<!-- Title: Day shift. Wed Dec 10, 2003 -->
<!-- SciCo: Vaia Papadimitriou -->
<!-- DAQAce: Taka Moruyama -->
<!-- MonAce: Dale Stentz -->
<!-- CO: Fiodar Prakoshyn -->
<!-- OpManager: Mary Convery+Kaori -->
<!-- Notes: -->
<pre>Take all calibrations and cosmics and exercise the DAQ. 
Around noon we will take calibrtaions and cosmics with silicon 
which will be ready to be tested</pre>

<!-- Date: Wed Dec 10 08:34:10.308 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=175982">175982</A>
TERMINATE: End the run to start calibration.
<!-- Author: Taka x2080 -->


<!-- Date: Wed Dec 10 08:34:52.762 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=175983">175983</A>
ACTIVATE: Calorimeter QIE calibration.
<!-- Author: Taka x2080 -->
<!-- Comment: Wed Dec 10 09;21;14 comment by...V Tiwari -->
Several PHA PIN diode channels show drift in pedestals. These channels are not used for data taking so it should not be of much concern. However, I would like the shift crew to take another CAL QIE calibration after making sure that the Plug Laser is off. Thanks.
<!-- Date: Wed Dec 10 08:36:24.846 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=175983">175983</A>
TERMINATE: End the Calorimeter QIE calibration.
<!-- Author: Taka x2080 -->


<!-- Date: Wed Dec 10 08:38:33.091 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=175984">175984</A>
ACTIVATE: Start CLC QIE calibration.
<!-- Author: Taka x2080 -->


<!-- Date: Wed Dec 10 08:39:24.506 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=175984">175984</A>
TERMINATE: End the run for CLC QIE calibration.
<!-- Author: Taka x2080 -->


<!-- Date: Wed Dec 10 08:45:18.861 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=175985">175985</A>
ACTIVATE: Start BSC QIE calibaration.
<!-- Author: Taka x2080 -->


<!-- Date: Wed Dec 10 08:45:33.080 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=175985">175985</A>
TERMINATE: End the run for BSC QIE calibration.
<!-- Author: Taka x2080 -->


<!-- Date: Wed Dec 10 08:47:42.455 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=175987">175987</A>
ACTIVATE: TOF TAC calibration prior to working on TOF system during supervised access.
<!-- Author: m jones x2080 -->


<!-- Date: Wed Dec 10 08:47:43.300 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=175988">175988</A>
ACTIVATE: Start the Roman Pot SMQIE.
<!-- Author: Taka x2080 -->


<!-- Date: Wed Dec 10 08:48:19.129 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=175988">175988</A>
TERMINATE: End the run Roman POT SMQIE.
<!-- Author: Taka x2080 -->



<!-- Date: Wed Dec 10 08:54:01.170 2003 -->
<br><b>Diode Tests:</b> 
<br>Tested diode connectivity and everything appears ok.  We 
want to check the connectivity after the plug closes to confirm 
that nothing happened during the closing. 
<br>
<!-- Author: R.J. Tesarek -->

<!-- Date: Wed Dec 10 08:55:25.047 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=175990">175990</A>
ACTIVATE: Start the CEM LED calibration
<!-- Author: Taka x2080 -->


<!-- Date: Wed Dec 10 08:57:05.502 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=175990">175990</A>
TERMINATE: End the run (CEM LED calib)
<!-- Author: Taka x2080 -->


<!-- Date: Wed Dec 10 09:06:49.123 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=175992">175992</A>
ACTIVATE: Start the muon calibration run.
<!-- Author: Taka x2080 -->
<!-- Comment: Wed Dec 10 09;32;09 comment by...Dale -->
<pre>For the run:

SW (168-72) BMU is off
CMX was all on</pre>
<!-- Date: Wed Dec 10 09:08:50.288 2003 -->
I've started fiddling with the BMU HV in anticipation 
of going in on an access to fix BMU-HV-84 HV section. 
Please ignore BMU alarms and leave alone the HV 
until further notice.
<!-- Author: Camille Ginsburg -->

<!-- Date: Wed Dec 10 09:10:59.921 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=175992">175992</A>
TERMINATE: End the run for muon calibration.
<!-- Author: Taka x2080 -->


<!-- Date: Wed Dec 10 09:13:14.020 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=175994">175994</A>
ACTIVATE: Start SMX QIE Default Nonstandard. (note: plug is maybe turned off)
<!-- Author: Taka x2080 -->


<!-- Date: Wed Dec 10 09:15:00.826 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=175994">175994</A>
TERMINATE: Wrong DAQ mode (change from Generic to QIE) 
<!-- Author: Taka x2080 -->


<!-- Date: Wed Dec 10 09:17:45.029 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=175995">175995</A>
ACTIVATE: Start the run for SMX QIE Default NonStandard.
<!-- Author: Taka x2080 -->


<!-- Date: Wed Dec 10 09:17:46.369 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=175995">175995</A>
TERMINATE: End the run for SMX QIE Default NonStandard.
<!-- Author: Taka x2080 -->




<!-- Date: Wed Dec 10 09:21:17.047 2003 -->
Dee disabled VESDA for the work of Dan and Camille.
<!-- Author: vaia -->


<!-- Date: Wed Dec 10 09:25:14.489 2003 -->
<pre>Unable calibrations right now. 
 
