<!-- Title: Day shift. Sat Aug 25, 2007 -->
<!-- SciCo: Petar Maksimovic -->
<!-- DAQAce: Ye Li -->
<!-- MonAce: &nbsp; -->
<!-- CO: &nbsp; -->
<!-- OpManager: Steve Hahn -->
<!-- Notes: -->
<pre>Finish ISL work (1/8), and then take calibrations and cosmics, 
and then torture the Level 2. 
</pre>

<!-- Date: Sat Aug 25 08:17:21.259 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248028>248028</A>
ACTIVATE: RunSet: CALIB_QIE_FWD 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 25 08:17:22.534 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248028>248028</A>
CALIBRATION: <U><B>bscqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#bscqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 25 08;18;47 comment by...Ye Li -->
okay for the only in-use BSC-1
<!-- Date: Sat Aug 25 08:17:24.857 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248028>248028</A>
TERMINATE: done
<!-- Author: Petteri -->



<!-- Date: Sat Aug 25 08:19:44.255 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248029>248029</A>
CALIBRATION: <U><B>tofqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#tofqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 25 08;21;52 comment by...Ye Li -->
not okay, one channel in wcal04 off in plot avg pedestal RMS
<!-- Date: Sat Aug 25 08:19:45.822 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248029>248029</A>
ACTIVATE: RunSet: ACE_TOF_QIE_NEW 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 25 08:19:47.624 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248029>248029</A>
TERMINATE: done
<!-- Author: Petteri -->



<!-- Date: Sat Aug 25 08:22:42.693 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248030>248030</A>
CALIBRATION: <U><B>cemled</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cemled target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 25 08;25;16 comment by...Ye Li -->
the same problem as Petteri reported last eve shift
<!-- Date: Sat Aug 25 08:22:43.571 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248030>248030</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 08:22:45.250 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248030>248030</A>
TERMINATE: done
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 08:24:02.582 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248031>248031</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 08:24:03.215 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248031>248031</A>
TERMINATE: done
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 08:24:04.663 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248031>248031</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 25 08;26;12 comment by...Ye Li -->
okay
<!-- Date: Sat Aug 25 08:25:05.304 2007 -->
The emergency Diesel generator went into alarm around 8:00 this morning, probably during its weekly self-test. The generator had just been repaired and put back into operation. We have notified Steve Hahn.
<!-- Author: Peter Wilson, Ulrich Husemann -->



<!-- Date: Sat Aug 25 08:31:38.596 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248032>248032</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 08:31:39.739 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248032>248032</A>
TERMINATE: done
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 08:31:40.319 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248033>248033</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL 
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 08:37:32.319 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248033>248033</A>
TERMINATE: done
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 08:43:56.323 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248034>248034</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 08:46:02.572 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248034>248034</A>
CALIBRATION: <U><B>pluglaser</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#pluglaser target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 25 08;51;50 comment by...Ye Li -->
okay
<!-- Date: Sat Aug 25 08:46:03.835 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248034>248034</A>
TERMINATE: done
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 08:47:47.176 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248035>248035</A>
ACTIVATE: RunSet: BOTHPLUGS_MM 
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 08:51:15.980 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248035>248035</A>
TERMINATE: done
<!-- Author: Ye Li -->



<!-- Date: Sat Aug 25 08:58:04.575 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248036>248036</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 09:00:21.690 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248036>248036</A>
TERMINATE: done
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 09:03:13.291 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248037>248037</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 09:03:15.177 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248036>248036</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 25 09;05;35 comment by...Ye Li -->
bad on a large number of channels
<!-- Date: Sat Aug 25 09:03:16.085 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248037>248037</A>
TERMINATE: done
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 09:03:17.751 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248037>248037</A>
CALIBRATION: <U><B>calxef</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calxef target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 25 09;06;15 comment by...Ye Li -->
like before, things havn't changed for 2 weeks
<!-- Date: Sat Aug 25 09:05:06.632 2007 -->
<p>Peter called me and told me the emergency diesel generator had an "overcrank 
alarm", which I've never heard of before.  Presumably, this happened this morning 
when the automatic one hour test of the generator started at 0800.  We have just 
recently had this generator returned to service after its voltage regulator was replaced; 
it passed load tests earlier this week with no problem.</p> 
 
