<!-- Title: 09:57:22  Mon Apr 12 1999 -->
<!-- Crewchief: Steve Hahn -->
<!-- Op1: &nbsp; -->
<!-- Op2: &nbsp; -->
<!-- Op3: &nbsp; -->
<!-- Op4: &nbsp; -->
<!-- Op5: &nbsp; -->
<!-- Op6: &nbsp; -->
<!-- Notes: -->
<center><H1>Wedge Test Activities</H1></center> 
<p> 
Welcome to the Wedge Test electronic logbook (e-log for short)!


<!-- Date: Mon Apr 12 10:01:51 1999 -->
To add an entry, click on the "Add Entry" button at the top of your screen (in IE, you may have to click somewhere else on the menu bar frame at the top first).  You will be required to enter your name and some text.  Please preview your entry by clicking on the "HTML" or "Plain" radio button below the text box; you can then correct any mistakes before actually submitting the entry.  You <b>can</b> add HTML tags, but please use sparingly or not at all.  
<!-- Author: Steve Hahn -->
<!-- Comment: Mon May 17 10;10;03 comment by...jpp -->
test comment
<!-- Date: Mon Apr 12 10:10:07 1999 -->
OK, some real information...Dean Beckner and Bruce Vollmer have been working on installing the gasworks on the NW arch for central muon and shower max systems.  They should finish with the NW arch today or tomorrow.  When they are done, the NW arch will be pushed in and the NE arch pulled out in preparation for the same work on the NE arch.  Don Mizicko estimates they should be done by Friday, but this work may go into next week.<p> 
The consequences are that HV on the NW arch may be off during the day (7:00 AM - 3:30 PM approximately; this has been the case last Friday and over the weekend), and the NW arch will not be isolated from the floor when it is pushed in.  We will study if this results in noisier conditions.   
<!-- Author: Steve Hahn -->


<!-- Date: Mon Apr 12 10:13:47 1999 -->
One more reminder: the wedgetest schedule page is <a href=http://hahn.fnal.gov/schedule.html>here</a>.  No one has given me any requests for either of the TDC crates, so if you do have such requests please let me know.
<!-- Author: Steve Hahn -->


<!-- Date: Tue Apr 13 17:37:25 1999 -->
Frank Chlebana asked for more info on when Dean Beckner has been turning the HV on and off.  Deans says he usually turns it off when he starts at 7:00 AM in the morning, and back on when he quits sometime between 5:15-5:30 PM.<p> 
 
Also, Frank requested that the TESTCLK clock signals be restored as the clock into the TRACER (as opposed to the CLOCK signal from upstairs).  Done.  
<!-- Author: Steve Hahn -->


<!-- Date: Tue Apr 13 17:42:53 1999 -->
One more note: the guide rails for the NW arch were installed today this afternoon.  This means the NW arch is no longer isolated from the transporter it rest on.  Tomorrow or Thursday when Dean is done installing gasworks on the NW arch, the NW arch will be pushed in and the NE arch pulled out.  
<!-- Author: Steve Hahn -->


<!-- Date: Tue Apr 13 18:24:59 1999 -->
I attempted to run with the ADMEMs using the 2301 (WEGTST01).  I found that all the values were way off in all four cards, so I decided to recalibrate them all.  The recalibration and downloading proceeded OK, but when I tried to take pedestals again, the values were still all over (not near 50).<p> 
 
I then went to the the MVME162 (B0TS02).  This seemed to work correctly for slot 16 and 17.  Slot 16, QIE 6 appears to be dead.  This is new since last week.  
<!-- Author: Steve -->


<!-- Date: Wed Apr 14 19:49:37 1999 -->
<ul> 
<li> 
Was having trouble running QIETEST on 2301 yesterday.&nbsp; Did calibrations 
and took pedestals on all four ADMEMs this morning using 162.&nbsp; Went 
to Hao Wei to demonstrate problems with 2301, and he had no problems.</li> 
 
<li> 
Found channel 6 (of 0-19) was dead in both QIE calibration and pedestals.&nbsp; 
Went downstairs to replace it, and found all fans had been unplugged for 
the VME crate and power supplies.&nbsp; This probably happened because 
someone had to get access underneath the arch stand, and these power cables 
get in the way of the limited access between the rack and the 90 degree 
side.&nbsp; <b><font color="#FF0000">This is very bad!</font></b>&nbsp; 
This means crate was without cooling for an undetermined amount of time.&nbsp; 
When I removed the ADMEMs, they were very hot.&nbsp; No doubt this contributed 
to the dead Caf&eacute; card.</li> 
 
<li> 
Decided to replace channel 12 also.&nbsp; This is the channel which has 
a current channel pedestal much more negative than the other channels.&nbsp; 
Whereas the other channels become positive with the DC calibrator circuit 
set to 100, this channel is not positive till a setting of 400 or more.&nbsp; 
Lou DelMonte gave me two replacement channels for these two Caf&eacute; 
cards.</li> 
 
<li> 
Hao Wei and I went downstairs to reinsert ADMEM016 into slot 16, and remove 
the MVME162 from the crate to give to Lee Holloway, per his request to 
Aesook.&nbsp; After replacing the card, the problems I had seen yesterday 
returned in force.&nbsp; We successfully calibrated and downloaded the 
FRAMs on all four ADMEMs, but when we tried to read pedestals using the 
lookup tables, QIETEST returned nonsense.&nbsp; It looks like some problem 
either with memory allocation, or mismatched versions of software (ROBIN, 
FISION, VxWorks?).&nbsp; We also tried to run calibGUI, but it appears 
the setups Hao Wei and I know are now out of date.&nbsp; Will consult with 
Frank tomorrow.</li> 
</ul>
<!-- Author: Steve Hahn -->


<!-- Date: Fri Apr 16 14:56:56 1999 -->
Problems of Wednesday continued Thursday and Friday: 
<ul> 
<li> 
Thursday morning Frank showed me some very peculiar pedestals taken Wednesday 
night late over a 20 minute interval: histograms of each channel had a 
similar structure and the RMSs were 10-15 counts.&nbsp; Note this is not 
the same as the 1000s of counts we had seen with QIEtest Tuesday.</li> 
 
<li> 
Thursday morning they started moving out the NE arch.&nbsp; It is out now 
with a slight offset (not as far as) the NW arch.&nbsp; I went in the pit 
about 11:00 AM as they were finishing up to reconnect the external trigger 
LEMO cable and Category 5 CLOCK cable to the NIM bin in the NE arch lower 
rack.&nbsp; Ilya Kravchenko was still logged onto WEGTST01.&nbsp; I noticed 
that the CLOCK cable was plugged into the TRACER, not the TESTCLK cable 
which I had inserted the night before.&nbsp; I also noticed that the yellow 
WAIT light was lit on the TRACER.&nbsp; On a lark, I hit the TESTCLK RESET 
button.&nbsp; The WAIT light cleared and, to my surprise, the crate started 
taking events.&nbsp; My surprise doubled when I found that the NIM bin 
was not on, and so there were no CLOCK signals into the TRACER.&nbsp; <b>Why 
is data-taking possible without clock signals?</b></li> 
 
<li> 
After reconnecting cables and repowering the NE arch rack, I tried running 
again with ADMEM016 in slot 16 with new Caf&eacute; cards in channels 6 
and 12 (of 0-19).&nbsp; These Caf&eacute; cards behaved exactly as those 
I had removed and replaced: channel 6 read out 0 for pedestal instead of 
about 50, and channel 12 had a large negative offset reading out the current 
channel compared to the the other channels.</li> 
 
<li> 
I removed ADMEM016 from slot 16 in anticipation of taking it to Lee Scott 
Friday morning (sorry, Cristoph!).&nbsp; However, Wei Hao convinced me 
to check the PMT output signals first.&nbsp; This morning he and I went 
down and looked at signals.&nbsp; Both CEM channels 6 (21W T3R) and 12 
(21W T6R) had normal voltages (0 mV) and currents (100-200 pA with higher 
excursions due to cosmics), and we could see cosmic rays on the Illinois 
digital scope (<font color="#006600">with some aliasing problems; gee, 
it sure would be nice to have an Infinium</font>) although they seemed 
small in T3R.&nbsp; After plugging everything back in and recalibrating 
and redownloading, we found that channel 6 had returned to life.&nbsp; 
The fact that it had a large negative offset (so the pedestal was below 
zero) but the calibration worked implies a large pedestal shift.&nbsp; 
However, why such would occur and why it went away while we were plugging 
and unplugging cables is unclear.&nbsp; Channel 12, on the other hand, 
has some real problem.&nbsp; When the PMT is plugged in, the current offset 
is present, and when it's unplugged, the offset is not present.</li> 
</ul> 
 
</body> 
</html>
<!-- Author: Steve Hahn -->
<!-- Comment: Fri Apr 16 19;34;15 comment by...Steve Hahn -->
I forgot to add that Frank ran on Thursday after I reconnected cables and the pedestal histograms looked fine.


<!-- Date: Fri Apr 16 19:45:21 1999 -->
<ul> 
<li> 
Today, the TSI return cable (Category 5 from the wedgetest TRACER up to 
the TSI crate in the trigger room) and the DEBUG monitor cable (from the 
wedgetest 2301 to the VT200 in the trigger room) were run.&nbsp; This means 
we now have all necessary clock, TS, and VRB cables into the TRACER for 
full DAQ readout!</li> 
 
<li> 
Unfortunately, the DEBUG cable (to reboot the 2301 remotely from the VT200 
monitor in the trigger room) hangs the 2301 when it's plugged in.&nbsp; 
This does not make sense!&nbsp; So, currently, this is not plugged in.</li> 
 
<li> 
Andrew Martin, Nathan Eddy, Greg Feild, and I worked on making the TS crate 
and the wedgetest crate read out simultaneously.&nbsp; I connected the 
TS fiber from the rear of the VRB crate on the 3rd floor down to the trigger 
XPT in the TS crate.&nbsp; The TSI return cable is in one of the TSI return 
transition cards eight inputs, though we did not use this.&nbsp; When Andrew 
read out the TS crate and wedgetest crate simultaneously after sending 
start scans we saw the difference between the bunch ID in a given L2 buffer 
and that in the TS stay fixed as the bunch IDs rotated around the bunchs 
(0-159 or 0x0-0x9F).&nbsp; It appears to work!&nbsp; Next: using Frank's 
readout code modified to not use the TESTCLK in conjunction with this.</li> 
</ul>  
<!-- Author: Steve Hahn -->


<!-- Date: Fri Apr 16 19:46:52 1999 -->
I have restored the TRACER in the wedge test crate to using the TSI input cable from the TESTCLK instead of the TS crate.  The clock input to the TRACER is from the CLOCK in the trigger room.  This in preparation for overnight VRB tests.
<!-- Author: Steve Hahn -->


<!-- Date: Mon May 10 09:59:45 1999 -->
On Friday May 7th 1 SMD card with 24 channels of real  
SMQIE was installed,  along with the SMC controller. 
The SMXR was moved and plugged into slot 3,  where the 
transition card was plugged into the back.  Readout 
was attempted using simple ANL VISION/C code that 
had worked on the ANL crate previously.  In this case,  
however,  the SMC DTR light came on but no L1A light,  
while the SMD L1a light came on but no DTR.   
Probably need a scope...
<!-- Author: Steve Kuhlmann -->



<!-- Date: Mon Jun 14 12:46:59 1999 -->
<!doctype html public "-//w3c//dtd html 4.0 transitional//en"> 
<html> 
<head> 
   <meta http-equiv="Content-Type" content="text/html; charset=iso-8859-1"> 
   <meta name="Author" content="Steve Hahn"> 
   <meta name="GENERATOR" content="Mozilla/4.51 (Macintosh; I; PPC) [Netscape]"> 
   <title>Untitled</title> 
</head> 
<body> 
 
<ul> 
<li> 
06/07/99 - Ed McKigney left me the 2nd prototype Penn CALIB card; this 
does LED and xenon calibrations only in the B inputs on the transition 
card.</li> 
 
<li> 
06/08/99 - Lee Scott has given me one more production (V4) ADMEM and Caf&eacute; 
cards for both this ADMEM and the one which replaced with the first production 
ADMEM in slot 16.&nbsp; So, I have completely reassigned cards and slots 
as follows:</li> 
 
<table BORDER > 
<caption>Before rearrangement</caption> 
 
<tr> 
<td>Slot</td> 
 
<td>ADMEM card</td> 
 
<td>Caf&eacute; cards</td> 
 
<td>PROMs</td> 
</tr> 
 
<tr> 
<td>16</td> 
 
<td>ADMEM020 (V4)</td> 
 
<td>V4</td> 
 
<td>460A</td> 
</tr> 
 
<tr> 
<td>17</td> 
 
<td>ADMEM019 (V2)</td> 
 
<td>V4</td> 
 
<td>470</td> 
</tr> 
 
<tr> 
<td>18</td> 
 
<td>ADMEM016 (V2)</td> 
 
<td>V4</td> 
 
<td>470</td> 
</tr> 
 
<tr> 
<td>19</td> 
 
<td>ADMEM018 (V2)</td> 
 
<td>V2</td> 
 
<td>470</td> 
</tr> 
 
<tr> 
<td>20</td> 
 
<td>&oacute;</td> 
 
<td>&oacute;</td> 
 
<td>&oacute;</td> 
</tr> 
 
<tr> 
<td>21</td> 
 
<td>&oacute;</td> 
 
<td>&oacute;</td> 
 
<td>&oacute;</td> 
</tr> 
</table> 
 
<table BORDER > 
<caption>After rearrangement</caption> 
 
<tr> 
<td>Slot</td> 
 
<td>ADMEM card</td> 
 
<td>Caf&eacute; cards</td> 
 
<td>PROMs</td> 
</tr> 
 
<tr> 
<td>16</td> 
 
<td>ADMEM020 (V4)</td> 
 
<td>V4</td> 
 
<td>460A</td> 
</tr> 
 
<tr> 
<td>17</td> 
 
<td>ADMEM022 (V4)</td> 
 
<td>V4</td> 
 
<td>470</td> 
</tr> 
 
<tr> 
<td>18</td> 
 
<td>ADMEM016 (V2)</td> 
 
<td>V4</td> 
 
<td>470</td> 
</tr> 
 
<tr> 
<td>19</td> 
 
<td>ADMEM007 (V1?)</td> 
 
<td>V4</td> 
 
<td>470</td> 
</tr> 
 
<tr> 
<td>20</td> 
 
<td>ADMEM019 (V2)</td> 
 
<td>V4</td> 
 
<td>470</td> 
</tr> 
 
<tr> 
<td>21</td> 
 
<td>ADMEM018 (V2)</td> 
 
<td>V2</td> 
 
<td>470</td> 
</tr> 
</table> 
 
<li> 
All cards were successfully calibrated and downloaded after this rearrangement.</li> 
 
