<!-- Title: Eve shift. Sun Jan 12, 2003 -->
<!-- SciCo: Ted Liu -->
<!-- DAQAce: Jean-Francois/Albert -->
<!-- MonAce: Oliver/Luis -->
<!-- CO: Kihyeon Cho -->
<!-- OpManager: Eric James -->
<!-- Notes: -->
<pre>(1) Bill Foster left, disappointed. MCR shift crew lost their 
last chance to win the $20/person "Foster award" for breaking 
the record for single-shift delivered luminosity on his shift. 
We concluded that $20 is just not good enough to bribe our good 
machine physicists. 
(2) instead, MCR shift crew issued the LONG (3 weeks) beam  
abort gap signal right before Bill came to shift this morning. 
(3) the leading edge of that long beam abort gap signal  
has turned the control room/B0 into a zoo... 
(4) expect some sili-people here at 7pm. try not to feed them</pre>

<!-- Date: Sun Jan 12 16:26:31 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156523">156523</A>
TERMINATE: Plug Sourcing, tubes 55&56, HA, Layers 0&6, sector 2.1
<!-- Author: sedov -->


<!-- Date: Sun Jan 12 16:36:28 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156524">156524</A>
TERMINATE: Plug Sourcing, tubes 61&62, EM, Layers 4&6, sector 3.0
<!-- Author: sedov -->


<!-- Date: Sun Jan 12 16:47:26 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156525">156525</A>
TERMINATE: Plug Sourcing, tubes 64&65, HA, Layers 0&6, sector 3.0
<!-- Author: sedov -->


<!-- Date: Sun Jan 12 17:43:46 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156511">156511</A>
 Terminated at 2003.01.12 17:38:20
<!-- Author: RunControl -->


<!-- Date: Sun Jan 12 17:43:46 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156511">156511</A>
TERMINATE: end test run for plug laser calibration
<!-- Author: Luis/Alberto -->


<!-- Date: Sun Jan 12 17:43:46 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156526">156526</A>
ACTIVATE: Plug Laser Calibration, normal mode.
<!-- Author: Luis/Alberto -->


<!-- Date: Sun Jan 12 17:50:42 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156526">156526</A>
TERMINATE: End laser plug calibration, normal mode.
<!-- Author: Luis/Alberto -->
<!-- Comment: Sun Jan 12 17;59;01 comment by...Willis -->
Plug laser runs 156526 and 156527 are with LER=1
<!-- Date: Sun Jan 12 17:54:43 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156527">156527</A>
ACTIVATE: Plug laser calibration, myron mode.
<!-- Author: Luis/Alberto -->


<!-- Date: Sun Jan 12 17:55:06 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156527">156527</A>
TERMINATE: End of laser plug calibration, Myron mode.
<!-- Author: Luis/Alberto -->



<!-- Date: Sun Jan 12 18:17:22 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156529">156529</A>
TERMINATE: Plug Sourcing, tubes 70&71, EM, Layers 4&6, sector 3.1
<!-- Author: sedov -->
<!-- Comment: Sun Jan 12 18;25;54 comment by...Alexei Sedev, Willis -->
Possible problem with run: b0pcal00 was apparently in a "hung"
state at the end of this run. These pokes have been redone:
run 156532. 
<!-- Date: Sun Jan 12 18:19:09 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156530">156530</A>
ACTIVATE: Central Hadron Laser B off pipeline length 26

<!-- Author: Luis/Alberto -->


<!-- Date: Sun Jan 12 18:22:51 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156530">156530</A>
TERMINATE: Myron Mode
<!-- Author: Luis/Alberto -->



<!-- Date: Sun Jan 12 18:28:40 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156531">156531</A>
ACTIVATE: Central Laser run, Myron mode, B off HrdcEVB, pipeline length 28
<!-- Author: Luis/Alberto -->


<!-- Date: Sun Jan 12 18:31:17 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156531">156531</A>
TERMINATE: Triggering on calorimeter
<!-- Author: Luis/Alberto -->


