<!-- Title: 10:20:15  Mon Jul  7 2008 -->
<!-- Crewchief: Sebastian Carron (MPC) -->
<!-- Op1: Sasha Sukhanov (SPL) -->
<!-- Op2: &nbsp; -->
<!-- Op3: &nbsp; -->
<!-- Op4: &nbsp; -->
<!-- Op5: &nbsp; -->
<!-- Op6: &nbsp; -->
<!-- Notes: -->
Start of Shift, Taking Data, no problems at the moment

<!-- Date: Mon Jul 7 11:55:46 2008 -->
<p><font size=+1> D-mode Calibration Job Finished </font>
<p><pre>
This is a computer-generated silicon D-mode calibration status report.
Silicon D-mode calibration job 509 FAILED.

Job info
--------
Calibration type : standard
DPS-off run      : 264170
DPS-on run       : 264169
Requested by     : timo
Submitted        : 2008-07-07 06:23:19
Completed        : 2008-07-07 11:55:29
Status           : FAILURE
Comment          : frozen lnx8-machine

Last processing stage reported : comparing to the reference calibration
Failure diagnostics :
The calibration failed the following reference comparison criteria:

            Cut Name    Sensor Set   Cut Value   Data Value
--------------------    ----------   ---------   ----------
    bad_chan rel max    SVX_Z        0.3         0.55795   
bad_fraction abs max    SVX_Z        0.025       0.044179  

Reference calibration runs are 263610 (DPS off) and 263609 (DPS on)
</pre>
<!-- Author: D-mode script run_calib_analysis.tcl -->



<!-- Date: Mon Jul 7 12:53:46 2008 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12727&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12727  width='300' align='top'></a></td><td>This ladder SB0W6L4 has had erratic behavious since the big shutdown in 2007, we had noticed this before already nbut I found it unignored and in pinky state in my shift. I am ignoring it again because increasing the BIAS limit does not make any sense in this case it is an erratic bias behavious and the problem should be investigated firther. Perhaps it has a bad cable connection or bad cables, perhaps we can try another Power Supply in its place.</td></tr></table>
<!-- Author: Sebastian -->


<!-- Date: Mon Jul 7 15:08:46 2008 -->
Auto-hockerization of CAEN crate #2. SAR started recovery procedure, but svxinter became frozen. I have stopped PSgui (actually, killed it) and SAR, reboot svxinter and restarted PSgui and SAR. Re-enabled SAR auto-recovery function and SAR recovered all ladders from PS#2. While I was making this entry, another auto-hockerization happened to PS crates #3 and 4. SAR recovered.
<!-- Author: Sebastian and SashaS -->

<!-- Date: Tue Jul 8 06:58:11 2008 -->
<p><font size=+1> D-mode Calibration Job Started </font>
<p><pre>
Job id            : 510
Calibration type  : standard
Run period tag    : Jul_2008
DPS-off run       : 264339
DPS-on run        : 264338
Requested by      : David Cox
Main process id   : 5643
</pre>
<!-- Author: CDF silicon D-mode calibration job server v1.4 -->


<!-- Date: Tue Jul 8 08:00:49 2008 -->
Entered in Silicon database: <ul>
<li>Ladder: IB3W0L0 
<li>problem n: 2310 
<li>Status: fixed
<li>Type: BIAS
<li>Action: IMON_limits
<li>entered by: carron 
<li>description is: Chenaged Bias0 limit from 210 to 250<table border=1>
<tr><th>IMON LIMITS</th><th>Old</th><th>New</th></tr>
<tr><td><b>Nominal</b></td><td>120</td><td>120</td></tr>
<tr><td><b>Lower</b></td><td>0.5</td><td>0.5</td></tr>
<tr><td><b>Upper</b></td><td>210</td><td>250</td></tr></table> 
<li>See <a href="http://b0sili01.fnal.gov/cgi-bin/pdb/DBreport.pl?detsys=ISL&wedge=B3W0&layer=L0">report</a></ul>
<!-- Author: Problem DB -->



<!-- Date: Tue Jul 8 08:10:34 2008 -->
There was a resonance during the night, I saved the trace and rearmed the logic analizer
<!-- Author: Sebastian -->


