<!-- Title: Eve shift. Fri Sep 21, 2007 -->
<!-- SciCo: Phil Schlabach -->
<!-- DAQAce: Francesco Crescioli -->
<!-- MonAce: &nbsp; -->
<!-- CO: &nbsp; -->
<!-- OpManager: Phil Schlabach -->
<!-- Notes: -->
<pre>a couple of sillis here, muon scint. done for the day</pre>


<!-- Date: Fri Sep 21 16:06:35.995 2007 -->
<pre>HV status 
--------------- 
 
ON: CEM CHA WHA GAM 
STBY: PEM PHA PSH 
OFF: COT TOF L00 CMU CMP CMX BMU CES CPR CSW MSK CSX CSP BSU TSU CLC MNP BSC 
RED: SVX ISL (but off, silicon ppl here so no problem) 
 
Global alarms: 
All green except PSM. Known problem. 
GAM, ICI and INH light green.</pre>
<!-- Author: Francesco -->
<!-- Comment: Fri Sep 21 20;04;35 comment by...Francesco -->
Disregard information about CLC MNP BSC being OFF. Now they are ON and I saw that at least BSC was ON on the previous shift so maybe it's me looking at the bars instead that the "stat val" value.
<!-- Date: Fri Sep 21 16:07:59.603 2007 -->
I have given the lift key to Ignacio.
<!-- Author: Phil -->

<!-- Date: Fri Sep 21 16:38:58.140 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249932>249932</A>
 Terminated at 2007.09.21 16:38:52
<!-- Author: RunControl -->


<!-- Date: Fri Sep 21 16:39:23.065 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249932>249932</A>
TERMINATE: done; to rerun TOF_TAC_CALIB which didn't give results in the previous shift
<!-- Author: Francesco Crescioli -->



<!-- Date: Fri Sep 21 16:48:03.027 2007 -->
TOF_TAC_CALIB is stuck at partition time waiting for sevb to reply. I've already seen this in my previous bunch of shifts and I'll try the "restart softevb proxy" solution. Why this is happening?
<!-- Author: Francesco -->
<!-- Comment: Fri Sep 21 16;50;31 comment by...Francesco -->
Like the previous times stopping and restarting softevb proxy solved the problem. Note: the Partition_0 box did never went red, neither when I stopped softevb proxy.
<!-- Date: Fri Sep 21 16:50:30.720 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249945>249945</A>
ACTIVATE: RunSet: TOF_TAC_CALIB 
<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 16:50:31.693 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249930>249930</A>
CALIBRATION: <U><B>toftac</B></U> calibration FAILED, please investigate
<!-- Author: CDFCALIB -->




<!-- Date: Fri Sep 21 16:52:25.391 2007 -->
Got a brief "node unreachable" alarm for CSW MSK CSX CSP BSU TSU. Now it's back to green but heartbeat still missing.
<!-- Author: Francesco :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249945' target=_top>249945</a>) -->
<!-- Comment: Fri Sep 21 17;00;04 comment by...Francesco -->
Went to check the PC downstairs which is running fine. We'll send an email to the expert.
<!-- Comment: Fri Sep 21 17;03;05 comment by...Francesco -->
Everything is back to normal. Someone is working on the machine downstair.
<!-- Date: Fri Sep 21 16:53:24.490 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249945>249945</A>
TERMINATE: done; went to idle state smoothly this time
<!-- Author: Francesco Crescioli -->




<!-- Date: Fri Sep 21 17:06:43.852 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249948>249948</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [19,952,639] changed trigger table according to wb (L3 was 633 in the default)
<!-- Author: Francesco Crescioli -->



<!-- Date: Fri Sep 21 17:08:49.347 2007 -->
Got this L2 error at the very beginning of L2: 
<pre> 
b0clifford.fnal.gov:TEDCpp:Sep 21, 2007 05:07:00 PM->{ 17:7:0, 9/21/2007 }: L2-Pulsar-Linux: Control node is unable to contact decision node process in node  #0 at address 192.168.0.10; Decision node may be in trouble.  -->   
 Additional Information:  
 
