<!-- Title: Day shift. Sun Oct  9, 2005 -->
<!-- SciCo: Rick St. Denis -->
<!-- DAQAce: Sergo Jindariani -->
<!-- MonAce: John Freeman -->
<!-- CO: Anton Anastassov -->
<!-- OpManager: Aron Soha -->
<!-- Notes: -->
<pre>Running smoothly, Beam dump at 11, Access for TDC replacement</pre>


<!-- Date: Sun Oct 9 08:44:26.851 2005 -->
Eric James Called and will head in now. He will contact Jianrong.
<!-- Author: Rick St. Denis :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205324' target=_top>205324</a>) -->


<!-- Date: Sun Oct 9 08:45:37.015 2005 -->
 Got Busy TO for crate b0clc00 , followed immediately by crates b0cot08 and b0cot09. HRR recovered it  
<!-- Author: Sergo :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205324' target=_top>205324</a>) -->


<!-- Date: Sun Oct 9 08:46:14.251 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-18263&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-18263' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-18264&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-18264' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-18265&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-18265' width='300'></a></center>
<!-- Author: Freeman -->
<!-- Comment: Sun Oct  9 08;46;46 comment by...Freeman -->
Hourly plots, 0730-0830


<!-- Date: Sun Oct 9 08:47:12.358 2005 -->
<pre>MLE) b0svx00:Messenger:Oct 9, 2005 8:37:45 AM->Silicon Resonator: Sl 10 Ch 2 -> e022: Bytes: 00066 
 
(MLE) b0svx01:Messenger:Oct 9, 2005 8:37:45 AM->Silicon Resonator: Sl 10 Ch 2 -> e122: Bytes: 00056 
 
(MLE) b0svx05:Messenger:Oct 9, 2005 8:37:44 AM->Silicon Resonator: Sl 10 Ch 2 -> e522: Bytes: 00226 
 
(MLE) b0svx04:Messenger:Oct 9, 2005 8:37:44 AM->Silicon Resonator: Sl 16 Ch 3 -> e463: Bytes: 00084 
 
 
 Hrr-ed </pre>
<!-- Author: Sergo :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205324' target=_top>205324</a>) -->


<!-- Date: Sun Oct 9 09:03:09.089 2005 -->
Busy TO - crate b0l2pu01 . HRRed. Getting close to having too few  farms in L3. One-two more HRRs and I'll have to restart the run 
<!-- Author: Sergo :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205324' target=_top>205324</a>) -->
<!-- Comment: Sun Oct  9 09;08;11 comment by...Sergo -->
I meant buffers sorry

<!-- Date: Sun Oct 9 09:13:38.301 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205324">205324</A>
 Terminated at 2005.10.09 09:13:00
<!-- Author: RunControl -->


<!-- Date: Sun Oct 9 09:13:39.376 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205324">205324</A>
TERMINATE: too few buffers left in L3 
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 09:17:16.937 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205325">205325</A>
 Activated at 2005.10.09 09:16:34
<!-- Author: RunControl -->


<!-- Date: Sun Oct 9 09:17:17.588 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205325">205325</A>
ACTIVATE: RunSet: AAA_CURRENT  Trigger: PHYSICS_3_07 [2,713,594] 
<!-- Author: Sergo x2080 -->



<!-- Date: Sun Oct 9 09:18:12.716 2005 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2005&action=view&page=-18266&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2005&action=view&page=-18266  width='300' align='top'></a></td><td>Channel that briefly went pink</td></tr></table>
<!-- Author: Freeman -->
<!-- Comment: Sun Oct  9 09;19;01 comment by...Freeman -->
Hard to see the name on the plot - it's Silicon AVDD, B3W7L2
<!-- Date: Sun Oct 9 09:19:01.810 2005 -->
Busy TO crates b0svx06 b0l2pu01 during last run - HRR  leading to no buffers in L3 - had to restart the run. 
<!-- Author: Sergo -->



