<!-- Title: Eve shift. Sat May 22, 2004 -->
<!-- SciCo: Carlsmith -->
<!-- DAQAce: Corrinne Mills -->
<!-- MonAce: Dae Jong Kung -->
<!-- CO: Kyung Kwang Joo -->
<!-- OpManager: jj schmidt -->
<!-- Notes: -->
<pre>Stack = 94, accrt =8 b0lum = 0 
waiting for shot setup in roughly 1/2 hour</pre>

<!-- Date: Sat May 22 16:08:36.540 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183043">183043</a>
 Terminated at 2004.05.22 16:08:11
<!-- Author: RunControl -->


<!-- Date: Sat May 22 16:09:52.966 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183043">183043</a>
TERMINATE: End of PMT spike run.  Avg. L1 rate 74.5 Hz for this run.
<!-- Author: corrinne x2080 -->


<!-- Date: Sat May 22 16:16:30.850 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183044">183044</a>
ACTIVATE: L2 torture, to keep the system busy.  No quiet time to run cosmics w/ silicon.
<!-- Author: corrinne x2080 -->



<!-- Date: Sat May 22 16:24:40.748 2004 -->
All of subfarm 13 has turned gold in the L3 subfarm monitor.  This just means that monitoring is lost for that subfarm, according the the L3 troubleshooting webpage.
<!-- Author: corrinne :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183044' target=_top>183044</a>) -->


<!-- Date: Sat May 22 16:29:17.172 2004 -->
I tried to make BMU-E trip.  I was unable to make BMU-E-72 
trip.  It does not have any strange behavior that I can tell. 
BMU-E-12 tripped once on the anode.  It is drawing 0.3 uA, 
compared to the usual 0.0 or 0.1 uA (still not much). 
I did not see any growing current draw before the trip. 
It is not very reproducible at all.  Shiftcrew: please 
try the usual 3 times to recover from trips.  If unable 
to recover, please page BMU.
<!-- Author: Camille Ginsburg -->

<!-- Date: Sat May 22 16:35:00.066 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183044">183044</a>
TERMINATE: Ending to start cosmics for shot setup
<!-- Author: corrinne x2080 -->


<!-- Date: Sat May 22 16:40:54.430 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183045">183045</a>
 Activated at 2004.05.22 16:39:50
<!-- Author: RunControl -->


<!-- Date: Sat May 22 16:40:55.181 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183045">183045</a>
ACTIVATE: Start cosmics run for shot setup.  
<!-- Author: corrinne x2080 -->



<!-- Date: Sat May 22 17:09:57.498 2004 -->
<pre>PSM Alarm 
1st floor - 1RR18I_1 :  returned valule : 6.06V 
 
It returned to normal quickely.  
                 </pre>
<!-- Author: Daejung :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183045' target=_top>183045</a>) -->

<!-- Date: Sat May 22 17:33:20.433 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183045">183045</a>
 Terminated at 2004.05.22 17:32:59
<!-- Author: RunControl -->


<!-- Date: Sat May 22 17:33:40.048 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183045">183045</a>
TERMINATE: They're injecting final protons.  Switching to AAA_Shotsetup.
<!-- Author: corrinne x2080 -->



<!-- Date: Sat May 22 17:39:16.673 2004 -->
 
<TABLE bgcolor=#cdc9a5> 
<TR><TH>Date</th><th>Time</th><th>BLM</th><th colspan=2>Dose</th></tr> 
<tr><td>2004.05.22</td><td>17:35:40</td><td>W Inner BLM</td><td>9.43</td><td>RADS</td></tr> 
<tr><td>2004.05.22</td><td>17:35:40</td><td>W Outer BLM</td><td>2.19</td><td>RADS</td></tr> 
<tr><td>2004.05.22</td><td>17:35:40</td><td>E Inner BLM</td><td>5.65</td><td>RADS</td></tr> 
<tr><td>2004.05.22</td><td>17:35:40</td><td>E Outer BLM</td><td>10.19</td><td>RADS</td></tr> 
</table> 
<strong>Integrated dosage</strong> 
  