 Attention !!!!.  L2PULSAR_NODELOST !!!! 
</pre> 
An HRR seems to have done the job, as noted in previous shifts. Everything is normal now.
<!-- Author: Francesco :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249948' target=_top>249948</a>) -->


<!-- Date: Fri Sep 21 17:13:55.399 2007 -->
<center><a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-6504&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-6505' ></a></center>
<!-- Author: Francesco -->
<!-- Comment: Fri Sep 21 17;16;42 comment by...Francesco -->
The L2 panel look a bit weird. In the upper part there are the current triggers working (we are in L2_TORTURE mode now) but below there are some leftover from the triggers in the previous cosmics run (that 32 Hz is the smoking gun, I remember also the position from before).


<!-- Comment: Fri Sep 21 17;56;44 comment by...Francesco -->
Under Donatella suggestion I restarted the scaler monitor and now everything is fine. Maybe when we change run type something is not clearing the columns correctly, but it's just some aesthetic problem.
<!-- Date: Fri Sep 21 17:30:16.850 2007 -->
I got an RTSERVER and IMON error, both related to b0dasvx02. I've asked Ignacio who said that they are working on it and maybe everything will be back to normal for the end of the day, so I'll just ignore this error and go on. 
I'll try to trace down the RTSERVER error which is maybe fixable.
<!-- Author: Francesco :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249948' target=_top>249948</a>) -->
<!-- Comment: Fri Sep 21 17;36;06 comment by...Francesco -->
Logged into b0dasvx02 following instruction from the webpage for diagnose of rtserver problems but the last column, which should show the offending pid is "-". No luck here. I'll just wait and see if it will come back green by itself when IMON problems on that machine are solved.

<!-- Comment: Fri Sep 21 17;42;43 comment by...Francesco -->
RTSERVER come back to green, by herself. IMON still red, but this is expected (b0dasvx02 has crashed IMON).


<!-- Comment: Fri Sep 21 18;47;43 comment by...Francesco -->
IMON is ok now, seems that problem on b0dasvx02 has been fixed.
<!-- Date: Fri Sep 21 18:27:57.973 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249948>249948</A>
TERMINATE: done; now I'll start some calibrations
<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 18:29:14.074 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249961>249961</A>
TERMINATE: done
<!-- Author: Francesco Crescioli -->
<!-- Comment: Fri Sep 21 18;32;03 comment by...Francesco -->
somehow went directly from partition to idle to abort, but maybe it's my fault (clicked something by mistake), because second try seems to run smoothly
<!-- Date: Fri Sep 21 18:31:17.064 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249962>249962</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Francesco Crescioli -->



<!-- Date: Fri Sep 21 18:32:26.822 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249961>249961</A>
CALIBRATION: <U><B>calqie</B></U> calibration FAILED, please investigate
<!-- Author: CDFCALIB -->


<!-- Date: Fri Sep 21 18:32:27.340 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249962>249962</A>
TERMINATE: done
<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 18:35:00.365 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249963>249963</A>
ACTIVATE: RunSet: CALIB_CLC_QIE 
<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 18:37:10.028 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249963>249963</A>
TERMINATE: done
<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 18:37:11.925 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249963>249963</A>
CALIBRATION: <U><B>clcqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#clcqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Sep 21 18;44;21 comment by...Francesco -->
good
<!-- Date: Fri Sep 21 18:39:37.244 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249964>249964</A>
ACTIVATE: RunSet: CALIB_QIE_FWD 
<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 18:39:38.788 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249964>249964</A>
CALIBRATION: <U><B>bscqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#bscqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Sep 21 18;46;38 comment by...Francesco -->
good
<!-- Date: Fri Sep 21 18:39:52.849 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249964>249964</A>
TERMINATE: done

<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 18:44:06.979 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249965>249965</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL 
<!-- Author: Francesco Crescioli -->





<!-- Date: Fri Sep 21 18:49:49.972 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249965>249965</A>
TERMINATE: done
<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 18:52:17.349 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249966>249966</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 18:54:29.766 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249966>249966</A>
TERMINATE: done
<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 18:54:30.655 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249966>249966</A>
CALIBRATION: <U><B>pluglaser</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#pluglaser target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Sep 21 19;01;04 comment by...Francesco -->
good, second timing at the edge of the limit (580.6ns). Emailed to experts.
<!-- Date: Fri Sep 21 18:56:05.170 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249967>249967</A>
ACTIVATE: RunSet: BOTHPLUGS_MM 
<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 19:00:17.024 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249967>249967</A>
TERMINATE: done
<!-- Author: Francesco Crescioli -->