<!-- Date: Sun Oct 9 09:24:57.443 2005 -->
Flying wire
<!-- Author: Sergo :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205325' target=_top>205325</a>) -->


<!-- Date: Sun Oct 9 09:37:24.637 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-18267&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-18267' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-18268&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-18268' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-18269&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-18269' width='300'></a></center>
<!-- Author: Freeman -->
<!-- Comment: Sun Oct  9 09;37;40 comment by...Freeman -->
0830-0930
<!-- Date: Sun Oct 9 09:37:25.261 2005 -->
<table border=0><td><tr><td bgcolor=#e0e030>Xiaojian Zhang and the primary CMP pager will be out of town through Tuesday. CMP pager coverage will be provided by Catalin (LRP 630-218-4121).</td></tr></td></table>
<!-- Author: Phil -->



<!-- Date: Sun Oct 9 09:45:28.068 2005 -->
<pre>Got b0clc00 Done TO and  
 crate/s: b0clc00(118),  Done TO. [RXPT] 
(MLE) b0dap73.fnal.gov:Thread-27:Oct 9, 2005 9:43:53 AM->Requested Halt-Recover-Run issued (errmon) 
(MLE) b0svx03:Messenger:Oct 9, 2005 9:43:59 AM->Silicon Resonator: Sl 10 Ch 7 -> e332: Bytes: 00072 
 
(MLE) b0svx04:Messenger:Oct 9, 2005 9:43:59 AM->Silicon Resonator: Sl 18 Ch 9 -> e494: Bytes: 00092 
 
(MLE) b0svx00:Messenger:Oct 9, 2005 9:43:59 AM->Silicon Resonator: Sl 16 Ch 9 -> e074: Bytes: 00146 
 
(MLE) b0svx08:Messenger:Oct 9, 2005 9:43:59 AM->Silicon Resonator: Sl 20 Ch 0 -> f220: Bytes: 00006 
 
HRR.</pre>
<!-- Author: Sergo :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205325' target=_top>205325</a>) -->


<!-- Date: Sun Oct 9 09:51:43.086 2005 -->
Done TO b0clc00. HRR
<!-- Author: Sergo :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205325' target=_top>205325</a>) -->
<!-- Comment: Sun Oct  9 09;52;23 comment by...Sergo -->
Somehow automatic HRR wasn't issued. I did it manually.

<!-- Comment: Sun Oct  9 09;53;36 comment by...Sergo -->
<pre>more info, in case someone finds it useful

MLE) b0clc00:Messenger:Oct 9, 2005 9:51:00 AM->TDC Header Mismatch evt 249952 slot 21: static/fifo 0x00800E19/0x00800D83 [10C,114]
(MLE) b0clc00:Messenger:Oct 9, 2005 9:51:00 AM->TDC evt 249952 slot 21 static/fifo: ModID 1/1 L2B 0/0 Hits 14/13 BC 25/131
(MLE) b0clc00:Messenger:Oct 9, 2005 9:51:00 AM->Event 249952: Bunchcounters in slot 16 (BC=25) and slot 21 (BC=131) disagree (L2B=0) 
(MLE) b0clc00:Messenger:Oct 9, 2005 9:51:00 AM->Runtime Error 4, Event 249952: Bunch counter mismatch, mismatch count = 1
(</pre>


<!-- Date: Sun Oct 9 09:57:03.883 2005 -->
boclc00 again, exactly same thing 
HRR helped but this is second time in the last ~5 minutes 
<!-- Author: Sergo :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205325' target=_top>205325</a>) -->
<!-- Comment: Sun Oct  9 10;08;42 comment by...Sergo -->
Happened again, I'm shepherding the crate

<!-- Comment: Sun Oct  9 10;09;48 comment by...Sergo -->
Shepherd was successful - resume the run


<!-- Date: Sun Oct 9 10:16:02.954 2005 -->
<pre>Done TO from clc00 . HRR and we're back to normal.  
Second Done TO immediately after that.- HRR</pre>
<!-- Author: Freeman -->
<!-- Comment: Sun Oct  9 10;21;39 comment by...Sergo -->
And again. I'll try shepherding it one more time. Shepherded it - running now.


