<!-- Title: Day shift. Tue Feb 12, 2008 -->
<!-- SciCo: Kevin McFarland -->
<!-- DAQAce: Martin Frank -->
<!-- MonAce: &nbsp; -->
<!-- CO: Gianluca de Lorenzo -->
<!-- OpManager: Massimo Casarsa -->
<!-- Notes: -->
<pre>inherit smooth running</pre>


<!-- Date: Tue Feb 12 08:03:47.972 2008 -->
CER_SVXMON error.  HRR.  Everything ok again.
<!-- Author: Martin :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257858' target=_top>257858</a>) -->


<!-- Date: Tue Feb 12 08:04:57.958 2008 -->
L2 Decision TO (waiting for L2CAL 3).  HRR.  Everything ok again.
<!-- Author: Martin :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257858' target=_top>257858</a>) -->
<!-- Comment: Tue Feb 12 09;27;16 comment by...Martin -->
Again (waiting for RECES).
<!-- Comment: Tue Feb 12 09;33;10 comment by...Martin -->
Again (waiting for RECES).
<!-- Date: Tue Feb 12 08:16:55.472 2008 -->
Bob Wagner calls with a gentle reminder that the XFT group should unmask the wires affected by yesterday's TDC swap between stores.
<!-- Author: KSM :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257858' target=_top>257858</a>) -->


<!-- Date: Tue Feb 12 08:25:58.601 2008 -->
I am still getting the connection error to HTRDATA in iFix.  Is that still being worked on?
<!-- Author: Martin :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257858' target=_top>257858</a>) -->


<!-- Date: Tue Feb 12 09:30:03.792 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257858>257858</A>
general: <tt>good run.</tt>
<!-- Author: Consumer Operator( gianluca) -->




<!-- Date: Tue Feb 12 09:49:38.591 2008 -->
The dead cells in the COT have been fixed. I'm updating the deadwire list. I'll when I'm ready to restart the run.
<!-- Author: Jason :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257858' target=_top>257858</a>) -->
<!-- Comment: Tue Feb 12 09;59;03 comment by...Jason -->
All set! The deadwire list has been updated. I unmasked the wires in SL3 Cells 104-111. The changes will be picked up when you restart the run. 
<!-- Date: Tue Feb 12 09:58:30.298 2008 -->
Jason calls.  New XFT masks are in place to be loaded at the next run start. <p>We will start a new run when Laura is ready to test the new PreFred board.
<!-- Author: KSM :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257858' target=_top>257858</a>) -->


<!-- Date: Tue Feb 12 10:19:50.350 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257858>257858</A>
 Terminated at 2008.02.12 10:19:32
<!-- Author: RunControl -->


<!-- Date: Tue Feb 12 10:20:17.430 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257858>257858</A>
TERMINATE: swapping PreFred board in slot 10
<!-- Author: Martin -->



<!-- Date: Tue Feb 12 10:28:02.082 2008 -->
A proton only store to the end of the low beta squeeze has been added to the between stores plan. Note that Salah had planned to be in shot setup by noon.
<!-- Author: Phil -->

<!-- Date: Tue Feb 12 10:35:34.006 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257866>257866</A>
 Activated at 2008.02.12 10:35:19
<!-- Author: RunControl -->


<!-- Date: Tue Feb 12 10:35:35.357 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257866>257866</A>
ACTIVATE: RunSet: AAA_CURRENT  Trigger: PHYSICS_5_01 [4,1032,717] 
<!-- Author: Martin -->
<!-- Comment: Tue Feb 12 10;36;55 comment by...KSM -->
should have new XFT masks; have asked Gianluca (CO) to check out COT and XFT plots ASAP.  


<!-- Date: Tue Feb 12 10:44:07.277 2008 -->
<pre>We are testing new fermware for a prefred board (b0l1gl00, slot 10) in preparation fro the new L1 Met trigger. The new trigger bits will replaced the old GAP triggers that are not used anymore.</pre>
<!-- Author: Laura, carla :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257866' target=_top>257866</a>) -->


<!-- Date: Tue Feb 12 10:46:31.778 2008 -->
<center><a href='/cgi/elog/elog.pl?nb=2008&action=view&page=-885&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2008&action=view&page=-885' width='300'></a><a href='/cgi/elog/elog.pl?nb=2008&action=view&page=-886&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2008&action=view&page=-886' width='300'></a><a href='/cgi/elog/elog.pl?nb=2008&action=view&page=-887&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2008&action=view&page=-887' width='300'></a></center>
<!-- Author: gianluca -->
<!-- Comment: Tue Feb 12 10;47;36 comment by...gianluca -->
COT and XFT.  Still low statistic.  Nothing strange up to now.  I'll keep an eye on both.