<!-- Date: Fri Sep 21 19:06:53.401 2007 -->
During two of the three pluglaser calibrations (BOTHPLUGS_NORMAL and BOTHPLUGS_MM) I got errors from CEQ_DAQERRMON_WRONG_MODE_ERROR and when I checked concon TrigMon was red. Related? Anyhow I restarted (these two times) and everything seems fine. 
<p> 
The error from the console the last time was: 
<pre> 
 XFTTrigger : beginRun()  
 file name = 249967XTRPSim_Summary.txt 
 XTRPSimModule::getFromD applying conditions from run 249967 
terminate called after throwing an instance of 'std::ou 
t_of_range'  what():  basic_string::substr 
%ERLOG-s Abnormal termination: caused by signal:  6 Framework abortJob()  
         21-Sep-2007 18:57:04 CDT run = 249967 event = 0 
</pre>
<!-- Author: Francesco -->
<!-- Comment: Fri Sep 21 20;07;15 comment by...Phil -->
XTRPSim doesn't like plug calibrations?
<!-- Date: Fri Sep 21 19:09:30.082 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249968>249968</A>
 Activated at 2007.09.21 19:09:14
<!-- Author: RunControl -->


<!-- Date: Fri Sep 21 19:09:31.342 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249968>249968</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [7,957,639] shutdown condition
<!-- Author: Francesco Crescioli -->



<!-- Date: Fri Sep 21 19:13:19.887 2007 -->
L2 Pulsar in daqmon went red. Seems that b012pu04 is not responsive. I'm investigating. No alarm sounded.
<!-- Author: Francesco :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249968' target=_top>249968</a>) -->
<!-- Comment: Fri Sep 21 19;21;41 comment by...Phil -->
Called L2 main pager, Laura shows up in 30s.
<!-- Comment: Fri Sep 21 19;31;08 comment by...Francesco -->
Laura come and fixed the problem. Now we are running cosmics without errors.
<!-- Date: Fri Sep 21 19:17:30.298 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249968>249968</A>
 Terminated at 2007.09.21 19:17:17
<!-- Author: RunControl -->


<!-- Date: Fri Sep 21 19:20:29.757 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249968>249968</A>
TERMINATE: done
<!-- Author: Francesco Crescioli -->
<!-- Comment: Fri Sep 21 19;21;39 comment by...Francesco -->
L1A rate was 49Hz


<!-- Date: Fri Sep 21 19:25:54.562 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249973>249973</A>
TERMINATE: done; some tests on b012pu04
<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 19:30:28.396 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249975>249975</A>
 Activated at 2007.09.21 19:28:48
<!-- Author: RunControl -->


<!-- Date: Fri Sep 21 19:30:29.267 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249975>249975</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,954,639] to test the b012pu04 problem
<!-- Author: Francesco Crescioli -->



<!-- Date: Fri Sep 21 19:33:17.554 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249975>249975</A>
 Terminated at 2007.09.21 19:32:59
<!-- Author: RunControl -->



<!-- Date: Fri Sep 21 19:33:20.240 2007 -->
Silicon finished in the collision hall for the day. Ignacio returns the lift key.
<!-- Author: Phil -->

<!-- Date: Fri Sep 21 19:35:28.642 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249975>249975</A>
TERMINATE: done; b012pu04 is responding and running fine now, so I'll continue with calibrations
<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 19:37:17.860 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249977>249977</A>
CALIBRATION: <U><B>tofqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#tofqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Sep 21 20;07;44 comment by...Francesco -->
bad; known problem with wcal04
<!-- Date: Fri Sep 21 19:39:58.578 2007 -->
<b>PS replacement work:</b> 
Today we finished installing and testing crates 9 and 10 (SVX). Tomorrow we will move to the SE side to work on the remaining east SVX crates.
<!-- Author: The Silis -->

