<!-- Title: Day shift. Mon Sep  8, 2008 -->
<!-- SciCo: William Badgett -->
<!-- DAQAce: Stefano Camarda -->
<!-- MonAce: &nbsp; -->
<!-- CO: Derek Thompson -->
<!-- OpManager: Gene Flanagan -->
<!-- Notes: -->
<pre>Inherit run 266929 and store 6406 running smoothly.   Stash  
is at 292E10, so perhaps we will have a shot setup during  
this shift.</pre>


<!-- Date: Mon Sep 8 08:09:50.933 2008 -->
Current instantaneous lumi is 82.5 × 10<SUP>30</SUP>  
cm<SUP>-2</SUP> sec<SUP>-1</SUP>
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266929' target=_top>266929</a>) -->


<!-- Date: Mon Sep 8 08:11:32.773 2008 -->
CER_SVXMON_HALT_RECOVER_RUN_ERROR  
Stuck Cellid S/B0/W0/L4/C7-13 
AUTO HRR was issued
<!-- Author: Stefano :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266929' target=_top>266929</a>) -->
<!-- Comment: Mon Sep  8 09;48;54 comment by...Stefano -->
again the same error, followed by L2 Timeout and double HRR
<!-- Comment: Mon Sep  8 09;56;24 comment by...Stefano -->
It happened three more times at 09:49, at 09:51 and at 09:54, trying to restart svxmon
<!-- Comment: Mon Sep  8 10;01;41 comment by...Stefano -->
while restarting we had CER_SVXMON_HALT_RECOVER_RUN_ERROR
Pipeline out of synch in 490 silicon readout chips.
AUTO HRR was issued
<!-- Date: Mon Sep 8 08:27:59.782 2008 -->
TOF HV Trip. Turned On HV Power
<!-- Author: Stefano :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266929' target=_top>266929</a>) -->


<!-- Date: Mon Sep 8 08:31:01.876 2008 -->
Scaler Monitor Error: 
"FERML_HIGH_DEADTIME 
Trigger/DAQ Deadtime is too high (at 61.32197 %)" But last 60s average is below 1.5%
<!-- Author: Stefano :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266929' target=_top>266929</a>) -->
<!-- Comment: Mon Sep  8 08;36;24 comment by...Donatella -->
I'll increase the 'wait' time after an Inibit to reduce these spurious error messages from ScalerMon
<!-- Date: Mon Sep 8 08:33:31.076 2008 -->
SVT stable at 55.
<!-- Author: Derek Thompson :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266929' target=_top>266929</a>) -->



<!-- Date: Mon Sep 8 08:59:55.011 2008 -->
Diff tracks stable at 55.
<!-- Author: Derek Thompson :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266929' target=_top>266929</a>) -->


<!-- Date: Mon Sep 8 09:04:20.850 2008 -->
RXPT Error for crate svx02, manual HRR issued. Then "FEML_SRC_FATALITY Error, SRC Fatality Error fo svx02: SI 6 Resonance Detected" and again a RXPT Error for the same crate. One more HRR was issued. Then a L2 TO, and a third HRR fixed the problem.
<!-- Author: Stefano :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266929' target=_top>266929</a>) -->
<!-- Comment: Mon Sep  8 09;36;34 comment by...Stefano -->
Again: First a RXPT error for crate svx02 and a manual HRR. Then a "SRC Fatality Error for b0svx02: Sl 6 Resonance Detected". As soon as Run was recovered a L2 Timeout and an auto HRR. Then L2 Trigger rate dropped to 0 and the third HRR fixed the problem.
 