<!-- Date: Tue Feb 12 11:10:15.659 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257858>257858</A>
general: <tt>last check for this run: everything's good.</tt>
<!-- Author: Consumer Operator( gianluca) -->



<!-- Date: Tue Feb 12 11:15:14.996 2008 -->
I killed and restarted the plug HV monitoring in preparation for the next store.  I did this while we were still taking data.  Is that ok?
<!-- Author: Martin :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257866' target=_top>257866</a>) -->


<!-- Date: Tue Feb 12 11:16:24.138 2008 -->
MCR calls with 15 minutes end of store warning
<!-- Author: KSM :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257866' target=_top>257866</a>) -->


<!-- Date: Tue Feb 12 11:18:33.026 2008 -->
<center><a href='/cgi/elog/elog.pl?nb=2008&action=view&page=-891&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2008&action=view&page=-891' width='300'></a><a href='/cgi/elog/elog.pl?nb=2008&action=view&page=-890&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2008&action=view&page=-890' width='300'></a><a href='/cgi/elog/elog.pl?nb=2008&action=view&page=-889&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2008&action=view&page=-889' width='300'></a><a href='/cgi/elog/elog.pl?nb=2008&action=view&page=-888&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2008&action=view&page=-888' width='300'></a></center>
<!-- Author: gianluca -->
<!-- Comment: Tue Feb 12 11;18;54 comment by...gianluca -->
XFT looks all right.


<!-- Date: Tue Feb 12 11:20:49.081 2008 -->
<center><a href='/cgi/elog/elog.pl?nb=2008&action=view&page=-892&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2008&action=view&page=-892' width='300'></a><a href='/cgi/elog/elog.pl?nb=2008&action=view&page=-893&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2008&action=view&page=-893' width='300'></a></center>
<!-- Author: gianluca -->
<!-- Comment: Tue Feb 12 11;21;12 comment by...gianluca -->
COT also looks OK.

<!-- Date: Tue Feb 12 11:22:09.651 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257866>257866</A>
 Terminated at 2008.02.12 11:21:35
<!-- Author: RunControl -->


<!-- Date: Tue Feb 12 11:22:10.235 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257866>257866</A>
TERMINATE: end of store
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 11:32:14.902 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257876>257876</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 11:33:22.799 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257876>257876</A>
TERMINATE: Done.
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 11:33:42.942 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257876>257876</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Tue Feb 12 11;45;10 comment by...gianluca -->
ok.
<!-- Date: Tue Feb 12 11:34:30.050 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257877>257877</A>
ACTIVATE: RunSet: CALIB_CLC_QIE 
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 11:35:40.277 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257877>257877</A>
TERMINATE: Done.
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 11:35:41.863 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257877>257877</A>
CALIBRATION: <U><B>clcqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#clcqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Tue Feb 12 11;46;08 comment by...gianluca -->
ok.
<!-- Date: Tue Feb 12 11:38:22.801 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257878>257878</A>
ACTIVATE: RunSet: CALIB_QIE_FWD 
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 11:40:29.635 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257878>257878</A>
TERMINATE: Done.
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 11:40:31.452 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257866>257866</A>
general: <tt>last check on this run: everything's good.</tt>
<!-- Author: Consumer Operator( gianluca) -->


<!-- Date: Tue Feb 12 11:42:12.085 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257879>257879</A>
 Activated at 2008.02.12 11:40:25
<!-- Author: RunControl -->


<!-- Date: Tue Feb 12 11:42:13.722 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257878>257878</A>
CALIBRATION: <U><B>bscqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#bscqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Tue Feb 12 11;53;02 comment by...gianluca -->
few hot channel with rms sig between 5 and 10.
CHA 13Wch6 has a rms=12.59.
see image below
<!-- Comment: Tue Feb 12 11;55;20 comment by...gianluca -->
sorry, I commented the wrong calib!!   I was reffering to the calqie calibration...
<!-- Date: Tue Feb 12 11:43:13.898 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257879>257879</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [7,1023,706] 
<!-- Author: Martin -->




<!-- Date: Tue Feb 12 11:52:18.379 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257879>257879</A>
 Terminated at 2008.02.12 11:51:49
<!-- Author: RunControl -->