<!-- Author: Daejung -->

<!-- Date: Sat May 22 17:40:20.615 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183046">183046</a>
ACTIVATE: AAA_SHOTSETUP... by now we are loading antiprotons.
<!-- Author: corrinne x2080 -->



<!-- Date: Sat May 22 17:46:37.134 2004 -->
Pbars being loaded. Paged Si primary and 218-9520 and 722-7483
<!-- Author: carlsmith :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183046' target=_top>183046</a>) -->


<!-- Date: Sat May 22 18:01:12.963 2004 -->
Final pbars loaded
<!-- Author: carlsmith -->

<!-- Date: Sat May 22 18:05:27.379 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183046">183046</a>
TERMINATE: At flat top.  Will load AAA_NOSILICON
<!-- Author: corrinne x2080 -->



<!-- Date: Sat May 22 18:17:13.275 2004 -->
Scraping complete. Turning on CLC.
<!-- Author: Carlsmith -->


<!-- Date: Sat May 22 18:18:44.419 2004 -->
Store 3528, initial lum 57e30
<!-- Author: carlsmith -->


<!-- Date: Sat May 22 18:27:47.605 2004 -->
Tevpr=9063e9, Tevpb=753, lostp<10kHz, mcr noticed lostp dropped the instant CLC came on...
<!-- Author: carlsmith -->

<!-- Date: Sat May 22 18:29:14.850 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183047">183047</a>
 Activated at 2004.05.22 18:28:59
<!-- Author: RunControl -->


<!-- Date: Sat May 22 18:30:26.022 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183047">183047</a>
ACTIVATE: Start physics run with AAA_NOSILICON[1,475,455].  TOF HV would not come all the way up so we have masked it off.  This run will be bad for TOF
<!-- Author: corrinne x2080 -->
<!-- Comment: Sat May 22 18;32;16 comment by...Daejung -->
<pre>When starting power up, TOF was stucked, 
I tried power on several time and turning off and on.
But I didn't work, 
I checked TOF PC, it worked.</pre>
<!-- Date: Sat May 22 18:31:16.445 2004 -->
Trouble getting TOF on. Page 722-6176. JJ's decision is to mask TOF off and run w/o si since , well, general concerns about recent experience and see what else might be wrong with CDF.
<!-- Author: carlsmith -->



<!-- Date: Sat May 22 19:05:45.766 2004 -->
First I got a "L2 trigger rate has dropped to zero Hz" error, which was proceeded by a <br><br> 
<pre> 
(MLE) b0l3pcom1.fnal.gov:main:6:48:12 PM->Host b0eb16.fnal.gov, task tRec_0 
SCPU-P0-E-VrbReadFailed: Error reading VRB in slot 10 for event 165762. 
VRB_BUS_ERROR: Alignment or VME bus error. 
Transferred 0 bytes before the error. 
 
I paused, which didn't help.  
So I halted and then got 
 
 
52'30" b0l3pcom1.fnal.gov:main:6:52:30 PM->Host b0eb16.fnal.gov, task tRec_0 
SCPU-P0-E-CantResetVrb: Reset of VRB in slot 10 failed. 
VRB_BUS_ERROR: Alignment or VME bus error. --> 
 Additional Information: 
 
 Attention!!!. Event Builder SCPU_CANT_RESET_VRB Error !!! 
 THE BAD VRB CRATE is: b0eb16 
 THE BAD VRB MODULE is in SLOT: 10 
 
</pre> 
So I followed the recommended procedure (reset the VRB in slot 10), recovered and ran, and things are fine now. 
<br><br> 
Got some minor errors after that, a "High Deadtime" warning and a L2 decision timeout.  All is fine now
<!-- Author: corrinne :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183047' target=_top>183047</a>) -->


<!-- Date: Sat May 22 19:07:01.083 2004 -->
Also we got a PSM alarm from 1RR18E_1 , the +5VDC (channel 0) was at +6V.  Got a corresponding trigger inhibit.  This alarm recovered itself quickly
<!-- Author: corrinne :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183047' target=_top>183047</a>) -->
<!-- Comment: Sat May 22 19;16;22 comment by...Daejung -->
same alarm, 1RR18E_1 : 6.01V
<!-- Date: Sat May 22 19:10:25.764 2004 -->
Reached Mathew Jones for TOF, passed on to Stefano Giagu,left message on his cell and home phones.
<!-- Author: carlsmith -->


