<!-- Title: Day shift. Fri Oct 26, 2007 -->
<!-- SciCo: Jim Patrick -->
<!-- DAQAce: Claudia Marino -->
<!-- MonAce: &nbsp; -->
<!-- CO: Timo Aaltonen -->
<!-- OpManager: Steve Hahn -->
<!-- Notes: -->
<pre>non-quiet time</pre>


<!-- Date: Fri Oct 26 08:09:09.571 2007 -->
<P> 
Have rebuilt the acnet package to point to the  
new AcNet receiver node.  The old node was due14; the  
new node is dce34.   The acnet package version is v5_1 
<P> 
The switch happened sometime yesterday;  Benedetto first  
noticed it from the SVT spy_manager process which  
was crashing. 
<P>
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252687' target=_top>252687</a>) -->
<!-- Comment: Fri Oct 26 08;09;33 comment by...badgett -->
and have restarted AcnetScaler, AcnetMonitor and DownTime 
on b0server00


<!-- Date: Fri Oct 26 08:11:17.050 2007 -->
<pre>start of shift status: 
 
- cosmics running on partition 0 (should we use partition 3?) 
 
- no alarms 
 
- HV summary: end of store state 
 
- non-quiet time 
 
- misc:  
  - SVT SPY Monitor grey in ProcMon 
  - TrigMon works but slower, Adrian wrote email to expert 
  - ICICLE status not checked correctly in ProcMon, IFIX  
 
- shift plan: switch between l2 torture and cosmics, do non-quiet time calibrations</pre>
<!-- Author: Claudia -->

<!-- Date: Fri Oct 26 08:14:53.264 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252687>252687</A>
 Terminated at 2007.10.26 08:14:45
<!-- Author: RunControl -->


<!-- Date: Fri Oct 26 08:15:11.935 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252687>252687</A>
TERMINATE: end to start non-quiet time calibrations
<!-- Author: Adrian Buzatu -->


<!-- Date: Fri Oct 26 08:17:19.459 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252688>252688</A>
ACTIVATE: RunSet: ACE_TOF_QIE_NEW 
<!-- Author: Adrian Buzatu -->


<!-- Date: Fri Oct 26 08:17:55.575 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252688>252688</A>
TERMINATE: done
<!-- Author: Adrian Buzatu -->


<!-- Date: Fri Oct 26 08:17:56.610 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252688>252688</A>
CALIBRATION: <U><B>tofqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#tofqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Oct 26 08;24;17 comment by...Timo -->
Fine.
<!-- Comment: Fri Oct 26 08;25;56 comment by...Timo -->
JJ, I commented documentation for you.
<!-- Date: Fri Oct 26 08:20:06.105 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252689>252689</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Adrian Buzatu -->


<!-- Date: Fri Oct 26 08:21:20.065 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252689>252689</A>
TERMINATE: done
<!-- Author: Adrian Buzatu -->


<!-- Date: Fri Oct 26 08:23:27.555 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252690>252690</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: Adrian Buzatu -->



<!-- Date: Fri Oct 26 08:25:26.261 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252690>252690</A>
CALIBRATION: <U><B>cemled</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cemled target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Oct 26 08;30;43 comment by...Timo -->
Fine.
<!-- Date: Fri Oct 26 08:25:27.259 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252690>252690</A>
TERMINATE: done
<!-- Author: Adrian Buzatu -->


<!-- Date: Fri Oct 26 08:25:28.280 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252689>252689</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Oct 26 08;33;03 comment by...Timo -->
Fine.
<!-- Date: Fri Oct 26 08:26:06.555 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252691>252691</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Adrian Buzatu -->


<!-- Date: Fri Oct 26 08:26:24.961 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252691>252691</A>
TERMINATE: done
<!-- Author: Adrian Buzatu -->


<!-- Date: Fri Oct 26 08:26:47.715 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252691>252691</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Oct 26 08;35;24 comment by...Timo -->
Too many bad channels.
<!-- Date: Fri Oct 26 08:26:54.017 2007 -->
Bob Wagner will be working on COT low voltage for the next 10 minutes. It may generate an alarm.
<!-- Author: Jim Patrick -->
<!-- Comment: Fri Oct 26 08;47;01 comment by...J. Patrick -->
Bob is now done.
<!-- Date: Fri Oct 26 08:30:40.963 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252692>252692</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: Adrian Buzatu -->



