<!-- Title: Eve shift. Tue Nov 13, 2001 -->
<!-- SciCo: none -->
<!-- DAQAce: Ayana Holloway -->
<!-- MonAce: same -->
<!-- CO: none -->
<!-- OpManager: Kevin Burkett -->
<!-- Notes: -->
<pre>Starting with a cosmics run attempt.</pre>


<!-- Date: Tue Nov 13 16:09:27 2001 -->
Cosmics run started::  131911 
T
<!-- Author: Ayana :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=131911' target=_top>131911</a>) -->


<!-- Date: Tue Nov 13 16:37:46 2001 -->
PSM alarm in the west central arches; probably someone is working but calling to check.
<!-- Author: Ayana -->
<!-- Comment: Tue Nov 13 16;42;33 comment by...ath -->
Indeed, we should ignore these alarms.


<!-- Date: Tue Nov 13 16:47:35 2001 -->
We check the DC connection test  of The Pha channels for 
W0 2-19 and W1 2-9. Channel W0-PHA-12 is noisy and 
W0-pha-4,W1-pha-4,6 are slightly noisy. We disconnected 
the cable at the PMT box.  We see 3 volts of the ADMEM  
and .2-.3 nf of the pmt test for all these channels. 
The connectors look OK.          
<!-- Author: Howard Budd -->


<!-- Date: Tue Nov 13 17:17:34 2001 -->
Ending run to give Alexander the CLC_00 crate.  Now might be a good time to look for sevb errors running a calibration.
<!-- Author: Ayana :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=131911' target=_top>131911</a>) -->


<!-- Date: Tue Nov 13 17:26:52 2001 -->
Doing muon calibration for Trevor (after Abraham and Muge power-cycle CMX crates)
<!-- Author: ath -->


<!-- Date: Tue Nov 13 18:24:13 2001 -->
<pre>Calibrations:	131923 	cmx-only muon calibration for muge and abe.  	(would have been all muons, but imu00 DTOed) 
		131926	cmp-imu muon calibration to see if rebooting imu00 worked while muge/abe turned off cmx power supply. 
	</pre>
<!-- Author: ayana :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=131926' target=_top>131926</a>) -->


<!-- Date: Tue Nov 13 18:36:50 2001 -->
Tried to do normal muon calibration; (run 131930)  now CMP00 is 
giving DTO.  
<!-- Author: Ayana -->


<!-- Date: Tue Nov 13 18:50:41 2001 -->
trevor's muon calibration finished.
<!-- Author: Ayana :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=131932' target=_top>131932</a>) -->


<!-- Date: Tue Nov 13 19:01:42 2001 -->
taking all ccal and wcal and l1 to run for a bit.  got SMXR errors in ccal 09,10,11,12 so removed all cards from our crates. 
  
<!-- Author: Level 2 -->


<!-- Date: Tue Nov 13 19:29:19 2001 -->
Running cosmics with the few leftover crates.
<!-- Author: Ayana :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=0x20360 (131936)' target=_top>0x20360 (131936)</a>) -->


<!-- Date: Tue Nov 13 19:31:36 2001 -->
<i><font size=-2>Oops, didn't think of the run summary DB: if you want to click on the run number here it is again.</font></i>
<!-- Author: Ayana :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=131936' target=_top>131936</a>) -->


<!-- Date: Tue Nov 13 21:56:06 2001 -->
CMU HV trip. Expert is going to look.
<!-- Author: ath -->
<!-- Comment: Tue Nov 13 22;02;51 comment by...ath -->
...Hyunsoo reset and lowered HV a little.


<!-- Date: Tue Nov 13 22:14:12 2001 -->
<table border=0><td><tr><td bgcolor=#e0e030>In a horrifying display of agression, the L2 crate fights back, resulting in the loss of Monica's one contact lens!</td></tr></td></table>
<!-- Author: h -->


<!-- Date: Tue Nov 13 22:32:13 2001 -->
Put muons into the calib_continuous cosmic run.
<!-- Author: ath :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=131939' target=_top>131939</a>) -->


<!-- Date: Tue Nov 13 22:39:25 2001 -->
Vision-impared or not, I trust the l2 ladies can handle the rest of the shift (thanks!) 
 
Pseudo-summary: no calorimiter calibrations this shift.  Cosmics running ok.  Didn't have the sevb problems with the muon calibrations, at least.  
<!-- Author: ayana -->


<!-- Date: Tue Nov 13 23:34:30 2001 -->
Rack Trip at 11:24<BR> 
Looks like COSET had a drip failure which tripped  
COSEB, CONET, CONEB as well.  Techs unresponsive, but 
heather reports "someone on the third floor doing something"
<!-- Author: SCN -->
<!-- Comment: Tue Nov 13 23;37;06 comment by...SCN -->
Person on third floor says he is responsible for the 
voltman heartbeat type problems but not the rack trips.


<!-- Comment: Tue Nov 13 23;44;17 comment by...SCN -->
No Techs, but Gerry Bauer is kinda interested in these crates.
Anyway, I silenced the warning and will wake up Kevin to see if he cares...

<!-- Comment: Tue Nov 13 23;54;06 comment by...Inspector Clouseau -->
On advice of Dr. Burkett, I reset the racks and they come back.
In the meantime, we find a culprit, a certain MIT Grad student by the name of Mike who upon learning what a drip sensor looked like fessed up that he more than likely was the cause of the trip.


<!-- Date: Wed Nov 14 00:49:42 2001 -->
IMU Work summary : Reprogrammed all IMU tdcs with the new mezzanine card registers in anticipation of looking at bmu trigger. Taking a long run overnite, with the 2 imu crates to have enough data to find some stubs tomorrow!  TDCtest results (had some difficulty with running - DTO) on the web. <font color=red><cite>(entry outside this shift's time range )</cite></font>
<!-- Author: dan -->


<!-- Date: Wed Nov 14 03:18:08 2001 -->
<P> Done testing TOF L1 Trigger cables.  Looks like they all survived installation.  Hooray!  Magnetic Monopoles here we come!!! </P><font color=red><cite>(entry outside this shift's time range )</cite></font>
<!-- Author: mulhearn -->


<!-- Date: Wed Nov 14 05:31:44 2001 -->
done ith CLIST tests. Found and solved multiple problems in the system which were causing 0 MB words passed to the alpha. 
Observed some loss-of-sync errors still not understood. 
Last run taken is 131955.<font color=red><cite>(entry outside this shift's time range )</cite></font>
<!-- Author: monica and heather -->


<!-- Date: Wed Nov 14 08:24:50 2001 -->
<font color=red>Shift Summary: </font>Found cosmics run 131939 still running.  4.6M events.   
<!-- Author: Abraham Gallas -->