<!-- Date: Tue Jul 8 09:07:20 2008 -->
After goin to standby because of the end of store there was a problem with a trip on LB1W5L3, SAR was not able to recover the trip. I tried a few things and eventually was able to recover after hockerizing crate 15. Now all of the silicon is safely on standby.
<!-- Author: Sebastian -->

<!-- Date: Tue Jul 8 16:55:29 2008 -->
<p><font size=+1> D-mode Calibration Job Finished </font>
<p><pre>
This is a computer-generated silicon D-mode calibration status report.
Silicon D-mode calibration job 510 FAILED.

Job info
--------
Calibration type : standard
DPS-off run      : 264339
DPS-on run       : 264338
Requested by     : David Cox
Submitted        : 2008-07-08 06:57:21
Completed        : 2008-07-08 16:55:23
Status           : FAILURE

Last processing stage reported : starting histogram transfer
Failure diagnostics :
child process exited abnormally
</pre>
<!-- Author: D-mode script run_calib_analysis.tcl -->
<!-- Comment: Wed Jul  9 06;28;16 comment by...timo -->
<pre>jippii....</pre>

<!-- Date: Wed Jul 9 06:50:58 2008 -->
<p><font size=+1> D-mode Calibration Job Started </font>
<p><pre>
Job id            : 511
Calibration type  : standard
Run period tag    : Jul_2008
DPS-off run       : 264339
DPS-on run        : 264338
Requested by      : timo
Main process id   : 31928
Comment           : please please please... b0sili01 hick-up
</pre>
<!-- Author: CDF silicon D-mode calibration job server v1.4 -->



<!-- Date: Wed Jul 9 11:22:03 2008 -->
Anti-protons are finally in after a long time without taking data because of Tevatron problems. Waiting for ramping and then scraping.
<!-- Author: Sebastian -->


<!-- Date: Wed Jul 9 11:56:35 2008 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12728&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12728  width='300' align='top'></a></td><td>Store 6278 is in, no problems turning silicon on, initial luminosity of about 200E30. This is the usual SpyMon plot, looking ok. Also the SvxMon plots look ok. We are taking data with no problems so far.</td></tr></table>
<!-- Author: Sebastian  -->

<!-- Date: Wed Jul 9 15:13:57 2008 -->
<p><font size=+1> D-mode Calibration Job Finished </font>
<p><pre>
This is a computer-generated silicon D-mode calibration status report.
Silicon D-mode calibration job 511 FAILED.

Job info
--------
Calibration type : standard
DPS-off run      : 264339
DPS-on run       : 264338
Requested by     : timo
Submitted        : 2008-07-09 06:50:43
Completed        : 2008-07-09 15:12:58
Status           : FAILURE
Comment          : please please please... b0sili01 hick-up

Last processing stage reported : starting histogram transfer
Failure diagnostics :
child process exited abnormally
</pre>
<!-- Author: D-mode script run_calib_analysis.tcl -->



<!-- Date: Wed Jul 9 20:21:09 2008 -->
There was a spontaneous swith of in crate 5, followd by an auto hockerization and a successfull SAR recovery. Also the cryo guys silenced the silicon alarm, so I unignored it now. Sar is great I have to say.
<!-- Author: Sebastian -->


<!-- Date: Thu Jul 10 10:46:45 2008 -->
<pre>We did some quiet time studies on f533 e250 and e251. For f533 we were unable to reproduce the error in SVXDaq, so nothing we can do for now. Appearance of bit errors in e250 and e251 is correlated with the PS change (also an offset in Doim voltage is observed before and after the PS swap), so we decided not to go into access, but wait until we raplace the PS and then see what happens with these 2 ladders.</pre>
<!-- Author: SashaS and Sergo -->

<!-- Date: Thu Jul 10 11:20:47 2008 -->
<p><font size=+1> D-mode Calibration Job Finished </font>
<p><pre>
This is a computer-generated silicon D-mode calibration status report.
Silicon D-mode calibration job 511 FAILED.