(1) Plug Laser.   (HV is turned off and being fixed.) 
(2) TOF QIE       (Mattew worked for WCAL) 
(3) COT           (shouldn't be; due to Silicon workings) 
(4) CEM XEF       (very known problem) 
(5) TOF TAC       (Mattew worked for WCAL) 
(6) PMT spike run (Mattew worked for WCAL) 
 
We can do (2), (5) and (6) later. But, (1),(3) and (4) is hopeless 
for a while.</pre>
<!-- Author: Taka -->



<!-- Date: Wed Dec 10 09:33:16.279 2003 -->
<pre>Gas alarms.  O2 analyzer alarm (O2 alarm summary).  Silenced by process system tech.</pre>
<!-- Author: Dale -->
<!-- Comment: Wed Dec 10 09;48;17 comment by...Dale -->
Appears to have gone away.
<!-- Date: Wed Dec 10 09:37:37.661 2003 -->
<pre>We will turn off b0wcal04 to replace ADMEM S/N 369 in slot 9 
with ADMEM 049.</pre>
<!-- Author: M Jones -->

<!-- Date: Wed Dec 10 09:43:56.192 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=175996">175996</A>
ACTIVATE: Start the run for cosmic no track.

<!-- Author: Taka x2080 -->




<!-- Date: Wed Dec 10 09:51:50.702 2003 -->
<pre>PSM alarm - End Wall - CONET _1.  All voltages are 0 with black background.  Will ingore for now.</pre>
<!-- Author: Dale -->
<!-- Comment: Wed Dec 10 10;34;49 comment by...Dale -->
opps - CONET_4 not _1
<!-- Date: Wed Dec 10 09:54:29.818 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=175996">175996</A>
TERMINATE: End the run to take short another calibration for CAL QIE.

<!-- Author: Taka x2080 -->



<!-- Date: Wed Dec 10 09:54:30.254 2003 -->
<pre>Network connection appears to be down for the 'acnet' PC on the west wall (2RR03G).  Expert suggested to restart PC.</pre>
<!-- Author: Dale -->
<!-- Comment: Wed Dec 10 10;10;59 comment by...Dale -->
No effect.  Still no network connection.
<!-- Date: Wed Dec 10 09:56:41.255 2003 -->
Serguei Bourov reported that upstairs (3rd fl) computer room 
temp. was going up and higher than it should be.  Dervin and 
Serguei looked into more (also moved fans around).  The temp. went up due to some work went on earlier this morning (door 
was open for a while), but now it is around 75F and not going 
up.  So, it is all right. 
<!-- Author: kaori -->


<!-- Date: Wed Dec 10 09:57:36.106 2003 -->
We cannot take calorimeter QIE calibration  
right now since Mattew's works. 
<!-- Author: Taka -->

<!-- Date: Wed Dec 10 09:58:28.598 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=175997">175997</A>
ACTIVATE: Start the run cosmic no tracks. 
<!-- Author: Taka x2080 -->



<!-- Date: Wed Dec 10 10:02:53.738 2003 -->
<table border=0><td><tr><td bgcolor=#e0e030><pre>Status of Accelerator Complex 
 
Fix of the spool pieces in TeV on track.   Should finish mechanical work by noon thursday, then leak check begins 
 
TeV rolls in C section done today/tomorrow 
 
Booster fixed some of its alignment problems and losses are better 
 
Recycler could not find its "vacuum leak".  They hooked up an RGA  
and did not see the helium or oxygen.  The gasses they measure  
look like what you expect for vacuum.  So, perhaps they have a  
sector that was not fully baked out.  They fired the sublimation 
pumps and moved on.  Will monitor this section and leave the RGA 
attached.  (if it were Mike and I, we would have baked it out  
again right now) 
 
Tevatron guys still working on a plan to move our low beta  
quad positions.  Plan will be presented tomorrow (hopefully).   
They feel that they can wait until Sunday if necessary and still 
complete the moves that they currently have in mind.  To be 
continued....</pre></td></tr></td></table>
<!-- Author: Mike and Rob -->


<!-- Date: Wed Dec 10 10:03:00.349 2003 -->
Taking COT crates for checkout.
<!-- Author: ambrose -->
<!-- Comment: Wed Dec 10 10;06;15 comment by...ambrose -->
<pre>COT plug-ins look good, ready for Silicon cool-down.</pre>
<!-- Date: Wed Dec 10 10:04:24.718 2003 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2003&action=view&page=-13158&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-13158  width='300' align='top'></a></td><td>This plot seems to be different from the reference. Besides, there is no comparable references for that calibration plots in appropriate location</td></tr></table>
<!-- Author: Fiodar -->
<!-- Comment: Wed Dec 10 11;34;03 comment by...convery -->
This is because the version of DBANA changed yesterday.  I will update references as soon as possible, but not today.  This plot looks fine, by the way.
<!-- Date: Wed Dec 10 10:14:33.520 2003 -->
<pre>Update on BMU work: 
 
BMU SW HV remains off. 
 