<!-- Date: Sun Oct 9 10:29:00.265 2005 -->
clc00 kept giving Done TO errors. We did visiondemo reset.
<!-- Author: Sergo & Jianrong :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205325' target=_top>205325</a>) -->


<!-- Date: Sun Oct 9 10:37:48.256 2005 -->
Btw, I noticed that even though there were many HRRs - number of L3 buffers is still high.
<!-- Author: Sergo -->


<!-- Date: Sun Oct 9 10:51:58.912 2005 -->
MCR Called: plan to terminate store at 11:00.We should call when ready with HV
<!-- Author: Rick St. Denis :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205325' target=_top>205325</a>) -->

<!-- Date: Sun Oct 9 10:52:38.101 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205325">205325</A>
 Terminated at 2005.10.09 10:52:11
<!-- Author: RunControl -->


<!-- Date: Sun Oct 9 10:52:40.858 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205325">205325</A>
TERMINATE: End of store.
<!-- Author: Sergo x2080 -->



<!-- Date: Sun Oct 9 11:00:01.021 2005 -->
Voltages Down, Called MCR to notify them that they can dump.
<!-- Author: Rick St. Denis :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205325' target=_top>205325</a>) -->


<!-- Date: Sun Oct 9 11:07:23.705 2005 -->
Starting calibrations.
<!-- Author: Sergo -->

<!-- Date: Sun Oct 9 11:12:43.401 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205326">205326</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Sergo x2080 -->
<!-- Comment: Sun Oct  9 14;02;07 comment by...Anton -->
There are additional bad channels compared to the ref. These were seen earlier (Oct 1) and comments posted in the elog.

<!-- Date: Sun Oct 9 11:12:44.188 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205326">205326</A>
TERMINATE: Calibr.Done
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 11:12:53.948 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205327">205327</A>
ACTIVATE: RunSet: CALIB_CLC_QIE 
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 11:13:39.044 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205327">205327</A>
TERMINATE: Calibr. Done
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 11:21:19.295 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205328">205328</A>
ACTIVATE: RunSet: CALIB_QIE_FWD 
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 11:21:20.786 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205328">205328</A>
TERMINATE: Calibr.Done
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 11:23:26.295 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205329">205329</A>
ACTIVATE: RunSet: CALIB_RP_SMQIE 
<!-- Author: Sergo x2080 -->
<!-- Comment: Sun Oct  9 13;15;33 comment by...Anton -->
checked: OK
<!-- Date: Sun Oct 9 11:23:41.640 2005 -->
MCR Called: ready for access
<!-- Author: Rick St. Denis -->

<!-- Date: Sun Oct 9 11:24:41.420 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205329">205329</A>
TERMINATE: Calibr.Done
<!-- Author: Sergo x2080 -->



<!-- Date: Sun Oct 9 11:25:24.942 2005 -->
Silicon and TDC preparing to go in.  Silicon will do a fiber swap to diagnose a ladder.  May swap a FIB.
<!-- Author: Rick St. Denis -->