<!-- Date: Tue Feb 12 11:52:19.243 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257879>257879</A>
TERMINATE: Done.  No Beam during entire run.
<!-- Author: Martin -->
<!-- Comment: Tue Feb 12 11;52;59 comment by...Martin -->
Average L1A for run = 44.0 Hz



<!-- Comment: Tue Feb 12 13;17;28 comment by...Martin -->
No spikes in YMon Calorimeter Occupancy Plots.
<!-- Date: Tue Feb 12 11:53:33.785 2008 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2008&action=view&page=-894&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-894  width='300' align='top'></a></td><td>badpeds list CALQIE</td></tr></table>
<!-- Author: gianluca -->
<!-- Comment: Tue Feb 12 15;09;17 comment by...gianluca -->
Mailed Mark Mattson about the CHA 13W ch6.  He answered data should still be good, but we should take an eye on that channel to see if it gets worse.
<!-- Date: Tue Feb 12 11:55:05.189 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257883>257883</A>
ACTIVATE: RunSet: ACE_TOF_QIE_NEW 
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 11:55:20.115 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257883>257883</A>
TERMINATE: Done.
<!-- Author: Martin -->



<!-- Date: Tue Feb 12 11:55:30.985 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257883>257883</A>
CALIBRATION: <U><B>tofqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#tofqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->


<!-- Date: Tue Feb 12 11:57:19.348 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257884>257884</A>
ACTIVATE: RunSet: ACE_TOF_QIE_NEW 
<!-- Author: Martin -->
<!-- Comment: Tue Feb 12 12;23;28 comment by...Martin -->
This was supposed to be a SMX QIE calibration, but I forgot to change the run setting.  Please see Run 257891 for the proper TOF QIE calibration.
<!-- Date: Tue Feb 12 11:57:51.074 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257884>257884</A>
CALIBRATION: <U><B>tofqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#tofqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->


<!-- Date: Tue Feb 12 11:58:39.577 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257884>257884</A>
TERMINATE: Done.
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 11:59:47.473 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257885>257885</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 12:01:15.562 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257885>257885</A>
TERMINATE: Done.
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 12:01:16.034 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257885>257885</A>
CALIBRATION: <U><B>cemled</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cemled target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Tue Feb 12 12;46;08 comment by...gianluca -->
ok.
<!-- Date: Tue Feb 12 12:03:08.695 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257886>257886</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 12:03:09.838 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257886>257886</A>
TERMINATE: Done.
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 12:03:10.288 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257886>257886</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Tue Feb 12 12;47;25 comment by...gianluca -->
ok.
<!-- Date: Tue Feb 12 12:04:13.318 2008 -->
Hockerized Silicon Power Supply Crate #15. Recovered ISL ladder IB3W2L2 which was tripping yeterday on Bias1 over-voltage. Cycled this ladder STBY->ON->STBY 2 times. It is fine now. Set it online in HDWDB.
<!-- Author: SashaS -->

<!-- Date: Tue Feb 12 12:06:51.060 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257887>257887</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 12:07:14.509 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257887>257887</A>
TERMINATE: Done.
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 12:07:33.848 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257887>257887</A>
CALIBRATION: <U><B>calxef</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calxef target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Tue Feb 12 12;49;17 comment by...gianluca -->
ok.
<!-- Date: Tue Feb 12 12:11:02.241 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257888>257888</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 12:11:44.677 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257888>257888</A>
TERMINATE: bocmp00 had a Done TO, so I will redo this calibration
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 12:12:54.320 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257889>257889</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Martin -->
<!-- Comment: Tue Feb 12 12;16;15 comment by...Martin -->
Done.
<!-- Date: Tue Feb 12 12:15:14.705 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257890>257890</A>
ACTIVATE: RunSet: TOF_TAC_CALIB 
<!-- Author: Martin -->




<!-- Date: Tue Feb 12 12:16:56.627 2008 -->
MCR called: they're injecting proton beam.
<!-- Author: gianluca :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257890' target=_top>257890</a>) -->

<!-- Date: Tue Feb 12 12:21:53.062 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257890>257890</A>
TERMINATE: Done.
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 12:21:54.399 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257890>257890</A>
CALIBRATION: <U><B>toftac</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#toftac target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Tue Feb 12 12;50;55 comment by...gianluca -->
ok.
<!-- Date: Tue Feb 12 12:23:14.660 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257891>257891</A>
ACTIVATE: RunSet: ACE_TOF_QIE_NEW 
<!-- Author: Martin -->