Dan and Camille were able to find the bad chamber via quick binary search.  Their repair work is underway.</pre>
<!-- Author: Dale -->


<!-- Date: Wed Dec 10 10:16:05.415 2003 -->
<pre>Sometime ago we put the CPR HV on for cosmics.  The rest of shower max is off.  (See entry in last shift)</pre>
<!-- Author: Dale -->
<!-- Comment: Wed Dec 10 10;22;31 comment by...Dale -->
<a href=
'/cgi/elog/elog.pl?nb=2003&action=view&page=1029&frame=2&anchor=08:27:17.094'>
See here. </a>
<!-- Date: Wed Dec 10 10:17:22.104 2003 -->
VxWorks Monitor shows b0wcal04 is red. 
May be due to experts' workings.
<!-- Author: Taka -->
<!-- Comment: Wed Dec 10 10;20;13 comment by...Taka -->
<pre>fixed.</pre>
<!-- Date: Wed Dec 10 10:18:23.914 2003 -->
<pre>BIGBROTHER:  b0dap52 still error: 
 
red Wed Dec 10 10:17:43 CST 2003 ERROR: Can't connect to 131.225.236.162</pre>
<!-- Author: Taka -->
<!-- Comment: Wed Dec 10 10;19;17 comment by...Taka -->
<pre>also b0dau35 disk:

yellow Wed Dec 10 10:14:09 CST 2003 - Disk on b0dau35,fnal,gov at WARNING level</pre>


<!-- Date: Wed Dec 10 10:19:53.470 2003 -->
Trigger rate is about 45Hz
<!-- Author: Taka -->



<!-- Date: Wed Dec 10 10:22:34.073 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=175999">175999</A>
ACTIVATE: TAC calibration after replacing ADMEM in b0wcal04 slot 9.
<!-- Author: m jones x2080 -->



<!-- Date: Wed Dec 10 10:22:49.472 2003 -->
Kaori wants to check the consumer monitor with current configuration. We will keep this situation as long as possible.
<!-- Author: Taka -->

<!-- Date: Wed Dec 10 10:26:13.064 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176000">176000</A>
ACTIVATE: TOF QIE calibration after replacing ADMEM in b0wcal04 slot 9.
<!-- Author: m jones x2080 -->



<!-- Date: Wed Dec 10 10:33:09.908 2003 -->
<pre>Problems with "beam divsions" network (PC at 2RR03G and the beam division web page) have suddenly gone away after paging CDF online expert. 
 
ACNET up and seems to be running fine.</pre>
<!-- Author: Dale -->


<!-- Date: Wed Dec 10 10:34:42.705 2003 -->
<pre>Paged online computers because we could not access any 
acnet devices. At the same time I could not access the 
BEams Dovision home page. May be these two things were related. 
Sergei called but the we gained access again to ACNET etc right before 
the phone rang.</pre>
<!-- Author: vaia -->



<!-- Date: Wed Dec 10 10:39:20.015 2003 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2003&action=view&page=-13159&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-13159  width='300' align='top'></a></td><td>on 21E we have Sigma > 10</td></tr></table>
<!-- Author: Fiodar -->
<!-- Comment: Wed Dec 10 10;43;30 comment by...V Tiwari -->
This channel is a PIN diode channel. Only used for diagnostics and not for data-taking. Usually these PIN diodes go nuts during the access. So this is not a problem. 

<!-- Comment: Wed Dec 10 10;44;37 comment by...V Tiwari -->
In PHA, channels 0 and 1 are PIN diode channels (true for all the wedges) and that is why you see so many of them.

<!-- Date: Wed Dec 10 10:55:37.403 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=175997">175997</A>
TERMINATE: end the run
<!-- Author: Taka x2080 -->


<!-- Date: Wed Dec 10 10:56:44.912 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176001">176001</A>
ACTIVATE: TOF QIE calibration after reseating the QIE card for channel 14 on ADMEM S/N 049.
<!-- Author: m jones x2080 -->



<!-- Date: Wed Dec 10 10:56:52.552 2003 -->
CMX trip on NW wedge 20.  Brought back up.
<!-- Author: Dale -->

<!-- Date: Wed Dec 10 11:03:12.183 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176004">176004</A>
ACTIVATE: TOF TAC calibration after replacing ADMEM in b0wcal04 slot 9 and reseating CAFE card in channel 14.
<!-- Author: m jones x2080 -->


<!-- Date: Wed Dec 10 11:04:55.344 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176003">176003</A>
ACTIVATE: start the cosmic ray (with no track) run again.
Jonathan wants to have 2 crates for b0pcal01 and 1 cot crate.
(we removed these two)
<!-- Author: Taka x2080 -->
<!-- Comment: Wed Dec 10 11;07;56 comment by...Taka -->
<pre>Actually, Jonathan wants only "resource"s for these two crates.</pre>
<!-- Comment: Wed Dec 10 11;13;07 comment by...Taka -->
<pre>Before activate this run, we shepharded the b0svt06.
The shepharding was succeeded.</pre>
<!-- Date: Wed Dec 10 11:04:56.259 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176004">176004</A>
TERMINATE: Didn't pass activate transition.
<!-- Author: m jones x2080 -->


<!-- Date: Wed Dec 10 11:07:56.350 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176008">176008</A>
ACTIVATE: TOF TAC calibration after replacing ADMEM in b0wcal04 slot 9 and reseating CAFE card for ch. 14.
<!-- Author: m jones x2080 -->




<!-- Date: Wed Dec 10 11:16:46.881 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176003">176003</A>
TERMINATE: End of run to remove the inhibit crate.
<!-- Author: Taka x2080 -->



<!-- Date: Wed Dec 10 11:17:02.844 2003 -->
Dervin repaired the cable of West Plug, crate 2. 
Willis should check it out. 
I am paging him.
<!-- Author: vaia -->
<!-- Comment: Wed Dec 10 11;40;40 comment by...Willis -->
The plug HV CAEN crate 2 HV cable bundle repairs seems to have
worked. CAEN crate 2 came back up, and I ramped the HV to
Standby and then full On without any problems. I have left
the plug HV at full ON, and would like it left on for the 
next several hours if possible.
<!-- Comment: Wed Dec 10 11;43;28 comment by...vaia -->
<pre>Willis says teh Plug HV should saty on for now.
HE checked the repaired channel and it looks good.