<!-- Date: Mon Sep 8 09:26:05.920 2008 -->
L2 Timeout. Auto HRR
<!-- Author: Stefano :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266929' target=_top>266929</a>) -->



<!-- Date: Mon Sep 8 09:42:13.803 2008 -->
pcal02 failed, shepherded
<!-- Author: Stefano :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266929' target=_top>266929</a>) -->



<!-- Date: Mon Sep 8 09:48:58.547 2008 -->
Have added SILI_BIAS_SCAN[4,1136,706] to the list of current tables on the white board. This table should be updated whenever the other default tables are updated.
<!-- Author: Gene -->


<!-- Date: Mon Sep 8 09:49:43.754 2008 -->
<table><tr><td><a href=/elog/notebooks/2008/Mon_Sep__8_2008_Day.094937.ps align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-4572  align='top'></a></td><td>Occurrence of known error.</td></tr></table>
<!-- Author: Derek Thompson -->

<!-- Date: Mon Sep 8 09:50:35.374 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266929>266929</A>
smx: <tt>Another o</tt>
<!-- Author: Consumer Operator( Greg Thompson) -->
<!-- Comment: Mon Sep  8 10;17;08 comment by...Derek Thompson -->
This is a mistake.  Comment began/aborted by CO Derek Thompson, not Greg.
<!-- Date: Mon Sep 8 09:50:36.035 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266929>266929</A>
general: <tt>CO Checklist complete.  No new problems.<br>
Diff tracks stable at 55.</tt>
<!-- Author: Consumer Operator( Greg Thompson) -->
<!-- Comment: Mon Sep  8 10;17;41 comment by...Derek Thompson -->
Checklist completed by Derek Thompson, not Greg.
<!-- Date: Mon Sep 8 09:59:10.321 2008 -->
Expect to drop the store 11-12 oclock
<!-- Author: Gene -->
<!-- Comment: Mon Sep  8 10;13;50 comment by...phil -->
<a href=
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=operations&action=view&page=207&frame=2&anchor=10:13:07'>
news</a> from the AD ops meeting




<!-- Date: Mon Sep 8 10:22:41.251 2008 -->
Diff tracks stable at 55.
<!-- Author: Derek Thompson :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266929' target=_top>266929</a>) -->

<!-- Date: Mon Sep 8 10:40:35.835 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266929>266929</A>
 Terminated at 2008.09.08 10:39:37
<!-- Author: RunControl -->



<!-- Date: Mon Sep 8 10:41:08.419 2008 -->
Tevatron quench.   We end run and begin post-quench check out.
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266929' target=_top>266929</a>) -->
<!-- Comment: Mon Sep  8 10;45;08 comment by...badgett -->
There was NO dose alarm during this abort.


<!-- Date: Mon Sep 8 10:49:16.425 2008 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2008&action=view&page=-4574&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-4574  width='300' align='top'></a></td><td>Abort Data plot for store 6406</td></tr></table>
<!-- Author: Stefano -->


<!-- Date: Mon Sep 8 10:52:14.017 2008 -->
MCR says, "It will be a few hours before we can put beam  
back in the machine"    We will do quiet time calibrations  
now...
<!-- Author: badgett -->

<!-- Date: Mon Sep 8 10:58:31.158 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266931>266931</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Stefano -->


<!-- Date: Mon Sep 8 10:58:32.278 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266931>266931</A>
TERMINATE: done
<!-- Author: Stefano -->


<!-- Date: Mon Sep 8 11:01:19.230 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266932>266932</A>
ACTIVATE: RunSet: CALIB_CLC_QIE 
<!-- Author: Stefano -->


<!-- Date: Mon Sep 8 11:01:20.939 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266931>266931</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Mon Sep  8 12;08;29 comment by...Derek Thompson -->
calqie OK.
<!-- Date: Mon Sep 8 11:01:33.920 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266932>266932</A>
TERMINATE: done
<!-- Author: Stefano -->


<!-- Date: Mon Sep 8 11:02:05.234 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266932>266932</A>
CALIBRATION: <U><B>clcqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#clcqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Mon Sep  8 12;10;13 comment by...Derek Thompson -->
clcqie OK.
<!-- Date: Mon Sep 8 11:06:03.246 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266933>266933</A>
ACTIVATE: RunSet: CALIB_QIE_FWD 
<!-- Author: Stefano -->


<!-- Date: Mon Sep 8 11:06:55.933 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266933>266933</A>
CALIBRATION: <U><B>bscqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#bscqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Mon Sep  8 12;10;34 comment by...Derek Thompson -->
bscqie OK.
<!-- Date: Mon Sep 8 11:06:57.608 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266933>266933</A>
TERMINATE: done
<!-- Author: Stefano -->