<!-- Date: Tue Feb 12 12:24:08.683 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257891>257891</A>
CALIBRATION: <U><B>tofqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#tofqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Tue Feb 12 12;51;32 comment by...gianluca -->
ok.
<!-- Date: Tue Feb 12 12:24:09.899 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257891>257891</A>
TERMINATE: Done.
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 12:25:27.979 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257892>257892</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 12:26:30.519 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257892>257892</A>
TERMINATE: I am not sure whether CES HV was completely off when I started the calibration, so I will do it again.
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 12:27:29.471 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257893>257893</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 12:28:45.214 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257893>257893</A>
TERMINATE: Done.
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 12:31:34.235 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257892>257892</A>
CALIBRATION: <U><B>smx</B></U> calibration FAILED, please investigate
<!-- Author: CDFCALIB -->


<!-- Date: Tue Feb 12 12:36:19.799 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257894>257894</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [19,1021,705] 
<!-- Author: Martin -->







<!-- Date: Tue Feb 12 12:54:24.440 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257894>257894</A>
TERMINATE: redo smx calibrations
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 12:57:31.639 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257895>257895</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 13:00:46.484 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257895>257895</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Tue Feb 12 13;17;32 comment by...gianluca -->
ok.
<!-- Date: Tue Feb 12 13:01:44.816 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257895>257895</A>
TERMINATE: Done.
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 13:04:42.102 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257896>257896</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [19,1021,705] 
<!-- Author: Martin -->




<!-- Date: Tue Feb 12 13:19:40.107 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257858>257858</A>
<HTML>Run <A HREF=http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN=257858 target=_runSummary>257858</A> has not yet been checked for goodrun status; please do so.</HTML>
<!-- Author: RunControl -->


<!-- Date: Tue Feb 12 14:02:57.761 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257896>257896</A>
TERMINATE: Dr. Badgett will reboot a few crates.
<!-- Author: Martin -->



<!-- Date: Tue Feb 12 14:05:58.440 2008 -->
<P> 
Rebooted all the b0pcal crates b0pcal00...b0pcal11;  they  
all pick up the non-optimized version of the readout  
code now.   If one crate is already running this version,  
they all might as well be, and then we have a better chance  
of catching a code crash. 
<P>
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257896' target=_top>257896</a>) -->

<!-- Date: Tue Feb 12 14:09:13.877 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257897>257897</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [19,1021,705] 
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 14:17:36.903 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257897>257897</A>
TERMINATE: Donatella wants to test a new version of ScalerMon
<!-- Author: Martin -->
<!-- Comment: Tue Feb 12 15;21;00 comment by...Donatella T. -->
the "DPS disabled" warning window popped-up correctly but the main GUI did not update accordingly. So it needs a bit more work...
<!-- Date: Tue Feb 12 14:21:02.719 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257898>257898</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_5_01 [4,1032,717] 
<!-- Author: Martin -->


<!-- Date: Tue Feb 12 14:23:08.777 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257866>257866</A>
<HTML>Run <A HREF=http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN=257866 target=_runSummary>257866</A> has not yet been checked for goodrun status; please do so.</HTML>
<!-- Author: RunControl -->



<!-- Date: Tue Feb 12 14:35:44.037 2008 -->
MCR calls: final protons to be loaded shortly
<!-- Author: KSM :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257898' target=_top>257898</a>) -->


<!-- Date: Tue Feb 12 14:56:04.728 2008 -->
b0cot16 is giving us a bunch of Done TOs.  We shephered twice successfully, but when we go back into the active state, we immediately get a Done TO.
<!-- Author: Britney/Martin :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257898' target=_top>257898</a>) -->
<!-- Comment: Tue Feb 12 14;59;02 comment by...KSM -->
paged TDC twice w/in past five minutes.  no answer yet.

<!-- Comment: Tue Feb 12 15;07;54 comment by...Britney/Martin -->
Bill stopped the run and ran a DAQTEST with b0cot16 alone and it worked fine.  We then started a new run (AAA_SHOTSETUP) and now everything is fine again.
<!-- Date: Tue Feb 12 15:04:52.597 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257905>257905</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_5_01 [4,1032,717] 
<!-- Author: Britney/Martin -->





<!-- Date: Tue Feb 12 15:09:28.828 2008 -->
<P> 
The b0cot16 crate was giving done timeouts and also pulling  
the crate error line (the hardware error line fed across  
the backplane and through the Tracer to the error crosspoint). 
There was no indication on the front-end console or in the  
error logger as to why this was happening. 
<P> 
I took a DAQTEST run with just b0cot16 and had no problems  
in the readout.     We went back to AAA_SHOTSETUP and  
the problem was gone.  The voo doo that you do... 
<P>  
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257905' target=_top>257905</a>) -->