</pre>
<!-- Date: Wed Dec 10 11:21:03.134 2003 -->
<pre>I removed the inhibit crate for an expert. 
Also, removed ccal00 and cmp00 for the expert.</pre>
<!-- Author: Taka -->

<!-- Date: Wed Dec 10 11:23:13.416 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176009">176009</A>
ACTIVATE: Start the cosmic ray run with no track.
<!-- Author: Taka x2080 -->



<!-- Date: Wed Dec 10 11:26:43.715 2003 -->
We removed following crates from normal cosmic with no tracks: 
 
(1) b0wcal**  (Mattew Jones) 
(2) b0inh00, b0ccal00, b0cmp00  (for an inhibit expert) 
(3) b0pcal01 (1 COT crate)      (for Jonathan) 
  
<!-- Author: Taka -->


<!-- Date: Wed Dec 10 11:32:05.739 2003 -->
<pre>The plug appear to becoming back to life (at stand-by).  Willis is here and wants us to keep the HV on (being brought up now).</pre>
<!-- Author: Dale -->
<!-- Comment: Wed Dec 10 11;34;30 comment by...Dale -->
<pre>Trip on PEM on CAEN 2 - wedge 6 on the west (same as before).  Alarm has gone away.  </pre>


<!-- Date: Wed Dec 10 11:37:29.657 2003 -->
<pre>We are going to turn of b0wcal03.</pre>
<!-- Author: M Jones -->

<!-- Date: Wed Dec 10 11:38:44.269 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176009">176009</A>
TERMINATE: Finished the run to take CEM LED run in D-mode for EMTiming system
<!-- Author: Taka x2080 -->


<!-- Date: Wed Dec 10 11:40:39.048 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176010">176010</A>
ACTIVATE: CEM LED run in D-mode for EMTiming system.
<!-- Author: Max x2120 -->



<!-- Date: Wed Dec 10 11:41:32.777 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176010">176010</A>
TERMINATE: Finished CEM LED run
<!-- Author: Max x2120 -->




<!-- Date: Wed Dec 10 11:49:29.583 2003 -->
Plug HV is now on. Willis fixed the problems.
<!-- Author: Taka -->
<!-- Comment: Wed Dec 10 11;59;20 comment by...Dale -->
We should keep the plug on for a least a couple of hours.
<!-- Comment: Wed Dec 10 12;20;39 comment by...Taka -->
Why?
<!-- Date: Wed Dec 10 11:50:18.313 2003 -->
<pre>We are finished working on TOF for today.  Here's what we 
did: 
 
1.  Replaced ADMEM in b0wcal04 slot 9 S/N 369 with S/N 049. 
2.  Inspected the VTT enable cable in b0wcal01. 
 
We did not work on TOAD0 on the TOMAIN board in b0wcal03 slot 9 
because the north west CMX arch is closed.</pre>
<!-- Author: M Jones/G Bauer/G Gomez -->


<!-- Date: Wed Dec 10 11:56:39.986 2003 -->
<pre>Dan and Camille (IMU experts) have made a repair to the BMU.  HV is on every where with logical current values on both the anode and cathode.  They are on their way up from b0.</pre>
<!-- Author: Dale -->



<!-- Date: Wed Dec 10 12:01:35.352 2003 -->
Eric James needs the hadron TDC crate and the muon trigger crate 
for studies that will last about 15 minutes. 
We also give to Vivek and Fotis time to for a test to determine 
if a certain problem at WHA 16E 8L is electronics related
<!-- Author: vaia -->


<!-- Date: Wed Dec 10 12:04:45.721 2003 -->
<pre>w-cdf-detector1 and w-cdf-detector2 have been pluged in to allow the wireless network in b0.</pre>
<!-- Author: Dale -->


<!-- Date: Wed Dec 10 12:06:17.455 2003 -->
<pre>Camille and Dan finishe shuttering. 
VESDA is eneabled by Dee.</pre>
<!-- Author: vaia -->

<!-- Date: Wed Dec 10 12:08:28.193 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176013">176013</A>
 Activated at 2003.12.10 12:07:49
<!-- Author: RunControl -->


<!-- Date: Wed Dec 10 12:08:30.719 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176013">176013</A>
ACTIVATE: Spike run to debug WHA 16E, 8L.
<!-- Author: Max x2120 -->
<!-- Comment: Wed Dec 10 12;27;51 comment by...Taka -->
<pre>Because Mattew et al finished their working, we can include
b0wcal** from now. Calorimeter experts want to work with these
modules so that we paged them.</pre>
<!-- Date: Wed Dec 10 12:10:32.783 2003 -->
IMU Access Report : HV trip in SW84 of BMU last night was isolated with Dale's help upstairs to SW91. The high voltage board was replaced and all BMU sectors are now holding HV as of now. 
<!-- Author: C. Ginsburg, Dale S. D. Cyr -->
<!-- Comment: Wed Dec 10 12;12;24 comment by...dan -->
Note to shift crew - returning control of the HV to you, please inform Camille or Dan if problems occur.