<!-- Date: Mon Sep 8 11:10:05.297 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266934>266934</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL 
<!-- Author: Stefano -->


<!-- Date: Mon Sep 8 11:16:42.109 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266934>266934</A>
TERMINATE: done
<!-- Author: Stefano -->


<!-- Date: Mon Sep 8 11:18:00.203 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266935>266935</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: Stefano -->


<!-- Date: Mon Sep 8 11:21:41.815 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266935>266935</A>
TERMINATE: done
<!-- Author: Stefano -->


<!-- Date: Mon Sep 8 11:21:42.723 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266935>266935</A>
CALIBRATION: <U><B>pluglaser</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#pluglaser target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Mon Sep  8 12;12;23 comment by...Derek Thompson -->
pluglaser OK.
<!-- Date: Mon Sep 8 11:23:43.026 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266936>266936</A>
ACTIVATE: RunSet: BOTHPLUGS_MM 
<!-- Author: Stefano -->


<!-- Date: Mon Sep 8 11:24:05.941 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266936>266936</A>
TERMINATE: done
<!-- Author: Stefano -->



<!-- Date: Mon Sep 8 11:26:06.214 2008 -->
MCR offers us a 1 to 1.5 hour access, if we need it
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266936' target=_top>266936</a>) -->


<!-- Date: Mon Sep 8 11:26:27.955 2008 -->
Stefano rebooted muon3 HV PC per Phil's request
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266936' target=_top>266936</a>) -->
<!-- Comment: Mon Sep  8 15;36;30 comment by...phil -->
thanks
<!-- Date: Mon Sep 8 11:29:08.380 2008 -->
The Consumer experts, Koji, has left the consumer pager  
at home.   Please call his cell phone at 630-770-5021  
instead until later today.
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266936' target=_top>266936</a>) -->
<!-- Comment: Mon Sep  8 14;24;15 comment by...Koji -->
I picked up it. so please page consumer as usual in a case of  problem. Thanks,
<!-- Date: Mon Sep 8 11:31:12.389 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266937>266937</A>
 Activated at 2008.09.08 11:30:55
<!-- Author: RunControl -->


<!-- Date: Mon Sep 8 11:31:13.654 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266937>266937</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [7,1134,706] 
<!-- Author: Stefano -->
<!-- Comment: Mon Sep  8 11;35;57 comment by...Stefano -->
PMT Spike run running with no beam
<!-- Comment: Mon Sep  8 11;51;48 comment by...Stefano -->
L1 Average trigger rate for the run is 43.9
<!-- Date: Mon Sep 8 11:31:19.128 2008 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2008&action=view&page=-4575&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-4575  width='300' align='top'></a></td><td>Network usage plot of solenoid2.  Whether these network  
outages are a cause of the ICICLE problems and/or the usual 
reboot afterward is not clear to me.  I should be able to find 
an error log for ICICLE, also.</td></tr></table>
<!-- Author: Steve Hahn -->
<!-- Comment: Mon Sep  8 11;49;39 comment by...Steve Hahn -->
I see some errors in the "Event Viewer" on solenoid2 which imply
that solenoid2 could not talk to the domain controller (in FCC)
during the second period in the daily plot above from 0400-0500.
But there are also similar entries from 0500-0600.  Maybe I'm
misinterpreting this.
<!-- Date: Mon Sep 8 11:35:50.388 2008 -->
CDF declines access
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266937' target=_top>266937</a>) -->


<!-- Date: Mon Sep 8 11:42:33.189 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266937>266937</A>
 Terminated at 2008.09.08 11:42:28
<!-- Author: RunControl -->



<!-- Date: Mon Sep 8 11:46:07.271 2008 -->
<table><tr><td><a href=/elog/notebooks/2008/Mon_Sep__8_2008_Day.114602.ps align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-4577  align='top'></a></td><td>High RMS is Channel 17 W.  Expert notified.</td></tr></table>
<!-- Author: Derek Thompson -->
<!-- Comment: Mon Sep  8 11;47;29 comment by...badgett -->
Paging ADMEM expert