<li> 
06/09/99 - Lee would like to change the ID PROMs on all the ADMEMs to reflect 
the PROM versions in the above table.&nbsp; We will have to remove all 
the cards and pop in new ID PROMs at some point.</li> 
</ul> 
 
</body> 
</html>  
<!-- Author: Steve Hahn -->


<!-- Date: Mon Jun 14 12:51:53 1999 -->
<pre>We have received the new prototype central linear power supply from PEI.  We are currently doing some last minute checks on the old (Condor  
linear analog and Astec digital) power supplies for comparison.  The power lugs on the back of the VME crate are quite noisy, similar to what we  
saw on the 14th floormore than 100 mV p-p in noise burst characteristic of the Astec.   I've taken pictures with the Infinium.  Removing all  
external connections to the crate had no effect, nor did removing all cards but the 6 ADMEMs, the MVME, the TRACER, and the TESTCLK.</pre>
<!-- Author: Steve Hahn -->


<!-- Date: Tue Jun 15 10:16:54 1999 -->
Note: the previous entry illustrates why you should never enter an entry as "Plain" as opposed to "HTML".  Note that it does not word wrap to the size of the browser window.  It may be useful for copy-and-pasting some tabbed files, but normally do not use it.
<!-- Author: Steve Hahn -->


<!-- Date: Fri Jul 2 12:16:17 1999 -->
Found that fans were not on in the wedge rack.  The circuit breakers for the oulets had been turned off.  Since there are now outlets in the back of the rack we moved the fan cords and both power supply cords to the rack outlets.  These are all switched by a single breaker. 
 
Currently turned on with Condor Linears and ASTEC switcher (+5V digital).  This has been the configuration since NW arch power was turned on Wednesday June 30. 
 
Yesterday afternoon (Thursday July 1) +5V analaog fuses were blown on the ADMEMs in slots 16 and 18.  These ADMEMs were taken to the high-rise by Bob DeMaat for inspection by  Terri Shaw.  We believe they blew on a power up cycle.  
<!-- Author: Peter Wilson -->


<!-- Date: Fri Jul 2 12:27:49 1999 -->
For reference of data taken over past 2 weeks this is a chronology of when the prototype PEI linear was connected to the wedge crate. 
<ul> 
<li>June 17, 18 Initial tests by Steve Hahn.  End of day June 18 take to Proton pole building.  Reconnected Condor/ASTEC supplies.  
<li> June 23-26 reconnected PEI supply on wedge crate.  Powered off every evening. 
<li> June 26-28 Operate PEI continuously with local smoke detector rack protection. 
<li> June 28 Remove PEI to east clean room while NW arch electrical work is in progress 
<li> June 30 Restore operation with Condor/ASTEC combination. 
</ul> 
<!-- Author: Peter Wilson -->


<!-- Date: Wed Jul 14 11:42:05 1999 -->
<!doctype html public "-//w3c//dtd html 4.0 transitional//en"> 
<html> 
<head> 
   <meta name="Author" content="Steve Hahn"> 
   <meta name="GENERATOR" content="Mozilla/4.51 (Macintosh; I; PPC) [Netscape]"> 
</head> 
<body> 
 
<ul> 
<li> 
After being down for two days straight due to repeated problems with new 
prototype PEI power supply for central, we have gone back to using the 
Astec (digital switcher) and Condor (analog linears) power supplies.</li> 
 
<li> 
Note that the upper backplane (P1-P0-P2) has been replaced due to components 
on the old one fried during testing with the PEI.&nbsp; The lower backplane 
(thru P3) is the same.</li> 
 
<li> 
I have tested using calibrations, TEST pedestals, and TEST_TSIE pedestals 
in QIETEST for all six ADMEMs.&nbsp; No ADMEMs have been moved.&nbsp; Calibrations 
look fine, but pedestals are not near 50 counts; I probably calibrated 
and downloaded too soon after turning the crate back on.&nbsp; TEST pedestals 
are quite noisy (6--8 counts RMS), but TEST_TSIE peds are OK (2-3 counts); 
this with lots of electrical activity in the hall.</li> 
 
<li> 
Currently, the crate is setup with the external clock (from the real clock 
fanout crate!) plugged into the TRACER, the external trigger coming from 
an asynchronous clock at about 50 Hz, and TS input, DATA out, and BUSY 
out connected to the fiber bundle going to the VRB crate on the 3rd floor.</li> 
</ul> 
 
</body> 
</html>
<!-- Author: Steve Hahn -->


<!-- Date: Sun Jul 18 14:39:54 1999 -->
<!doctype html public "-//w3c//dtd html 4.0 transitional//en"> 
<html> 
<head> 
   <meta name="Author" content="Steve Hahn"> 
   <meta name="GENERATOR" content="Mozilla/4.51 (Macintosh; I; PPC) [Netscape]"> 
</head> 
<body> 
I have started a <a href="http://hahn.fnal.gov/Home/Run2_Guide.html">CDF 
Run 2 idiot's guide</a> for online DAQ for my own faulty memory. 
<p>Hopefully, you may find it useful, also. 
</body> 
</html>
<!-- Author: Steve Hahn -->


<!-- Date: Sun Jul 18 14:51:38 1999 -->
<!doctype html public "-//w3c//dtd html 4.0 transitional//en"> 
<html> 
<head> 
   <meta name="Author" content="Steve Hahn"> 
   <meta name="GENERATOR" content="Mozilla/4.51 (Macintosh; I; PPC) [Netscape]"> 
</head> 
<body> 
Friday afternoon and evening, Nathan Eddy and I suffered the frigid trigger 
room in a partially successful attempt to read out both the wedge test 
crate downstairs along with the TDC crate (containing the muon TDC) in 
the trigger room. 
<p>Nathan, in fact, had succeeded in doing this once before.&nbsp; I thought 
he had just read out the TDC crate while generating pulses in the wedge 
test crate on the muon ASD card, but he had actually read out both crates, 
also.&nbsp; We were taking this one step further by attempting to actually 
use the return XPT to get the AND of DONE from both the wedge test crate 
and the TDC crate.&nbsp; He had only used the DONE from the TDC crate directly 
into the front of the TS. 
<p>We had several false starts, mostly due to not completely resetting 
the TSI panel every time and not having a clock into the TDC crate TRACER 
(aside: it's very annoying that the TRACER creates what looks like a clock 
when no clock is plugged in--we were fooled even with the logic analyzer 
hooked up).&nbsp; Finally, we managed to get a few events (the TS status 
panel said 55, while VRC_status said 32/33).&nbsp; Our impression was the 
the return (a.k.a. DONE-ERROR-BUSY) XPT was not really doing the AND correctly 
every time.&nbsp; We'll have to get back to Andrew on this one. 
<p>Note, in the idiot's guide noted in the last entry is the instructions 
for setting up the TSI crate. 
</body> 
</html>  
<!-- Author: Steve Hahn -->


<!-- Date: Sun Jul 18 15:16:05 1999 -->
<!doctype html public "-//w3c//dtd html 4.0 transitional//en"> 
<html> 
<head> 
   <meta name="Author" content="Steve Hahn"> 
   <meta name="GENERATOR" content="Mozilla/4.51 (Macintosh; I; PPC) [Netscape]"> 
</head> 
<body> 
Pedro Amaral is coming Monday and the remainder of next week expecting 
to read out the L1 crate in the trigger room in conjunction with the wedge 
test crate downstairs.&nbsp; To minimize confusion (and since we're still 
having problems reading out two crate under control of the TSI), he intends 
to use the TESTCLK in the L1 crate to control both wedges.&nbsp; Here are 
the steps I think will be necessary after discusssions with the involved 
parties: 
<br>&nbsp; 
<ul> 
<li> 
According to Steve Chappa, two clock signals split from the TESTCLK are 
unneccesary and probably would not work.&nbsp; LVDS clock cables longer 
than 100' are not guaranteed to work due to slewing and signal loss.&nbsp; 
However, the signals from the CDF clock are in sync at the fanout crates; 
that is, the clock coming from the wedge test fanout crate is exactly (within 
1 ns) in sync with that from the fanout crate on the 2nd floor.&nbsp; If 
you use different length cables from the fanout crate to the two TRACERs, 
your signals will differ by that delay.&nbsp; The standard length cable 
for this purpose is a 65' length, which is what all the cables out of the 
fanout in the trigger room are.&nbsp; Right now, we are using a very short 
length in the wedge test, but we have 65' lengths in our cabinet downstairs, 
and I have another one in my office (I'll leave it in front of my computer).</li> 
 
<li> 
If the triggers have to syncronized with the clock then some logic will 
have to be put together to do that.&nbsp; I don't know of any logic upstairs, 
but downstairs we have a NIM bin above the clock fanout rack.&nbsp;&nbsp; 
If you use this, this implies that the TESTCLK in the wedge test&nbsp; 
crate will control the L1 crate instead of the other way around.&nbsp; 
Generating this sync should be easy, since there is a NIM output on the 
clock fanout module copyback signal.&nbsp; Make sure this signal is selected 
to copy back the first input cable; there is a small selector wheel and 
indicators near the top of the fanout module (I have verified the wedge 
test module is set correctly with a scope).</li> 
 
<li> 
The TS signal from the controlling TESTCLK, whether upstairs or down, must 
be put into a fiber optic splitter and then sent to the TRACER in that 
same crate and the TRACER in the other crate.&nbsp; There is an active 
splitter (TAXI fanout) in the NIM bin on the second floor.&nbsp; JJ Schmidt 
also has some passive splitters; though this will result in a small signal 
loss it may be less prone to faults.&nbsp; Ideally, the length of this 
delay to both crates should be the same.&nbsp; In my office are two 200' 
lengths of dual optic fiber with ST connectors; one will just go back to 
the originating crate, but the other must be strung from the splitter to 
upstairs (or downstairs).&nbsp; We could just throw this down as we have 
other test cables; it should be simple enough, but if in doubt, consult 
Dervin Allen.</li> 
 
<li> 
I don't remember if Pedro is qualified to go in the pit; at the very least, 
he may need a buddy knowledgable in the wedge test to change out cards.&nbsp; 
Peter Wilson and I both think it would be a good idea if Philippe Gris 
got involved with this effort next week, so someone here could gain experience 
in using the L1 system.</li> 
</ul> 
 
<p><br>Good luck, Steve! 
</body> 
</html>  
<!-- Author: Steve Hahn -->


<!-- Date: Mon Jul 19 18:59:35 1999 -->
Wanted to do readout tests today, which failed due to unknown reason. First suspicion was a hardware/setup problem with the 
testclock. Philippe went down, we tried a different slot etc. -> no cure. Took the testclock to the trigger room, where it worked 
fine (it is now again in the wedge crate). 
Arnd.
<!-- Author: A.Meyer -->


<!-- Date: Tue Jul 20 15:23:21 1999 -->
Unsuccesfull attempt to put the linear power supply. The Wedge test is back with the old one.
<!-- Author: Ph.Gris -->


<!-- Date: Fri Jul 30 15:32:40 1999 -->
Installed PEI power supply in NW arch with the returns jumpered together.  All voltages ok.  Did not connect -5.2v to v1 and v2 for U of I until correct polarity can be determined and documented.   
 
Tried to connect sense cable from back of supply to VME crate, but, noticed fault lights on most (if not all) supplies.  This may be due to NOT having PEI front panel breaker tunred on all the way!  I did not want to take a chance on causing a problem, so, I connected original sense conector to supply (sense wires connected directly to back of supply).   
 