<p>I've called the MCR and talked to the crew chief Brian Schupbach.  Currently, 
the lab has no duty mechanical or electrical techs on off-hours or weekends (which 
is under review given the last few days).  However, I told Brian this was an  
emergency; if we have a power outage right now, we will have emergency and 
critical systems only as long as various UPSs last, usually less than half an hour. 
He calling on his emergency list and I've given him my cell phone number to call 
back if there are any questions.</p>  
<!-- Author: Steve Hahn :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248033' target=_top>248033</a>) -->
<!-- Comment: Sat Aug 25 09;08;22 comment by...Steve Hahn -->
<table border=0><td><tr><td bgcolor=#e0e030><p>Given the current circumstances, I would advise everyone to evacuate B0 if
there was a power outage (including the shift crew).  Phone numbers to reach 
people working in the bores in the detector are located
<a href=
'/cgi/elog/elog.pl?nb=2007&action=view&page=691&frame=2&anchor=10:08:10.734'>here</a>.</p> 
</td></tr></td></table>
<!-- Date: Sat Aug 25 09:08:07.549 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248038>248038</A>
ACTIVATE: RunSet: TOF_TAC_CALIB 
<!-- Author: Ye Li -->



<!-- Date: Sat Aug 25 09:12:38.573 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248038>248038</A>
TERMINATE: done
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 09:12:39.164 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248038>248038</A>
CALIBRATION: <U><B>toftac</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#toftac target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 25 09;18;41 comment by...Ye Li -->
 Wcal00 is empty and slot9 of wcal04 is removed

<!-- Date: Sat Aug 25 09:19:47.155 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248039>248039</A>
 Activated at 2007.08.25 09:19:42
<!-- Author: RunControl -->


<!-- Date: Sat Aug 25 09:35:12.646 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248039>248039</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [7,957,639] 
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 09:35:13.814 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248039>248039</A>
 Terminated at 2007.08.25 09:28:57
<!-- Author: RunControl -->


<!-- Date: Sat Aug 25 09:35:15.828 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248039>248039</A>
TERMINATE: done
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 09:35:16.011 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248040>248040</A>
 Activated at 2007.08.25 09:32:50
<!-- Author: RunControl -->


<!-- Date: Sat Aug 25 09:35:17.859 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248040>248040</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,954,639] 
<!-- Author: Ye Li -->



<!-- Date: Sat Aug 25 10:50:53.603 2007 -->
memory allocation error for muon3, ces cmu cmp cmx bmu lost communication. Restart muon3
<!-- Author: Ye Li :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248040' target=_top>248040</a>) -->


<!-- Date: Sat Aug 25 11:26:50.231 2007 -->
<p>I have just talked to the tech MCR called in to look at our 
emergency diesel generator out in our west parking lot.   
The fuel pump for this unit has 
failed badly; he is currently looking about site for a  
replacement.  I hope he's more successful than FESS was looking 
for a replacement voltage regulator.</p>
<!-- Author: Steve Hahn :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248040' target=_top>248040</a>) -->


<!-- Date: Sat Aug 25 12:58:35.986 2007 -->
<B> ISL Cooling Repair Update: </B> This morning we started wih an inspection of the epoxy deposition from Friday.   The fourth tube (right-back) of the return manifold was coated, a couple of spots on the left-back tube were touched up and the front tubes were given a recoat since the coat from Friday was a bit sparse.   Overal the return looked quite good.  On the supply manifold the initial epoxy on Friday was cold when mixed.  The tubes done with the cold epoxy did not lay down as evenly.  There are some pits which did not get coated.   We will recoat these tubes on Monday.    
<p> 
Overall, very good progress.
<!-- Author: Peter for the ISL repair crew -->


<!-- Date: Sat Aug 25 13:02:19.618 2007 -->
<p>Peter and I just talked to Rich from FESS and the tech from 
Illini Power/GenPower.  They managed to replace the fuel pump 
with a +10 V pump, and they are test-running the diesel 
generator now.  They will order a fuel pump next week.</p>
<!-- Author: Steve Hahn :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248040' target=_top>248040</a>) -->


<!-- Date: Sat Aug 25 13:06:27.259 2007 -->
<table border=0><td><tr><td bgcolor=#e0e030><p>Please, keep checking the dew points during the weekend.  
 
