicon (All HV standby even for COT)
- 08:42 try to cold start for SVX, both current step down. Maybe there is something affecting this ladder if RC or other detectors are configuring?

p.s. When I tried to cold start, b0fib06 failed to config.
- tried to reset and shepherde from RC but failed
- tried to vxlogin and soft-reboot the crate then re-partition/config. It worked.
- took sometime to figure out how to do, and that's why AVDD stayed low for a while. It came back after I activated the run.
</pre>

<!-- Comment: Thu Jan 29 10;03;18 comment by...rainer -->
shepherding a fib crate almost never helps.  that's because  the errors ist most often caused by a ladder not initializing:
<pre>
(MLE) b0fib06:Messenger:8:44:27 AM->Error Initializing HDI Slot 9 Chan 7: SVX B3
W7
(MLE) b0fib06:Messenger:8:44:27 AM->Statechange failed
(MLE) b0fib06:Messenger:8:46:43 AM->Error Initializing HDI Slot 9 Chan 7: SVX B3
W7
(MLE) b0fib06:Messenger:8:46:43 AM->Statechange failed
CT: 2004.01.29 08:48:12
48'12" b0dap42.fnal.gov:Thread-77:8:48:12 AM->Failed Shepherding client FIB_SVX_
06 (b0fib06)[MLE]
(RC)   9:7:14 Config -> READY
(RC)   9:7:29 Activate -> ACTIVE
</pre>
SB3W7L2 failed to readback. this is a ladder which caused trouble earlier. it might have a flaky readback due to AVDD2. you should mask the readback of this ladder.

<!-- Comment: Thu Jan 29 10;07;33 comment by...rainer -->
your observation of high DVDD current is consistent with what was observed 
<a href=
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=182&frame=2&anchor=03:54:49'>
here </a> (trip on coldstart while taking dmod, twice).





<!-- Date: Thu Jan 29 11:53:23 2004 -->
<pre>e372 readback masked in hardware database.</pre>
<!-- Author: Rong -->


<!-- Date: Thu Jan 29 12:36:55 2004 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3438&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3439  width='300' align='top'></a></td><td>SVT SpyMon plot</td></tr></table>
<!-- Author: Rong -->


<!-- Date: Thu Jan 29 15:29:42 2004 -->
Got paged due to SVXMon error message of 6 chips unbiased.<a href='http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2004&action=view&page=86&frame=2&anchor=14:45:26.470'> here</a>. 
<br>
SB1W3L3 doesn't have entry in Problem DB <br>
IB4W1L3 had bias current readback problem history also FE ISEL bits setting changed after beam incident<a href="http://www-cdfonline.fnal.gov/internal/usr/lmiller/cgi-bin/newproblemdb/DBreport.pl?detsys=ISL&wedge=B4W1&layer=L3"> here </a> <br>
IB2w1l0 also has FE bit setting changed<a href="http://www-cdfonline.fnal.gov/internal/usr/lmiller/cgi-bin/newproblemdb/DBreport.pl?detsys=ISL&wedge=B2W1&layer=L0"> here </a>. 

<!-- Author: Rong -->


<!-- Date: Thu Jan 29 15:46:38 2004 -->
<center><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3440&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3441' ></a><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3442&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3443' ></a><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3444&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3445' ></a><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3446&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3447' ></a></center>
<!-- Author: David -->
<!-- Comment: Thu Jan 29 16;13;23 comment by...David -->
<p><b>SB0W1L3 Chip 86 investigation</b><P>

Plots of phi cluster position are numbered from left to right, top to bottom. Below are the ranges of runs for which the above behaviors are seen.
<ol>
<li>Physics seen on all chips before 12/20 quench:<br>
last Run=176696, 12/20/03, Last run before quench

<li>No Physics from chip 86 after quench:<br>
1st Run=177199, 12/31/03, 16:24:34-16:31:29 1st run since 12/20 quench
<br>last Run=177315, 1/3/04, 08:33:30-12:43:58

<li>Some physics with hot channels:<br>
1st Run=177316, 1/3/04, 13:10:45-20:09:25
<br>last Run=177345, 1/4/04, 16:29:06-00:36:20