<!-- Comment: Mon Sep  8 11;51;01 comment by...badgett -->
Shalhout calls back and points out that these are PIN diode 
reference channels, and not part of the calorimeter 
readout.   So we don't care.   I asked him to update 
the reference table.
<!-- Comment: Mon Sep  8 12;18;25 comment by...m mattson (ADMEM) -->
<br>Reference page states explicitly that those PIN diodes are to be ignored (once for the reference plot, once under "Notes").  Also, updating the reference pedestals will do absolutely nothing for the hard-coded limit of 5 for the RMS.
<!-- Comment: Mon Sep  8 12;21;58 comment by...badgett -->
You don't expect us to read the fine print?  
I meant update the reference table graphics file that 
the C.O. uses to compare, not the 
reference pedestals.

<!-- Date: Mon Sep 8 11:50:13.149 2008 -->
the PSH crate 07 is in overV. I tried to kill and restart the Plug HV monitor programs. Checking Group Consistency reports the following error: 
 
Cr:03 Gr "All Channels" Sl:1 - Channels present in Configuration and NOT in CAEN: 1 
Cr:03 Gr "BothPlugs" Sl:1 - Channels present in Configuration and NOT in CAEN: 1 
Cr:03 Gr "WestPlug" Sl:1 - Channels present in Configuration and NOT in CAEN: 1 
Cr:03 Gr "ECal" Sl:1 - Channels present in Configuration and NOT in CAEN: 1 
Cr:03 Gr "WestECal" Sl:1 - Channels present in Configuration and NOT in CAEN: 1 
 
The Parameters Consistency chek reports:  
Cr:03 Sl:1 Ch:01 has inconsistent parameter(s): VMAXcaen/VMAXconf 852.0/822.0 V0caen/V0conf 852.0/822.0 V1caen/V1conf 511.0/493.0  
<!-- Author: Stefano :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266937' target=_top>266937</a>) -->
<!-- Comment: Mon Sep  8 12;12;41 comment by...Stefano -->
First tried to reset crate 07, and then I took a 5 min power cycle of the crate. This last action fixed the Over V Problem. The Crate 3 error is still present
<!-- Comment: Mon Sep  8 12;16;18 comment by...Stefano -->
A reset of the crate 3 fixed the consistency problem. I can proceed with calibrations
<!-- Date: Mon Sep 8 11:50:45.435 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266937>266937</A>
TERMINATE: done
<!-- Author: Stefano -->




<!-- Date: Mon Sep 8 11:55:49.287 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266938>266938</A>
CALIBRATION: <U><B>tofqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#tofqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Mon Sep  8 12;15;33 comment by...Derek Thompson -->
tofqie OK.
<!-- Date: Mon Sep 8 11:58:34.258 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266938>266938</A>
ACTIVATE: RunSet: ACE_TOF_QIE_NEW 
<!-- Author: Stefano -->


<!-- Date: Mon Sep 8 11:58:35.159 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266938>266938</A>
TERMINATE: done
<!-- Author: Stefano -->









<!-- Date: Mon Sep 8 12:18:24.946 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266939>266939</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Stefano -->



<!-- Date: Mon Sep 8 12:19:25.409 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266939>266939</A>
TERMINATE: done
<!-- Author: Stefano -->


<!-- Date: Mon Sep 8 12:21:24.970 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266940>266940</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: Stefano -->



<!-- Date: Mon Sep 8 12:22:42.430 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266940>266940</A>
TERMINATE: done
<!-- Author: Stefano -->


<!-- Date: Mon Sep 8 12:22:43.230 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266940>266940</A>
CALIBRATION: <U><B>cemled</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cemled target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Mon Sep  8 12;57;40 comment by...Derek Thompson -->
cemled OK.
<!-- Date: Mon Sep 8 12:22:47.354 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266939>266939</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Mon Sep  8 12;46;44 comment by...Derek Thompson -->
Questionable bad channels in PPR.  Marked OK.  Emailed willis@fnal.gov.
<!-- Date: Mon Sep 8 12:24:12.101 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266941>266941</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Stefano -->


<!-- Date: Mon Sep 8 12:24:25.606 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266941>266941</A>
TERMINATE: done
<!-- Author: Stefano -->