<!-- Date: Sat May 22 19:10:44.660 2004 -->
<pre>After turned off TOF HV, 
I tried to turn on TOF HV, it had same problem. 
I found crate1 had yellow color on TOF PC, the other crate2 and crate3 has green color, tried to restart smacs, but same problem. 
</pre>
<!-- Author: Daejung -->

<!-- Date: Sat May 22 19:12:32.723 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183047">183047</a>
 Terminated at 2004.05.22 19:12:16
<!-- Author: RunControl -->


<!-- Date: Sat May 22 19:12:45.148 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183047">183047</a>
TERMINATE: bad for TOF.  We are going to bring in the silicon
<!-- Author: corrinne x2080 -->



<!-- Date: Sat May 22 19:18:24.363 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183053">183053</a>
 Activated at 2004.05.22 19:17:56
<!-- Author: RunControl -->


<!-- Date: Sat May 22 19:19:26.207 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183053">183053</a>
ACTIVATE: Start AAA_SILICON with the ok of Reiner et al.  TOF still troubled, so TOF inhibits still maksed.  
<!-- Author: corrinne x2080 -->



<!-- Date: Sat May 22 19:20:43.054 2004 -->
<center><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7644&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7644' width='300'></a><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7646&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7646' width='300'></a><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7645&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7645' width='300'></a><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7647&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7647' width='300'></a></center>
<!-- Author: Daejung -->
<!-- Comment: Sat May 22 19;21;29 comment by...Daejung -->
Hourly Plots 17:30 - 19:00


<!-- Date: Sat May 22 19:22:39.027 2004 -->
B0Lum 47, D0Lum=38. Page CLC 218-4352
<!-- Author: carlsmith -->


<!-- Date: Sat May 22 19:23:27.314 2004 -->
Difference between B0 and D0 luminosity seems larger than normal. 
I ask Duncan to page CLC to take a look.
<!-- Author: jj -->
<!-- Comment: Sat May 22 19;39;20 comment by...jj -->
Sasha calls in but is on the road and two hours away.
<!-- Date: Sat May 22 19:25:41.183 2004 -->
Got a 
<pre> 
21'37" b0dap84.fnal.gov:ConsumerErrorRe:7:21:37 PM->Runtime Error 1, Event 5, RunNum 183053: SvxMon Halt Recover Run: Pipeline out of synch in 122 silicon readout chips . --> 
 Additional Information: 
 
 Attention !!!. CER_SVXMON_HALT_RECOVER_RUN_ERROR !!! 
 
 Pipeline out of synch in 122 silicon readout chips . 
 HRR is STRONGLY RECOMMENDED. 
 
</pre> 
Did an HRR, and b0pcal07 wouldn't come along for the ride.  SO I shepherded it, now we are fine.<br><br> 
Did an additional HRR at Reiner's request.  
<!-- Author: corrinne :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183053' target=_top>183053</a>) -->


<!-- Date: Sat May 22 19:27:22.832 2004 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2004&action=view&page=-7649&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2004&action=view&page=-7649  width='100' align='top'></a></td><td>... would not send any update initially, but then changed its mind.</td></tr></table>
<!-- Author: Corinne, Rainer, Bill da Cat -->
<!-- Comment: Sat May 22 19;36;01 comment by...dr -->
DuPage County?  Or Bloom County.....
<!-- Date: Sat May 22 19:31:33.429 2004 -->
<pre>BMU Alarm 
NE 12, red 
I turn on again. 
Now Ok.. 
</pre>
<!-- Author: Daejung :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183053' target=_top>183053</a>) -->
<!-- Comment: Sat May 22 20;22;42 comment by...Daejung -->
same alarm. fixed by turing on HV
<!-- Date: Sat May 22 19:33:38.916 2004 -->
<pre>PSM Alarm 
1RR18D_1  : 6.01V. 
It turned back to normal, 5.13 V</pre>
<!-- Author: Daejung :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183053' target=_top>183053</a>) -->


<!-- Date: Sat May 22 19:35:28.097 2004 -->
Mathew Jones from TOF will be in in 20 minutes
<!-- Author: carlsmith -->
<!-- Comment: Sat May 22 20;37;10 comment by...jj -->
Matt has been here a while working on TOF HV system although
he says Stefano Giagu would be a better choice. We are happy
Matt is here!
<!-- Date: Sat May 22 19:37:09.417 2004 -->
Got first simultaneous done timeouts from b0fcal00, b0mutr00, and b0svx02.  Then I got  
<pre> 
Attention !!!. FERML_SRC_FATALITY ERROR !!! 
 