<!-- Date: Wed Dec 10 12:25:56.922 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176013">176013</A>
 Terminated at 2003.12.10 12:24:55
<!-- Author: RunControl -->


<!-- Date: Wed Dec 10 12:25:57.274 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176013">176013</A>
TERMINATE: Spike run ends. waiting for consumers.
<!-- Author: Vivek x2120 -->



<!-- Date: Wed Dec 10 12:29:50.706 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176015">176015</A>
 Activated at 2003.12.10 12:28:56
<!-- Author: RunControl -->


<!-- Date: Wed Dec 10 12:29:52.765 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176015">176015</A>
ACTIVATE: Another spike sun to debug WHA 16E, 8L.
<!-- Author: Vivek x2120 -->


<!-- Date: Wed Dec 10 12:42:46.371 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176015">176015</A>
 Terminated at 2003.12.10 12:41:57
<!-- Author: RunControl -->


<!-- Date: Wed Dec 10 12:42:47.458 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176015">176015</A>
TERMINATE: End spike run.
<!-- Author: Vivek x2120 -->


<!-- Date: Wed Dec 10 12:45:06.936 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176016">176016</A>
ACTIVATE: CAL QIE calib without pcal01 and ccal00.
<!-- Author: Vivek x2120 -->


<!-- Date: Wed Dec 10 12:45:17.357 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176016">176016</A>
TERMINATE: End CAL QIE.
<!-- Author: Vivek x2120 -->



<!-- Date: Wed Dec 10 12:54:18.574 2003 -->
Eric James finished his working.  
<!-- Author: Taka -->


<!-- Date: Wed Dec 10 12:59:13.111 2003 -->
We have finished with the hardware check-out of the silicon.  We probed the junction card connections directly and didn't find any connection problems.  We are ready to cool down and to start DAQ testing of the system to make sure we are in a better state than before!  (More details in the  
<a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=176&frame=2&anchor=12:53:15'> 
silicon e-log</a>.)
<!-- Author: Matt H., Jason N. -->


<!-- Date: Wed Dec 10 13:00:32.271 2003 -->
Start cooling down the silicon.  You will have alarms until it's finished.
<!-- Author: Andy H. -->

<!-- Date: Wed Dec 10 13:01:33.224 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176017">176017</A>
ACTIVATE: TOF QIE run (needed?)
<!-- Author: Vivek x2120 -->


<!-- Date: Wed Dec 10 13:01:34.982 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176017">176017</A>
TERMINATE: end the run
<!-- Author: Vivek x2120 -->



<!-- Date: Wed Dec 10 13:02:33.130 2003 -->
We have noticed that WHA 16E, 8L is dead in the laser runs. However the PMT is alive, electronics is fine and also the COSMIC runs show that the PMT is alive. The suspicion was that the electronics was not behaving properly at high ADC counts (~7000 counts for laser runs). We have faked the channel by turning the full scale charge injection ON and have verified that the output of the channel is indeed maxminmum energy. So the cafe-card is behaving as it should.
<!-- Author: Foris, Vivek -->

<!-- Date: Wed Dec 10 13:03:50.652 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176019">176019</A>
ACTIVATE: TOF TAC calibration run
<!-- Author: Taka x2120 -->


<!-- Date: Wed Dec 10 13:05:50.858 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176019">176019</A>
TERMINATE: end the run (TOF TAC)
<!-- Author: Taka x2120 -->



<!-- Date: Wed Dec 10 13:07:08.444 2003 -->
<pre>Adjusted the timing of hadron tdc primitives arriving 
at the muon trigger crate.  </pre>
<!-- Author: Eric James -->


<!-- Date: Wed Dec 10 13:08:48.732 2003 -->
I did TOF and TOF TAC calibration. These were not done in the  
previous calibrations in the morning.
<!-- Author: Taka -->


<!-- Date: Wed Dec 10 13:11:44.237 2003 -->
Florencia and Julia are coming for Silicon DAQ Testa at 3:00 pm. 
 
 
 
 
 
 
  
<!-- Author: vaia -->


<!-- Date: Wed Dec 10 13:22:06.554 2003 -->
Called Jonathan because the ACES could not do a plug laser calibration 
and the trigger type they neede dfor that was not available.
<!-- Author: vaia -->

<!-- Date: Wed Dec 10 13:30:28.481 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176021">176021</A>
ACTIVATE: plug laser calibration BOTHPLUGS_NORMAL (but pcal01 is out).
<!-- Author: Taka x2120 -->
<!-- Comment: Wed Dec 10 13;33;17 comment by...Dale and Taka -->
<pre>This is using the new TT COSMICS[12,358,405].  This has been saved (before we removed pcal01).  We will save the other plug laser calibrations.</pre>

<!-- Date: Wed Dec 10 13:37:06.423 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176021">176021</A>
TERMINATE: Ending BOTHPLUGS_NORMAL
<!-- Author: Taka x2120 -->


<!-- Date: Wed Dec 10 13:41:29.448 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176023">176023</A>
ACTIVATE: Starting BOTHPLUGS_MM_C <p>With new trigger table COSMICS[12,358,405] (saved) and w/o pcal01</p>
<!-- Author: Taka x2120 -->
<!-- Comment: Wed Dec 10 13;44;55 comment by...Dale -->
<pre>The orginal 'old' trigger table was COSMICS[11,284,369].  JDL figured out that the new L3 tag set had to be 405 ("null v2" in discription).  He pointed out that the L2 tag was not important for the calibration.</pre>
<!-- Date: Wed Dec 10 13:42:36.370 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176023">176023</A>
TERMINATE: end the run
<!-- Author: Taka x2120 -->