<!-- Date: Sun Oct 9 11:26:28.217 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205330">205330</A>
ACTIVATE: RunSet: ACE_TOF_QIE_NEW 
<!-- Author: Sergo x2080 -->
<!-- Comment: Sun Oct  9 13;16;06 comment by...Anton -->
checked: OK
<!-- Date: Sun Oct 9 11:27:33.397 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205330">205330</A>
TERMINATE: Calibr. Done
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 11:35:21.299 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205331">205331</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 11:35:22.033 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205331">205331</A>
TERMINATE: Calibr_load didn't go through
<!-- Author: Sergo x2080 -->
<!-- Comment: Sun Oct  9 13;43;36 comment by...Sergo -->
calib_load was requested by expert for testing purposes
<!-- Date: Sun Oct 9 11:37:41.077 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205332">205332</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 11:42:30.019 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205332">205332</A>
TERMINATE: same 
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 11:45:44.126 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205333">205333</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 11:52:54.854 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205333">205333</A>
TERMINATE: same 
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 11:54:27.073 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205334">205334</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 11:57:08.837 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205334">205334</A>
TERMINATE: same
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 11:59:15.308 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205335">205335</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Sergo x2080 -->
<!-- Comment: Sun Oct  9 13;34;34 comment by...Anton -->
looks OK, but there are additional bad channels compared to the ref.
<!-- Date: Sun Oct 9 11:59:16.777 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205335">205335</A>
TERMINATE: Calibr. DOne
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 12:02:14.122 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205336">205336</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: Sergo x2080 -->
<!-- Comment: Sun Oct  9 14;32;38 comment by...Anton -->
checked: OK
<!-- Date: Sun Oct 9 12:04:04.483 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205336">205336</A>
TERMINATE: Calibr.Done
<!-- Author: Sergo x2080 -->



<!-- Date: Sun Oct 9 12:05:17.718 2005 -->
<pre>Attempted to add the same delays Admem uses when loading flashram 
to SMX code,  didn't work.   PCAL01 hung twice,  and the first time 
CCAL09 gave a heap corrupt error,  second time CCAL06 gave a heap 
corrupt error.   Noticed when spying on PCAL01 the second time 
that it said "crate_wait 5 ns".  Will ask Admem experts if this makes sense.  </pre>
<!-- Author: Kuhlmann -->

<!-- Date: Sun Oct 9 12:06:42.507 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205337">205337</A>
ACTIVATE: RunSet: CAL_NOM 
<!-- Author: jameseb -->


<!-- Date: Sun Oct 9 12:06:43.905 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205337">205337</A>
TERMINATE: Ambrose Calibration - Seems ok.
<!-- Author: jameseb -->


<!-- Date: Sun Oct 9 12:10:21.457 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205338">205338</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 12:12:11.726 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205338">205338</A>
TERMINATE: Calibr. done
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 12:14:30.309 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205339">205339</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Sergo x2080 -->
<!-- Comment: Sun Oct  9 13;10;04 comment by...Anton -->
checked: OK
<!-- Date: Sun Oct 9 12:14:31.980 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205339">205339</A>
TERMINATE: Calibr. Done
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 12:25:17.148 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205340">205340</A>
TERMINATE: got heap corrupt error for cot12. Bad.
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 12:25:18.678 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205341">205341</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Sergo x2080 -->
<!-- Comment: Sun Oct  9 13;14;40 comment by...Anton -->
checked: OK
<!-- Date: Sun Oct 9 12:25:19.887 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205341">205341</A>
TERMINATE: Calibr. done 
<!-- Author: Sergo x2080 -->



<!-- Date: Sun Oct 9 12:25:20.721 2005 -->
COT calibration after the change of card in crate 2 slot 15 
Calibration will be rerun. 
---------------------------------------------------------------- 
 
COT Calibration 12:12:16 on 2005-10-09 
aveStart:                        210.14    
total numBad:                    356     - o.k. 
   numBad  cells:                  2       
   numBad cables:                  0       
   numBad slots:                   0       
 
aveDrift:                       0.055     
max number drifted in crate:    80        
total numDrifted:              193  
   number of drifted  cells        0       
   number of drifted cables        0      
   number of drifted  slots        2       
 
 
This calibration is o.k. 
*** Timing has drifted, constants updated (i.e new COMPLETE file written to DB)*** 
  
Out of the last 8 calibrations, constants have been updated 7 times 
 
 
 
Expert Only: 
file: /cdf/onln/home/cdfdaq/consumers/log/runCotCtt_0_2005-10-09_12:12:16.log 
 