<!-- Date: Fri Oct 26 08:32:24.137 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252692>252692</A>
CALIBRATION: <U><B>calxef</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calxef target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Oct 26 08;37;27 comment by...Timo -->
Fine.
<!-- Date: Fri Oct 26 08:32:25.525 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252692>252692</A>
TERMINATE: done
<!-- Author: Adrian Buzatu -->





<!-- Date: Fri Oct 26 08:36:03.399 2007 -->
COT in alarm in IFIX, known, expert does some work
<!-- Author: Claudia -->
<!-- Comment: Fri Oct 26 08;45;41 comment by...Claudia -->
<a href=
'/cgi/elog/elog.pl?nb=2007&action=view&page=830&frame=2&anchor=08:26:54.017'>
see here </a> 
<!-- Comment: Fri Oct 26 08;48;22 comment by...Claudia -->
COT green again in global alarms, I do another COT calibration as Bob requested
<!-- Date: Fri Oct 26 08:37:56.313 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252693>252693</A>
TERMINATE: done
<!-- Author: Adrian Buzatu -->


<!-- Date: Fri Oct 26 08:37:57.570 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252693>252693</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1, usual error message and Consumer Error
<!-- Author: Adrian Buzatu -->
<!-- Comment: Fri Oct 26 08;39;18 comment by...Claudia -->
<pre>Error message:
Strange (but not necessarily fatal) Run Configuration selections:
Crate IMU_00 is missing card TDC_BMU_03 in slot 14</pre>

<!-- Date: Fri Oct 26 08:40:02.207 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252694>252694</A>
ACTIVATE: RunSet: TOF_TAC_CALIB 
<!-- Author: Adrian Buzatu -->


<!-- Date: Fri Oct 26 08:45:22.054 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252694>252694</A>
TERMINATE: done
<!-- Author: Adrian Buzatu -->


<!-- Date: Fri Oct 26 08:45:41.856 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252694>252694</A>
CALIBRATION: <U><B>toftac</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#toftac target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Oct 26 13;30;18 comment by...Claudia -->
GOOD
<!-- Date: Fri Oct 26 08:48:22.177 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252695>252695</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Adrian Buzatu -->



<!-- Date: Fri Oct 26 08:48:52.745 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252695>252695</A>
TERMINATE: done
<!-- Author: Adrian Buzatu -->


<!-- Date: Fri Oct 26 08:48:53.721 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252695>252695</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Oct 26 08;56;04 comment by...Claudia -->
BAD, too many bad channels, I wrote email to Eiko and paged COT calibration expert
<!-- Comment: Fri Oct 26 08;58;41 comment by...Claudia -->
Eiko replied it is a known problem
<!-- Comment: Fri Oct 26 09;41;47 comment by...J. Patrick -->
Eiko calls and says she will come in between 11 and 11:30 to look at this in more detail, following Bob Wagner's work.
<!-- Date: Fri Oct 26 08:49:43.245 2007 -->
My phone is 
x8414
<!-- Author: Timo :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252695' target=_top>252695</a>) -->

<!-- Date: Fri Oct 26 08:51:17.374 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252696>252696</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [19,986,705] 
<!-- Author: Adrian Buzatu -->
<!-- Comment: Fri Oct 26 09;00;59 comment by...Claudia -->
I dont know why there is still Adrians name, I changed personal client name ... ???
<!-- Date: Fri Oct 26 09:00:57.057 2007 -->
MCR calls and says they will have 1.5 hours of quiet time as soon as they dump the current beam. SVX paged, they will make use of the time.
<!-- Author: Jim Patrick -->



<!-- Date: Fri Oct 26 09:06:22.860 2007 -->
Phil calls from the 9 a.m. accelerator meeting. D0 will be doing an access, the quiet time is now expected to be 3 hours. SVX is beginning a noise bias scan.
<!-- Author: Jim Patrick -->

<!-- Date: Fri Oct 26 09:10:09.517 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252696>252696</A>
TERMINATE: end to do quiet time calibrations
<!-- Author: Claudia Marino -->


<!-- Date: Fri Oct 26 09:14:03.511 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252697>252697</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Oct 26 09;27;01 comment by...Claudia -->
GOOD
<!-- Date: Fri Oct 26 09:14:04.743 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252697>252697</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Claudia Marino -->