<!-- Date: Mon Sep 8 12:27:42.352 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266941>266941</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Mon Sep  8 12;49;16 comment by...Derek Thompson -->
cot OK.
<!-- Date: Mon Sep 8 12:29:50.984 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266942>266942</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: Stefano -->


<!-- Date: Mon Sep 8 12:30:43.660 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266942>266942</A>
CALIBRATION: <U><B>calxef</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calxef target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Mon Sep  8 12;52;53 comment by...Derek Thompson -->
calxef OK.
<!-- Date: Mon Sep 8 12:32:44.171 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266942>266942</A>
TERMINATE: done
<!-- Author: Stefano -->


<!-- Date: Mon Sep 8 12:34:24.248 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266943>266943</A>
TERMINATE: done
<!-- Author: Stefano -->


<!-- Date: Mon Sep 8 12:34:50.350 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266943>266943</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Stefano -->


<!-- Date: Mon Sep 8 12:36:59.286 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266944>266944</A>
ACTIVATE: RunSet: TOF_TAC_CALIB 
<!-- Author: Stefano -->



<!-- Date: Mon Sep 8 12:40:08.860 2008 -->
<table><tr><td><a href=/elog/notebooks/2008/Mon_Sep__8_2008_Day.124005.ps align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-4579  align='top'></a></td><td>Bad channels in SMX plots</td></tr></table>
<!-- Author: Derek Thompson -->
<!-- Comment: Mon Sep  8 12;40;30 comment by...Derek Thompson -->
Expert emailed.

<!-- Date: Mon Sep 8 12:42:36.956 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266944>266944</A>
TERMINATE: done
<!-- Author: Stefano -->


<!-- Date: Mon Sep 8 12:43:13.615 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266944>266944</A>
CALIBRATION: <U><B>toftac</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#toftac target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Mon Sep  8 12;54;57 comment by...Derek Thompson -->
toftac OK.
<!-- Date: Mon Sep 8 12:47:27.240 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266945>266945</A>
 Activated at 2008.09.08 12:47:08
<!-- Author: RunControl -->


<!-- Date: Mon Sep 8 12:47:28.277 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266945>266945</A>
ACTIVATE: RunSet: ACE_COSMIC_ALL_L1MET  Trigger: COSMICS [15,1130,706] 
<!-- Author: Stefano -->




<!-- Date: Mon Sep 8 12:51:48.935 2008 -->
<table><tr><td><a href=/elog/notebooks/2008/Mon_Sep__8_2008_Day.125146.ps align=top><img src=/cgi/elog/elog.pl?nb=2008&action=view&page=-4581  align='top'></a></td><td>Occurrence of known error.</td></tr></table>
<!-- Author: Derek Thompson -->




<!-- Date: Mon Sep 8 13:13:29.985 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266945>266945</A>
 Terminated at 2008.09.08 13:12:42
<!-- Author: RunControl -->


<!-- Date: Mon Sep 8 13:19:20.372 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266946>266946</A>
 Activated at 2008.09.08 13:18:40
<!-- Author: RunControl -->


<!-- Date: Mon Sep 8 13:19:21.207 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266946>266946</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: TEST_FPS [1,1128,706] 
<!-- Author: Stefano -->
<!-- Comment: Mon Sep  8 13;25;03 comment by...Donatella -->
with Sparky system (Sparky1 cable in Fred NIM0, Sparky2 in Fred NIM1)
<!-- Date: Mon Sep 8 13:21:39.062 2008 -->
Donatella has arrived to do some work on dynamic  
prescales (FUPS)
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266946' target=_top>266946</a>) -->



<!-- Date: Mon Sep 8 13:33:20.946 2008 -->
Alcohol Alarm: Alcohol outside Tank Level at 2.2LL
<!-- Author: Stefano :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266946' target=_top>266946</a>) -->
<!-- Comment: Mon Sep  8 13;33;47 comment by...Stefano -->
now turned green, level at 2.0LL

<!-- Comment: Mon Sep  8 13;36;38 comment by...badgett -->
Cryo says he will refill the tank tomorrow, and don't worry 
about the alarm.
<!-- Date: Mon Sep 8 13:36:30.318 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266946>266946</A>
 Terminated at 2008.09.08 13:36:28
<!-- Author: RunControl -->