<li>Physics returns to chip 86:<br>
1st Run=177415, 1/6/04, 19:22:18-22:41:30
<br>latest run=178066, 1/18/04
</ol>
There was no significant happenings in the e-logs between run 177315 and 177316 on 1/4/04 that suggest a possible cause for the chip's "partial recovery". But, between 1/4 and 1/6 we had our first instance of a fatal SRC error due to resonance (Jan 5) and high spikes in the proton losses which required taking the silicon out of running for a while on the 6th. It's not clear if either of these incidents had anything to do with the recovery of chip 86. This chip should be monitored to see if errors return.
<!-- Date: Thu Jan 29 15:58:15 2004 -->
Entered in Silicon database: <ul>
<li>Ladder: SB5W11L4 
<li>problem n: 1200 
<li>Status: needs_access
<li>Type: Optical_other
<li>Action: Incident
<li>entered by: rong 
<li>description is: B0H since run 178540. study during quiet time, swap cables, swap RX. 
<li>See <a href="http://www-cdfonline.fnal.gov/internal/usr/lmiller/cgi-bin/newproblemdb/DBreport.pl?detsys=SVX&wedge=B5W11&layer=L4">report</a></ul>
<!-- Author: Problem DB -->




<!-- Date: Thu Jan 29 16:34:15 2004 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3448&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3449  width='300' align='top'></a></td><td>Ibias history for SB1W3L3. The PS Gui shows bias current at 
10uA which is close to the history.</td></tr></table>
<!-- Author: Rong -->


<!-- Date: Thu Jan 29 16:35:33 2004 -->
<center><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3450&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3451' ></a><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3452&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3453' ></a><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3454&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3455' ></a><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3460&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3461' ></a></center>
<!-- Author: Rong -->
<!-- Comment: Thu Jan 29 17;15;37 comment by...Rong -->
I Bias for IB4W1L3 bias0 and 1 and IB2W1L3 bias 0 and 1.<br>
IB4W1L3 has current readback problem and shows 0 on PSGui <br>
IB2W1L3 shows 15uA and 13uA on PSGui.
<!-- Date: Thu Jan 29 16:38:27 2004 -->
<center><a href='/elog/notebooks/silicon/Thu_Jan_29_2004_163757.ps'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3457' ></a><a href='/elog/notebooks/silicon/Thu_Jan_29_2004_163809.ps'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3459' ></a></center>
<!-- Author: Rainer -->
<!-- Comment: Thu Jan 29 16;48;13 comment by...Rainer -->
Occupancies of f413 of the incriminated run 

<a href=
'/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=182&frame=2&anchor=15:29:42'>178593</a>, to be compared with 
<a href=
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=177&frame=2&anchor=10:46:26'>176731</a> before the ISL bit <a href=' http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=copyentry&page=179&time=16:38:37'> change </a>. I conclude that maybe the occupancies are sufficiently different such that the bias dog barks. we could study the ladder in the usual way  in quiet time just to be sure - since the bias currents are (and have been)  reading zero. 

<!-- Comment: Thu Jan 29 17;06;53 comment by...Rainer -->
<pre>This is probably a fluke due to the release change - if it happens again, we should have a closer look ...</pre>



<!-- Date: Fri Jan 30 07:27:59 2004 -->
<h3> Controlled Access </h3> 
<p> 
two groups: 
<li> Stefano and Julia at b0fib02/SWB e5b4 Slot#19 HDI#9 (swap w/HDI#8): 
     wedges SB0W4-9,SB1W4-9,SB2W4-9 
<p> 
A proposal of The Plan: (Ms Cleo Mail) <p> 
 
1) Turn off ladders connected to b0fib02, which is BH(0,1,2) Wedges (4,5,6,7,8,9 
)<p> 
2) Execute "killAll" on b0fib02 CPU (to ensure there is auto-attempt to resynch 
the FFO, which may 
    trip the other ladders) <p> 
3) Stefano and <somebody> go in with a spare FFO just in case 
<p> 
4) Stefano reprograms the FFO, hits reset on the crate CPU in the same crate. <p>  
The CPU should 
boot, and shortly (~ 30 sec) after the FFO resynch-you will see the VME access L 
EDs on the FIBs and FFO when 
it happens.<p> 
5) Somebody upstairs turns on one wedge in this crate, takes some standard SVXDA 
Q data with it, 
makes sure it is all good (a pre-access test wouldn't hurt either, but if the FFO for some reason 
is screwed up, you probably won't even get the wedge on.  I would have a hard time believing <p> 
the FFO could cause eg. a new bit error on one ladder) 
6) Rest of the wedges powered up, perhaps some data taken through SVXMon just to 
 see it is ok although 
I don't consider it necessary so skip it if time is short. <p> 
7) Stefano and <whoever> exit the Collision Hall 
<p> 
<li> Rong and Rainer at b0fib04/NET: 
     wedges SB3W0-3,SB0W10-11,SB4W0-3,SB4W10-11,SB5W0-3,SB5W10-11 
     IB5W0  
<!-- Author: Rainer -->