Job info
--------
Calibration type : standard
DPS-off run      : 264339
DPS-on run       : 264338
Requested by     : timo
Submitted        : 2008-07-09 06:50:43
Completed        : 2008-07-10 11:20:19
Status           : FAILURE
Comment          : please please please... b0sili01 hick-up

Last processing stage reported : filtering dead channels
Failure diagnostics :
child process exited abnormally
</pre>
<!-- Author: D-mode script  -->
<!-- Comment: Thu Jul 10 12;34;23 comment by...timo -->
<pre>Only theory I can give, is that there has been high load on fcdflnx8 machine. Same scripts randomly crash at different points...

Testing again, less load at a moment...</pre>

<!-- Date: Thu Jul 10 19:15:36 2008 -->
<p><font size=+1> D-mode Calibration Job Finished </font>
<p><pre>
This is a computer-generated silicon D-mode calibration status report.
Silicon D-mode calibration job 510 FAILED.

Job info
--------
Calibration type : standard
DPS-off run      : 264339
DPS-on run       : 264338
Requested by     : David Cox
Submitted        : 2008-07-08 06:57:21
Completed        : 2008-07-10 19:14:48
Status           : FAILURE

Last processing stage reported : comparing to the reference calibration
Failure diagnostics :
child process exited abnormally
</pre>
<!-- Author: D-mode script  -->



<!-- Date: Fri Jul 11 04:58:04 2008 -->
I was paged because of shot setup was injecting pbars, the  
shot setup must have been really fast because when I arrived  
the scraping had been finished, When I arrived I immediately  
turned on the silicon, but we are not taking data yet because  
of a COT trip, once the COT gets fixed we can start taking data. But we lost about 5 minutes of data taking time because of the 
time I arrived at, sorry for that, that was a fast shot  
setup.
<!-- Author: Sebastian -->
<!-- Comment: Fri Jul 11 05;03;46 comment by...Sebastian -->
Ok the COT tripped twice, but now it is good and we are taking data. I take the blame for about 5 minutes of the delay in taking data. That was a very fast pbar injection and scraping. 


<!-- Date: Fri Jul 11 05:09:56 2008 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12729&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12729  width='300' align='top'></a></td><td>After some tribulations at the start of the store we are taking data for store 6284. SvxMON plots look good and this SpyMon plot looks good. </td></tr></table>
<!-- Author: Sebastian -->

<!-- Date: Fri Jul 11 05:36:29 2008 -->
Entered in Silicon database: <ul>
<li>Ladder: SB3W11L2 
<li>problem n: 1522 
<li>Status: fixed
<li>Type: BIAS
<li>Action: IMON_limits
<li>entered by: carron 
<li>description is: Increased bias limit due to normal aging<table border=1>
<tr><th>IMON LIMITS</th><th>Old</th><th>New</th></tr>
<tr><td><b>Nominal</b></td><td>200</td><td>200</td></tr>
<tr><td><b>Lower</b></td><td>0.5</td><td>0.5</td></tr>
<tr><td><b>Upper</b></td><td>265</td><td>285</td></tr></table> 
<li>See <a href="http://b0sili01.fnal.gov/cgi-bin/pdb/DBreport.pl?detsys=SVX&wedge=B3W11&layer=L2">report</a></ul>
<!-- Author: Problem DB -->


<!-- Date: Fri Jul 11 05:42:25 2008 -->
Entered in Silicon database: <ul>
<li>Ladder: SB2W2L4 
<li>problem n: 672 
<li>Status: fixed
<li>Type: BIAS
<li>Action: IMON_limits
<li>entered by: carron 
<li>description is: Changed Bias limit due to normal aging<table border=1>
<tr><th>IMON LIMITS</th><th>Old</th><th>New</th></tr>
<tr><td><b>Nominal</b></td><td>32.28</td><td>32.28</td></tr>
<tr><td><b>Lower</b></td><td>0.5</td><td>0.5</td></tr>
<tr><td><b>Upper</b></td><td>150</td><td>150</td></tr></table> 
<li>See <a href="http://b0sili01.fnal.gov/cgi-bin/pdb/DBreport.pl?detsys=SVX&wedge=B2W2&layer=L4">report</a></ul>
<!-- Author: Problem DB -->