<!-- Date: Tue Feb 12 15:11:22.928 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257768>257768</A>
RUNSTATUS: <PRE>Marked Bad, explanation:
CSMX 
</PRE><a href=
'http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2008&action=view&page=125&frame=2&anchor=11:22:57.354'>
CSMX power supply problem</a>
<!-- Author: cdfscico -->




<!-- Date: Tue Feb 12 15:21:00.906 2008 -->
injecting final protons
<!-- Author: KSM :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257905' target=_top>257905</a>) -->


<!-- Date: Tue Feb 12 15:25:13.411 2008 -->
Donatella told us to restart the Error Handler because the error log file was getting too big.
<!-- Author: Britney/Martin :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257905' target=_top>257905</a>) -->

<!-- Date: Tue Feb 12 15:30:45.091 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257905>257905</A>
TERMINATE: Bill will reboot some crates.
<!-- Author: Britney/Martin -->



<!-- Date: Tue Feb 12 15:30:45.656 2008 -->
b0cmx00 Done TO.  HRR.  Everything ok again.
<!-- Author: Britney/Martin :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257905' target=_top>257905</a>) -->


<!-- Date: Tue Feb 12 15:34:00.938 2008 -->
<P> 
Have rebooted the b0pcal00...b0pcal11 crates again to  
pick up a new code snippet that dumps the Messenger  
and Readout task status to a log file in the  
/cdf/onln/data/vxworks/fer directory when the CPU  
is rebooted. 
<P>
<!-- Author: badgett] :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257907' target=_top>257907</a>) -->

<!-- Date: Tue Feb 12 15:35:19.020 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257907>257907</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_5_01 [4,1032,717] 
<!-- Author: Britney/Martin -->



<!-- Date: Tue Feb 12 15:42:15.021 2008 -->
iFix is telling me that it cannot connect to CDFS6.
<!-- Author: Britney/Martin :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257907' target=_top>257907</a>) -->


<!-- Date: Tue Feb 12 15:53:00.552 2008 -->
l2ctrlb busy timeout. auto HRR. recovered
<!-- Author: chiara :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=257907' target=_top>257907</a>) -->

<!-- Date: Tue Feb 12 15:55:06.214 2008 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>PhysicsTable</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L3Accepts</TH>
<TH>DelivLumi nb<SUP>-1</SUP></TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN=257858 target=_runSummary>257858</A> x3EF42</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_5_01 [4,1032,717]</FONT></TD>
<TD ALIGN=CENTER>00:09:46</TD>
<TD ALIGN=CENTER>10:19:32</TD>
<TD ALIGN=CENTER>09:32:33</TD>
<TD ALIGN=RIGHT>   3,589,975</TD>   <TD ALIGN=RIGHT>  2,461.33</TD>
<TD ALIGN=RIGHT>  2,303.29</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN=257866 target=_runSummary>257866</A> x3EF4A</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_5_01 [4,1032,717]</FONT></TD>
<TD ALIGN=CENTER>10:35:19</TD>
<TD ALIGN=CENTER>11:21:35</TD>
<TD ALIGN=CENTER>00:42:22</TD>
<TD ALIGN=RIGHT>     213,707</TD>   <TD ALIGN=RIGHT>    130.46</TD>
<TD ALIGN=RIGHT>    117.41</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>15:55:01</TD>
<TD ALIGN=CENTER>10:14:55</TD>
<TD ALIGN=RIGHT>     3,803,682</TD> <TD ALIGN=RIGHT>  2,591.79</TD>
<TD ALIGN=RIGHT>  2,420.70</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Tue Feb 12 15:58:05.251 2008 -->
<font color=red>Shift Summary: </font><pre>Inherit store 5901, run 257858 
 
0816 Bob Wagner reminds us to have XFT masks updated after TDC replacement.  Will page XFT expert at 1000. 
 
1035 Start run 257866 to test new PreFred firmware and with new XFT masks (after TDC replacement) 
 
1130 Store 5901 is ended.  Move to quiet time calibrations, etc. 
 
1518 Injecting final protons for store 5903. 
1528 Aborted these protons.  Try again. 
1540 Aborted again.  When at first you don't succeed... 
</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   257858, 257866
Delivered Luminosity   593  
Acquired Luminosity    516  
Efficiency             87.0%
   
</td></tr></table></pre>
<!-- Author: KSM -->