 SRC Fatal Error from b0svx02: Sl 5 Gap Error 
</pre> 
Went to HRR, but b0fcal00 wouldn't come along for the ride, so I shepherded it.  Got some RXPT errors, too, just for good measure.   
<br><br> 
Then I got some (3) done timeouts from b0cmx01, coincident with the PSM alarm from its rack. Shepherded to recover. 
<!-- Author: corrinne :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183053' target=_top>183053</a>) -->



<!-- Date: Sat May 22 19:41:28.434 2004 -->
<center><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7651&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7651' width='300'></a><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7650&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7650' width='300'></a></center>
<!-- Author: jj -->
<!-- Comment: Sat May 22 19;42;35 comment by...JJ -->
CDF and D0 luminosity for beginning of this store and all
of store 3513. (Last "normal" store this week....)<br>
Vertical scales are the same. 


<!-- Comment: Sat May 22 20;00;05 comment by...jj -->
Comparing a run from store 3513 and 3528 shows L2 trigger bit
cross sections uniformly lower by 10-20%. This implies our
luminosity may be reading too high.
<!-- Date: Sat May 22 19:44:57.650 2004 -->
Busy timeout, EVB VrbBadByteCount error in b0eb19, slot 16.  HRR recovered.  I realized that Auto-HRR wasn't enabled, and enabled it.
<!-- Author: corrinne :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183053' target=_top>183053</a>) -->


<!-- Date: Sat May 22 19:49:49.145 2004 -->
<pre>CMP HV Alarm !! 
HV issued, Now Ok</pre>
<!-- Author: Daejung :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183053' target=_top>183053</a>) -->


<!-- Date: Sat May 22 19:51:44.178 2004 -->
PHA - two dead channels. Ref plot has one dead channel at different location. Page PHA 218-3896
<!-- Author: carlsmith -->



<!-- Date: Sat May 22 20:00:17.525 2004 -->
Left message with ROberto Rossin at 4166941 re CLC
<!-- Author: carlsmith -->


<!-- Date: Sat May 22 20:05:35.510 2004 -->
Reached ROberto Rossin on his cell 212-3261
<!-- Author: carlsmith -->
<!-- Comment: Sat May 22 20;12;26 comment by...jj -->
Roberto is coming in to look at HV for CLC.
<!-- Comment: Sat May 22 20;36;00 comment by...jj -->
Roberto is here to check on CLC.
<!-- Date: Sat May 22 20:09:58.358 2004 -->
Busy timeout from b0svx02, and it is ignoring the halt transition. From the errorlog: 
<pre> 
MLE) b0dap73.fnal.gov:Thread-26:8:06:22 PM->Requested Halt-Recover-Run issued [errmon] 
(MLE) b0dap73.fnal.gov:Thread-26:8:06:22 PM->Busy Timeout: VRB_SVX_02 
(MLE) b0svx06:Messenger:8:06:24 PM->Silicon Timeout:BUSY- Slots:  10:fa20 20:f840 08:fa00 18:f820 16:f800 12:fa40 
 