<!-- Date: Fri Oct 26 09:14:23.495 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252697>252697</A>
TERMINATE: done
<!-- Author: Claudia Marino -->


<!-- Date: Fri Oct 26 09:17:27.076 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252698>252698</A>
ACTIVATE: RunSet: CALIB_CLC_QIE 
<!-- Author: Claudia Marino -->


<!-- Date: Fri Oct 26 09:18:33.697 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252698>252698</A>
TERMINATE: done
<!-- Author: Claudia Marino -->


<!-- Date: Fri Oct 26 09:18:34.768 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252698>252698</A>
CALIBRATION: <U><B>clcqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#clcqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Oct 26 09;27;43 comment by...Claudia -->
GOOD
<!-- Date: Fri Oct 26 09:20:16.391 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252699>252699</A>
ACTIVATE: RunSet: CALIB_QIE_FWD 
<!-- Author: Claudia Marino -->


<!-- Date: Fri Oct 26 09:21:09.922 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252699>252699</A>
CALIBRATION: <U><B>bscqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#bscqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Oct 26 09;28;50 comment by...Claudia -->
GOOD
<!-- Date: Fri Oct 26 09:22:02.264 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252699>252699</A>
TERMINATE: done
<!-- Author: Claudia Marino -->



<!-- Date: Fri Oct 26 09:22:52.391 2007 -->
I just saw on Process System Alarms that GAS is yellow and silenced. However, I did not get an voice alarm. Alcohol Use Rat is in ALARM -> now it just turned back to be OK, and GAS is unsilenced 
<!-- Author: Claudia -->


<!-- Date: Fri Oct 26 09:25:28.580 2007 -->
For a very short time TOF went in WRN state in HV summary, now it is back, so it seems that everything is fine
<!-- Author: Claudia -->

<!-- Date: Fri Oct 26 09:25:49.921 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252700>252700</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL 
<!-- Author: Claudia Marino -->





<!-- Date: Fri Oct 26 09:34:15.174 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252700>252700</A>
TERMINATE: done
<!-- Author: Claudia Marino -->


<!-- Date: Fri Oct 26 09:35:51.704 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252702>252702</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: Claudia Marino -->


<!-- Date: Fri Oct 26 09:37:56.623 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252702>252702</A>
TERMINATE: done
<!-- Author: Claudia Marino -->


<!-- Date: Fri Oct 26 09:37:57.654 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252702>252702</A>
CALIBRATION: <U><B>pluglaser</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#pluglaser target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Oct 26 12;14;38 comment by...Claudia -->
BAD, East 21: wedge 5-8 and 16 - 18, 20 are bad (red), TDC mean E - PHA is below 550 ns, <a href=
'/cgi/elog/elog.pl?nb=2007&action=view&page=830&frame=2&anchor=11:57:49.590'>
see also </a> 
<!-- Date: Fri Oct 26 09:40:07.450 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252703>252703</A>
ACTIVATE: RunSet: BOTHPLUGS_MM 
<!-- Author: Claudia Marino -->



<!-- Date: Fri Oct 26 09:42:34.786 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252703>252703</A>
TERMINATE: done
<!-- Author: Claudia Marino -->


<!-- Date: Fri Oct 26 09:54:32.699 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252704>252704</A>
ACTIVATE: RunSet: BOTHPLUGS_MM, second time, but just to figure out some TrigMon problems, jdl doing some test to find out why TrigMon crashes
<!-- Author: Claudia Marino -->


<!-- Date: Fri Oct 26 09:56:41.946 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252704>252704</A>
TERMINATE: done
<!-- Author: Claudia Marino -->



<!-- Date: Fri Oct 26 10:02:00.437 2007 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2007&action=view&page=-6631&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2007&action=view&page=-6631  width='300' align='top'></a></td><td><p>While investigating heartbeat alarms with ICICLE from  
the SOLENOID2 PC last night, I discovered there seems to 
be something wrong with the smart hub for iFIX.  Every 
node on this hub sees the exact same periods with no  
traffic in or out as shown here; this cannot be real, since 
no problems were reported in these periods (especially the 
long period from 1330-1530 or so yesterday).  I'll check 
with JJ.</p></td></tr></table>
<!-- Author: Steve Hahn -->