<!-- Date: Mon Sep 8 13:40:10.268 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266929>266929</A>
<HTML>Run <A HREF=http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN=266929 target=_runSummary>266929</A> has not yet been checked for goodrun status; please do so.</HTML>
<!-- Author: RunControl -->
<!-- Comment: Mon Sep  8 13;49;20 comment by...badgett -->
done.   it's all good.
<!-- Date: Mon Sep 8 13:43:46.962 2008 -->
Updated clc acceptances. 3% increase in lumi for the next store.
<!-- Author: Yuri -->


<!-- Date: Mon Sep 8 13:48:32.322 2008 -->
Pasha requests to use the DAQ system for event builder  
tests.   He will return in one hour.
<!-- Author: badgett -->


<!-- Date: Mon Sep 8 13:53:03.528 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266947>266947</A>
 Activated at 2008.09.08 13:52:18
<!-- Author: RunControl -->


<!-- Date: Mon Sep 8 13:53:04.954 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266947>266947</A>
ACTIVATE: RunSet: COSMICS_TEST_FUPS_5  Trigger: COSMICS_TEST_FUPS [5,1126,706] 
<!-- Author: donatella -->
<!-- Comment: Mon Sep  8 14;10;29 comment by...Donatella -->
need to look at logfiles. Restored everything for regular data taking
<!-- Date: Mon Sep 8 13:58:40.390 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266947>266947</A>
 Terminated at 2008.09.08 13:57:52
<!-- Author: RunControl -->




<!-- Date: Mon Sep 8 14:14:41.636 2008 -->
Allison and one of the Cox twins are swapping two  
XFT stereo finder boards &minus; expect b0xft06 and b0xft07  
PSM alarms
<!-- Author: badgett -->



<!-- Date: Mon Sep 8 14:42:21.278 2008 -->
<pre>Looks like there was a power glitch across the  
accelerator complex.   From MCR eLog: 
 
"Systems affected by glitch: 
 
    * MAC room appears to have lost power. 
          o MI & TCLK FEs all rebooted 
          o all TCLK crates lost power as well. 
          o Controls experts recovered all this. 
    * Linac: 
          o Saw glitch, but recovered quickly on its own. 
    * Booster: 
          o GMPS tripped, required local attention. 
          o RF tripped, recovered. 
          o Used downtime to address B:SEXSB rack fan, fix timer card for BRF16.  
    * Main Injector: 
          o Front ends all went down. 
          o PSLOOP dropped, several PSs tripped as a result. 
          o I:LAM52 tripped an internal breaker. 
    * Tevatron: 
          o Quench at D4, E1, D0 low betas. Recovery underway. 
          o Vacuum at E0 took a hit due to the beam loss. All IPs were off, and reset"</pre>
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266949' target=_top>266949</a>) -->

<!-- Date: Mon Sep 8 14:43:05.644 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266949>266949</A>
ACTIVATE: RunSet: AAA_SHOTSETUP_L1MET  Trigger: PHYSICS_5_01 [9,1129,725] 
<!-- Author: Pedro/Heather -->


<!-- Date: Mon Sep 8 14:48:53.218 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266948>266948</A>
ACTIVATE: RunSet: AAA_SHOTSETUP_L1MET  Trigger: PHYSICS_5_01 [9,1129,725] 
<!-- Author: Stefano  -->



<!-- Date: Mon Sep 8 14:51:02.056 2008 -->
<pre>We swapped 2 stereo finders. 
b0xft06 slot 19 (bunch counters at L2 different from rest of event) with slot 18 (good). 
</pre>
<!-- Author: David Cox + Alison  -->

<!-- Date: Mon Sep 8 14:59:10.045 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266950>266950</A>
ACTIVATE: RunSet: AAA_SHOTSETUP_L1MET  Trigger: PHYSICS_5_01 [9,1129,725] 
<!-- Author: Pedro/Heather -->



<!-- Date: Mon Sep 8 15:13:18.524 2008 -->
Tested L2 executable that dumps data when there are XFT data corruptions or filar errors.
<!-- Author: Heather (and Pedro) -->


<!-- Date: Mon Sep 8 15:29:29.171 2008 -->
MCR called and stated they will start injecting protons  
in a few minutes.  After a proton-only store, they will  
start shot setup.   Estimate 1 - 2 hours before shot setup. 
  