<!-- Date: Fri Jul 11 08:22:05 2008 -->
Today is definitely a bad day for me, I set up my alarm clock wrong and missed 
the 8 oclock meeting,  just when I had some things to explain from last night. 
I just woke up to realize the 8 am meeting is over...
<!-- Author: Sebastian -->


<!-- Date: Fri Jul 11 11:14:19 2008 -->
Tevatron quenched, I am going to the crontrol room. From The Scico Description silicon seems to be ok.
<!-- Author: Sebastian -->
<!-- Comment: Fri Jul 11 11;45;29 comment by...Sebastian -->
Looking at the radiation monitors ther was no sudden spike of radiation in the quench. The silicon is ok.


<!-- Comment: Sun Jul 13 22;20;12 comment by...Satyajit -->
Did we do a COLDSTART? ACE doesn't seems to have done the rest of
the post-quench checkout.
<!-- Comment: Mon Jul 14 08;41;50 comment by...Sebastian -->
Yes I did
<!-- Date: Fri Jul 11 17:46:01 2008 -->
JJ woked on the B0control16 computer to have a display broadcast to the ACE, for that time SAR had to be stopped and restarted in b0dasvx01, now that JJ finished we are running SAR again in its usual place.
<!-- Author: Sebastidx -->

<!-- Date: Sun Jul 13 04:31:25 2008 -->
Entered in Silicon database: <ul>
<li>Ladder: SB4W3L2 
<li>problem n: 1789 
<li>Status: fixed
<li>Type: BIAS
<li>Action: IMON_limits
<li>entered by: carron 
<li>description is: NOrmal aging of the bias<table border=1>
<tr><th>IMON LIMITS</th><th>Old</th><th>New</th></tr>
<tr><td><b>Nominal</b></td><td>110</td><td>110</td></tr>
<tr><td><b>Lower</b></td><td>0.5</td><td>0.5</td></tr>
<tr><td><b>Upper</b></td><td>150</td><td>180</td></tr></table> 
<li>See <a href="http://b0sili01.fnal.gov/cgi-bin/pdb/DBreport.pl?detsys=SVX&wedge=B4W3&layer=L2">report</a></ul>
<!-- Author: Problem DB -->



<!-- Date: Sun Jul 13 04:34:48 2008 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12730&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12730  width='300' align='top'></a></td><td>Store 6290 is in no problems turning on the silicon, we are taking data, Initial Luminosity of 295E30. SvxMon plots look good, SpyMon plot here looks good. However we got a lot of pinkies at the begining of the store, so I will spend some time clearing them out.</td></tr></table>
<!-- Author: Sebastian -->

<!-- Date: Sun Jul 13 04:38:45 2008 -->
Entered in Silicon database: <ul>
<li>Ladder: SB0W0L2 
<li>problem n: 2574 
<li>Status: fixed
<li>Type: BIAS
<li>Action: IMON_limits
<li>entered by: carron 
<li>description is: Normal Bias Aging <table border=1>
<tr><th>IMON LIMITS</th><th>Old</th><th>New</th></tr>
<tr><td><b>Nominal</b></td><td>28.8</td><td>28.8</td></tr>
<tr><td><b>Lower</b></td><td>0.5</td><td>0.5</td></tr>
<tr><td><b>Upper</b></td><td>120</td><td>150</td></tr></table> 
<li>See <a href="http://b0sili01.fnal.gov/cgi-bin/pdb/DBreport.pl?detsys=SVX&wedge=B0W0&layer=L2">report</a></ul>
<!-- Author: Problem DB -->


<!-- Date: Sun Jul 13 04:40:49 2008 -->
Entered in Silicon database: <ul>
<li>Ladder: SB3W10L2 
<li>problem n: 2626 
<li>Status: fixed
<li>Type: BIAS
<li>Action: IMON_limits
<li>entered by: carron 
<li>description is: Changed Bias limit due to normal aging<table border=1>
<tr><th>IMON LIMITS</th><th>Old</th><th>New</th></tr>
<tr><td><b>Nominal</b></td><td>160.45</td><td>160.45</td></tr>
<tr><td><b>Lower</b></td><td>0.5</td><td>0.5</td></tr>
<tr><td><b>Upper</b></td><td>220.0</td><td>250.0</td></tr></table> 
<li>See <a href="http://b0sili01.fnal.gov/cgi-bin/pdb/DBreport.pl?detsys=SVX&wedge=B3W10&layer=L2">report</a></ul>
<!-- Author: Problem DB -->