I tried to adjust the 5 voltages from the front panel, but, only the +5v digital and -15v adjustments worked!  The 15v supply did not need adjusting, but, the -5.2 and the +5v analog was off by ~100mv.  I will look into fixing the sense connection early next week  
<!-- Author: Fred Lewis -->


<!-- Date: Mon Aug 2 08:02:51 1999 -->
PEI supply has been running all weekend with no problem.  Right now I am reloading FRAMs with new QIE constants, since I don't believe this has been done so far.  The old V1 ADMEMs with old Cafe cards in slot 20 and 21 have been removed and moved up to the new CLC crate in the 1st floor counting room.
<!-- Author: Steve Hahn -->


<!-- Date: Thu Aug 12 18:19:04 1999 -->
The PEI all-linear 400 Hz power supply has been returned to PEI as of last Friday (OK, this is a bit retroactive).  Friday afternoon, the Condor linear 60 Hz power supply and Astec switcher digital power supply were re-installed.  Also, at that same time, a new fire protection box (BiRa) was installed which will power off both the power supplies if the smoke detector goes off.  So, as with the PEI, we no longer have to turn off these power supplies overnight.  The PEI supply is scheduled to reappear the beginning of next week.
<!-- Author: Steve Hahn -->


<!-- Date: Thu Aug 12 18:24:02 1999 -->
Today at about 4:00 PM, I rewired the HV cables on the NW arch such that only wedges 21W, 22W, and 23W have HV.  Thus, the CEM PMT modifications can proceed on the other wedges (these wedges are completed) while we run on these wedges which are connected to the wedge test crate. 
<p>I did this by disconnecting the CEM and CHA HV inputs from the movable cable tray into the Pisabox for wedge 05W at the beginning of the HV daisy chain, adding long HV cables with SHV barrels to these cables, and connecting the long cables at the other end into wedge 23W.  Then I unplugged the connections between 21W and 20W.  I checked that only wedges 23W, 22W, and 21W were receiving HV.
<!-- Author: Steve Hahn -->


<!-- Date: Mon Aug 16 16:20:17 1999 -->
Please use caution if you choose to put HTML into an entry.  Forgetting to use a close tag for each open tag will certainly cause problems. (click to clear this message)  
<!-- Author: Ph. Gris -->


<!-- Date: Mon Aug 16 16:21:49 1999 -->
New configuration for the PMT voltages of 21W: they are now directly connected to the first floor by two cables.
<!-- Author: Ph. Gris -->


<!-- Date: Wed Sep 1 19:01:24 1999 -->
<!doctype html public "-//w3c//dtd html 4.0 transitional//en"> 
<html> 
<head> 
   <meta name="Author" content="Steve Hahn"> 
   <meta name="GENERATOR" content="Mozilla/4.51 (Macintosh; I; PPC) [Netscape]"> 
</head> 
<body> 
Yesterday, Charlie Nelson gave me 44 V5 Caf&eacute; cards; enough for 2 
ADMEMs leaving 4 spares.&nbsp; This afternoon, I installed and tested these 
Caf&eacute; cards in the two production (V4) ADMEMs we have in slots 16 
and 17 in the wedge test crate.&nbsp; In anticipation of wiring the new 
production transition boards for 23W, I also removed two ADMEMs from the 
CLC crate in the 1st floor counting room; one of these already had V4 Caf&eacute; 
cards, while the other I had to fill with V4 Caf&eacute; cards that I had 
just removed from one of the V4 ADMEMs. 
<p>Before addition of new cards (and Caf&eacute; cards): 
<table BORDER > 
<tr> 
<td>Slot</td> 
 
<td>ADMEM Type</td> 
 
<td>ADMEM #</td> 
 
<td>Caf&eacute; Type</td> 
</tr> 
 
<tr> 
<td>16</td> 
 
<td>Prod V4</td> 
 
<td>020</td> 
 
<td>V4</td> 
</tr> 
 
<tr> 
<td>17</td> 
 
<td>Prod V4</td> 
 
<td>022</td> 
 
<td>V4</td> 
</tr> 
 
<tr> 
<td>18</td> 
 
<td>Prod V2</td> 
 
<td>016</td> 
 
<td>V4</td> 
</tr> 
 
<tr> 
<td>19</td> 
 
<td>Prod V2</td> 
 
<td>018</td> 
 
<td>V4</td> 
</tr> 
 
<tr> 
<td>20</td> 
 
<td>empty</td> 
 
<td></td> 
 
<td></td> 
</tr> 
 
<tr> 
<td>21</td> 
 
<td>empty</td> 
 
<td></td> 
 
<td></td> 
</tr> 
</table> 
 
<p>After addition of new cards (and Caf&eacute; cards): 
<table BORDER > 
<tr> 
<td>Slot</td> 
 
<td>ADMEM Type</td> 
 
<td>ADMEM #</td> 
 
<td>Caf&eacute; Type</td> 
</tr> 
 
<tr> 
<td>16</td> 
 
<td>Prod V4</td> 
 
<td>020</td> 
 
<td>V5</td> 
</tr> 
 
<tr> 
<td>17</td> 
 
<td>Prod V4</td> 
 
<td>022</td> 
 
<td>V5</td> 
</tr> 
 
<tr> 
<td>18</td> 
 
<td>Prod V2</td> 
 
<td>016</td> 
 
<td>V4</td> 
</tr> 
 
<tr> 
<td>19</td> 
 
<td>Prod V2</td> 
 
<td>018</td> 
 
<td>V4</td> 
</tr> 
 
<tr> 
<td>20</td> 
 
<td>Prod V2</td> 
 
<td>019</td> 
 
<td>V4</td> 
</tr> 
 
<tr> 
<td>21</td> 
 
<td>Prod V2</td> 
 
<td>017</td> 
 
<td>V4</td> 
</tr> 
</table> 
 
<p>I recalibrated and redownloaded all cards; however, the pedestals differed 
considerably from 50 counts.&nbsp; I think I did not let them warm up sufficiently.&nbsp; 
Hao Wei's new version of QIETEST to download the new FRAMs on the V5 Caf&eacute; 
cards did not work correctly; however, the chip erase in the old version 
worked just fine. 
</body> 
</html>  
<!-- Author: Steve Hahn -->


<!-- Date: Fri Sep 3 08:32:25 1999 -->
Moved B Fanout cable on Fanout in Slot 10 to Fanout in Slot 13 (B). Connected Tracer Clock Input for TSI Crate to Channel B of Slot 10 Fanout.
<!-- Author: Steve V. -->


<!-- Date: Fri Sep 3 14:38:38 1999 -->
<HTML> 
<HEAD> 
   <META NAME="Author" CONTENT="Steve Hahn"> 
   <META NAME="GENERATOR" CONTENT="Mozilla/4.05C-SGI [en] (X11; I; IRIX 6.5 IP32) [Netscape]"> 
</HEAD> 
<BODY> 
Rearranged cards in wedge test crate so they are more like complete final 
crate: 
<BR>&nbsp; 
<DT> 
Before:</DT> 
 
<UL> 
<LI> 
Slot 1 - MVME 2301</LI> 
 
<LI> 
Slot 2 - empty</LI> 
 
<LI> 
Slot 3 - empty</LI> 
 
<LI> 
Slot 4 - TESTCLK</LI> 
 
<LI> 
Slot 5 - TRACER</LI> 
</UL> 
After 
<UL> 
<LI> 
Slot 1 - MVME 2301</LI> 
 
<LI> 
Slot 2 - TRACER</LI> 
 
<LI> 
Slot 3 - TESTCLK</LI> 
 
<LI> 
Slot 4 - empty (for now)</LI> 
 
<LI> 
Slot 5 - SMXR (not in before)</LI> 
</UL> 
Also, I plugged the hadron ASD&nbsp;into slot 12. 
 
<P>After doing this, we rechecked the VME crate voltages: 
 
<P>+5 digital - 5.09 V 
<BR>+5 analog - 4.82 V 
<BR>-5 analog - -4.86 V 
<BR>+15 analog - 14.95 V 
<BR>-15 analog - -15.00 V 
 
<P>We will tweak these up to more correct values. 
</BODY> 
</HTML>
<!-- Author: Steve Hahn -->


<!-- Date: Tue Sep 7 19:57:08 1999 -->
<!doctype html public "-//w3c//dtd html 4.0 transitional//en"> 
<html> 
<head> 
   <meta name="GENERATOR" content="Mozilla/4.61 [en] (X11; I; Linux 2.0.36 i686) [Netscape]"> 
</head> 
<body> 
Today, Fred and company put remote sensing on all the Condor linear supply 
outputs (but not the Astec digital power).&nbsp; All the voltages are correct 
to within a few hundredths of a volt except digital power, which is at 
+5.1 V.&nbsp; We had some problems with getting the FRAMs downloaded correctly 
after coming back up; curiously, the oldest cards/Cafe cards in slot 20 
and 21 had to be <b>erased</b> several times in succesion before being 
correct (all 0xffff).&nbsp; I have no put all the cards back in the crate 
and recalibrated and redownload all the cards.&nbsp; The noise looks remarkably 
quiet, better than I've seen it in a long time.&nbsp; For example, here's 
slot 16 (ADMEM V4/Caf&eacute; V5): 
<p><tt>&nbsp;Statistics of the last 1000 events (taken in 1 seconds):</tt> 
<br><tt>&nbsp;&nbsp; QIE&nbsp;&nbsp; #Events&nbsp;&nbsp; L2_0(mean +/- 
RMS)&nbsp; L2_1(mean +/- RMS)&nbsp; L2_2(mean +/- RMS)&nbsp; L2_3(mean 
+/- RMS)</tt> 
<br><tt>&nbsp;&nbsp;&nbsp; 0&nbsp;&nbsp;&nbsp; 1000&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.7&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.9&nbsp;&nbsp;&nbsp;&nbsp; 1.1&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.4&nbsp;&nbsp;&nbsp;&nbsp; 1.1&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.8&nbsp;&nbsp;&nbsp;&nbsp; 1.2</tt> 
<br><tt>&nbsp;&nbsp;&nbsp; 1&nbsp;&nbsp;&nbsp; 1000&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.7&nbsp;&nbsp;&nbsp;&nbsp; 1.1&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.6&nbsp;&nbsp;&nbsp;&nbsp; 1.1&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.6&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.6&nbsp;&nbsp;&nbsp;&nbsp; 1.1</tt> 
<br><tt>&nbsp;&nbsp;&nbsp; 2&nbsp;&nbsp;&nbsp; 1000&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.7&nbsp;&nbsp;&nbsp;&nbsp; 1.3&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.2&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.2&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.9&nbsp;&nbsp;&nbsp;&nbsp; 1.1</tt> 
<br><tt>&nbsp;&nbsp;&nbsp; 3&nbsp;&nbsp;&nbsp; 1000&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.0&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.3&nbsp;&nbsp;&nbsp;&nbsp; 1.1&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.3&nbsp;&nbsp;&nbsp;&nbsp; 1.3&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.2&nbsp;&nbsp;&nbsp;&nbsp; 1.1</tt> 
<br><tt>&nbsp;&nbsp;&nbsp; 4&nbsp;&nbsp;&nbsp; 1000&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.9&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.7&nbsp;&nbsp;&nbsp;&nbsp; 1.4&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.2&nbsp;&nbsp;&nbsp;&nbsp; 1.1&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.7&nbsp;&nbsp;&nbsp;&nbsp; 1.2</tt> 
<br><tt>&nbsp;&nbsp;&nbsp; 5&nbsp;&nbsp;&nbsp; 1000&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.7&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.0&nbsp;&nbsp;&nbsp;&nbsp; 1.1&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.6&nbsp;&nbsp;&nbsp;&nbsp; 1.1&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.4&nbsp;&nbsp;&nbsp;&nbsp; 1.2</tt> 
<br><tt>&nbsp;&nbsp;&nbsp; 6&nbsp;&nbsp;&nbsp; 1000&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.4&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.9&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.7&nbsp;&nbsp;&nbsp;&nbsp; 1.1&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.8&nbsp;&nbsp;&nbsp;&nbsp; 1.1</tt> 
<br><tt>&nbsp;&nbsp;&nbsp; 7&nbsp;&nbsp;&nbsp; 1000&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.8&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.0&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.8&nbsp;&nbsp;&nbsp;&nbsp; 1.1&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.2&nbsp;&nbsp;&nbsp;&nbsp; 1.1</tt> 
<br><tt>&nbsp;&nbsp;&nbsp; 8&nbsp;&nbsp;&nbsp; 1000&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.0&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.9&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.3&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.1&nbsp;&nbsp;&nbsp;&nbsp; 1.2</tt> 
<br><tt>&nbsp;&nbsp;&nbsp; 9&nbsp;&nbsp;&nbsp; 1000&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.2&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.3&nbsp;&nbsp;&nbsp;&nbsp; 1.1&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.7&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.0&nbsp;&nbsp;&nbsp;&nbsp; 1.1</tt> 
<br><tt>&nbsp;&nbsp; 10&nbsp;&nbsp;&nbsp; 1000&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.0&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.9&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.7&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.9&nbsp;&nbsp;&nbsp;&nbsp; 1.2</tt> 
<br><tt>&nbsp;&nbsp; 11&nbsp;&nbsp;&nbsp; 1000&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.6&nbsp;&nbsp;&nbsp;&nbsp; 1.3&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.9&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.3&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.2&nbsp;&nbsp;&nbsp;&nbsp; 1.2</tt> 
<br><tt>&nbsp;&nbsp; 12&nbsp;&nbsp;&nbsp; 1000&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.5&nbsp;&nbsp;&nbsp;&nbsp; 1.3&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.9&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.1&nbsp;&nbsp;&nbsp;&nbsp; 1.1&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.9&nbsp;&nbsp;&nbsp;&nbsp; 1.2</tt> 
<br><tt>&nbsp;&nbsp; 13&nbsp;&nbsp;&nbsp; 1000&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.3&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.9&nbsp;&nbsp;&nbsp;&nbsp; 1.1&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.4&nbsp;&nbsp;&nbsp;&nbsp; 1.1&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.9&nbsp;&nbsp;&nbsp;&nbsp; 1.5</tt> 
<br><tt>&nbsp;&nbsp; 14&nbsp;&nbsp;&nbsp; 1000&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.1&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.0&nbsp;&nbsp;&nbsp;&nbsp; 1.1&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.8&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.4&nbsp;&nbsp;&nbsp;&nbsp; 1.1</tt> 
<br><tt>&nbsp;&nbsp; 15&nbsp;&nbsp;&nbsp; 1000&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.4&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.5&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.4&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.3&nbsp;&nbsp;&nbsp;&nbsp; 1.2</tt> 
<br><tt>&nbsp;&nbsp; 16&nbsp;&nbsp;&nbsp; 1000&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.0&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.4&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.4&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.2&nbsp;&nbsp;&nbsp;&nbsp; 1.2</tt> 
<br><tt>&nbsp;&nbsp; 17&nbsp;&nbsp;&nbsp; 1000&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.2&nbsp;&nbsp;&nbsp;&nbsp; 1.3&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.2&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.8&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.1&nbsp;&nbsp;&nbsp;&nbsp; 1.1</tt> 
<br><tt>&nbsp;&nbsp; 18&nbsp;&nbsp;&nbsp; 1000&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.3&nbsp;&nbsp;&nbsp;&nbsp; 1.3&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.5&nbsp;&nbsp;&nbsp;&nbsp; 1.1&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.1&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.6&nbsp;&nbsp;&nbsp;&nbsp; 1.2</tt> 
<br><tt>&nbsp;&nbsp; 19&nbsp;&nbsp;&nbsp; 1000&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.0&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.4&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
50.7&nbsp;&nbsp;&nbsp;&nbsp; 1.2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
49.8&nbsp;&nbsp;&nbsp;&nbsp; 1.3</tt> 
<br>&nbsp; 
<br>&nbsp; 
<br>&nbsp; 
</body> 
</html>
<!-- Author: Steve Hahn -->