<!-- Date: Wed Dec 10 13:46:23.023 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176025">176025</A>
ACTIVATE: start the run for BOTHPLUGS_MM with new trigger table but w/o b0pcal01. 
<!-- Author: Taka x2120 -->
<!-- Comment: Wed Dec 10 13;47;11 comment by...Taka -->
With new trigger table COSMICS[12,358,405] (saved) and w/o pcal01
<!-- Date: Wed Dec 10 13:47:03.954 2003 -->
Fotis and I checked the transition card for wcal06, slot 20 (specially ch 4 on which WHA 16E, 8L is fed). Didn't find anything wrong with with it. All the connections are good and there are no bent pins. Fotis turned on the laser and we see signal on all the PMTs except WHA 16E, 8L. Now Fotis and Nickolay are going to check the PMT.
<!-- Author: Fotis, Vivek -->


<!-- Date: Wed Dec 10 13:47:42.287 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176025">176025</A>
TERMINATE: End of the run 
<!-- Author: Taka x2120 -->


<!-- Date: Wed Dec 10 13:54:04.918 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176026">176026</A>
 Activated at 2003.12.10 13:53:11
<!-- Author: RunControl -->


<!-- Date: Wed Dec 10 13:54:19.906 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176026">176026</A>
ACTIVATE: Start the run for the spike run. (w/o b0ccal00, b0pcal01, b0inh00)
<!-- Author: Taka x2120 -->


<!-- Date: Wed Dec 10 13:54:46.601 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176026">176026</A>
 Terminated at 2003.12.10 13:54:14
<!-- Author: RunControl -->


<!-- Date: Wed Dec 10 13:54:47.524 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176026">176026</A>
TERMINATE: End the run since plug laser is on. (junk run)
<!-- Author: Taka x2120 -->
<!-- Comment: Wed Dec 10 13;59;20 comment by...Taka -->
<pre>I turned off the plug laser. 
 </pre>

<!-- Date: Wed Dec 10 14:07:45.509 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176027">176027</A>
 Activated at 2003.12.10 14:07:18
<!-- Author: RunControl -->



<!-- Date: Wed Dec 10 14:08:05.295 2003 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2003&action=view&page=-13160&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-13160  width='300' align='top'></a></td><td>real-time plotter of silicon cooling [almost at 0.0 C]</td></tr></table>
<!-- Author: Dale -->
<!-- Comment: Wed Dec 10 14;24;57 comment by...Dale -->
(<a href=
'/cgi/elog/elog.pl?nb=2003&action=view&page=1030&frame=2&anchor=13:00:32.271'>
silicon is cooling)</a> 
<p>
Cooling continues - chiller return was slowly reaching the setpoint of 0.0 C.  It has been set for -6.0 C since reaching this limit.  Currently just under 0.0 C.  We will page Andy once the chiller return is ~ -6 C.
</p>
<!-- Date: Wed Dec 10 14:08:42.190 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176027">176027</A>
ACTIVATE: Start the run for PMT spike run (w/o b0pcal01, b0ccal00, b0inh00)
L1 rate is about 120Hz.
<!-- Author: Taka x2120 -->



<!-- Date: Wed Dec 10 14:11:01.528 2003 -->
Unmask PHA eta=1 phi=3 for Vivek to check
<!-- Author: smaria -->
<!-- Comment: Wed Dec 10 14;16;40 comment by...mark, vivek -->
Can't check. somebody has pcal01.
<!-- Date: Wed Dec 10 14:11:49.494 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176027">176027</A>
 Terminated at 2003.12.10 14:11:27
<!-- Author: RunControl -->


<!-- Date: Wed Dec 10 14:16:39.042 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176027">176027</A>
TERMINATE: End of the PMT spike run.
Average of L1 rate in this run is 123.7Hz.
<!-- Author: Taka x2120 -->
<!-- Comment: Wed Dec 10 14;18;12 comment by...mark, vivek -->
PMT spike run shows DCAS-ADMEM mismatches from eta=8, phi=0,1,2. This was expected because b0ccal00 was out of the run.

<!-- Date: Wed Dec 10 14:20:18.452 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176028">176028</A>
ACTIVATE: start the cosmic with no track. (w/o b0pcal01, b0ccal00, b0inh00)
<!-- Author: Taka x2120 -->
<!-- Comment: Wed Dec 10 14;26;22 comment by...Taka -->
<pre>Dale suggested that we might remove b0inh00 in cosmic ray run.
We can remove this crate in next time.</pre>
<!-- Date: Wed Dec 10 14:22:06.603 2003 -->
<pre>Jonathan has crate pcal01 and MArk and Vivek want the same crate 
to investigate a hot tower. 
Jonathan thinks he will be done in about 30 minutes from now.</pre>
<!-- Author: vaia -->




<!-- Date: Wed Dec 10 14:33:59.532 2003 -->
<pre>Busy timeouts on WCAL_05 and WCAL_06 [b0wacal_06].  PSM alarm.</pre>
<!-- Author: Dale -->
<!-- Comment: Wed Dec 10 14;35;03 comment by...Dale and Taka -->
<pre>Error massage:


  State       ACTIVATE
1 crate/s: b0wcal05(21),  Done TO.2 crate/s: b0wcal05(21), b0wcal06(22),  busy.1 crate/s: b0wcal05(21),  in error.[RXPT]
(MLE) b0dap73.fnal.gov:Thread-10262:2:29:35 PM-&gt;Done Timeout: WCAL_05 
(MLE) b0dap73.fnal.gov:Thread-10262:2:29:35 PM-&gt;FrontEnd Crate Error Condition from:
WCAL_06