<li> COT DEWPT should be <b>smaller</b> than <b>0 C</b> 
<li> SIBAG, SVX and ISL DEWPT should be <b>smaller</b> than <b>-10 C</b>  
 
 
<p>Page silicon if the values go closer to the limits mentioned above Pager 722 8740, pager and numbers are on the white board. </td></tr></td></table>
<!-- Author: The Silis -->

<!-- Date: Sat Aug 25 13:36:46.909 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248040>248040</A>
 Terminated at 2007.08.25 13:36:15
<!-- Author: RunControl -->


<!-- Date: Sat Aug 25 13:36:47.133 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248040>248040</A>
TERMINATE: end for L2 torture
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 13:54:14.357 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248041>248041</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [19,952,633] 
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 15:06:52.997 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248041>248041</A>
TERMINATE: calibrations
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 15:20:22.598 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248042>248042</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 15:20:23.101 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248042>248042</A>
TERMINATE: done
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 15:20:24.308 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248042>248042</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 25 16;08;09 comment by...Petteri -->
ok
<!-- Date: Sat Aug 25 15:20:25.652 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248043>248043</A>
ACTIVATE: RunSet: CALIB_CLC_QIE 
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 15:20:27.533 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248043>248043</A>
TERMINATE: done
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 15:20:28.201 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248043>248043</A>
CALIBRATION: <U><B>clcqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#clcqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 25 16;08;23 comment by...Petteri -->
ok
<!-- Date: Sat Aug 25 15:20:29.233 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248044>248044</A>
ACTIVATE: RunSet: CALIB_QIE_FWD 
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 15:20:30.699 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248044>248044</A>
TERMINATE: done
<!-- Author: Ye Li -->


<!-- Date: Sat Aug 25 15:20:31.681 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248044>248044</A>
CALIBRATION: <U><B>bscqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#bscqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 25 16;08;45 comment by...Petteri -->
ok
<!-- Date: Sat Aug 25 15:20:32.006 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248045>248045</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 25 15:22:02.468 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248045>248045</A>
TERMINATE: done
<!-- Author: Petteri -->


<!-- Date: Sat Aug 25 15:24:14.664 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248046>248046</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 25 15:31:48.925 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248046>248046</A>
TERMINATE: done
<!-- Author: Petteri -->


<!-- Date: Sat Aug 25 15:31:49.837 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248046>248046</A>
CALIBRATION: <U><B>pluglaser</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#pluglaser target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 25 16;09;02 comment by...Petteri -->
ok
<!-- Date: Sat Aug 25 15:31:50.862 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248047>248047</A>
ACTIVATE: RunSet: BOTHPLUGS_MM 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 25 15:31:51.479 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248047>248047</A>
TERMINATE: done
<!-- Author: Petteri -->


<!-- Date: Sat Aug 25 15:32:54.289 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248048>248048</A>
 Activated at 2007.08.25 15:32:51
<!-- Author: RunControl -->


<!-- Date: Sat Aug 25 15:34:27.939 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248048>248048</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [7,957,639] 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 25 15:42:37.409 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248048>248048</A>
 Terminated at 2007.08.25 15:42:12
<!-- Author: RunControl -->


<!-- Date: Sat Aug 25 15:43:16.253 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248048>248048</A>
TERMINATE: done, L1A 54.1Hz
<!-- Author: Petteri -->