<!-- Date: Thu Sep 9 18:16:48 1999 -->
<!doctype html public "-//w3c//dtd html 4.0 transitional//en"> 
<html> 
<head> 
   <meta name="Author" content="Steve Hahn"> 
   <meta name="GENERATOR" content="Mozilla/4.51 (Macintosh; I; PPC) [Netscape]"> 
</head> 
<body> 
Fred and his new tech have adjusted the +5 and -5.2 analog power supplies 
to test whether low voltages cause the "roll-over" we say in the I/2 range 
of the QIEs last week.&nbsp; The value below is lowest we could set the 
+5 V analog power supply with the pot; the available range was from +5.08 
V to +5.3 V.&nbsp;&nbsp; So, the voltages are NOT correct now, but instead 
are set at: 
<table BORDER > 
<tr> 
<td> 
<center>Power supply</center> 
</td> 
 
<td> 
<center>Actual voltage</center> 
</td> 
</tr> 
 
<tr> 
<td> 
<center>+5 analog</center> 
</td> 
 
<td> 
<center>+5.08 V</center> 
</td> 
</tr> 
 
<tr> 
<td> 
<center>-5.2 analog</center> 
</td> 
 
<td> 
<center>-4.80 V</center> 
</td> 
</tr> 
 
<tr> 
<td> 
<center>+15 analog</center> 
</td> 
 
<td> 
<center>+14.99 V</center> 
</td> 
</tr> 
 
<tr> 
<td> 
<center>-15 analog</center> 
</td> 
 
<td> 
<center>-15.00 V</center> 
</td> 
</tr> 
 
<tr> 
<td> 
<center>+5 digital</center> 
</td> 
 
<td> 
<center>+5.08 V</center> 
</td> 
</tr> 
</table> 
 
<p>Tomorrow, after running tests we will set them back (and understand 
the +5 analog problem).&nbsp; We may remove remote sensing as a test, also 
(but put it back when done).<br> 
<BR> 
</body> 
</html>  
<!-- Author: Steve Hahn -->


<!-- Date: Tue Sep 14 11:28:27 1999 -->
<pre>It's now Tuesday and the Condor power supplies are still not 
adjusted back to the correct voltages. 
 
It appears there is a new bad channel in slot 16 (Cafe V5): 
 
 Statistics of the last 1000 events (taken in 1 seconds): 
   QIE   #Events   L2_0(mean +/- RMS)  L2_1(mean +/- RMS)  L2_2(mean +/- RMS)  L2_3(mean +/- RMS) 
    0    1000         47.4     6.3         48.8     6.4         49.8     6.4         50.8     6.5 
    1    1000         47.2     6.6         49.3     6.6         49.5     6.5         51.4     6.5 
    2    1000         48.7     6.0         50.0     6.2         50.3     5.9         51.5     6.1 
    3    1000         47.4     6.0         49.1     5.9         49.8     6.2         51.4     5.7 
    4    1000         47.6     6.2         49.6     6.6         50.2     6.3         51.1     6.0 
    5    1000         47.2     5.8         49.5     5.7         49.7     6.3         51.8     5.5 
    6    1000         47.7     5.8         50.7     6.0         50.2     6.0         51.4     5.6 
    7    1000         47.7     5.2         50.4     5.4         49.8     5.4         51.2     5.1 
    8    1000         47.0     5.8         50.6     6.0         50.3     6.0         52.0     5.7 
    9    1000         47.6     5.7         50.2     5.6         49.3     5.8         50.8     5.4 
   10    1000         47.0     5.5         50.2     5.5         49.6     5.6         50.8     5.3 
   11    1000         48.1     5.2         50.2     4.9         49.7     4.9         50.7     4.9 
   12    1000         47.7     5.7         50.9     5.6         49.3     5.8         51.3     5.4 
   13    1000         47.8     5.8         48.9     5.4         50.4     5.5         51.8     5.3 
   14    1000         48.3     6.0         50.8     5.6         49.4     5.8         51.4     5.6 
   15    1000         48.7     5.7         50.7     5.4         49.9     5.7         51.4     5.3 
   16    1000         46.7     6.0         51.4     5.6         48.7     6.1         51.2     5.6 
   17    1000         47.1     5.4         51.4     5.2         49.8     5.5         51.4     5.2 
   18    1000         47.0     5.5         49.6     5.2         48.7     5.4         51.3     5.2 
   19    1000     262136.0     0.0         51.2     5.9         49.4     6.4         52.1     5.6 
 
I will now try recalibrating to see if this fixes the problem. 
</pre>
<!-- Author: Steve Hahn -->


<!-- Date: Tue Sep 14 11:35:02 1999 -->
<pre>Recalibrating fixed bad QIE 19: 
 
 Statistics of the last 1000 events (taken in 42 seconds): 
   QIE   #Events   L2_0(mean +/- RMS)  L2_1(mean +/- RMS)  L2_2(mean +/- RMS)  L2_3(mean +/- RMS) 
    0    1000         50.3     6.0         49.7     6.0         49.9     5.9         50.2     6.1 
    1    1000         50.4     6.2         49.7     6.4         49.9     6.1         50.2     6.2 
    2    1000         50.4     5.6         50.0     5.6         50.1     5.5         50.5     5.7 
    3    1000         50.3     5.6         49.8     5.5         50.1     5.6         50.3     5.5 
    4    1000         50.6     5.8         50.0     5.9         50.2     5.8         50.6     5.9 
    5    1000         50.2     5.6         49.7     5.7         49.9     5.5         50.2     5.7 
    6    1000         50.1     5.7         49.7     5.6         49.9     5.4         50.1     5.6 
    7    1000         50.4     5.1         49.9     5.2         50.2     5.1         50.3     5.1 
    8    1000         50.7     5.7         50.2     5.6         50.5     5.5         50.8     5.6 
    9    1000         50.5     5.4         50.1     5.4         50.3     5.3         50.5     5.4 
   10    1000         50.6     5.2         50.1     5.2         50.4     5.1         50.6     5.3 
   11    1000         50.4     4.7         50.1     4.9         50.2     4.7         50.4     4.8 
   12    1000         50.4     5.5         50.0     5.5         50.3     5.3         50.3     5.5 
   13    1000         50.2     5.2         49.9     5.3         50.0     5.0         50.2     5.1 
   14    1000         50.5     5.6         50.1     5.6         50.3     5.4         50.6     5.6 
   15    1000         50.1     5.3         49.8     5.4         50.0     5.2         50.2     5.3 
   16    1000         50.2     5.5         49.8     5.6         50.1     5.4         50.3     5.6 
   17    1000         50.3     5.1         50.0     5.1         50.2     4.9         50.2     5.1 
   18    1000         50.2     5.2         49.9     5.2         50.1     5.0         50.2     5.1 
   19    1000         50.3     5.5         50.0     5.6         50.0     5.4         50.3     5.5 
 
Presumably, FRAM got clobbered, but how? 
</pre>
<!-- Author: Steve Hahn -->


<!-- Date: Wed Sep 15 18:23:01 1999 -->
<pre>Clock fanout crate has been moved to upper NE corner of endwall 
on east side.  Clock inputs have been rerouted to this crate, and 
clock fanout cable now comes down 90 degree side of NW arch from 
fanout crate location.  We were running with TESTCLK clock 
emulator the last day or two; I have just plugged in clock fanout 
cable.  Peds looks good: 
 
 Statistics of the last 1000 events (taken in 18 seconds): 
   QIE   #Events   L2_0(mean +/- RMS)  L2_1(mean +/- RMS)  L2_2(mean +/- RMS)  L2_3(mean +/- RMS) 
    0    1000         49.7     3.9         49.7     3.9         49.5     3.9         49.6     3.9 
    1    1000         49.7     4.2         49.9     4.2         49.7     4.0         49.8     4.2 
    2    1000         50.3     4.0         50.2     3.9         50.2     4.1         50.0     4.2 
    3    1000         49.9     4.0         49.9     4.0         49.7     3.8         49.8     3.9 
    4    1000         50.4     4.0         50.4     3.8         50.2     3.8         50.1     4.0 
    5    1000         49.7     3.9         49.8     3.8         49.5     3.8         49.8     3.8 
    6    1000         49.8     3.8         49.8     3.7         49.8     3.8         49.7     3.8 
    7    1000         50.2     3.5         50.1     3.4         50.0     3.4         50.1     3.4 
    8    1000         49.9     3.9         49.9     3.8         49.8     3.9         49.7     4.0 
    9    1000         50.0     3.7         49.9     3.6         50.0     3.6         49.9     3.6 
   10    1000         50.2     4.2         50.1     4.1         50.1     4.1         49.9     4.1 
   11    1000         49.9     3.1         49.8     3.3         49.7     3.2         49.8     3.2 
   12    1000         50.3     3.8         50.2     3.8         50.3     3.8         50.0     3.8 
   13    1000         49.8     3.5         49.8     3.5         49.7     3.5         49.7     3.5 
   14    1000         50.4     4.2         50.3     4.2         50.3     4.2         50.2     4.3 
   15    1000         50.7     3.9         50.5     3.9         50.4     3.9         50.5     3.9 
   16    1000         50.1     3.8         50.0     3.8         50.0     3.9         49.9     3.7 
   17    1000         50.1     4.1         49.9     4.2         50.0     4.3         50.0     4.2 
   18    1000         48.6     3.4         48.5     3.5         48.5     3.5         48.5     3.5 
   19    1000         51.3     4.1         51.1     4.1         51.2     4.1         51.1     4.2 
 
</pre>
<!-- Author: Steve Hahn -->


<!-- Date: Thu Sep 16 10:15:05 1999 -->
Since one week problem with the calibration card: blown fuse (that corresponding to VCC15) each time the card was plugged (21W). After investigations with Ed, came to the conclusion it was the cable from the transition card to the LED box that was not good.  
I had another problem with the same cable of the wedge 22W a few weeks ago. When plugging the card, this cable became extremely hot and a burnt smelling near the crate was detectable. 
However I never managed (up to now) to reproduce this phenomenon. 
The conclusion from all this is that these cables are not really reliable. Fortunately, for run II, new cables will take place...
<!-- Author: Ph. Gris -->


<!-- Date: Thu Sep 16 14:46:26 1999 -->
&
<!-- Author: {Quick_Entry} -->


<!-- Date: Mon Sep 20 12:01:50 1999 -->
Manage to flash LEDs in the RunControl framework for 21W (took link cable form transition card to LED box from 20W). 
Problem with 22W: changing the dac value does not change the ouput pulse. This problem is also there when using CosmicReadout.
<!-- Author: Ph. Gris -->


<!-- Date: Thu Sep 23 12:24:26 1999 -->
This morning: runs with Calibration Card (ADMEM 16 Wedge 21W). ADMEM and Calcard buffers read through RunControl. Had problems with one channel (QIE 3 of the ADMEM 16): always gave 0 counts. Made a QIE calibration+FlashRam loading. New calibration run with Calcard. All ADMEM buffers were in a strange state (always the same number of counts for the 4 L2 buffers for a given channel). Made a reset of ADMEM+TCLK. Run with Calcard. Then all channel gave 0 counts for all L2 buffers. Reboot the crate. Works fine then (channel 3 of ADMEM 16 cured).
<!-- Author: Ph.Gris -->


<!-- Date: Thu Sep 23 13:35:54 1999 -->
<pre>Wanted to make comparisons of Calcard runs between RunControl and CosmicReadout.  
Have some problems with the new startup script(boot/front-end/startup.all) and CosmicReadout.  
There seems to be no communication between the GUI and the processor.  
Decided to take the old startup script (boot/front-end/lists/startup.allCdf).  
Then CosmicReadout works fine. 
 