CT: 2004.05.22 20:06:30 
6'30" 1 crate/s: b0svx02(112),  busy.[RXPT] 
</pre> 
Fortunately there are several silicon experts sitting right here, so they are looking into it.  
<!-- Author: corrinne :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183053' target=_top>183053</a>) -->
<!-- Comment: Sat May 22 20;19;10 comment by...corrinne -->
Tried to abort the run, and got
<pre>
 Attention!!!. SM_SCPU_UNRESPONSIVE Error !!!.
 One or more SCPUs failed to respond to Delete OR Define Partition.
 We STRONGLY RECOMMEND the following steps:
 - End the run, Reset
 - Clean up EVB
 - Start run again.
[MLE]
</pre>
as well as
<pre>
Attention!!!. SM_SCPU_UNRESPONSIVE Error !!!.
 One or more SCPUs failed to respond to Delete OR Define Partition.
 We STRONGLY RECOMMEND the following steps:
 - End the run, Reset
 - Clean up EVB
 - Start run again.
</pre>
Marcel and Rong reset b0svx02 downstairs.  Then I cleaned up the evb, and am restarting the run
<!-- Date: Sat May 22 20:12:26.098 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183053">183053</A>
 Terminated at 2004.05.22 20:12:16
<!-- Author: RunControl -->



<!-- Date: Sat May 22 20:13:27.654 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183053">183053</A>
TERMINATE: b0svx02 got stuck in the middle of the run - would not respond have to abort
<!-- Author: corrinne x2080 -->



<!-- Date: Sat May 22 20:14:14.533 2004 -->
<pre>PSM Alarm !! 
1RR18I_1 : 6.06V 
Now OK : 5.11 V</pre>
<!-- Author: Daejung -->


<!-- Date: Sat May 22 20:15:23.538 2004 -->
After some from home checking, ROberto Rossin will be in 15 min to see if perhaps the clc HV is not set correctly producing a bolum 15% too high. JJ claims the l2 x's are consistently low while bolum is high so... 
 
Rob Roser called to check in.
<!-- Author: carlsmith -->


<!-- Date: Sat May 22 20:17:18.794 2004 -->
<center><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7652&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7652' width='300'></a><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7653&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7653' width='300'></a><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7654&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7654' width='300'></a><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7655&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7655' width='300'></a></center>
<!-- Author: Daejung -->
<!-- Comment: Sat May 22 20;17;53 comment by...Daejung -->
Hourly Plots 19:00 - 20:00


<!-- Date: Sat May 22 20:18:21.637 2004 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2004&action=view&page=-7656&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2004&action=view&page=-7656  width='300' align='top'></a></td><td>Curious feature of proton losses/halo at beginning of store.</td></tr></table>
<!-- Author: jj -->
<!-- Comment: Sat May 22 20;21;46 comment by...jj -->
This strange behavior of proton losses from BSC and the
proton halo from halo counters is unexplained by either
us or MCR. I checked collimator positions that are 
predefined by Rick Tesarek in ACNET plots and did not
see anything that was still retracting. <br>
The MCR would like to correlate this with our CLC turning
on but I can not imagine any possible connection. The 
detailed timing if you plot b0ilum and the above do not
really line up as well.<br>


<!-- Date: Sat May 22 20:23:25.201 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183054">183054</A>
 Activated at 2004.05.22 20:23:05
<!-- Author: RunControl -->


<!-- Date: Sat May 22 20:23:43.879 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183054">183054</A>
ACTIVATE: Restart physics run with silicon good (and TOF bad) after b0svx02 problem.
<!-- Author: corrinne x2080 -->


<!-- Date: Sat May 22 20:31:59.730 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183053">183053</A>
central: <tt>In the beginning of run, HV problem in TOF. YMON #84 TOF (number of hits<br>
distribution) is different from ref. plot</tt>
<!-- Author: Consumer Operator( Kyung Kwang Joo) -->


<!-- Date: Sat May 22 20:32:00.112 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183053">183053</A>
cal: <tt>- PHATowerCanvas (3 dead ch in ref. plot, but found 2 and location<br>
 &nbsp;of dead channels is also different from ref plot, ie, phi wedges is <br>
 &nbsp;15 in the above plot)<br>
- WHATowerCanvas (2 dead ch in the ref plot, but too many in the <br>
 &nbsp;actual plot) &nbsp;</tt>
<!-- Author: Consumer Operator( Kyung Kwang Joo) -->



<!-- Date: Sat May 22 20:32:52.758 2004 -->
<pre>GAS Alamrm !! 
SHED ALARM SUMMARY   Mix & Storage Shed Alarm Summary   Alarm</pre>
<!-- Author: Daejung :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183054 ' target=_top>183054 </a>) -->
<!-- Comment: Sat May 22 20;33;13 comment by...Daejung -->
2 more times,