(MLE) b0dap73.fnal.gov:Thread-10262:2:29:35 PM-&gt;FrontEnd Error: WCAL_06 
(MLE) b0dap73.fnal.gov:Thread-10262:2:29:35 PM-&gt;Busy Timeout: WCAL_05 WCAL_06 
(MLE) b0tsi02:Messenger:2:29:35 PM-&gt;BUSY timeout
(MLE) b0tsi02:Messenger:2:29:39 PM-&gt;DONE timeout
(MLE) b0tsi02:Messenger:2:29:39 PM-&gt;BUSY timeout</pre>


<!-- Date: Wed Dec 10 14:37:32.743 2003 -->
<pre>PSM alarm COSET_4, COSEB_3.  
This looks due to experts' working.</pre>
<!-- Author: Taka -->
<!-- Comment: Wed Dec 10 14;39;41 comment by...Taka -->
<pre>This is the source of done timeout. We will Recover and Run
immidiately after this is cleared.</pre>

<!-- Date: Wed Dec 10 14:45:46.447 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176028">176028</A>
TERMINATE: Abort the run since we have rc stucked.
<!-- Author: Taka x2120 -->



<!-- Date: Wed Dec 10 14:57:17.839 2003 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2003&action=view&page=-13161&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2003&action=view&page=-13161  width='300' align='top'></a></td><td>Getting close to -6.0 C. (Now about ~-5 C)</td></tr></table>
<!-- Author: Dale -->


<!-- Date: Wed Dec 10 15:00:45.504 2003 -->
I have fixed the reset lines for b0cot17 and b0pcal01.  cot17 just had a flaky connector.  For pcal01, both the regular cable and spare cable are bad.  Reterminating did not help.  I kludged in a new line by taking over one of the former diagnostic ethernet cables.
<!-- Author: jdl -->


<!-- Date: Wed Dec 10 15:02:01.639 2003 -->
<pre>Some improvements to inhibit gui: 
   1) A help button responds. 
   2) System is *supposed*  to know what is in partition when 
      started from run control (not standalone). 
   3) This is done during a partition and will change as 
      elements are or are not included in partition. 
   4) Help button actually brings up some information. 
 
Work remains however: 
   1) There is a request to have system start standalone in 
       a different form.  
   2) Undefined signals are a problem. 
   3) Aesthetic issues. 
   These will be fixed tomorrow. 
  In meantime, it would be useful if shift crew starts it 
   up (with partition as an argument) and, if not bored,  
   see if colors and changes to them are sensible.</pre>
<!-- Author: steve v. -->

<!-- Date: Wed Dec 10 15:05:47.552 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176029">176029</A>
 Activated at 2003.12.10 15:05:16
<!-- Author: RunControl -->


<!-- Date: Wed Dec 10 15:06:37.374 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176029">176029</A>
ACTIVATE: Spike run with only Plug crates. Plug HV is ON.
<!-- Author: Taka x2120 -->


<!-- Date: Wed Dec 10 15:10:53.679 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176029">176029</A>
 Terminated at 2003.12.10 15:09:30
<!-- Author: RunControl -->


<!-- Date: Wed Dec 10 15:10:54.286 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176029">176029</A>
TERMINATE: End run. PHA eta=1, phi=3 shows mismatches on DCAS-ADMEM plot.
<!-- Author: Taka x2120 -->



<!-- Date: Wed Dec 10 15:13:50.639 2003 -->
<pre>Spike run taken. 
Mismatches seens for PHA, eta=1, phi=3. 
Vivek and MArk investigating.</pre>
<!-- Author: vaia -->


<!-- Date: Wed Dec 10 15:13:53.841 2003 -->
Silicon is cold again.  Treat alarms as real (although I imagine the test-festers will be causing a few...)
<!-- Author: Andy H. -->


<!-- Date: Wed Dec 10 15:15:46.091 2003 -->
<pre>Will start taking cosmics. 
The silicon testers are almost here. 
They will first use their standalone DAQ and then take 
calibrations. After that they should be integrated in cosmic runs.</pre>
<!-- Author: vaia -->


<!-- Date: Wed Dec 10 15:16:45.430 2003 -->
<pre>We have error: 
 
Failed to Book: 
MUON_CAMAC_00 
MUON_SCINT_00 
MUON_TRIGGER_00 
</pre>
<!-- Author: Taka -->
<!-- Comment: Wed Dec 10 15;17;35 comment by...Taka -->
Looks like Jonathan uses these one.


<!-- Comment: Wed Dec 10 15;49;44 comment by...Dale and Taka -->
<pre>We are not sure what were going on this.
But, Jonathan's partition#15 somehow have the these crates.
(Someone wrongly pushed the button??)
Anyway, we released all of modules from Jonathan's partition 
with Jonathan's agreement.
</pre>
<!-- Date: Wed Dec 10 15:27:21.183 2003 -->
We removed b0pcal01, b0ccal05 and b0ccal06. 
These are all due to experts' workings.
<!-- Author: Taka -->


<!-- Date: Wed Dec 10 15:29:04.067 2003 -->
It looks like b0ccal05 and b0ccal06 were released by experts. 
We will include these.
<!-- Author: Taka and Dale. -->


<!-- Date: Wed Dec 10 15:29:24.913 2003 -->
<pre>Silicon is cool. The silicon crew is here and start their tests.</pre>
<!-- Author: vaia -->

<!-- Date: Wed Dec 10 15:33:41.255 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176032">176032</A>
ACTIVATE: Central laser run - testing WHA 16E-8L
<!-- Author: Taka x2120 -->