Comparison of the modules loaded for each startup scripts: 
 
with boot/front-end/startup.all (new):  
 
MODULE NAME     MODULE ID  GROUP #    TEXT START DATA START  BSS START 
--------------- ---------- ---------- ---------- ---------- ---------- 
      libxdr.so   0xec60a0          2   0xebb028   0xebcaf4   0xebcb48 
  libutilsip.so   0xeae428          3   0xe9cae8   0xea7028   0xea724c 
       tools.lo   0xeba860          4   0xeb0e60   0xeb38fc   0xeb3a78 
      rshell.lo   0xeba4d0          5   0xeb52a8   0xeb65e0   0xeb6638 
libVISIONserver.so   0xeb9cc0          6   0xe85088   0xe908e4   0xe913c0 
   VISIONserver   0xe9c620          7   0xebd198   0xec0430   0xec04ac 
     VISIONdemo   0xebcc40          8   0xe92490   0xe96130   0xe962e8 
    VISIONtimer   0xeb66d0          9   0xeb0180   0xeb0650   0xeb0650 
VISIONinterrupt   0xeb3e68         10   0xeaf010   0xeaf810   0xeaf810 
    ROBINserver   0xe96410         11   0xe3e4c0   0xe446d0   0xe448c4 
       CrateVme   0xe23c00         12   0xe1f970   0xe22740   0xe22858 
    librtutil.x   0xeb7798         13   0x8f8868   0x95dac0   0x96afb8 
     librtipc.x   0xeb7960         14   0x9acf10   0xa47ee4   0xa47fa4 
      FER_Smart   0xe0d5c0         15   0xdf18f0   0xe067d4   0xe06918 
value = 0 = 0x0 
 
 
with boot/front-end/lists/startup.allCdf (old): 
 
MODULE NAME     MODULE ID  GROUP #    TEXT START DATA START  BSS START 
--------------- ---------- ---------- ---------- ---------- ---------- 
libVISIONserver.so   0xea9c30          2   0xe9b650   0xea6eac   0xea7988 
   VISIONserver   0xebfa88          3   0xeb5638   0xeb88d0   0xeb894c 
     VISIONdemo   0xebf338          4   0xe96030   0xe99cd0   0xe99e88 
    VISIONtimer   0xeb9b80          5   0xeb40b0   0xeb4580   0xeb4580 
VISIONinterrupt   0xeb94e0          6   0xeb2f40   0xeb3740   0xeb3740 
    ROBINserver   0xeb3780          7   0xe50c08   0xe56e18   0xe5700c 
       CrateVme   0xec51e0          8   0xe35790   0xe38560   0xe38678 
    librtutil.x   0xe83cc8          9   0x90e9f8   0x973c50   0x981148 
     librtipc.x   0xe487e0         10   0x9c3870   0xa5e844   0xa5e904 
    librtutil.x   0x9bf1a0         11   0x7f09d8   0x855c30   0x863128 
     librtipc.x   0xa842b8         12   0x868238   0x90320c   0x9032cc 
      FER_Smart   0x9c33c0         13   0xe1f5d8   0xe344bc   0xe34600 
value = 0 = 0x0 
 
As it can be noticed the differences are: 
- more files .so and .lo with the new startup script. 
- librtutil.x and librtipc.x are loaded twice with the old script. 
 
An interesting point is that even if I do not see anything on the processor, I can see LED flashing with the scope. Thus the registers seem to be correctely written. 
 
I would like to add another point not related to this problem but related to the startup scripts. They both load FER_Smart. This is not really useful when people wants to make some tests with RunControl because in that case one must unload FER_Smart to put its own version. 
</pre>
<!-- Author: Ph.Gris -->


<!-- Date: Wed Oct 20 07:40:39 1999 -->
<pre>Comment on above loading of smart sockets twice. 
Best Guess:  startup.allCdf loacs one set of smartsockets 
libraries, then the custom startup which loads FER_Smart 
loads the other set.  Not sure how it goes in the "new" startup 
scripts</pre>
<!-- Author: Steve N (that&#39;s an N) ahn -->


<!-- Date: Tue Oct 26 13:33:36 1999 -->
Only a test
<!-- Author: Steve Hahn -->


<!-- Date: Wed Oct 27 10:59:35 1999 -->
For Ray Culbertson and Pedro Amaral, I have plugged in the temporary L1 cables into the ADMEMs in slot 16 and 17 in the wedge test crate.  In slot 16 (CEM 21W), cables 1-7 are plugged into the L1 outputs on the front; in slot 17 (CHA 21W), cables 8-13 are plugged in the first six of the seven outputs.  I believe all these cards are configured as plug cards, so trigger sums (which are not the correct sums for central) appear on every output. 
<p> 
Some of the connector ends of these cables are broken.  Cable 3 had a bent connector casing which prevented it from engaging properly with the L1 output on the ADMEM.  I attempted a repair with needlenose pliers; it seems better, but not good.  Cable 8 had the shrinkwrap and ground braid pulled free from the connector casing, so only the individual wires to pins were holding the cable on.
<!-- Author: Steve Hahn -->


<!-- Date: Fri Nov 5 15:33:00 1999 -->
Found that +15 V light on front panel of ADMEM in slot 16 was dim...sure enough, fuse F5 was blown.  Replaced, now card calibrates OK.  Also, with Pedro Amaral's help, found all ADMEM transition cards were not fully seated (20 and 21 were obvious, but the rest were not).  Now all transition cards are in place.
<!-- Author: Steve Hahn -->


<!-- Date: Wed Dec 1 09:30:25 1999 -->
Changed the configuration of the crate CPU to boot from Flash RAM. Included in it are: kernel, smartsockets, RTIlib, FISION 2_12e-beta, ROBIN v2_4_10, 
CrateVme v1_2, rdate to set the on-board clock, flashProg for Flash RAM programming, vxhttpd (simple http demon), and cdfInit (a C version of the startup script). Please report any problems. Note that in automated scripts something like "moduleShow" will now lead to different results - I'd recommend using e.g. symFindByName().  
<!-- Author: Arnd. -->


<!-- Date: Wed Dec 8 13:19:00 1999 -->
It appears the -5.2 Condor linear power supply for the NW central crate (a.k.a. the wedge test crate) is not working.  I could find no obvious problems (like disconnected wires).  Until we resolve this please leave the NW crate off (using circuit breaker #12 in the 60 Hz breaker box).
<!-- Author: Steve Hahn -->


<!-- Date: Wed Dec 8 16:01:50 1999 -->
<!DOCTYPE HTML PUBLIC "-//W3C//DTD HTML 4.0 Transitional//EN"> 
<HTML> 
<HEAD> 
   <META HTTP-EQUIV="Content-Type" CONTENT="text/html; charset=iso-8859-1"> 
   <META NAME="Author" CONTENT="Richard J. Tesarek"> 
   <META NAME="GENERATOR" CONTENT="Mozilla/4.07 [en] (X11; I; Linux 2.0.36 i686) [Netscape]"> 
</HEAD> 
<BODY> 
Began a noise source study yesterday afternoon (12/7/99) involving the 
north east central calorimeter 
<BR>crate (b0ccal14).&nbsp; The study involved taking pedestal data using 
an older version of Hao Wei's qietest 
<BR>program (~weihao/cal_calib/src) under differing conditions.&nbsp; The 
conditions and output files are listed 
<BR>in the table below. The goal is to locate and quantify possible noise 
sources for this crate.&nbsp; The conditions 
<BR>common for the whole study are: 
<UL> 
<LI> 
standard TRACER/MVME connections.</LI> 
 
<LI> 
only data from ADMEM in slot 16.</LI> 
 
<LI> 
ADMEM operated in "passthrough" mode.</LI> 
 
<LI> 
samples taken for each condition.</LI> 
 
<LI> 
1000 events for each sample.</LI> 
</UL> 
 
<CENTER><TABLE BORDER WIDTH="100%" NOSAVE > 
<CAPTION>Table of output file and conditions for various pedestal runs.</CAPTION> 
 
<TR> 
<TD>I.</TD> 
 
<TD>pedtest00-04.txt</TD> 
 
<TD> 
<UL> 
<LI> 
no inputs to ADMEM</LI> 
 
<LI> 
no outputs from ADMEM</LI> 
</UL> 
</TD> 
</TR> 
 
<TR> 
<TD>II.</TD> 
 
<TD>pedtest05-09.txt</TD> 
 
<TD> 
<UL> 
<LI> 
PMT inputs (no HV) to ADMEM</LI> 
 
<LI> 
no outputs from ADMEM</LI> 
</UL> 
</TD> 
</TR> 
 
<TR> 
<TD>III.</TD> 
 
<TD>pedtest10-14.txt</TD> 
 
<TD> 
<UL> 
<LI> 
PMT inputs</LI> 
 
<LI> 
trigger outputs from ADMEM</LI> 
</UL> 
</TD> 
</TR> 
 
<TR> 
<TD>IV.</TD> 
 
<TD>pedtest15-19.txt</TD> 
 
<TD> 
<UL> 
<LI> 
PMT inputs</LI> 
 
<LI> 
trigger outputs from ADMEM</LI> 
 
<LI> 
SMXR backplane connected (to wedge VME crate)</LI> 
</UL> 
</TD> 
</TR> 
 
<TR> 
<TD>V.</TD> 
 
<TD>pedtest20-24.txt</TD> 
 
<TD> 
<UL> 
<LI> 
PMT inputs</LI> 
 
<LI> 
trigger outputs from ADMEM</LI> 
 
<LI> 
SMXR backplane connected</LI> 
 
<LI> 
SMXR front panel connected</LI> 
</UL> 
</TD> 
</TR> 
 
<TR> 
<TD>VI.</TD> 
 
<TD>pedtest25-29.txt</TD> 
 
<TD> 
<UL> 
<LI> 
PMT inputs</LI> 
 
<LI> 
trigger outputs from ADMEM</LI> 
 
<LI> 
SMXR front/back connected</LI> 
 
<LI> 
Hadron ASD dynodes connected*</LI> 
</UL> 
</TD> 
</TR> 
 
<TR> 
<TD>VII.</TD> 
 
<TD>pedtest30-34.txt</TD> 
 
<TD> 
<UL> 
<LI> 
PMT inputs</LI> 
 
<LI> 
trigger outputs from ADMEM</LI> 
 
<LI> 
SMXR front/back connected</LI> 
 
<LI> 
Hadron ASD dynodes connected*</LI> 
 
<LI> 
Hadron ASD TDC cables connected *</LI> 
</UL> 
</TD> 
</TR> 
</TABLE></CENTER> 
 
<P>*Note:&nbsp; This is a plug hadron ASD and the internal wiring is not 
compatable with the central cables. 
<P>On completion of this study all cables were returned to their previous 
location <B>except</B> the Morse cables 
<BR>connecting the ADMEM in slot 16 with the TSCR. 
<BR> 
<HR WIDTH="100%"> 
<BR>Below is a summary of the data obtained.&nbsp; The Figure 1 below summarizes 
typical output; different colors indicate different capacitors on the QIE.&nbsp;&nbsp;&nbsp; 
Using the statistics on the RMS plots from the 5 samples taken under common 
conditions, a mean and standard deviation was calculated.&nbsp; Figure 
2 shows the ADMEM average 
<BR>pedestal (and uncertainty) for pedestals taken undera a given condition. 
<P>Figure 1 and Figure 2 immediately follow this note. 
<BR>&nbsp; 
<P><B>Conclusions:</B> 
<BR>The data indicate&nbsp; that the ADMEM has a typical pedestal RMS of 
approximately 1.06. ADC counts.&nbsp; After the PMT&nbsp;cables are connected 
the RMS from the ADMEM is 1.33 ADC counts.&nbsp; This means that we ave 
approximately 0.84 ADC counts of noise input to the system solely from 
the PMTs and their cables.&nbsp; Connecting the remaining cables to the 
crate does not presently show any enhancement in the noise. 
<BR>&nbsp; 
</BODY> 
</HTML>
<!-- Author: Rick Tesarek -->


<!-- Date: Wed Dec 8 16:14:17 1999 -->
<table><tr><td><a href=/elog/notebooks/wedge/Wed_Dec__8_1999_161406.ps align=top><img src=/elog/gifs/ghost64.gif  width='56' align='top'></td><td>Figure 1: Typical summary of pedestal sample.</a></td></tr></table>
<!-- Author: Rick Tesarek -->


<!-- Date: Wed Dec 8 16:15:27 1999 -->
<table><tr><td><a href=/elog/notebooks/wedge/Wed_Dec__8_1999_161521.ps align=top><img src=/elog/gifs/ghost64.gif  width='56' align='top'></td><td>Figure 2:  Summary of noise study.</a></td></tr></table>
<!-- Author: Rick Tesarek -->


<!-- Date: Wed Dec 8 16:53:22 1999 -->
<!doctype html public "-//w3c//dtd html 4.0 transitional//en"> 
<html> 
<head> 
   <meta name="Author" content="Steve Hahn"> 
   <meta name="GENERATOR" content="Mozilla/4.51 (Macintosh; I; PPC) [Netscape]"> 
</head> 
<body> 
Nathan Eddy and I checked the NW crate (wedge test crate) to see if we 
could find why I saw no -5.2 analog voltage on the ADMEMs earlier.&nbsp;&nbsp; 
First with no cards or transition cards, we saw no problem.&nbsp; Then 
we plugged back in the MVME, TRACER, muon ASD, and the ADMEM in slot 16.&nbsp; 
Still no problems.&nbsp; Then we plugged in all cards and transition cards.&nbsp; 
Still no problem seen at the power lugs.&nbsp; On the ADMEMs, all power 
LEDs came on except the +15 analog on the ADMEM in slot 19.&nbsp; Found 
the blown picofuse (F3) and replaced it. 
<p>However, found that after crate was powered back up there were many 
new bad channels.&nbsp; Here's the output from the error summary from Hao 
Wei's new version of QIETEST (found at /data1/haowei): 
<pre> 
*******************   Start of WARNING messages   ************************* 
* ADMEM=16 QIE=16 Warning 11: Sigma(offset) big              (    3.5033) * 
* ADMEM=16 QIE=16 Warning  8: Sigma(slope)/slope big         (    0.0068) * 
*******************   End of WARNING messages   *************************** 
 