<!-- Date: Sat May 22 20:35:34.033 2004 -->
We got a couple of brief GAS alarms in the control room. I went 
down and checked with KC in cryo. He said an Arcon/CO2 trailer 
is low on pressure (75psi) and he will keep an eye on it.
<!-- Author: jj -->



<!-- Date: Sat May 22 20:36:14.598 2004 -->
<pre>PSM Alarm !!  
1RR18I_1 : 6.09V  
Now OK : 5.07 V 
</pre>
<!-- Author: Daejung :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183054 ' target=_top>183054 </a>) -->
<!-- Comment: Sat May 22 20;38;31 comment by...Daejung -->
same alarm 


<!-- Date: Sat May 22 20:46:34.579 2004 -->
<pre>PSM Alarm  
1RR18D_1  : 6.03V.  
It turned back to normal, 5.13 V 
</pre>
<!-- Author: Daejung :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183054 ' target=_top>183054 </a>) -->
<!-- Comment: Sat May 22 21;17;54 comment by...183054 -->
same alarm
<!-- Date: Sat May 22 20:59:17.520 2004 -->
<table><tr><td><a href=/elog/notebooks/2004/Sat_May_22_2004_Eve.205912.ps align=top><img src=/cgi/elog/elog.pl?nb=2004&action=view&page=-7658  align='top'></a></td><td>pedestrian trigger cross sections for this and previous store with silicon in. clearly something wrong with this stores luminosity measurement - but I guess you knew this before ...</td></tr></table>
<!-- Author: Marcel, Rong, Rainer -->


<!-- Date: Sat May 22 21:01:05.096 2004 -->
D0 called to say they have looked hard at their cross sections and luminosity and all appear legit.
<!-- Author: carlsmith -->


<!-- Date: Sat May 22 21:04:51.298 2004 -->
<pre>Event Builder error 
Attention!! SCPU_BAD_VRB_BYTE_COUNT Error !! 
SCPU b0eb22 
 
Auto HRR will be issued, 
 
Auto HRR issued, Now Fine</pre>
<!-- Author: Daejung & corrinne :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183054' target=_top>183054</a>) -->


<!-- Date: Sat May 22 21:10:41.354 2004 -->
<center><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7659&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7659' width='300'></a><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7660&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7660' width='300'></a><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7661&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7661' width='300'></a><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7662&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7662' width='300'></a></center>
<!-- Author: Daejung -->
<!-- Comment: Sat May 22 21;11;16 comment by...Daejung -->
Hourly Plots 20:00 - 21:00


<!-- Date: Sat May 22 21:13:48.910 2004 -->
Willis say if plug hv problems arise, call calorimeter general pager. TUrns out his pager was dead but he will investigate two dead pha channels.
<!-- Author: carlsmith -->


<!-- Date: Sat May 22 21:14:29.890 2004 -->
Mike Mulhearn just responded to TOF pager. He is sorry to say 
that for HV problems, Stefano Giagu is indeed the expert and 
Matt Jones (who is here now) certainly knows more than 
Mulhearn, So we get no additional help here.
<!-- Author: jj -->



<!-- Date: Sat May 22 21:35:24.348 2004 -->
Attention!!!. SCPU_BAD_VRB_BYTE_COUNT Error !!! 
 Hardware EVB has detected a problem with data quality in 
 SCPU b0eb25.  .... Auto-HRR recovered
<!-- Author: corrinne :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183054' target=_top>183054</a>) -->


<!-- Date: Sat May 22 21:38:47.733 2004 -->
<br><br> 
Current plan from Tevatron Run Coordinator is to keep this 
store in for about 24 hours. That would mean another shot 
on Sunday evening.<br><br>
<!-- Author: jj -->


<!-- Date: Sat May 22 22:13:54.542 2004 -->
Busy timeout from b0svx06, which auto-HRR recovered.
<!-- Author: corrinne :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183054' target=_top>183054</a>) -->


<!-- Date: Sat May 22 22:20:59.779 2004 -->
<center><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7663&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7663' width='300'></a><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7664&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7664' width='300'></a><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7665&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7665' width='300'></a><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7666&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7666' width='300'></a></center>
<!-- Author: Daejung -->
<!-- Comment: Sat May 22 22;21;36 comment by...Daejung -->
Hourly Plots 21:00 - 22:00


<!-- Date: Sat May 22 22:39:27.915 2004 -->
<pre>PSM Alarm 
1RR18D_1 : 6.04V 
Now OK.</pre>
<!-- Author: Daejung :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183054' target=_top>183054</a>) -->


<!-- Date: Sat May 22 22:41:58.616 2004 -->
<pre>PSM Alarm  
1RR18E_1 : 6.06V  
Now OK. 
 