<!-- Date: Fri Sep 21 19:40:42.894 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249977>249977</A>
ACTIVATE: RunSet: ACE_TOF_QIE_NEW 
<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 19:40:43.299 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249977>249977</A>
TERMINATE: doe
<!-- Author: Francesco Crescioli -->



<!-- Date: Fri Sep 21 19:46:37.310 2007 -->
ifix on vnode1 signaled an heartbeat problem with ICICLE and in fact on SOLENOID2 iFix was giving a bounch of "Internal Error". We rebooted the machine and restarted ICICLE, no more errors but still hearbeat in purple. Maybe we just have to wait.
<!-- Author: Francesco :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249977' target=_top>249977</a>) -->
<!-- Comment: Fri Sep 21 19;52;24 comment by...Francesco -->
It went green. All ok.
<!-- Comment: Fri Sep 21 19;54;15 comment by...Phil -->
The heartbeat for icicle just went green.
<!-- Comment: Fri Sep 21 20;06;36 comment by...Phil -->
It looked to me like iFix had crashed.
<!-- Date: Fri Sep 21 19:51:42.426 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249980>249980</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 19:52:12.617 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249980>249980</A>
TERMINATE: done
<!-- Author: Francesco Crescioli -->



<!-- Date: Fri Sep 21 19:54:15.729 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249980>249980</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Sep 21 20;10;29 comment by...Francesco -->
good
<!-- Date: Fri Sep 21 19:55:22.312 2007 -->
Conor wants to do some quick tests with L3.
<!-- Author: Phil -->
<!-- Comment: Fri Sep 21 20;46;48 comment by...Conor -->
quick tests done. I have fixed the run-time errors with L3 startup that we were seeing before on SL4 nodes. However, SL4 converters are having trouble communicating with eventBuilder. Still work in progress.
<!-- Date: Fri Sep 21 20:13:21.476 2007 -->
Where are the results from TOC_TAC_CALIB run 249945 and ACE_QIE (calqie) run 249962 ? Both runs ended without problems but results are not in the elog.
<!-- Author: Francesco -->
<!-- Comment: Fri Sep 21 20;14;11 comment by...Francesco -->
It looks like the results are in the webpage, so it's just a problem in the calibration results done to elog comunication.


<!-- Date: Fri Sep 21 20:21:30.704 2007 -->
comments on calqie run 249962 
<p> 
WHA 10W ch4 is slightly over 5 for RMS (5.25) 
<p> 
CEM 5W ch19 are slightly over 5 for RMS (5.62 - 5.36) 
<p> 
good.
<!-- Author: Francesco :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249962' target=_top>249962</a>) -->


<!-- Date: Fri Sep 21 20:23:55.002 2007 -->
TOF_TAC_CALIB 
<p> 
White in wcal04 and some channels of wcal02 
<p> 
bad
<!-- Author: Francesco :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249945' target=_top>249945</a>) -->

<!-- Date: Fri Sep 21 20:44:12.955 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249981>249981</A>
 Activated at 2007.09.21 20:44:05
<!-- Author: RunControl -->


<!-- Date: Fri Sep 21 20:44:39.100 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249981>249981</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,954,639] test run, to check if everything is back to normal after l3 tests
<!-- Author: Francesco Crescioli -->



<!-- Date: Fri Sep 21 20:48:52.358 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249981>249981</A>
 Terminated at 2007.09.21 20:47:50
<!-- Author: RunControl -->


<!-- Date: Fri Sep 21 20:48:53.393 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249981>249981</A>
TERMINATE: done; everything was normal
<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 20:51:22.513 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249982>249982</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 20:53:33.241 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249982>249982</A>
TERMINATE: done
<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 20:53:34.068 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249982>249982</A>
CALIBRATION: <U><B>cemled</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cemled target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Sep 21 21;02;59 comment by...Francesco -->
bad; wedge 11 10 9 west all white, west 2R 6-5 white.
<!-- Date: Fri Sep 21 20:53:35.076 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249983>249983</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 20:53:36.837 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249983>249983</A>
TERMINATE: done
<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 21:00:27.111 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249983>249983</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Sep 21 21;03;49 comment by...Francesco -->
ok
<!-- Date: Fri Sep 21 21:00:29.654 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249984>249984</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 21:00:30.998 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249984>249984</A>
TERMINATE: done
<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 21:00:31.242 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249984>249984</A>
CALIBRATION: <U><B>calxef</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calxef target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Sep 21 21;05;00 comment by...Francesco -->
bad; 11 10 9 west all white
<!-- Date: Fri Sep 21 21:02:59.277 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249985>249985</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Francesco Crescioli -->