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266950' target=_top>266950</a>) -->


<!-- Date: Mon Sep 8 15:33:55.697 2008 -->
Pasha is fixing the EVB...
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266950' target=_top>266950</a>) -->



<!-- Date: Mon Sep 8 15:39:32.638 2008 -->
Laura and friends working on crate b0l2pu02
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=266950' target=_top>266950</a>) -->


<!-- Date: Mon Sep 8 15:47:52.398 2008 -->
Laura and I put a fourth slink rx card on the pulsar merger board in b0l2pu02 slot 18.  I'd like to run a shotsetup test or something to make sure it works.  We put the XFT1 fiber into this input.  After testing, Olga would like to go back to the previous configuration (XFT1 in "SPARE").  
<!-- Author: Heather -->
<!-- Comment: Mon Sep  8 17;53;10 comment by...Heather -->
XFT1 back to spare
<!-- Date: Mon Sep 8 15:55:22.085 2008 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>PhysicsTable</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L3Accepts</TH>
<TH>DelivLumi nb<SUP>-1</SUP></TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN=266929 target=_runSummary>266929</A> x412B1</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_5_01 [9,1129,725]</FONT></TD>
<TD ALIGN=CENTER>20:37:21</TD>
<TD ALIGN=CENTER>10:39:37</TD>
<TD ALIGN=CENTER>12:42:21</TD>
<TD ALIGN=RIGHT>   5,698,309</TD>   <TD ALIGN=RIGHT>  6,299.40</TD>
<TD ALIGN=RIGHT>  5,650.12</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>15:55:01</TD>
<TD ALIGN=CENTER>12:42:21</TD>
<TD ALIGN=RIGHT>     5,698,309</TD> <TD ALIGN=RIGHT>  6,299.40</TD>
<TD ALIGN=RIGHT>  5,650.12</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Mon Sep 8 15:58:44.321 2008 -->
<font color=red>Shift Summary: </font><pre>Inherit run 266929 and store 6406 running smoothly.   Stash   
is at 292E10, so perhaps we will have a shot setup during   
this shift.  Current instant lumi is 82.5E30 
 
08:22 TOF HV trip 
08:25 Pinkie, handled by Sasha 
09:55 Sticky SVXMon errors;  restart SVXMon 
10:00 Estimate end of store at 11:00 or 12:00 this morning 
10:40 Tevatron quench;  end run 266929, bring down voltages,  
      do post-quench check out  
10:45 MCR says no beam for a few hours 
10:50 Doing calibration, starting with quiet time 
11:10 Rebooted muon3 HV PC per Phil's request 
11:20 MCR offers us a 1 - 1.5 hour access;  CLC west muffin fan  
      and TOF requests are pending 
11:30 We decline access 
11:46 New bad channel in PHA with very high RMS in CALQIE calibration 
      Expert says this is pin diode readout -- ignore 
13:00 Complete calibrations; start cosmic run 
13:20 Donatella does work on dynamic prescaling FUPS 
13:30 Gas alarm from outside alcohol tank;  Cryo will refill it tomorrow 
13:40 Yuri reboots b0clc crate to pick up new calibration constants 
13:48 Pasha request central DAQ system for EVB work;  return in 1 hour 
14:15 Allison and a Cox swapping XFT stereo cards in b0xft06, b0xft07 
14:30 Pedro and Heather testing a new L2 executable 
15:30 MCR calls and will start injecting test protons in a few minutes 
      If the proton-only store looks good, they will move into  
      shot setup right away.   Perhaps 1 - 2 hours 
15:35 Laura & company working on crate b0l2pu02 
 
MCR will shoot from current stash at 375E10 when ready. 
 
</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   266929
Delivered Luminosity   733.2  
Acquired Luminosity    663.7  
Efficiency             90.5%
   
</td></tr></table></pre>
<!-- Author: badgett -->