<!-- Date: Sun Jul 13 04:50:29 2008 -->
Entered in Silicon database: <ul>
<li>Ladder: SB5W0L2 
<li>problem n: 1756 
<li>Status: fixed
<li>Type: BIAS
<li>Action: IMON_limits
<li>entered by: carron 
<li>description is: Changed Bias limit due to normal aging<table border=1>
<tr><th>IMON LIMITS</th><th>Old</th><th>New</th></tr>
<tr><td><b>Nominal</b></td><td>105</td><td>105</td></tr>
<tr><td><b>Lower</b></td><td>0.5</td><td>0.5</td></tr>
<tr><td><b>Upper</b></td><td>180</td><td>180</td></tr></table> 
<li>See <a href="http://b0sili01.fnal.gov/cgi-bin/pdb/DBreport.pl?detsys=SVX&wedge=B5W0&layer=L2">report</a></ul>
<!-- Author: Problem DB -->


<!-- Date: Sun Jul 13 04:50:29 2008 -->
Entered in Silicon database: <ul>
<li>Ladder: SB5W0L2 
<li>problem n: 1756 
<li>Status: fixed
<li>Type: BIAS
<li>Action: IMON_limits
<li>entered by: carron 
<li>description is: Changed Bias limit due to normal aging<table border=1>
<tr><th>IMON LIMITS</th><th>Old</th><th>New</th></tr>
<tr><td><b>Nominal</b></td><td>105</td><td>105</td></tr>
<tr><td><b>Lower</b></td><td>0.5</td><td>0.5</td></tr>
<tr><td><b>Upper</b></td><td>180</td><td>210</td></tr></table> 
<li>See <a href="http://b0sili01.fnal.gov/cgi-bin/pdb/DBreport.pl?detsys=SVX&wedge=B5W0&layer=L2">report</a></ul>
<!-- Author: Problem DB -->



<!-- Date: Sun Jul 13 05:31:42 2008 -->
There was a clean abhort of the tevatron due to a power supply problem with them. Will see how long until we have a store
<!-- Author: Sebastian -->

<!-- Date: Sun Jul 13 13:18:20 2008 -->
<p><font size=+1> D-mode Calibration Job Started </font>
<p><pre>
Job id            : 512
Calibration type  : standard
Run period tag    : Jul_2008
DPS-off run       : 264170
DPS-on run        : 264169
Requested by      : timo
Main process id   : 18537
Comment           : double cheking
</pre>
<!-- Author: CDF silicon D-mode calibration job server v1.4 -->


<!-- Date: Sun Jul 13 15:23:26 2008 -->
Entered in Silicon database: <ul>
<li>Ladder: SB3W11L2 
<li>problem n: 1522 
<li>Status: fixed
<li>Type: BIAS
<li>Action: IMON_limits
<li>entered by: carron 
<li>description is: Needed to increase the limits just a little more<table border=1>
<tr><th>IMON LIMITS</th><th>Old</th><th>New</th></tr>
<tr><td><b>Nominal</b></td><td>200</td><td>200</td></tr>
<tr><td><b>Lower</b></td><td>0.5</td><td>0.5</td></tr>
<tr><td><b>Upper</b></td><td>285</td><td>300</td></tr></table> 
<li>See <a href="http://b0sili01.fnal.gov/cgi-bin/pdb/DBreport.pl?detsys=SVX&wedge=B3W11&layer=L2">report</a></ul>
<!-- Author: Problem DB -->