<!-- Date: Sun Jan 12 18:35:30 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156532">156532</A>
TERMINATE: Plug Sourcing, tubes 70&71, EM, layers 4&6, sector 3.1
<!-- Author: sedov -->


<!-- Date: Sun Jan 12 18:37:39 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156533">156533</A>
ACTIVATE: Central laser run, Myron mode, Triggering on Calorimeter, B off, HRDEVB, pipeline length 27
<!-- Author: Luis/Alberto -->


<!-- Date: Sun Jan 12 18:41:54 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156533">156533</A>
TERMINATE: done

<!-- Author: Luis/Alberto -->



<!-- Date: Sun Jan 12 18:42:43 2003 -->
Took 3 central laser runs in Myron mode triggering on calorimeter  
<br> 
runs 156530,156531 and 156533
<!-- Author: fotis -->

<!-- Date: Sun Jan 12 18:44:09 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156534">156534</A>
TERMINATE: Plug Sourcing, tubes 73&74, HA, layers 0&6, sector 3.1
<!-- Author: sedov -->


<!-- Date: Sun Jan 12 18:54:03 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156535">156535</A>
TERMINATE: Plug Sourcing, tubes 79&80, EM, layers 4&6, sector 4.0
<!-- Author: sedov -->


<!-- Date: Sun Jan 12 19:04:29 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156537">156537</A>
ACTIVATE: Silicon test trying to HRR with Calib Continuous trigger
<!-- Author: nielsenj -->


<!-- Date: Sun Jan 12 19:15:09 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156536">156536</A>
TERMINATE: Bad run
<!-- Author: sedov -->


<!-- Date: Sun Jan 12 19:15:09 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156537">156537</A>
TERMINATE: Trigger rate is too low; need to decrease Interval0
<!-- Author: nielsenj -->


<!-- Date: Sun Jan 12 19:16:48 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156538">156538</A>
TERMINATE: Bad run

<!-- Author: sedov -->


<!-- Date: Sun Jan 12 19:20:47 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156539">156539</A>
ACTIVATE: Searching for silicon HRR with CalibContinuous (Interval0=3000)
<!-- Author: nielsenj -->


<!-- Date: Sun Jan 12 19:28:58 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156539">156539</A>
TERMINATE: L1 Rate is only 50 Hz with Interval0=3000
<!-- Author: nielsenj -->


<!-- Date: Sun Jan 12 19:28:58 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156540">156540</A>
TERMINATE: Plug Sourcing, tubes 82&83, HA, layers 0&6, sector 4.0
<!-- Author: sedov -->


<!-- Date: Sun Jan 12 19:34:40 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156542">156542</A>
ACTIVATE: Silicon HRR test with CalibContinuous trigger (Interval0=100)
<!-- Author: nielsenj -->


<!-- Date: Sun Jan 12 19:39:53 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156541">156541</A>
TERMINATE: Plug Sourcing, tubes 95&96, EM, layers 4&6, sector 4.1
<!-- Author: sedov -->


<!-- Date: Sun Jan 12 19:47:34 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156543">156543</A>
TERMINATE: Plug Sourcing, tubes 95&96, HA, layers 0&6, sector 4.1
<!-- Author: sedov -->



<!-- Date: Sun Jan 12 19:48:44 2003 -->
WHA is yellow on HV summary and global alarms. 
Towers (Eta/Phi) -9/17R, -8/5RL, -6/17R, -6/5L and -6/3R 
are flashing red. Calorimeter expert (Fotis) was taking 
central calorimeter laser runs and he told us, that these channels may have floated and he will look into it.
<!-- Author: Oliver/Luis -->

<!-- Date: Sun Jan 12 19:58:21 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156544">156544</A>
TERMINATE: Plug Sourcing, tubes 101&102, EM, layers 4&6, sector 5.0
<!-- Author: sedov -->