<!-- Date: Sat Aug 25 15:51:51.179 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248049>248049</A>
ACTIVATE: RunSet: ACE_TOF_QIE_NEW 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 25 15:51:52.399 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248049>248049</A>
CALIBRATION: <U><B>tofqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#tofqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 25 16;09;50 comment by...Petteri -->
bad, one channel in wcal04 off in plot avg pedestal RMS
<!-- Date: Sat Aug 25 15:51:53.345 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248049>248049</A>
TERMINATE: done
<!-- Author: Petteri -->


<!-- Date: Sat Aug 25 15:51:54.012 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248050>248050</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 25 15:51:55.494 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248050>248050</A>
TERMINATE: done
<!-- Author: Petteri -->


<!-- Date: Sat Aug 25 15:51:56.097 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248051>248051</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 25 15:51:57.096 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248051>248051</A>
TERMINATE: done
<!-- Author: Petteri -->


<!-- Date: Sat Aug 25 15:51:59.802 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248051>248051</A>
CALIBRATION: <U><B>cemled</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cemled target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 25 16;10;49 comment by...Petteri -->
same dead channels as before
<!-- Date: Sat Aug 25 15:52:00.397 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248050>248050</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 25 16;12;00 comment by...Petteri -->
Wedges 10,33,39 are hot for CES, CPE and CCR.
<!-- Date: Sat Aug 25 15:54:09.530 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248052>248052</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 25 15:54:10.584 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248052>248052</A>
TERMINATE: done
<!-- Author: Petteri -->


<!-- Date: Sat Aug 25 15:54:11.552 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248052>248052</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 25 16;14;13 comment by...Petteri -->
total numBad: 455 - Too Many or Too Few Bad channels
<!-- Date: Sat Aug 25 16:00:24.583 2007 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>PhysicsTable</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L3Accepts</TH>
<TH>DelivLumi nb<SUP>-1</SUP></TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>15:55:01</TD>
<TD ALIGN=CENTER>::</TD>
<TD ALIGN=RIGHT></TD>               <TD ALIGN=RIGHT></TD>
<TD ALIGN=RIGHT></TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->


<!-- Date: Sat Aug 25 16:00:25.134 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248053>248053</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 25 16:00:27.500 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248053>248053</A>
TERMINATE: done
<!-- Author: Petteri -->


<!-- Date: Sat Aug 25 16:00:28.660 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248053>248053</A>
CALIBRATION: <U><B>calxef</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calxef target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 25 16;14;29 comment by...Petteri -->
no change
<!-- Date: Sat Aug 25 16:00:29.256 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248054>248054</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 25 16:00:30.825 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248054>248054</A>
TERMINATE: done
<!-- Author: Petteri -->


<!-- Date: Sat Aug 25 16:03:51.199 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248055>248055</A>
ACTIVATE: RunSet: TOF_TAC_CALIB 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 25 16:08:03.665 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248055>248055</A>
CALIBRATION: <U><B>toftac</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#toftac target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 25 16;14;56 comment by...Petteri -->
Wcal00 is empty and slot9 of wcal04 is removed
<!-- Date: Sat Aug 25 16:08:04.078 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248055>248055</A>
TERMINATE: done
<!-- Author: Petteri -->






<!-- Date: Sat Aug 25 16:09:32.756 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=248056>248056</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [19,952,633] 
<!-- Author: Petteri -->






<!-- Date: Sat Aug 25 16:13:12.959 2007 -->
<font color=red>Shift Summary: </font><pre>- The emergency diesel generator broke down during the automatic one hour test from 8am. Steve Hahn adivised us to evacuate B0 if there was a power outage. 
- We should keep checking the dew points during the weekend: 
  (1) COT DEWPT should be less than 0C 
  (2) SIBAG, SVX and ISL DEWPT should be less than -10C 
We are advised to page a silicon expert if the values go closer to 
the limit.</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   n/a
Delivered Luminosity   n/a  
Acquired Luminosity    n/a  
Efficiency             n/a
   
</td></tr></table></pre>
<!-- Author: Petar Maksimovic -->