<!-- Date: Sun Jul 13 15:25:30 2008 -->
Entered in Silicon database: <ul>
<li>Ladder: SB4W9L2 
<li>problem n: 2536 
<li>Status: fixed
<li>Type: BIAS
<li>Action: IMON_limits
<li>entered by: carron 
<li>description is: Changed Bias limit due to normal aging<table border=1>
<tr><th>IMON LIMITS</th><th>Old</th><th>New</th></tr>
<tr><td><b>Nominal</b></td><td>111</td><td>111</td></tr>
<tr><td><b>Lower</b></td><td>0.5</td><td>0.5</td></tr>
<tr><td><b>Upper</b></td><td>240</td><td>280</td></tr></table> 
<li>See <a href="http://b0sili01.fnal.gov/cgi-bin/pdb/DBreport.pl?detsys=SVX&wedge=B4W9&layer=L2">report</a></ul>
<!-- Author: Problem DB -->


<!-- Date: Sun Jul 13 15:26:40 2008 -->
Entered in Silicon database: <ul>
<li>Ladder: SB4W10L2 
<li>problem n: 1703 
<li>Status: fixed
<li>Type: BIAS
<li>Action: IMON_limits
<li>entered by: carron 
<li>description is: Changed Bias limit due to normal aging<table border=1>
<tr><th>IMON LIMITS</th><th>Old</th><th>New</th></tr>
<tr><td><b>Nominal</b></td><td>145</td><td>145</td></tr>
<tr><td><b>Lower</b></td><td>0.5</td><td>0.5</td></tr>
<tr><td><b>Upper</b></td><td>220</td><td>250</td></tr></table> 
<li>See <a href="http://b0sili01.fnal.gov/cgi-bin/pdb/DBreport.pl?detsys=SVX&wedge=B4W10&layer=L2">report</a></ul>
<!-- Author: Problem DB -->



<!-- Date: Sun Jul 13 15:28:44 2008 -->
<table><tr><td><a href=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12731&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=-12731  width='300' align='top'></a></td><td>Store 6292 is in with initial luminosity of 247E30, no problems at the begining. SpyMon plot is ok, SvxMon plots look ok also, veryu low p and pbar losses. We are taking data.</td></tr></table>
<!-- Author: Sebastian -->

<!-- Date: Sun Jul 13 18:06:50 2008 -->
<p><font size=+1> D-mode Calibration Job Finished </font>
<p><pre>
This is a computer-generated silicon D-mode calibration status report.
Silicon D-mode calibration job 512 finished successfully.

Job info
--------
Calibration type : standard
DPS-off run      : 264170
DPS-on run       : 264169
Requested by     : timo
Submitted        : 2008-07-13 13:18:17
Completed        : 2008-07-13 18:06:26
Status           : SUCCESS
Comment          : double cheking
</pre>
<!-- Author: D-mode script run_calib_analysis.tcl -->




<!-- Date: Mon Jul 14 01:08:57 2008 -->
<font size="3"> 
  
<h3> ADC Monitering Summary (Run264194-264535) </h3> 
I have checked ADCMon plots during this week from run264194 to 264535. 
<br> 
I will report the ladders which have a issue and intersting work.  
The other ladders look fine. 
<ul> 
<li> f533: This ladder shows unstable Bit Error (10-99) since Run264333. 
 
</ul> 
</font>
<!-- Author: Takayuki -->


<!-- Date: Mon Jul 14 07:57:37 2008 -->
There where 3 resonances in a row, but since twe where taking bad data with a COT problem, I asked for the silicon to be put to standby at around 6:30 am.  Now the store is about to be dumped
<!-- Author: Sebastian -->


<h2> Summary of the week </h2> 
<li> This week in general the silicon behaved VERY stably, but the tevatron had some problems</li> 


<h3> Monday  </h3> 
<ul> <li> Inherited store from previous shift</li> 
     <li> Today I found some erratic behaviour in some ladders that we had catched before but the ladders where unignored, we need to be more organized in keeping track of ignored ladders and what needs to be fixed</li> 
     <li> AutoHockerization of crate 2, SAR could not recover because svxinter was frozen, after restarting SVX inter and the PS gui it was all fine.</li>
     
</ul> 
 
<h3> Tuesday  </h3> 
<ul> <li> Resonance at the end of the store</li> 
     <li> D-mode calibration done</li> 
     <li> After going to standby at the end of the store, there was a trip in LB1W5L3, 
SAR did not recover the trip, eventually it was fixed with  a Hockerization of crate 15 </li>
</ul> 
 