*******************   Start of FAILURE messages   ************************* 
* ADMEM=16 QIE= 2 Failure 10: No fitting                     (    0.0000) * 
* ADMEM=18 QIE= 3 Failure 13: Slope too low                  (    0.0457) * 
* ADMEM=19 QIE= 0 Failure 10: No fitting                     (    0.0000) * 
* ADMEM=19 QIE= 1 Failure 10: No fitting                     (    0.0000) * 
* ADMEM=19 QIE= 2 Failure 10: No fitting                     (    0.0000) * 
* ADMEM=20 QIE= 0 Failure 10: No fitting                     (    0.0000) * 
*******************   End of FAILURE messages   *************************** 
 
*********************************************** 
Total Warnings: 2.  
Total Failures: 6.  
*********************************************** 
</pre> 
<p>Have tried recalibrating and reloading FRAMs with no luck.&nbsp; Will 
check with Charlie Nelson.&nbsp; Still do not understand what could have 
(temporarily) killed -5.2 analog power. 
</body> 
</html>  
<!-- Author: Steve Hahn -->


<!-- Date: Mon Dec 13 22:34:31 1999 -->
<!DOCTYPE HTML PUBLIC "-//W3C//DTD HTML 4.0 Transitional//EN"> 
<HTML> 
<HEAD> 
   <META HTTP-EQUIV="Content-Type" CONTENT="text/html; charset=iso-8859-1"> 
   <META NAME="Author" CONTENT="Richard J. Tesarek"> 
   <META NAME="GENERATOR" CONTENT="Mozilla/4.07 [en] (X11; I; Linux 2.0.36 i686) [Netscape]"> 
</HEAD> 
<BODY> 
&nbsp;&nbsp;&nbsp; This note reports measurements made by Philippe Gris 
and Rick Tesarek on 12/10/99 concerning the hadron ASD transition card.&nbsp;&nbsp; 
The goal was to look at the signal characteristics into these boards with 
2 tests 
<UL> 
<LI> 
Measure the amount of signal reflection on the hadron calorimeter dynode 
signal cables due to the&nbsp; hadron ASD transition board (Note:&nbsp; 
the dynode cables terminate on the hadron ASD board and not the transition 
board).</LI> 
 
<LI> 
Measure the noise levels on the dynode cables with the hadron ASD transition 
boards plugged into the PMT.&nbsp; The PMT was biased for this test.</LI> 
</UL> 
The Figure below shows the layout for the tests performed.&nbsp; The Figure 
also shows an approximate rendering of the 'scope trace during the first 
test.&nbsp; The peak-to-peak noise for the second test was measured at 
~3mV. 
</BODY> 
</HTML>  
<!-- Author: R. Tesarek, Ph. Gris -->


<!-- Date: Mon Dec 13 22:42:40 1999 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=wedge&action=view&page=-9&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=wedge&action=view&page=-9  width='250' align='top'></td><td>Diagram of the test setup used for the hadron ASD transition board signal reflection test.  An approximate rendering of the oscilloscope trace for this test is also shown.  The noise measurement tests were performed using a similar setup as shown here, but with a hadron PMT replacing the pocket pulser</a></td></tr></table>
<!-- Author: R.J. Tesarek, Ph. Gris -->


<!-- Date: Mon Dec 13 22:44:46 1999 -->
Installed muon ASD cards and transition boards to central calorimeter crate b0ccal14 (North East lowest crate).  All lights appear to be working.
<!-- Author: R.J. Tesarek, N. Eddy -->


<!-- Date: Mon Jan 3 16:31:02 2000 -->
Removed ADMEM in b0ccal14 slot 19 (ID=0141).  All the CAFE cards in this ADMEM read zeros during calibration and had fuse F4 blown. 
 
Removed ADMEM in boccal14 slot21 (ID=0129). The CAFE cards (channels 0-5,8,19) all had fuse F4 blown (CAFE IDs= 0927, 4496, 4411, 4181, 0797, 0884, 0935) and exhibited the symptoms described above.  The CAFE card in channel 8 exhibited the same symptoms but the fuse F4 is ok.
<!-- Author: R.J. Tesarek, S. Hahn -->


<!-- Date: Mon Jan 3 18:13:31 2000 -->
<HTML> 
<HEAD> 
   <META NAME="Author" CONTENT="Steve Hahn"> 
   <META NAME="GENERATOR" CONTENT="Mozilla/4.05C-SGI [en] (X11; I; IRIX 6.5 IP32) [Netscape]"> 
</HEAD> 
<BODY> 
Have replaced bad Cafe cards with new Cafes: 
 
<P><TT>QIE ID table:</TT> 
<BR><TT>QIE&nbsp;&nbsp;&nbsp; ADMEM: 16&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 17&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
18&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 19&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 20&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
21</TT> 
<BR><TT>&nbsp; 0&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
5554&nbsp;&nbsp;&nbsp;&nbsp; 860&nbsp;&nbsp;&nbsp; 4109&nbsp;&nbsp;&nbsp; 
3371&nbsp;&nbsp;&nbsp;&nbsp; 712&nbsp;&nbsp;&nbsp; 3479</TT> 
<BR><TT>&nbsp; 1&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
4444&nbsp;&nbsp;&nbsp;&nbsp; 771&nbsp;&nbsp;&nbsp; 4465&nbsp;&nbsp;&nbsp; 
4592&nbsp;&nbsp;&nbsp;&nbsp; 943&nbsp;&nbsp;&nbsp; 3112</TT> 
<BR><TT>&nbsp; 2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
4248&nbsp;&nbsp;&nbsp;&nbsp; 691&nbsp;&nbsp;&nbsp; 5767&nbsp;&nbsp;&nbsp; 
3487&nbsp;&nbsp;&nbsp; 4351&nbsp;&nbsp;&nbsp; 4654</TT> 
<BR><TT>&nbsp; 3&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
4078&nbsp;&nbsp;&nbsp;&nbsp; 759&nbsp;&nbsp;&nbsp; 4227&nbsp;&nbsp;&nbsp; 
3303&nbsp;&nbsp;&nbsp; 4498&nbsp;&nbsp;&nbsp; 4521</TT> 
<BR><TT>&nbsp; 4&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
925&nbsp;&nbsp;&nbsp; 4440&nbsp;&nbsp;&nbsp; 4170&nbsp;&nbsp;&nbsp; 3185&nbsp;&nbsp;&nbsp; 
6197&nbsp;&nbsp;&nbsp; 5461</TT> 
<BR><TT>&nbsp; 5&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
4279&nbsp;&nbsp;&nbsp; 4143&nbsp;&nbsp;&nbsp; 4494&nbsp;&nbsp;&nbsp; 5433&nbsp;&nbsp;&nbsp;&nbsp; 
842&nbsp;&nbsp;&nbsp; 2009</TT> 
<BR><TT>&nbsp; 6&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
4258&nbsp;&nbsp;&nbsp; 4052&nbsp;&nbsp;&nbsp; 4419&nbsp;&nbsp;&nbsp; 3553&nbsp;&nbsp;&nbsp; 
4497&nbsp;&nbsp;&nbsp; 4437</TT> 
<BR><TT>&nbsp; 7&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
4432&nbsp;&nbsp;&nbsp; 4049&nbsp;&nbsp;&nbsp; 4431&nbsp;&nbsp;&nbsp; 3171&nbsp;&nbsp;&nbsp;&nbsp; 
838&nbsp;&nbsp;&nbsp; 4256</TT> 
<BR><TT>&nbsp; 8&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
4245&nbsp;&nbsp;&nbsp; 4182&nbsp;&nbsp;&nbsp; 4378&nbsp;&nbsp;&nbsp; 3430&nbsp;&nbsp;&nbsp; 
4387&nbsp;&nbsp;&nbsp; 5237</TT> 
<BR><TT>&nbsp; 9&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
762&nbsp;&nbsp;&nbsp; 4309&nbsp;&nbsp;&nbsp; 4273&nbsp;&nbsp;&nbsp; 3444&nbsp;&nbsp;&nbsp;&nbsp; 
826&nbsp;&nbsp;&nbsp; 4488</TT> 
<BR><TT>&nbsp;10&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
4291&nbsp;&nbsp;&nbsp; 4357&nbsp;&nbsp;&nbsp; 5612&nbsp;&nbsp;&nbsp; 3358&nbsp;&nbsp;&nbsp; 
6346&nbsp;&nbsp;&nbsp; 4472</TT> 
<BR><TT>&nbsp;11&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
4086&nbsp;&nbsp;&nbsp;&nbsp; 711&nbsp;&nbsp;&nbsp; 4326&nbsp;&nbsp;&nbsp; 
4542&nbsp;&nbsp;&nbsp; 4464&nbsp;&nbsp;&nbsp; 6469</TT> 
<BR><TT>&nbsp;12&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
5747&nbsp;&nbsp;&nbsp; 4358&nbsp;&nbsp;&nbsp; 5765&nbsp;&nbsp;&nbsp; 3283&nbsp;&nbsp;&nbsp; 
6388&nbsp;&nbsp;&nbsp; 4162</TT> 
<BR><TT>&nbsp;13&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
4115&nbsp;&nbsp;&nbsp; 4439&nbsp;&nbsp;&nbsp; 5636&nbsp;&nbsp;&nbsp; 4660&nbsp;&nbsp;&nbsp;&nbsp; 
866&nbsp;&nbsp;&nbsp; 4448</TT> 
<BR><TT>&nbsp;14&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
4246&nbsp;&nbsp;&nbsp; 4236&nbsp;&nbsp;&nbsp; 5684&nbsp;&nbsp;&nbsp; 3498&nbsp;&nbsp;&nbsp;&nbsp; 
852&nbsp;&nbsp;&nbsp; 4484</TT> 
<BR><TT>&nbsp;15&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
4393&nbsp;&nbsp;&nbsp; 4289&nbsp;&nbsp;&nbsp; 4347&nbsp;&nbsp;&nbsp; 3095&nbsp;&nbsp;&nbsp; 
4316&nbsp;&nbsp;&nbsp; 6200</TT> 
<BR><TT>&nbsp;16&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
779&nbsp;&nbsp;&nbsp; 4068&nbsp;&nbsp;&nbsp; 5680&nbsp;&nbsp;&nbsp; 5267&nbsp;&nbsp;&nbsp; 
4292&nbsp;&nbsp;&nbsp; 4481</TT> 
<BR><TT>&nbsp;17&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
746&nbsp;&nbsp;&nbsp;&nbsp; 713&nbsp;&nbsp;&nbsp;&nbsp; 703&nbsp;&nbsp;&nbsp; 
3391&nbsp;&nbsp;&nbsp; 4508&nbsp;&nbsp;&nbsp;&nbsp; 829</TT> 
<BR><TT>&nbsp;18&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
755&nbsp;&nbsp;&nbsp; 4053&nbsp;&nbsp;&nbsp; 5523&nbsp;&nbsp;&nbsp; 3508&nbsp;&nbsp;&nbsp;&nbsp; 
832&nbsp;&nbsp;&nbsp;&nbsp; 953</TT> 
<BR><TT>&nbsp;19&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
736&nbsp;&nbsp;&nbsp; 4058&nbsp;&nbsp;&nbsp; 4098&nbsp;&nbsp;&nbsp; 4773&nbsp;&nbsp;&nbsp;&nbsp; 
897&nbsp;&nbsp;&nbsp; 4596</TT> 
 
<P>The ADMEMs are still the same numbers in the same slots: 
 
<P>&nbsp;In Slot16&nbsp;&nbsp; IDPROM = 0099 003 ADMEM_V4.0&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
470 
<BR>&nbsp;In Slot17&nbsp;&nbsp; IDPROM = 0101 003 ADMEM_V4.0&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
470 
<BR>&nbsp;In Slot18&nbsp;&nbsp; IDPROM = 0058 003 ADMEM_V4.0&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
470 
<BR>&nbsp;In Slot19&nbsp;&nbsp; IDPROM = 0141 003 ADMEM_V4.0&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
470 
<BR>&nbsp;In Slot20&nbsp;&nbsp; IDPROM = 0145 003 ADMEM_V4.0&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
470 
<BR>&nbsp;In Slot21&nbsp;&nbsp; IDPROM = 0129 003 ADMEM_V4.0&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
470 
 
<P>ADMEM slot 21, QIE&nbsp;channel 8 returns zero pedestals and QIE&nbsp;calibrations 
when transition card is plugged in, but does both correctly when unplugged.&nbsp; 
Have verified short at transition card for this channel; will repair tomorrow. 
 