<!-- Date: Sun Jan 12 20:09:03 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156542">156542</A>
TERMINATE: We have seen an HRR, so our test has succeeded.
<!-- Author: Jason Nielsen 2164 -->


<!-- Date: Sun Jan 12 20:09:03 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156545">156545</A>
TERMINATE: Plug Sourcing, tubes 104&105, HA, layers 0&6, sector 5.0
<!-- Author: sedov -->


<!-- Date: Sun Jan 12 20:19:44 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156546">156546</A>
TERMINATE: Plug Sourcing, tubes 110&111, EM, layers 4&6, sector 5.1
<!-- Author: sedov -->



<!-- Date: Sun Jan 12 20:24:16 2003 -->
<pre>Preparing for silicon warm-up to 16 deg C for access: 
 
1. Turn off detector power 
2. Silence all relevant cooling alarms 
3. Start warming up... 
</pre>
<!-- Author: Andy H. -->

<!-- Date: Sun Jan 12 20:30:31 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156547">156547</A>
TERMINATE: Plug Sourcing, tubes 113&114, HA, layers 0&6, sector 5.1
<!-- Author: sedov -->



<!-- Date: Sun Jan 12 20:35:38 2003 -->
We took a series of test runs with the silicon crates using a CalibContinuous trigger.  We were trying to reproduce the stuck pipeline cell ID error which generates an automatic HRR in SVXMon.  We were successful in generating at least one such error in run 156542.  This gives us some hope that the problem might be reproducible on the bench outside of B0. 
(See more details in <a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=138&frame=2&anchor=20:19:51'>this silicon e-log entry</a>.)
<!-- Author: Jason Nielsen, Rainer Wallny -->

<!-- Date: Sun Jan 12 20:40:23 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156548">156548</A>
TERMINATE: Plug Sourcing, tubes 21&22, EM, layers 4&6, sector 1.0
<!-- Author: sedov -->



<!-- Date: Sun Jan 12 20:42:19 2003 -->
Shut down TOF HV system in advance of tomorrow's 
power outage.
<!-- Author: gbauer -->

<!-- Date: Sun Jan 12 20:49:04 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156550">156550</A>
 Activated at 2003.01.12 20:47:39
<!-- Author: RunControl -->


<!-- Date: Sun Jan 12 20:49:04 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156549">156549</A>
TERMINATE: Plug Sourcing, tubes 24&25, HA, layers 0&6, sector 1.0
<!-- Author: sedov -->


<!-- Date: Sun Jan 12 20:49:49 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156550">156550</A>
ACTIVATE: Taking cosmics. Remove plug crates meanwhile the plug sourcing goes on.
<!-- Author: Jean-Francois -->


<!-- Date: Sun Jan 12 20:50:27 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156550">156550</A>
 Terminated at 2003.01.12 20:49:24
<!-- Author: RunControl -->


<!-- Date: Sun Jan 12 20:50:40 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156550">156550</A>
TERMINATE: End the run because of HEVNB error.
<!-- Author: Jean-Francois -->
<!-- Comment: Sun Jan 12 21;01;15 comment by...JFA -->
I tried to run cosmics during plug sourcing, but got a EVB error about scanner manager SCPU unresponsive. There might be a conflict with plug sourcing. Sedov saw the same error. I now cleaned up the EVB and won't try to run cosmics again.

<!-- Date: Sun Jan 12 21:10:41 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156553">156553</A>
TERMINATE: Plug Sourcing, tubes 34&35, EM, layers 4&6, sector 1.1
<!-- Author: sedov -->


<!-- Date: Sun Jan 12 21:20:04 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156554">156554</A>
TERMINATE: Plug Sourcing, tubes 37&38, HA, layers 0&6, sector 1.1
<!-- Author: sedov -->


<!-- Date: Sun Jan 12 21:33:33 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156555">156555</A>
TERMINATE: Plug Sourcing, tubes 28&29, showermax
<!-- Author: sedov -->