<!-- Date: Fri Jan 30 09:44:38 2004 -->
<pre>Firmware modification to FFO of fib02: 
 
ladders connected to fib02 were turned off, killall on fib02 CPU 
issued and Stefano reprogrammed the firmware. The CPU booted without problems and a wedge was successfully turned to STBY. 
All wedeges were turned to STBY and a coldstart was performed. 
Stefano also upgraded the spare FFO in the B0 TS.</pre>
<!-- Author: Stefano and Julia -->


<!-- Date: Fri Jan 30 09:52:43 2004 -->
<pre>Swap RX for SB5W11L4 (e5b4) bit0 high problem. 
- we did swap fiber between e5b3 and e5b4 then the bit0 high error stay with PIPE 
- Swap RX 1123 out and 0186 in. 
- Perfect data taken; also ran with all 10 ladders in this FTM to make sure everything is good.</pre>
<!-- Author: Rainer and Rong -->

<!-- Date: Fri Jan 30 09:56:39 2004 -->
Entered in Silicon database: <ul>
<li>Ladder: SB5W11L4 
<li>problem n: 1200 
<li>Status: fixed
<li>Type: RX
<li>Action: Swap_RX
<li>entered by: rong 
<li>description is: By swaping fiber with neighbor channel, bit0 high stay with DAQ PIPE9, so it's RX problem. We swap RX 0186 out and 1123 in. Data is perfect after that.<table border=1>
<tr><td><b>IN:</b></td><td><a href="http://www-cdfonline.fnal.gov/internal/usr/lmiller/cgi-bin/rxdb/DBreport.pl?serial=1123">1123</a></td></tr>
<tr><td><b>OUT:</b></td><td><a href="http://www-cdfonline.fnal.gov/internal/usr/lmiller/cgi-bin/rxdb/DBreport.pl?serial=0186">0186</a></td></tr></table> 
<li>See <a href="http://www-cdfonline.fnal.gov/internal/usr/lmiller/cgi-bin/newproblemdb/DBreport.pl?detsys=SVX&wedge=B5W11&layer=L4">report</a></ul>
<!-- Author: Problem DB -->


<!-- Date: Fri Jan 30 09:57:46 2004 -->
Entered in Silicon database: <ul>
<li>Ladder: SB5W11L4 
<li>problem n: 1200 
<li>Status: fixed
<li>Type: RX
<li>Action: Swap_RX
<li>entered by: rong 
<li>description is: Typo! it's 1123 out andd 0186 in.<table border=1>
<tr><td><b>IN:</b></td><td><a href="http://www-cdfonline.fnal.gov/internal/usr/lmiller/cgi-bin/rxdb/DBreport.pl?serial=0186">0186</a></td></tr>
<tr><td><b>OUT:</b></td><td><a href="http://www-cdfonline.fnal.gov/internal/usr/lmiller/cgi-bin/rxdb/DBreport.pl?serial=1123">1123</a></td></tr></table> 
<li>See <a href="http://www-cdfonline.fnal.gov/internal/usr/lmiller/cgi-bin/newproblemdb/DBreport.pl?detsys=SVX&wedge=B5W11&layer=L4">report</a></ul>
<!-- Author: Problem DB -->



<!-- Date: Fri Jan 30 16:29:38 2004 -->
<center><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3462&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3463' ></a><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3464&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3465' ></a></center>
<!-- Author: Rong -->
<!-- Comment: Fri Jan 30 16;31;43 comment by...Rong -->
Current of AVDD of SB1W5L4 vs days.
The mean value seems increasing gradually, and now we see pinky frequently.
What do we do for this? What's the policy to change IMon setting?
The IMon setting is at norminal 1035 with high-low end of 1020-1040, while current is pretty much stable at 1040V level.