<!-- Date: Fri Oct 26 10:32:43.012 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252706>252706</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: Claudia Marino -->


<!-- Date: Fri Oct 26 10:33:28.002 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252706>252706</A>
CALIBRATION: <U><B>cemled</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cemled target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Fri Oct 26 11;53;26 comment by...Claudia -->
GOOD
<!-- Date: Fri Oct 26 10:34:44.385 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252706>252706</A>
TERMINATE: done
<!-- Author: Claudia Marino -->



<!-- Date: Fri Oct 26 10:39:29.554 2007 -->
<pre>L3 cleanup: 
 
- I wanted to do PMT Spike run calibration 
- but I got the error message L3 cannot be booked 
- somehow I totally forgot about the silicon noise bias scan and cleaned up L3 farm 
- sili people had to restart their test: SORRY 
 
EVER CHECK before L3 cleanup, if some other partition uses it!!! 
 
</pre>
<!-- Author: Claudia -->


<!-- Date: Fri Oct 26 10:45:15.200 2007 -->
TrigMon keeps crashing, JDL did some tests, however, TrigMon is still not running
<!-- Author: Claudia -->
<!-- Comment: Fri Oct 26 12;49;59 comment by...Claudia -->
JDL undid his changes, TrigMon is now running while we are taking cosmics, we will keep an eye on it and see if it crashes again
<!-- Comment: Fri Oct 26 14;01;26 comment by...Claudia -->
<pre>TrigMon gives many error messages while running:
Fatal Error: SIXD_StorableBank   SIXD not found</pre>
<!-- Date: Fri Oct 26 10:55:16.308 2007 -->
I'm working on the tomcat web server. You may see occasional Procmon warnings and brief service interruptions. I am at x3986.
<!-- Author: Donatella T. -->
<!-- Comment: Fri Oct 26 13;20;47 comment by...Claudia -->
TOMCAT red in ProcMon
<!-- Comment: Fri Oct 26 14;27;02 comment by...Claudia -->
forgot to mention: was back after few minutes
<!-- Date: Fri Oct 26 11:05:54.815 2007 -->
<pre>We closed all Web Browser on machines with cdfdaq accounts, after Bill called back we opened them again. 
 
Messages: 
- Select User Profile (available profile b0control20):  
  b0control22, b0control14, b0control12, b0control21 
 
- Firefox is already running:  
  b0control13, b0control15</pre>
<!-- Author: Claudia -->
<!-- Comment: Fri Oct 26 11;07;43 comment by...Claudia -->
Bill called back and will have a look

<!-- Comment: Fri Oct 26 11;10;18 comment by...badgett -->
The runMozilal script is supposed to create a new configuration 
if it cannot find one based on the node name of the computer.
Not sure why this is failing...



<!-- Comment: Fri Oct 26 13;54;30 comment by...Claudia -->
I dont know how I should open a web browser on these machines, I could e. g. create new profile, I wrote email to Bill and asked
<!-- Comment: Fri Oct 26 15;30;31 comment by...badgett -->
That should be fine -- creating new configs as needed
<!-- Date: Fri Oct 26 11:21:58.717 2007 -->
Eike aked for COT crates (approx. 1h)
<!-- Author: Claudia -->

<!-- Date: Fri Oct 26 11:22:54.480 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252707>252707</A>
ACTIVATE: RunSet: COT_ACE 
<!-- Author: eikoyu -->



<!-- Date: Fri Oct 26 11:23:58.650 2007 -->
CRYO alarm, LLDEW is below lower limit, now at about 40.79% and lower limit is 45%, alarm is silenced so cryo techs know
<!-- Author: Claudia -->
<!-- Comment: Fri Oct 26 11;35;31 comment by...Claudia -->
<pre>Helium Flow Into Magnet now at 0 g/s, this alarm is also silenced from cryo techs, they called: 
solenoid trip, trip is due to wet engine problems, cold valve overhaul needed

Helium flow into magnet now rising</pre>


<!-- Comment: Fri Oct 26 11;57;26 comment by...Phil -->
Jim said half an hour or so to fix the wet engine.
<!-- Date: Fri Oct 26 11:37:55.428 2007 -->
<pre>COT in alarm on HV Monitor, Eiko informed me she is doing some work</pre>
<!-- Author: Claudia -->
<!-- Comment: Fri Oct 26 11;39;52 comment by...Shin-Shan Yu (Eiko) -->
I am leaving the +- 3.3 V low voltage power of SL3 LNW off for 30 mins. Will turn it back on around 12:10 pm. We want to see if the noise in b0cot00 slot 13 cable 1 will go away.