<P>ADMEM slot 19, QIE&nbsp;channel 0 shows zero QIE calibration but for 
I/2 range which is even sicker; with or without transition card.&nbsp; 
Fuses on this Cafe card also look good; will replace tomorrow. 
</BODY> 
</HTML>
<!-- Author: Steve Hahn, Rick Tesarek -->


<!-- Date: Mon Jan 3 18:32:59 2000 -->
The configuration of b0ccal14 is now also changed to boot from Flash RAM. 
Included in it are: kernel, smartsockets, RTIlib, FISION, ROBIN, rdate to set the 
on-board clock, flashProg for Flash RAM programming, vxhttpd (simple http demon), 
and cdfInit (a C version of the startup script). Please report any problems. 
Note that in automated scripts something like "moduleShow" will now lead to different 
results - I'd recommend using e.g. symFindByName().
<!-- Author: Arnd -->


<!-- Date: Tue Jan 4 16:44:57 2000 -->
<pre>Looked at the QIE 0 of the ADMEM 19. Fuses are OK. Tried a 
 swapping with QIE 18. Everything fine (ie correct results  
both in slots 0 and 18). Put it back in the old configuration. 
 The problem is still there (ie the slot 0 does not seem to see 
 any variation of the DAC, leading to 0 as slopes and offsets  
except for the range 0 where the slope is negative due to  
pedestals contribution). Tried a swapping between slots 0 and 1.  
Everything is fine. We suspect a board problem. Maybe a bad  
sudder (one connection seems strange). The board is now in its  
original configuration.</pre>
<!-- Author: R.Erbacher&Ph.Gris -->


<!-- Date: Tue Jan 4 19:01:17 2000 -->
<pre>Wanted to make some test with the "old" calcard. Put the  
 Testclock back in slot 3. Did not remove it (should not affect 
 anybody). Replugged the crate as it was</pre>
<!-- Author: Ph.Gris -->


<!-- Date: Thu Jan 6 18:34:53 2000 -->
Removed the hadron ASD transition board cables from the CANEB2 
(lowest wedges for NE arch).
<!-- Author: R.J. Tesarek, R. Erbacher -->


<!-- Date: Fri Jan 7 10:06:47 2000 -->
Tried to use main version of qietest in /data1/weihao.  Error given:  
-------------------------------------- 
0ccal14-> cd "/data1/weihao" 
value = 0 = 0x0 
b0ccal14-> ld < qietest 
Incorrect ELF header size: 2592 
ld error: error reading file (errno = 0x3d0001). 
value = 0 = 0x0 
-------------------------------------- 
Philippe says this has been the case since yesterday afternoon, so we are trying to use Philippe's personal (unofficial) version until we can find out the problem.  Rick and I think that there should always be a working "qietest" and that modifications should be made using a development version until such time as it works correctly, and then put into the main version.  This is necessary since several people rely on a functioning "qietest" to do work.  Perhaps there is something else wrong, and not a problem with "qietest" though?
<!-- Author: R. Erbacher and R. Tesarek -->
<!-- Comment: Mon Jan 10 09;58;43 comment by...R.J. Tesarek -->
Date: Mon, 10 Jan 2000 09:40:49 -0600 (CST)<BR>
From: WEIHAO@fnald<BR>
To: tesarek@fnal.gov<BR>
Cc: "ACDFT2::VAIA"@fnald<BR>
Subject: RE: Problems with your version of qietest<BR>
<P>


Hi,<BR>
  The problem is fixed.<BR>
haowei<BR>

<!-- Date: Fri Jan 7 16:07:04 2000 -->
Rick Tesarek was shocked by failing HV connector on CHA PMT for 20E 5L.  The center pin and shielding was pulling out of the barrel of the SHV connector.  After turning of all HV to NE arch, Rick cut off the center pin and wrapped it in electrical tape.  This will have to be repaired. 
<p> 
This is the most likely source of the shock, but we are not certain of this. 
<p> 
Missing bases in this region are CHA 20E 4L and CHA 19E 1R.  
<!-- Author: Steve Hahn, Rick Tesarek -->



<!-- Date: Mon Jan 10 18:49:01 2000 -->
Re-cable trigger inputs for central calorimeter crate b0ccal14.
<!-- Author: R. Erbacher, R.J. Tesarek -->


<!-- Date: Mon Jan 10 21:15:04 2000 -->
<HTML> 
<HEAD> 
   <META NAME="GENERATOR" CONTENT="Mozilla/4.05C-SGI [en] (X11; I; IRIX 6.5 IP32) [Netscape]"> 
</HEAD> 
<BODY> 
Performed tests to verify the performance of the hadron ASD transition 
board.&nbsp; For these tests 
<BR>only the dynodes for the hadronic sections of the central calorimeter 
in wedges 19R and 20L of the 
<BR>NE arch were connected to the hadron ASD transition board.<B></B> 
 
<P><B>Note: PMTs for 19R1 and 20L4 are missing and were not cabled...</B> 
 
<P>We measured peak-to-peak noise of approximately 4.5mV by eye into the 
hadron ASD transition 
<BR>boards.&nbsp; This is not significantly different than measured previously 
(see entry this log for 12/13/99 
<BR>by R.J. Tesarek and Ph. Gris). 
 
<P>In addition we measured the pedestal RMS for the ADMEMs operating in 
"passthrough" mode 
<BR>with and without the hadron PMT dynode cables installed (and with the 
cables plugged, but without 
<BR>the transition board plugged in).&nbsp;&nbsp;&nbsp; The data are located 
in the following files located on b0dau30: 
 
<P>~robine/pedrms01-05.vec&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; hadron ASD cables 
installed and plugged into VIPA crate. 
<BR>~robine/pedrms06-10.vec&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; hadron ASD cables 
installed but transition board not plugged into crate. 
<BR>~robine/pedrms11-15.vec&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; hadron ASD cables 
unplugged from PMTs and hadron ASD transition 
<BR>&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
board was unplugged from VIPA crate. 
 
<P>Analysis of the data taken this evening will follow in a subsequent 
entry. 
</BODY> 
</HTML>
<!-- Author: R. Erbacher, R.J. Tesarek -->


<!-- Date: Wed Jan 12 13:57:40 2000 -->
<pre>The testing program "qietest" is checked into the "fer" package. 
You should use the latest version at $FER_DIR/VxWorks-5.3/PPC 
after "setup fer devel". 
</pre>
<!-- Author: Wei Hao -->


<!-- Date: Thu Jan 13 18:36:31 2000 -->
Just a short entry...we tried QIEtest after powering b0ccal14 down and back up in preparation for bringing up b0ccal15.  Everything looks as before (including ADMEM slot 21, channel 8 still being bad only when transition card is plugged in).
<!-- Author: Steve Hahn, Rick Tesarek -->


<!-- Date: Fri Jan 14 09:53:16 2000 -->
New, Production PEI power supply installed into top crate of the lower NE arch <BR> rack. For low voltage monitoring purposes, 
the hardware addresses are: <BR> 
<BOLD> Analog address: 106 </BOLD><BR> 
<BOLD> Digital address: 107 </BOLD><BR>  
<!-- Author: Fred Lewis, Roberto -->


<!-- Date: Fri Jan 14 10:33:41 2000 -->
First attempt to calibrate ADMEMs in slots 18, 19, 20, and 21 (cards for slots 16 and 17 being used to align CANWB-U crate P3 backplane) in b0ccal15. 
<p> 
Results of error summary for first try are: 
<pre> 
*******************   Start of WARNING messages   ************************* 
* ADMEM=18 QIE= 9 Warning 11: Sigma(offset) big              (    1.5399) * 
* ADMEM=18 QIE= 9 Warning  8: Sigma(slope)/slope big         (    0.0031) * 
*******************   End of WARNING messages   *************************** 
 
*******************   Start of FAILURE messages   ************************* 
* ADMEM=18 QIE= 3 Failure 14: Sigma(offset) too big          (    9.6255) * 
* ADMEM=19 QIE= 0 Failure 10: No fitting                     (    0.0000) * 
* ADMEM=19 QIE= 1 Failure 10: No fitting                     (    0.0000) * 
* ADMEM=19 QIE= 2 Failure 10: No fitting                     (    0.0000) * 
* ADMEM=20 QIE= 0 Failure 10: No fitting                     (    0.0000) * 
*******************   End of FAILURE messages   *************************** 
 
*********************************************** 
Total Warnings: 2.  
Total Failures: 5.  
*********************************************** 
</pre>  
<!-- Author: Robin E., Rick T., Steve H. -->


<!-- Date: Fri Jan 14 11:34:23 2000 -->
<HTML> 
<HEAD> 
   <META NAME="GENERATOR" CONTENT="Mozilla/4.05C-SGI [en] (X11; I; IRIX 6.5 IP32) [Netscape]"> 
</HEAD> 
<BODY> 
Tested ADMEMs in slots 16-21 on b0ccal15 after power supply was checked 
out to have the proper 
<BR>voltages.&nbsp; Found that the board in slot 18 had cafe' cards in 
channels 3 and 9 improperly seated. 
<BR>Fixed problem by re-seating these.&nbsp; Found that slot 19 had cafe' 
cards in channels 0,1,2 with blown F4 
<BR>fuses.&nbsp; In addition, the ADMEM in slot 20 was found to have the 
same problem in channel 0: 
<UL> 
<LI> 
ADMEM slot 19 channel 0 (CAFE ID=5622)</LI> 
 
<LI> 
ADMEM slot 19 channel 1 (CAFE ID=5728)</LI> 
 
<LI> 
ADMEM slot 19 channel 2 (CAFE ID= 5658)</LI> 
 
<LI> 
ADMEM slot 20 channel 0 (CAFE ID= 5678)</LI> 
</UL> 
Will replace these four cards today. 
</BODY> 
</HTML>
<!-- Author: R.J. Tesarek, R. Erbacher -->


<!-- Date: Fri Jan 14 12:58:39 2000 -->
Finished working with b0ccal14 for today.  Had good sucess testing the Calcard with Xenon Flasher.  Note that Xenon box 
for Wedge 21 is missing and for Wedge 20 has a problem with 
the PIN diode readout.  Wedges 22 and 19 seem to work pretty 
well. 
  
<!-- Author: ed mckigney -->


<!-- Date: Fri Jan 14 17:43:24 2000 -->
<HTML> 
<HEAD> 
   <META NAME="GENERATOR" CONTENT="Mozilla/4.05C-SGI [en] (X11; I; IRIX 6.5 IP32) [Netscape]"> 
</HEAD> 
<BODY> 
<B>Crate: b0ccal15</B> 
<BR>The CAFE cards identified earlier today in ADMEMS in slots 19,20 were 
replace and they 
<BR>appear to check out fine.&nbsp; Mapping b0ccal15 now yields the following 
information: 
 
<P>&nbsp;In Slot16&nbsp;&nbsp; IDPROM = ..................{...........{. 
<BR>&nbsp;In Slot17&nbsp;&nbsp; IDPROM = ..................{...........{. 
<BR>&nbsp;In Slot18&nbsp;&nbsp; IDPROM = 0025 003 ADMEM_V4.0&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
470 
<BR>&nbsp;In Slot19&nbsp;&nbsp; IDPROM = 0027 003 ADMEM_V4.0&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
470 
<BR>&nbsp;In Slot20&nbsp;&nbsp; IDPROM = 0028 003 ADMEM_V4.0&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
470 
<BR>&nbsp;In Slot21&nbsp;&nbsp; IDPROM = 0034 003 ADMEM_V4.0&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
470 
 