<!-- Date: Fri Jan 30 17:16:37 2004 -->
<h3>Logic Analyzer Status</h3><ul> 
<li> Logic Analyzer connected to diagnostic crate on 2nd floor is set to trigger if the FFO sends the Readout Return signal for more than 500 nS.  This FFO has been updated with the latest firmware and hooked up to the SVXII SRC, so if one encounters the dreaded 
Stuck SRC Syndrome, it would be nice to see if both FFOs are stuck or just the one in the Collision Hall 
<li> Logic Analyzer connected to SRCs is set to look for RUN from the TS, then trigger on SRC pulling error, unless it sees HALT from the TS, in which case it will look for RUN again.  That is to say, it should only trigger on Errors while we are running.  The SRC in question is (unfortunately) the one in b0svx06, as the 
other SRC has old firmware which doesn't put Error on the logic analyzer pads.  So, if b0svx06 a.k.a. VRB_ISL_06 pulls CDFERROR, it would be good to have a look at this one. 
</ul> 
  
<!-- Author: Mr. Spock -->


<!-- Date: Sat Jan 31 03:15:40 2004 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3466&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3467  width='300' align='top'></a></td><td>SPYMon plots for store 3210</td></tr></table>
<!-- Author: Rong -->

<!-- Date: Sat Jan 31 14:55:20 2004 -->
Entered in Silicon database: <ul>
<li>Ladder: IB5W5L1 
<li>problem n: 1182 
<li>Status: needs_observation
<li>Type: Chip_bit_error
<li>Action: Diagnosis_update
<li>entered by: rainer 
<li>description is: bit#1 problem miraculously disappeared after run 177938. keep an eye on it. 
<li>See <a href="http://www-cdfonline.fnal.gov/internal/usr/lmiller/cgi-bin/newproblemdb/DBreport.pl?detsys=ISL&wedge=B5W5&layer=L1">report</a></ul>
<!-- Author: Problem DB -->


<!-- Date: Sat Jan 31 15:16:05 2004 -->
Entered in Silicon database: <ul>
<li>Ladder: SB3W8L2 
<li>problem n: 1191 
<li>Status: needs_observation
<li>Type: Optical_other
<li>Action: Diagnosis_update
<li>entered by: rainer 
<li>description is: bit problem seems to have disappeared on its own - keep an eye on it. 
<li>See <a href="http://www-cdfonline.fnal.gov/internal/usr/lmiller/cgi-bin/newproblemdb/DBreport.pl?detsys=SVX&wedge=B3W8&layer=L2">report</a></ul>
<!-- Author: Problem DB -->



<!-- Date: Sat Jan 31 22:01:35 2004 -->
<center><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3468&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3469' ></a><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3470&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3471' ></a></center>
<!-- Author: Rong -->
<!-- Comment: Sat Jan 31 22;03;23 comment by...Rong -->
<pre>Current plots for AVDD and DVDD of e224. Pretty stable during cosmic data taking. Ladder is removed from hardware database before store finished.</pre>


<!-- Date: Sun Feb 1 02:08:52 2004 -->
<b>b0svx02:Messenger:1:13:48 AM->SRC Fatal Error:Sl 5 Resonance Detected.</b> ACE tried HRR and DAQ hung during recovery. 
<a href= 'http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2004&action=view&page=94&frame=2&anchor=01:20:17.382'> see shift e-log </a> <br>
Ask ACE to abort the run then follow David's insturction and Rainer was on the phone. Flip FFO bit 8<->0 and twice config. 
Data taking recoverd.
<!-- Author: Rong -->
<!-- Comment: Sun Feb  1 02;15;37 comment by...rainer -->
link to LA snapshot
<a href=
'/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=174&frame=2&anchor=16:26:20'>
here </a> 



<!-- Comment: Sun Feb  1 02;48;29 comment by...rainer -->
<pre>I checked the log files - you did fine. excellent.</pre>


<!-- Date: Sun Feb 1 03:18:20 2004 -->
<center><a href='/elog/notebooks/silicon/Sun_Feb__1_2004_031758.ps'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3473' ></a><a href='/elog/notebooks/silicon/Sun_Feb__1_2004_031807.ps'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3475' ></a></center>
<!-- Author: Rong -->
<!-- Comment: Sun Feb  1 03;20;43 comment by...Rong -->
<pre>Snapshots of Logic Analysers on 1F (left plot) and 2F (right plot). Unfortunetly, the floopy disk used in scope can't be read.</pre>