<!-- Date: Sun Jan 12 21:49:19 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156558">156558</A>
TERMINATE: Plug Sourcing, tubes 30&31, showermax
<!-- Author: sedov -->


<!-- Date: Sun Jan 12 22:00:27 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156559">156559</A>
TERMINATE: Plug Sourcing, tubes 86&87, showermax
<!-- Author: sedov -->


<!-- Date: Sun Jan 12 22:12:08 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156560">156560</A>
TERMINATE: Plug Sourcing, tubes 88&89, showermax
<!-- Author: sedov -->


<!-- Date: Sun Jan 12 22:28:32 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156561">156561</A>
 Activated at 2003.01.12 22:25:41
<!-- Author: RunControl -->


<!-- Date: Sun Jan 12 22:28:32 2003 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=156561">156561</A>
ACTIVATE: Plug sourcing is finished, starting a cosmics run.
<!-- Author: Jean-Francois -->



<!-- Date: Sun Jan 12 22:46:31 2003 -->
Plug HV turned off and the plug HV control/monitor (HVSummary/iFix)  
program stopped in preparation for the 13Jan03/0700 power outage. 
All 12 plug CAEN crates have been remotely powered off via the 
control panel by the plug HV pc.
<!-- Author: Willis -->


<!-- Date: Sun Jan 12 23:11:03 2003 -->
<table border=0><td><tr><td bgcolor=#e0e030><p>Note for owl shift:  When silicon is off, the silicon PS GUI gets unhappy and runs verrrrry slow (you can see the "next update" countdown decrementing every few minutes instead of every second).  This causes iFix alarms ("GUI - iFix communication: NOT OK").  The cure is to exit the GUI and restart it following the 
instructions taped above the monitor (it will be just as slow 
responding to your commands, so patience is required).</p></td></tr></td></table>
<!-- Author: Andy H. -->
<!-- Comment: Sun Jan 12 23;13;00 comment by...Andy H. -->
P.S. This cure is not permanent, but my experience has been that it's
usually enough to get iFix to shut up for the remainder of your shift.


<!-- Date: Sun Jan 12 23:21:17 2003 -->
<pre>Plug Sourcing summary. 
 
PEM, PHA and showermax were sourced, 2 tubes per run, all sectors (and subsectors). 
 
PEM (layers 4 & 6): 
 
Run 156512 - subsector 0.0, 156514 - 0.1, 1565148 -  1.0, 156553 - 1.1, 156520 - 2.0, 156522 - 2.1, 156524 - 3.0, 156532 - 3.1, 156535 - 4.0, 156541 - 4.1, 156544 - 5.0, 156546 - 5.1 . 
 
 
PHA (layers 0 & 6): 
 
Run 156513 - subsector 0.0, 156515 - 0.1, 1565149 -  1.0, 156554 - 1.1, 156521 - 2.0, 156523 - 2.1, 156525 - 3.0, 156534 - 3.1, 156540 - 4.0, 156543 - 4.1, 156545 - 5.0, 156547 - 5.1 . 
 
Showermax. 
 
Run 156555 - tubes 28 & 29, 156558 - 30&31, 156559 - 86&87, 156560 - 88&89 . 
</pre>
<!-- Author: Alexei Sedov -->

<!-- Date: Sun Jan 12 23:56:10 2003 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>23:55:02</TD>                                                       
<TD ALIGN=CENTER>::</TD>                                                
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Mon Jan 13 00:00:33 2003 -->
<font color=red>Shift Summary: </font><pre>(1) started the shift at the beginning of a 
    LONG beam abort gap; 
(2) plug sourcing work continued through most of the shift; 
(3) a few sili people were here doing tests; 
(4) TOF HV turned off; 
(5) plug HV turned off; 
</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   0
Delivered Luminosity   0  
Acquired Luminosity    0  
Efficiency             -
   
</td></tr></table></pre>
<!-- Author: Ted Liu -->