<!-- Date: Wed Dec 10 15:36:55.426 2003 -->
<pre>Just as planed b0pcal01 alarmed when it was turned off by Vivek and mark for their work.  Voltage restored (alarm goes away).</pre>
<!-- Author: Dale and Taka -->
<!-- Comment: Wed Dec 10 15;37;17 comment by...Dale -->
mark ---> Mark


<!-- Date: Wed Dec 10 15:46:44.332 2003 -->
<pre>WHA global alarm is yellow. 
This is known problem by expert (Nikolay). 
</pre>
<!-- Author: Taka -->

<!-- Date: Wed Dec 10 15:48:59.546 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176033">176033</A>
 Activated at 2003.12.10 15:48:35
<!-- Author: RunControl -->



<!-- Date: Wed Dec 10 15:51:53.229 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176033">176033</A>
ACTIVATE: Spike run, after ADMEM swap between slots 15 and 17 in pcal01. Plug HV is ON. Laser in ON.
<!-- Author: Vivek x2120 -->


<!-- Date: Wed Dec 10 15:54:48.576 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176033">176033</A>
 Terminated at 2003.12.10 15:53:59
<!-- Author: RunControl -->


<!-- Date: Wed Dec 10 15:54:49.062 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176033">176033</A>
TERMINATE: PHA eta=1, phi=3 doesn't show mismatches.
<!-- Author: Vivek x2120 -->
<!-- Comment: Wed Dec 10 16;09;51 comment by...m mattson and v tiwari -->
<p>
We asked Maria S to unmask the
<a href=
'/cgi/elog/elog.pl?nb=2003&action=view&page=981&frame=2&anchor=03:59:26.691'>
hot PHA tower</a>, and the problem was still visible.  We swapped ADMEMs in neighboring wedges (eta 1, phi's 3 and 4), to see if the problem follows the electronics.  The problem has since gone away.<p>
We are hoping that it was a transient connection problem, but we will keep an eye out in case the problem comes back.

<!-- Date: Wed Dec 10 15:55:40.218 2003 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>15:55:01</TD>                                                       
<TD ALIGN=CENTER>::</TD>                                                
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->


<!-- Date: Wed Dec 10 15:56:50.793 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176034">176034</A>
ACTIVATE: CAL QIE only for plug with a possible download. 
<!-- Author: Vivek x2120 -->



<!-- Date: Wed Dec 10 16:02:28.623 2003 -->
<pre>Two trips to CMX SE HV on wedge 16-17.  The trips may have been due to "human interaction" or the many cables, cords, etc. in the area.  It came back up after I inspected it and double checked the anode fuse boxes and the cathode shv cables.  As the high voltage came up its proably fine.  As work continues in this area random trips should be expected.  Look for activity when they close up ~Thursday morning.</pre>
<!-- Author: Dale -->


<!-- Date: Wed Dec 10 16:06:23.361 2003 -->
<font color=red>Shift Summary: </font><pre>Plan: 
------ 
Do all calibrations and exercise DAQ. Be ready for silicon 
being used in calibrations and cosmics around noon. 
 
 
10:00 am: We did all calibrations except from plug laser,TOF QIE, 
TOF TAC, COT and CEM XEF because crates were used by experts. 
Spike run wa snot done for the same reason. 
Calibrations look mostly OK. 
 
Plug HV problem repaired by Dervin. 
Willis says to leave the voltage ON for now. 
 
Work on inhibit crate by Steve Vecjik 
 
CEM LED run taken by Max for timing studies. 
 
Took plug laser calibrations and TOF calibrations 
now that teh crates are available and they look good. 
 
 
The inhibit crate tests seem succesful so far. 
Several improvements. More work remaining. 
 
Took spike run. Mismatches seen for PHA, eta=1, phi=3. 
Either trigger problem or electronics. 
Mark and Vivek will try to isolate and we will need to take another  
spike run after they finsih their investigation 
 
Silicon will start tests areound 3:30 pm. 
Silicon is cool. 
First they will use their standalone DAQ. 
Then they will take silicon calibrations and after that they 
will be integrated in the cosmic runs. 
 
COT HV still off. Contact experts to see when to bring it back.</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   0
Delivered Luminosity   0  
Acquired Luminosity    0  
Efficiency             NA
   
</td></tr></table></pre>
<!-- Author: vaia -->

<!-- Date: Wed Dec 10 16:08:04.159 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176034">176034</A>
TERMINATE: End CAL QIE and download for Plug.
<!-- Author: Ace x2080 -->



<!-- Date: Wed Dec 10 16:10:59.240 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176035">176035</A>
 Activated at 2003.12.10 16:09:57
<!-- Author: RunControl -->


<!-- Date: Wed Dec 10 16:11:31.563 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176035">176035</A>
ACTIVATE: Spike run after the FRAM download in Plug. Plug HV is ON. Laser is ON in CHA, WHA.
<!-- Author: Vivek x2080 -->
<!-- Comment: Wed Dec 10 16;13;08 comment by...mark, vivek -->
Laser is what makes the rates slightly higher: 135 Hz instead of 90 Hz.


<!-- Date: Wed Dec 10 16:14:00.607 2003 -->
Replace the pmt on WHA 16E-8L. There is signal on the transition card and it looks ok for the laser. However we don't understand what happened with the old PMT since testing it upstairs it looks alive. One more mystery or an intermittent problem somewhere. It may show up again (at the end of the access probably)
<!-- Author: Nikolay and Fotis -->

<!-- Date: Wed Dec 10 16:14:36.571 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=176035">176035</A>
 Terminated at 2003.12.10 16:13:58
<!-- Author: RunControl -->