<P>QIE ID table: 
<BR>QIE&nbsp;&nbsp;&nbsp; ADMEM: 16&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 17&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
18&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 19&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 20&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
21 
<BR>&nbsp; 0&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5530&nbsp;&nbsp;&nbsp; 
5517&nbsp;&nbsp;&nbsp; 5683&nbsp;&nbsp;&nbsp; 3198&nbsp;&nbsp;&nbsp; 3302&nbsp;&nbsp;&nbsp; 
5699 
<BR>&nbsp; 1&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5497&nbsp;&nbsp;&nbsp; 
5493&nbsp;&nbsp;&nbsp; 5611&nbsp;&nbsp;&nbsp; 5214&nbsp;&nbsp;&nbsp; 5593&nbsp;&nbsp;&nbsp; 
5757 
<BR>&nbsp; 2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5520&nbsp;&nbsp;&nbsp; 
5544&nbsp;&nbsp;&nbsp; 5734&nbsp;&nbsp;&nbsp; 3242&nbsp;&nbsp;&nbsp; 5597&nbsp;&nbsp;&nbsp; 
5635 
<BR>&nbsp; 3&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5481&nbsp;&nbsp;&nbsp; 
5570&nbsp;&nbsp;&nbsp; 5645&nbsp;&nbsp;&nbsp; 5480&nbsp;&nbsp;&nbsp; 5760&nbsp;&nbsp;&nbsp; 
5564 
<BR>&nbsp; 4&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5508&nbsp;&nbsp;&nbsp; 
5496&nbsp;&nbsp;&nbsp; 5708&nbsp;&nbsp;&nbsp; 5550&nbsp;&nbsp;&nbsp; 5669&nbsp;&nbsp;&nbsp; 
5562 
<BR>&nbsp; 5&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5541&nbsp;&nbsp;&nbsp; 
5507&nbsp;&nbsp;&nbsp; 5630&nbsp;&nbsp;&nbsp; 5557&nbsp;&nbsp;&nbsp; 5741&nbsp;&nbsp;&nbsp; 
5629 
<BR>&nbsp; 6&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5516&nbsp;&nbsp;&nbsp; 
5509&nbsp;&nbsp;&nbsp; 5625&nbsp;&nbsp;&nbsp; 5502&nbsp;&nbsp;&nbsp; 5720&nbsp;&nbsp;&nbsp; 
5681 
<BR>&nbsp; 7&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5556&nbsp;&nbsp;&nbsp; 
5553&nbsp;&nbsp;&nbsp; 5676&nbsp;&nbsp;&nbsp; 5651&nbsp;&nbsp;&nbsp; 5746&nbsp;&nbsp;&nbsp; 
5647 
<BR>&nbsp; 8&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5484&nbsp;&nbsp;&nbsp; 
5531&nbsp;&nbsp;&nbsp; 5552&nbsp;&nbsp;&nbsp; 5653&nbsp;&nbsp;&nbsp; 5724&nbsp;&nbsp;&nbsp; 
5657 
<BR>&nbsp; 9&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5566&nbsp;&nbsp;&nbsp; 
5486&nbsp;&nbsp;&nbsp; 5672&nbsp;&nbsp;&nbsp; 5599&nbsp;&nbsp;&nbsp; 5571&nbsp;&nbsp;&nbsp; 
5759 
<BR>&nbsp;10&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5494&nbsp;&nbsp;&nbsp; 
5506&nbsp;&nbsp;&nbsp; 5642&nbsp;&nbsp;&nbsp; 5679&nbsp;&nbsp;&nbsp; 5646&nbsp;&nbsp;&nbsp; 
5688 
<BR>&nbsp;11&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5482&nbsp;&nbsp;&nbsp; 
5555&nbsp;&nbsp;&nbsp; 5701&nbsp;&nbsp;&nbsp; 5732&nbsp;&nbsp;&nbsp; 5705&nbsp;&nbsp;&nbsp; 
5609 
<BR>&nbsp;12&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5663&nbsp;&nbsp;&nbsp; 
5528&nbsp;&nbsp;&nbsp; 5756&nbsp;&nbsp;&nbsp; 5620&nbsp;&nbsp;&nbsp; 5573&nbsp;&nbsp;&nbsp; 
5714 
<BR>&nbsp;13&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5537&nbsp;&nbsp;&nbsp; 
5569&nbsp;&nbsp;&nbsp; 5572&nbsp;&nbsp;&nbsp; 5725&nbsp;&nbsp;&nbsp; 5643&nbsp;&nbsp;&nbsp; 
5608 
<BR>&nbsp;14&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5618&nbsp;&nbsp;&nbsp; 
5522&nbsp;&nbsp;&nbsp; 5709&nbsp;&nbsp;&nbsp; 5535&nbsp;&nbsp;&nbsp; 5626&nbsp;&nbsp;&nbsp; 
5586 
<BR>&nbsp;15&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5532&nbsp;&nbsp;&nbsp; 
5558&nbsp;&nbsp;&nbsp; 5561&nbsp;&nbsp;&nbsp; 5584&nbsp;&nbsp;&nbsp; 5718&nbsp;&nbsp;&nbsp; 
5524 
<BR>&nbsp;16&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5495&nbsp;&nbsp;&nbsp; 
5521&nbsp;&nbsp;&nbsp; 5514&nbsp;&nbsp;&nbsp; 5616&nbsp;&nbsp;&nbsp; 5758&nbsp;&nbsp;&nbsp; 
5623 
<BR>&nbsp;17&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5546&nbsp;&nbsp;&nbsp; 
5567&nbsp;&nbsp;&nbsp; 5539&nbsp;&nbsp;&nbsp; 5751&nbsp;&nbsp;&nbsp; 5692&nbsp;&nbsp;&nbsp; 
5754 
<BR>&nbsp;18&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5543&nbsp;&nbsp;&nbsp; 
5533&nbsp;&nbsp;&nbsp; 5490&nbsp;&nbsp;&nbsp; 5693&nbsp;&nbsp;&nbsp; 5723&nbsp;&nbsp;&nbsp; 
5703 
<BR>&nbsp;19&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5492&nbsp;&nbsp;&nbsp; 
5483&nbsp;&nbsp;&nbsp; 5595&nbsp;&nbsp;&nbsp; 5691&nbsp;&nbsp;&nbsp; 5527&nbsp;&nbsp;&nbsp; 
5590 
 
<P>Note that the ADMEMS in slots 16,17 do not read back their PROMID and 
ADMEM version 
<BR>numbers.&nbsp; Steve Hahn claims that the first few ADMEMs he received 
did not have their 
<BR>PROMs burned with this information.&nbsp; These ADMEMs should be replace 
on Monday (1/17). 
 
<P>Otherwise, everything appears to calibrate correctly with and without 
the transition boards 
<BR>plugged in. 
<BR>&nbsp; 
 
<P>Installed trigger cables for all ADMEMs in b0ccal15.  Maybe 
they're even done correctly. 
</BODY> 
</HTML>
<!-- Author: R.J. Tesarek, R. Erbacher -->


<!-- Date: Fri Jan 14 18:03:02 2000 -->
<HTML> 
<HEAD> 
   <META NAME="GENERATOR" CONTENT="Mozilla/4.05C-SGI [en] (X11; I; IRIX 6.5 IP32) [Netscape]"> 
</HEAD> 
<BODY> 
Checked some of NW arch PMT's.&nbsp; Found possible ringing problem in 
the following PMT's: 
<BR>Wedge&nbsp;&nbsp;&nbsp; Tower&nbsp;&nbsp;&nbsp; L/R&nbsp;&nbsp;&nbsp; 
H/EM 
<BR>20&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
6&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; L&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
H 
<BR>20&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
5&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; L&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
H 
<BR>20&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
7&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; L&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
H 
</BODY> 
</HTML>
<!-- Author: Olga Lobban -->


<!-- Date: Mon Jan 17 08:56:24 2000 -->
<HTML> 
<HEAD> 
   <META NAME="GENERATOR" CONTENT="Mozilla/4.05C-SGI [en] (X11; I; IRIX 6.5 IP32) [Netscape]"> 
</HEAD> 
<BODY> 
<B>crate b0ccal15:</B> 
<BR>Replaced ADMEMs occupying slots 16 and 17 in b0ccal15 with modules 
who know who they 
<BR>are (IDPROMS properly loaded).&nbsp; Below is the new mapping of the 
crate.&nbsp; I've also managed 
<BR>to set the new ADMEMs to the central crate trigger configuration 
 
<P>&nbsp;In Slot16&nbsp;&nbsp; IDPROM = 0263 003 ADMEM_V4.0&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
470 
<BR>&nbsp;In Slot17&nbsp;&nbsp; IDPROM = 0259 003 ADMEM_V4.0&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
470 
<BR>&nbsp;In Slot18&nbsp;&nbsp; IDPROM = 0025 003 ADMEM_V4.0&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
470 
<BR>&nbsp;In Slot19&nbsp;&nbsp; IDPROM = 0027 003 ADMEM_V4.0&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
470 
<BR>&nbsp;In Slot20&nbsp;&nbsp; IDPROM = 0028 003 ADMEM_V4.0&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
470 
<BR>&nbsp;In Slot21&nbsp;&nbsp; IDPROM = 0034 003 ADMEM_V4.0&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
470 
<BR>&nbsp; 
<BR>&nbsp; 
 
<P>QIE ID table: 
<BR>QIE&nbsp;&nbsp;&nbsp; ADMEM: 16&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 17&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
18&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 19&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 20&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 
21 
<BR>&nbsp; 0&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5815&nbsp;&nbsp;&nbsp; 
1882&nbsp;&nbsp;&nbsp; 5683&nbsp;&nbsp;&nbsp; 3198&nbsp;&nbsp;&nbsp; 3302&nbsp;&nbsp;&nbsp; 
5699 
<BR>&nbsp; 1&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5858&nbsp;&nbsp;&nbsp; 
5994&nbsp;&nbsp;&nbsp; 5611&nbsp;&nbsp;&nbsp; 5214&nbsp;&nbsp;&nbsp; 5593&nbsp;&nbsp;&nbsp; 
5757 
<BR>&nbsp; 2&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5063&nbsp;&nbsp;&nbsp; 
4205&nbsp;&nbsp;&nbsp; 5734&nbsp;&nbsp;&nbsp; 3242&nbsp;&nbsp;&nbsp; 5597&nbsp;&nbsp;&nbsp; 
5635 
<BR>&nbsp; 3&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 6059&nbsp;&nbsp;&nbsp; 
1881&nbsp;&nbsp;&nbsp; 5645&nbsp;&nbsp;&nbsp; 5480&nbsp;&nbsp;&nbsp; 5760&nbsp;&nbsp;&nbsp; 
5564 
<BR>&nbsp; 4&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 1642&nbsp;&nbsp;&nbsp; 
2051&nbsp;&nbsp;&nbsp; 5708&nbsp;&nbsp;&nbsp; 5550&nbsp;&nbsp;&nbsp; 5669&nbsp;&nbsp;&nbsp; 
5562 
<BR>&nbsp; 5&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 4184&nbsp;&nbsp;&nbsp; 
5379&nbsp;&nbsp;&nbsp; 5630&nbsp;&nbsp;&nbsp; 5557&nbsp;&nbsp;&nbsp; 5741&nbsp;&nbsp;&nbsp; 
5629 
<BR>&nbsp; 6&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5968&nbsp;&nbsp;&nbsp; 
1887&nbsp;&nbsp;&nbsp; 5625&nbsp;&nbsp;&nbsp; 5502&nbsp;&nbsp;&nbsp; 5720&nbsp;&nbsp;&nbsp; 
5681 
<BR>&nbsp; 7&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 2004&nbsp;&nbsp;&nbsp; 
5885&nbsp;&nbsp;&nbsp; 5676&nbsp;&nbsp;&nbsp; 5651&nbsp;&nbsp;&nbsp; 5746&nbsp;&nbsp;&nbsp; 
5647 
<BR>&nbsp; 8&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5924&nbsp;&nbsp;&nbsp; 
5972&nbsp;&nbsp;&nbsp; 5552&nbsp;&nbsp;&nbsp; 5653&nbsp;&nbsp;&nbsp; 5724&nbsp;&nbsp;&nbsp; 
5657 
<BR>&nbsp; 9&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 4383&nbsp;&nbsp;&nbsp; 
2069&nbsp;&nbsp;&nbsp; 5672&nbsp;&nbsp;&nbsp; 5599&nbsp;&nbsp;&nbsp; 5571&nbsp;&nbsp;&nbsp; 
5759 
<BR>&nbsp;10&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 1814&nbsp;&nbsp;&nbsp; 
5027&nbsp;&nbsp;&nbsp; 5642&nbsp;&nbsp;&nbsp; 5679&nbsp;&nbsp;&nbsp; 5646&nbsp;&nbsp;&nbsp; 
5688 
<BR>&nbsp;11&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 4215&nbsp;&nbsp;&nbsp; 
5992&nbsp;&nbsp;&nbsp; 5701&nbsp;&nbsp;&nbsp; 5732&nbsp;&nbsp;&nbsp; 5705&nbsp;&nbsp;&nbsp; 
5609 
<BR>&nbsp;12&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 2116&nbsp;&nbsp;&nbsp; 
6085&nbsp;&nbsp;&nbsp; 5756&nbsp;&nbsp;&nbsp; 5620&nbsp;&nbsp;&nbsp; 5573&nbsp;&nbsp;&nbsp; 
5714 
<BR>&nbsp;13&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 6017&nbsp;&nbsp;&nbsp; 
6045&nbsp;&nbsp;&nbsp; 5572&nbsp;&nbsp;&nbsp; 5725&nbsp;&nbsp;&nbsp; 5643&nbsp;&nbsp;&nbsp; 
5608 
<BR>&nbsp;14&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 1856&nbsp;&nbsp;&nbsp; 
1748&nbsp;&nbsp;&nbsp; 5709&nbsp;&nbsp;&nbsp; 5535&nbsp;&nbsp;&nbsp; 5626&nbsp;&nbsp;&nbsp; 
5586 
<BR>&nbsp;15&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 1875&nbsp;&nbsp;&nbsp; 
6016&nbsp;&nbsp;&nbsp; 5561&nbsp;&nbsp;&nbsp; 5584&nbsp;&nbsp;&nbsp; 5718&nbsp;&nbsp;&nbsp; 
5524 
<BR>&nbsp;16&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 1750&nbsp;&nbsp;&nbsp; 
6475&nbsp;&nbsp;&nbsp; 5514&nbsp;&nbsp;&nbsp; 5616&nbsp;&nbsp;&nbsp; 5758&nbsp;&nbsp;&nbsp; 
5623 
<BR>&nbsp;17&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 1915&nbsp;&nbsp;&nbsp; 
6038&nbsp;&nbsp;&nbsp; 5539&nbsp;&nbsp;&nbsp; 5751&nbsp;&nbsp;&nbsp; 5692&nbsp;&nbsp;&nbsp; 
5754 
<BR>&nbsp;18&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 4045&nbsp;&nbsp;&nbsp; 
1806&nbsp;&nbsp;&nbsp; 5490&nbsp;&nbsp;&nbsp; 5693&nbsp;&nbsp;&nbsp; 5723&nbsp;&nbsp;&nbsp; 
5703 
<BR>&nbsp;19&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp;&nbsp; 5928&nbsp;&nbsp;&nbsp; 
5954&nbsp;&nbsp;&nbsp; 5595&nbsp;&nbsp;&nbsp; 5691&nbsp;&nbsp;&nbsp; 5527&nbsp;&nbsp;&nbsp; 
5590 
<BR>&nbsp; 
<BR> 
</BODY> 
</HTML>
<!-- Author: R.J. Tesarek -->


<!-- Date: Thu Jan 20 10:20:13 2000 -->
<pre>  After "setup fer devel", you can use: 
 
qietest b0ccal14 load    => to load qietest in b0ccal14. 
qietest b0ccal14         => to run automatic AMMEM checkout 
qietest b0ccal14 calib   => do calibration and load FRAM. 
qietest b0ccal14 cem     => configure the ADMEMs as CEM board. 
...... 
 
  You can just type "qietest" to see what else you can do. 
</pre>
<!-- Author: wei hao -->