<!-- Comment: Fri Oct 26 12;18;41 comment by...Claudia -->
Eiko told me COT is back, now take alarms for real
<!-- Date: Fri Oct 26 11:57:49.590 2007 -->
I have adjusted the timing on the EASt and on the WEST plug lasers: 
 
WEST  (DDG1):  from 265 to 276 
East  (DDG0):  from 750 to 775 
 
This should bring the mean TDC averages back to ~ 565  (middle of the range 550 - 580)
<!-- Author: G. Pauletta -->


<!-- Date: Fri Oct 26 12:06:51.659 2007 -->
SVX people report they have finished the bias scans.
<!-- Author: J. Patrick -->


<!-- Date: Fri Oct 26 12:10:21.101 2007 -->
HVAC voice alarm, but HVAC looks fine in IFIX
<!-- Author: Claudia -->



<!-- Date: Fri Oct 26 12:17:13.511 2007 -->
RTServer became red in ProcMon, it got stuck due to copying some files, now it is back again, IMON was red, too and perhaps the reason why RTServer got stuck
<!-- Author: Claudia -->


<!-- Date: Fri Oct 26 12:17:49.529 2007 -->
We have finished with the Silicon Noise to Bias Scan. All resources have been released. Silicon is left to Stand-By.
<!-- Author: Oscar, Roberto and Ignacio -->



<!-- Date: Fri Oct 26 12:19:38.402 2007 -->
MCR calls, they will be injecting into the tevatron shortly.
<!-- Author: J. Patrick -->

<!-- Date: Fri Oct 26 12:27:40.358 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252711>252711</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->


<!-- Date: Fri Oct 26 12:30:14.436 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252712>252712</A>
 Activated at 2007.10.26 12:28:40
<!-- Author: RunControl -->


<!-- Date: Fri Oct 26 12:30:15.287 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252712>252712</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,985,706] 
<!-- Author: Claudia Marino -->



<!-- Date: Fri Oct 26 12:32:00.808 2007 -->
<pre>COT noise study of b0cot00 slot 13 cable 1: 
 
  This morning Bob Wagner power cycled the +-3.3 V power (SL3  
LNW) for front end electronics (off for 10 mins) and found this  
made the the number of bad channels in cable  1 wen down from 20 
 to 9. I did an ASDQ test around 11 am to see if there is any  
difference comparing to the result on October 24th (before power  
cycling). I found no difference in the occupancy, width RMS,  
width value and also drift time. There were even 4 additional  
bad channels from SL 8 cell 34. 
 
Then, I turned off +- 3.3 V for 30 mins and did an ASDQ and COT 
 calibration. ASDQ test result looked the same. The number of  
bad channels from cable 1 went up to 20 again. The dead channels 
in SL8 were gone now.  
 
Will make a low priority request to open the muon walls.  
</pre>
<!-- Author: Shin-Shan Yu (Eiko) -->




<!-- Date: Fri Oct 26 13:26:53.737 2007 -->
<table><tr><td><a href=/elog/notebooks/2007/Fri_Oct_26_2007_Day.132518.ps align=top><img src=/cgi/elog/elog.pl?nb=2007&action=view&page=-6633  align='top'></a></td><td>BAD plug laser calibration plot</td></tr></table>
<!-- Author: Claudia -->



<!-- Date: Fri Oct 26 13:50:24.203 2007 -->
On Level 3 Display we see s10p21 has four dead L3 filters
<!-- Author: Claudia -->
<!-- Comment: Fri Oct 26 14;10;35 comment by...Conor -->
It looks like this worker node lost connection with its output node, causing all the filter exes to crash. I see no evidence that this might be systematic, so I will attribute it to just being a glitch, but we'll keep an eye on this node in case it happens again. Thanks for pointing it out. This node should recover when you start a new run - if it does not, please let us know.
<!-- Comment: Fri Oct 26 14;10;49 comment by...Claudia -->
we ended cosmics and started l2 torture, s10p21 now fine
<!-- Date: Fri Oct 26 14:04:19.009 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252712>252712</A>
 Terminated at 2007.10.26 14:04:07
<!-- Author: RunControl -->