<h3> Wednesday </h3> 
<ul> <li> Lots of problems with the Tevatron since tuesday, eventually we got store 6278
at around noon, with initial luminosity of  200E30 </li> 
     <li> The Store did not last long, another D-mode Calibration done</li> 
     <li> Spontaneous swith of in crate 5, followd by an auto hockerization and a successfull 
SAR recovery, the cryo guys silenced the silicon alarm, so I unignored it later </li>
</ul> 
 
<h3> Thursday  </h3> 
<ul> <li> All day without a store with Tevatron problems</li> 
     <li> Sasha and Sergo did some quiet time studies on f533 e250 and e251 </li> 
     <li> More D mode calibrations, not much else to do anyway </li>
</ul>    
 
<h3> Friday  </h3> 
<ul> <li> We finally got a store in at 4 in the morning #6284, There was a little bit of
a miscomunication with the Scico resulting in me arriving a few minutes after the 
scaping was done, I turned the silicon on immediately, but the COT went through a 
few trips before we could take data. </li>
<li>  The store lasted only a few hours, as the tevatron quenched, the after quench
checkout revealed no problems. I did a coldstart after the quench with no problems,
but the ace seems to not have done the full quench checkout. </li> 
<li> During quiet time JJ worked on b0control16 to instal a video feed of the display
to the ace computer monitors.</li>
</ul>   
 
<h3> Saturday </h3> 
<ul> <li>  All of Saturday waiting for a store that never came</li> 
</ul> 
 
<h3> Sunday </h3> 
<ul> <li> Finally we got store #6290 at 4 in the morming, initial luminosity of 
295E30, but the store lasted only a little more than 1 hour, with a clean abhort 
by the Tovatron to fix a pump problem. </li> 
     <li> At around 3 pm the new store 6292 came in with initial luminosity of 247E30</li> 
     <li> Takayuki's Monitoring summary</li> 
</ul> 
 
<h3> Monday  </h3> 
<ul> <li> During the night we had 3 resonances, in a row, I was paged and I asked for the
silicon to be turned to standby, also because I learned that we where having problems
with the COT and taking bad COT data, so there was no reason to have the silicon in 
the first place</li> 
     <li> Will spend the rest of my last shift clearing ignored ladders.</li> 

<!-- Date: Mon Jul 14 08:57:39 2008 -->
Entered in Silicon database: <ul>
<li>Ladder: SB1W11L2 
<li>problem n: 1961 
<li>Status: fixed
<li>Type: BIAS
<li>Action: IMON_limits
<li>entered by: carron 
<li>description is: Normal aging bias limit redefinition<table border=1>
<tr><th>IMON LIMITS</th><th>Old</th><th>New</th></tr>
<tr><td><b>Nominal</b></td><td>??</td><td>??</td></tr>
<tr><td><b>Lower</b></td><td>??</td><td>??</td></tr>
<tr><td><b>Upper</b></td><td>150</td><td>180</td></tr></table> 
<li>See <a href="http://b0sili01.fnal.gov/cgi-bin/pdb/DBreport.pl?detsys=SVX&wedge=B1W11&layer=L2">report</a></ul>
<!-- Author: Problem DB -->


<!-- Date: Mon Jul 14 09:00:22 2008 -->
Entered in Silicon database: <ul>
<li>Ladder: SB3W3L0 
<li>problem n: 1852 
<li>Status: fixed
<li>Type: BIAS
<li>Action: IMON_limits
<li>entered by: carron 
<li>description is: Normal aging bias limit redefinition<table border=1>
<tr><th>IMON LIMITS</th><th>Old</th><th>New</th></tr>
<tr><td><b>Nominal</b></td><td>51.5</td><td>51.5</td></tr>
<tr><td><b>Lower</b></td><td>0.5</td><td>0.5</td></tr>
<tr><td><b>Upper</b></td><td>200</td><td>230</td></tr></table> 
<li>See <a href="http://b0sili01.fnal.gov/cgi-bin/pdb/DBreport.pl?detsys=SVX&wedge=B3W3&layer=L0">report</a></ul>
<!-- Author: Problem DB -->