</pre>
<!-- Author: Daejung :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183054' target=_top>183054</a>) -->
<!-- Comment: Sat May 22 23;48;19 comment by...Daejung -->
same alarm 6.02V
<!-- Date: Sat May 22 22:43:36.282 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183054">183054</A>
central: <tt>TOF HV off now. So YMON#84 TOF (1st plot is weird)</tt>
<!-- Author: Consumer Operator( Kyung Kwang Joo) -->


<!-- Date: Sat May 22 22:43:37.829 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183054">183054</A>
muons: <tt>- IMU CO Main BSU: 1st plot(no dead ch around phi=45 anymore), 4th plot(location of dead ch move from 25(ref plot) to 0(now). <br>
BMUCOCellsWest: 2nd plot(dead ch -25, 70 now, but 70 90 in the ref plot), 3rd plot(dead ch -25, 70 now, but 70 90 in the ref plot), 4th plot(dead ch -25 70, but 70, 90 in the ref plot) </tt>
<!-- Author: Consumer Operator( Kyung Kwang Joo) -->


<!-- Date: Sat May 22 22:43:38.115 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183054">183054</A>
trigger: <tt>-Xmon L2 prescaled X-section(bit=3 red). L3 x-section(bit 42 68 red)</tt>
<!-- Author: Consumer Operator( Kyung Kwang Joo) -->


<!-- Date: Sat May 22 22:50:38.661 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183054">183054</A>
 Terminated at 2004.05.22 22:50:04
<!-- Author: RunControl -->


<!-- Date: Sat May 22 22:51:54.755 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183054">183054</A>
TERMINATE: CLC has been fixed! TOF is still bad.
<!-- Author: corrinne x2080 -->



<!-- Date: Sat May 22 22:51:54.153 2004 -->
CLC folks report a cable connection was fixed and, looky here, bolum=dolum. Start a new run
<!-- Author: carlsmith -->

<!-- Date: Sat May 22 22:54:00.632 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183055">183055</A>
 Activated at 2004.05.22 22:53:48
<!-- Author: RunControl -->



<!-- Date: Sat May 22 22:55:06.487 2004 -->
<table border=0><td><tr><td bgcolor=#e0e030>Informed MCR about change in our luminosity.</td></tr></td></table>
<!-- Author: carlsmith :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183055' target=_top>183055</a>) -->

<!-- Date: Sat May 22 22:55:42.558 2004 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183055">183055</A>
ACTIVATE: AAA_CURRENT[1,475,455]: Silicon *IN*, CLC **FIXED**, TOF still _bad_news_.
<!-- Author: corrinne x2080 -->
<!-- Comment: Sat May 22 22;56;40 comment by...corrinne -->
Got the usual SRC fatal error L2A w/o L1A error.  Auto-HRR recovered


<!-- Date: Sat May 22 23:01:58.461 2004 -->
<p>We found a disconnected cable, which is used to send information 
about number of hits from CLC channels 92-95 to DPM. When we plugged 
the cable back CDF L = D0 L. We guess that this cable was occasionaly  
disconnected yesterday when power supply for the CLC01 crates was replaced. Luminosity during the store 3528 (runs 183053 and 183054) 
was overestimated by ~20%
<!-- Author: Roberto/Sasha -->


<!-- Date: Sat May 22 23:02:57.967 2004 -->
persistent busy timeout from b0cot12, quickly followed by a SCPU_CANT_RESET_VRB Error from VRB crate b0eb15, slot 14.  Brought the run to halt, then I and my trusty pen went downstairs to reset the VRB.  Recovered and ran, now things are fine.
<!-- Author: corrinne :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183055' target=_top>183055</a>) -->