<!-- Date: Fri Oct 26 14:04:23.653 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252712>252712</A>
TERMINATE: end to start L2 torture
<!-- Author: Claudia Marino -->


<!-- Date: Fri Oct 26 14:07:30.443 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252713>252713</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [19,986,705] 
<!-- Author: Claudia Marino -->
<!-- Comment: Fri Oct 26 14;15;26 comment by...Claudia -->
L2 Decision Timeout! -> auto HRR -> fine again

<!-- Comment: Fri Oct 26 14;19;32 comment by...Claudia -->
again

<!-- Comment: Fri Oct 26 14;22;48 comment by...Claudia -->
Done TO because of CMX_00

<!-- Comment: Fri Oct 26 14;23;24 comment by...Claudia -->
forgot to mention: auto HRR -> fine again

<!-- Date: Fri Oct 26 14:23:58.101 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252713>252713</A>
TERMINATE: end for some L3 work
<!-- Author: Claudia Marino -->



<!-- Date: Fri Oct 26 14:24:05.868 2007 -->
Conor is here to do some L3 work
<!-- Author: Claudia -->



<!-- Date: Fri Oct 26 14:35:30.189 2007 -->
<pre>COT calibration script change: 
   
  Since the noise in b0cot00 slot 13 cable 1 will not be understood  
until we open IMU or plugs, for the time being, calibration will  
ignore failure from this cable. Failures from the old and  
well-known bad cells b0cot02 slot 5 asdq 6 and b0cot03 slot 14 asdq  
3 will be ignored too.</pre>
<!-- Author: Shin-Shan Yu (Eiko) -->



<!-- Date: Fri Oct 26 15:33:46.197 2007 -->
Level3: Si found some ways to speed up END for Level3. It now takes only ~15s, as opposed to ~30s before. 
EVB: we also speeded up HRR for EVB, from 5s to 2s.
<!-- Author: Si and Conor :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252718' target=_top>252718</a>) -->

<!-- Date: Fri Oct 26 15:37:23.155 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252718>252718</A>
TERMINATE: end, did not come back to active, L2 trigger rate problems
<!-- Author: Claudia Marino -->


<!-- Date: Fri Oct 26 15:42:03.342 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252719>252719</A>
TERMINATE: forgot something
<!-- Author: Claudia Marino -->


<!-- Date: Fri Oct 26 15:43:38.148 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252720>252720</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_5_01 [1,987,708] , trigger table test for Tom and Dave
<!-- Author: Claudia Marino -->


<!-- Date: Fri Oct 26 15:51:48.198 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252720>252720</A>
TERMINATE: switch to testing old L2 tagset
<!-- Author: Claudia Marino -->


<!-- Date: Fri Oct 26 15:54:18.309 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252721>252721</A>
TERMINATE: test done
<!-- Author: Claudia Marino -->


<!-- Date: Fri Oct 26 15:55:48.134 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252721>252721</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_5_01 [1,949,708] 
<!-- Author: Claudia Marino -->


<!-- Date: Fri Oct 26 15:55:49.100 2007 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>PhysicsTable</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L3Accepts</TH>
<TH>DelivLumi nb<SUP>-1</SUP></TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>15:55:00</TD>
<TD ALIGN=CENTER>::</TD>
<TD ALIGN=RIGHT></TD>               <TD ALIGN=RIGHT></TD>
<TD ALIGN=RIGHT></TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->


<!-- Date: Fri Oct 26 15:57:00.948 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=252722>252722</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [19,986,705] 
<!-- Author: Claudia Marino -->



<!-- Date: Fri Oct 26 15:58:32.219 2007 -->
<font color=red>Shift Summary: </font><pre>Did all calibrations. 
During an access for D0 SVX noise scans were done. 
Eiko and Bob worked on COT noise but no improvement was made. 
 The calibration script will not complain about these channels 
 until they can be looked at in an access. 
The solenoid tripped due to wet engine troubles. It is still down. 
Conor and Si sped up the L3 end run transition. 
Cosmics and Level 2 torture were run as time permitted. 
Testing of physics trigger tables is underway.</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   0
Delivered Luminosity   0  
Acquired Luminosity    0  
Efficiency             0
   
</td></tr></table></pre>
<!-- Author: Jim Patrick -->