<!-- Date: Fri Sep 21 21:05:24.690 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249985>249985</A>
TERMINATE: done
<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 21:11:34.611 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249986>249986</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [19,952,633] 
<!-- Author: Francesco Crescioli -->
<!-- Comment: Fri Sep 21 21;13;47 comment by...Francesco -->
It did a couple of Done Timeout, at first in b0svt04 then after an HRR into a lot of crates (IMU_00, IMU_01, CMU_01, CMU_00, MUON_SCINT_00, CMX_00, CMX_01, CLC_00, HADRON_TIMING_00, EM_TIMING_00), the second HRR fixed it all and now is running without problems.


<!-- Date: Fri Sep 21 21:45:48.987 2007 -->
<pre>HV status 
--------------- 
 
ON: CEM CHA WHA and CPR CLC MNP BSC with bars off (known problem? at least for BSC it is) 
I noticed that Joachim found CPR on some shifts ago and turned it off. It has been on with bars off all day probably so it isn't anything critical and I will not make an attempt to mess up things at the end of the eve shift. Checked on CLC1 and it's everything ok except an log overflow error from the morning. 
NOT: GAM with bars up 
STBY: PEM PHA PSH 
OFF: COT TOF L00 SVX ISL CMU CMP CMX BMU CES CSW MSK CSX CSP BSU TSU  
 
Global alarms: 
All green except PSM. Known problem. 
GAM, ICI and INH light green.</pre>
<!-- Author: Francesco -->

<!-- Date: Fri Sep 21 21:47:54.254 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249986>249986</A>
TERMINATE: done; switch to cosmics for the night
<!-- Author: Francesco Crescioli -->


<!-- Date: Fri Sep 21 21:52:46.883 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249987>249987</A>
 Activated at 2007.09.21 21:51:20
<!-- Author: RunControl -->


<!-- Date: Fri Sep 21 21:52:47.487 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249987>249987</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,954,639] 
<!-- Author: Francesco Crescioli -->



<!-- Date: Fri Sep 21 22:00:10.420 2007 -->
<pre>Shift summary 
full set of calibrations + L2 torture + cosmics 
 
problems 
b0l2pul04 had to be fixed by Laura 
solenoid2 rebooted 
 
work done today 
muon scintillator:  done in east toroid 
SVX:  2 crates of PS installed and tested 
L3:  fixed some problems running under SL4 
 
to be done tomorrow 
muon scintillator:  top? 
SVX:  PS work on SE</pre>
<!-- Author: Phil Schlabach :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249987' target=_top>249987</a>) -->

<!-- Date: Fri Sep 21 23:56:12.295 2007 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>PhysicsTable</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L3Accepts</TH>
<TH>DelivLumi nb<SUP>-1</SUP></TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>23:55:01</TD>
<TD ALIGN=CENTER>::</TD>
<TD ALIGN=RIGHT></TD>               <TD ALIGN=RIGHT></TD>
<TD ALIGN=RIGHT></TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->


<!-- Date: Sat Sep 22 07:55:58.581 2007 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>PhysicsTable</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L3Accepts</TH>
<TH>DelivLumi nb<SUP>-1</SUP></TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>07:55:01</TD>
<TD ALIGN=CENTER>::</TD>
<TD ALIGN=RIGHT></TD>               <TD ALIGN=RIGHT></TD>
<TD ALIGN=RIGHT></TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Sat Sep 22 08:01:20.913 2007 -->
<font color=red>Shift Summary: </font><pre>Ending evening shift so I can start the day shift 
 
</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   0
Delivered Luminosity   0  
Acquired Luminosity    0  
Efficiency             0
   
</td></tr></table></pre>
<!-- Author: Jean slaughter -->