<!-- Date: Sat May 22 23:23:44.673 2004 -->
<center><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7667&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7667' width='300'></a><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7668&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7668' width='300'></a><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7669&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7669' width='300'></a><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7670&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7670' width='300'></a></center>
<!-- Author: Daejung -->
<!-- Comment: Sat May 22 23;24;15 comment by...Daejung -->
Hourly Plots 22:00 - 23:00


<!-- Date: Sat May 22 23:55:35.617 2004 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183047>
183047</A> 
x2CB07</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_2_05 [1,475,455]</TD>                                       
<TD ALIGN=CENTER>18:28:59</TD>                                          
<TD ALIGN=CENTER>19:12:16</TD>                                          
<TD ALIGN=CENTER>00:31:08</TD>                                          
<TD ALIGN=RIGHT>     30,351,066</TD>                                    
<TD ALIGN=RIGHT>       288,297</TD>                                     
<TD ALIGN=RIGHT>      73,238</TD>                                       
<TD ALIGN=RIGHT>   95.231</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183053>
183053</A> 
x2CB0D</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_2_05 [1,475,455]</TD>                                       
<TD ALIGN=CENTER>19:17:56</TD>                                          
<TD ALIGN=CENTER>20:12:16</TD>                                          
<TD ALIGN=CENTER>00:33:46</TD>                                          
<TD ALIGN=RIGHT>     31,169,309</TD>                                    
<TD ALIGN=RIGHT>       513,947</TD>                                     
<TD ALIGN=RIGHT>     129,971</TD>                                       
<TD ALIGN=RIGHT>   92.519</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183054>
183054</A> 
x2CB0E</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_2_05 [1,475,455]</TD>                                       
<TD ALIGN=CENTER>20:23:05</TD>                                          
<TD ALIGN=CENTER>22:50:04</TD>                                          
<TD ALIGN=CENTER>02:17:29</TD>                                          
<TD ALIGN=RIGHT>    129,804,254</TD>                                    
<TD ALIGN=RIGHT>     2,122,139</TD>                                     
<TD ALIGN=RIGHT>     506,878</TD>                                       
<TD ALIGN=RIGHT>  321.197</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=183055>
183055</A> 
x2CB0F</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_2_05 [1,475,455]</TD>                                       
<TD ALIGN=CENTER>22:53:48</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>00:52:11</TD>                                          
<TD ALIGN=RIGHT>     46,983,706</TD>                                    
<TD ALIGN=RIGHT>       696,172</TD>                                     
<TD ALIGN=RIGHT>     174,398</TD>                                       
<TD ALIGN=RIGHT>   92.192</TD>                                          
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER></TD>                                                  
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>23:55:04</TD>                                                       
<TD ALIGN=CENTER>04:14:35</TD>                                          
<TD ALIGN=RIGHT>     238,308,335</TD>                                   
<TD ALIGN=RIGHT>      3,620,555</TD>                                    
<TD ALIGN=RIGHT>       884,485</TD>                                     
<TD ALIGN=RIGHT>  601.137</TD>                                          
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Sat May 22 23:56:37.386 2004 -->
<font color=red>Shift Summary: </font><pre>First run after yesterday's power outage with COT@Full voltage, Si on after a bit, gas recirculating, TOF off due to bad CAEN. Luminosity was off due to CLC cable, fixed. Occasional BMU trips. 
Store 3528, bolum = 28, stack=71</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   183047-183054
Delivered Luminosity   807  
Acquired Luminosity    591  
Efficiency             73
   
</td></tr></table></pre>
<!-- Author: carlsmith -->


<!-- Date: Sun May 23 00:03:48.419 2004 -->
<center><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7671&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7671' width='300'></a><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7672&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7672' width='300'></a><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7673&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7673' width='300'></a><a href='/cgi/elog/elog.pl?nb=2004&action=view&page=-7674&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2004&action=view&page=-7674' width='300'></a></center>
<!-- Author: Daejung -->
<!-- Comment: Sun May 23 00;04;38 comment by...Daejung -->
Hourly Plots 23:00(May 22) - 00:00(May 23)