<!-- Comment: Sun Feb  1 13;16;45 comment by...rainer -->
as told, you need a windows machine, the format seems to be slightly idiosyncratic. 
<!-- Date: Sun Feb 1 12:50:57 2004 -->
<table><tr><td><a href=/elog/notebooks/silicon/Sun_Feb__1_2004_125048.ps align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3477  align='top'></a></td><td>scope traces of last night.</td></tr></table>
<!-- Author: rainer -->
<!-- Comment: Sun Feb  1 13;18;51 comment by...rainer -->
<b> do we learn anything from these traces ? </b>

<!-- Comment: Sun Feb  1 18;31;23 comment by...Rainer -->
reconstructed  envelope setting of the scope as best as I could. made a setting in a:\tek00000.set on the floppy to recall this setting. <p>
<li> trigger is set to channel#1. I assume this is the FFO readout signal (says  FFO status 3 on its connector). I hope this is correct.
<li> channel #2 is connected to  SRC TM. this is I hope as before.
<li> channel #3 is disconnected. I assume this was a second FFO we had hooked up earlier to look for coincidence between FFOs. I would assume this one is disconnected because a special connector was missing and was taken back by SCN. <p>

anyway we are probably good to go for catching another resonance error in this setting. for pager carriers, there is a printout of what to look for when the FFO goes bananas right next to the scope.
<p>
It would be good if someone could explain the pin outs and double check the setting.




<!-- Comment: Mon Feb  2 08;26;47 comment by...SCN -->
For the record, after the Friday access I plugged in the FFO in
b0fib02 into the SRCTM-M, thinking we want the one we modified to
be the one we use.  This FFO used to have my AMP-C connector on
it and was playing the role of the "2nd FFO".  As you cannot
be both 1st and 2nd FFO simultaneously, and I saw no point in
monitoring a different FFO with different firmware, I simply
removed (and reclaimed) my connector and plugged the FFO into
the SRCTM-M.
<!-- Date: Mon Feb 2 06:20:55 2004 -->
<center><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3482&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3483' ></a><a href='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3484&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-3485' ></a></center>
<!-- Author: Rong -->
<!-- Comment: Mon Feb  2 06;24;22 comment by...Rong -->
Tried to include e224 into data taking but only have 10 mins.
It tripped when power was turned on.
The DVDD current is high which is different compare to the case we run standard alone DAQ.<a href= 'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=182&frame=2&anchor=00:01:43'> see here</a> 



<!-- Date: Mon Feb 2 10:47:58 2004 -->
<h3>FFO consternation</h3> 
First, I installed in b0fib02 the ability to "reset the Readout Return Logic" by toggling bit 30 (high then low) of register 0x4C.  I believe this is set up to be done on any RECOVER, END, or ABORT transition.  It was tested to the extent that it does no harm, but of course we cannot test its effectiveness until we reproduce the problem, see the FFO holding the line high (during the HALT state)  and then issuing RECOVER. <P> 
In addition, the logic analyzer setup du jour... 
<ul> 
<li> 2nd floor LA is set up to trigger on the FFO in that crate 
setting the Readout Return signal (GBIN 0) high for &gt; 500 nS (as before). 
<li> 1st floor LA is set up to trigger on the GB pulling the BRAKE signal, i.e. when the Estring fires. 
<li> I cannot find any reference to how the expert wants the oscilloscope set up.  Inputs 1 and 2 are attached to some 
signals on the mezzanine card, it seems to be triggering on channel 1 edge with a timebase of 500 nS, and is in envelope mode.  I left it that way. 
</ul> 
IMHO<P> 
Should a Resonance Detected error occur, I would recommend  
going to the HALT state and trying to get some people to look 
at it.  If noone is found, or once they are done, attempt RECOVER and RUN, see if the reset code works.  If so, great, if not, 
follow normal recovery procedure. <P> 
This plan should be approved by Rainer before implementing it.
<!-- Author: SCN, Stefano, Marcel -->
<!-- Comment: Mon Feb  2 12;50;58 comment by...Rainer -->
I endorse the plan with wide open arms (scary, ey ?). my understanding is that the oscilloscope and LA traces will give the information needed to determine whether we still see the same error and give enough information for stefano to give him a lead. The oscilloscope settings as of now reflect my limited knowledge - experts welcome to humble me and double check that.