Drifted Slots: 
crate  2 slot 15: 77 
crate 15 slot 18: 79 
 
 
Bad Cells: 
crate  2 slot  5 asdq  6: 12 
crate  3 slot 14 asdq  3:  9 
 
 
 
  
<!-- Author: Anton :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205339' target=_top>205339</a>) -->

<!-- Date: Sun Oct 9 12:27:03.042 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205342">205342</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 12:31:28.096 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205342">205342</A>
TERMINATE: Calibr.done. System in standby, no beam in TeV
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 12:31:29.294 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205343">205343</A>
ACTIVATE: RunSet: TOF_TAC_CALIB 
<!-- Author: Sergo x2080 -->



<!-- Date: Sun Oct 9 12:31:34.677 2005 -->
<pre>Drifted cells are gone  
---------------------------------------------------- 
COT Calibration 12:22:09 on 2005-10-09 
aveStart:                        210.15    
total numBad:                    351     - o.k. 
   numBad  cells:                  2       
   numBad cables:                  0       
   numBad slots:                   0       
 
aveDrift:                       0.023     
max number drifted in crate:    28        
total numDrifted:               39  
   number of drifted  cells        0       
   number of drifted cables        0      
   number of drifted  slots        0       
 
 
This calibration is o.k. 
Timing has not changed, no new constants written 
  
Out of the last 8 calibrations, constants have been updated 6 times 
 
 
 
Expert Only: 
file: /cdf/onln/home/cdfdaq/consumers/log/runCotCtt_0_2005-10-09_12:22:09.log 
 
Bad Cells: 
crate  2 slot  5 asdq  6: 12 
crate  3 slot 14 asdq  3:  9 
 
 
</pre>
<!-- Author: Anton :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205341' target=_top>205341</a>) -->

<!-- Date: Sun Oct 9 12:39:59.811 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205343">205343</A>
TERMINATE: Calibr. Done
<!-- Author: Sergo x2080 -->
<!-- Comment: Sun Oct  9 13;08;41 comment by...Anton -->
checked: OK
<!-- Date: Sun Oct 9 12:48:21.252 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205344">205344</A>
TERMINATE: OPs. manager wants muon calibr done first
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 12:53:29.882 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205345">205345</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 - Redoing calibr after access
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 12:53:30.952 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205345">205345</A>
TERMINATE: Calibr Done. SYstem in STandby. No beam in TeV
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 12:55:49.250 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205346">205346</A>
 Activated at 2005.10.09 12:54:54
<!-- Author: RunControl -->


<!-- Date: Sun Oct 9 12:55:50.948 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205346">205346</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [5,720,540] 
<!-- Author: Sergo x2080 -->


<!-- Date: Sun Oct 9 13:03:17.216 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205346">205346</A>
 Terminated at 2005.10.09 13:02:55
<!-- Author: RunControl -->



<!-- Date: Sun Oct 9 13:03:18.548 2005 -->
MCR called to ask if we are done. We are done. 
TDC is fine, Silicon is no better, no worse.
<!-- Author: Rick St. Denis -->

<!-- Date: Sun Oct 9 13:04:11.085 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205346">205346</A>
TERMINATE: Calibration Done. L1 Rate ~ 60-70 Hz , no beam.
<!-- Author: Sergo x2080 -->



<!-- Date: Sun Oct 9 13:07:47.863 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-18270&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-18270' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-18271&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-18271' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-18272&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-18272' width='300'></a></center>
<!-- Author: Freeman -->
<!-- Comment: Sun Oct  9 13;08;14 comment by...Freeman -->
Apologies for the delay - plots are from 0930-1105






<!-- Date: Sun Oct 9 13:18:15.267 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205347">205347</A>
 Activated at 2005.10.09 13:18:07
<!-- Author: RunControl -->



<!-- Date: Sun Oct 9 13:18:17.778 2005 -->
<pre>Jianrong and I replaced the TDC in slot15 of b0cot02  
on access.  New board seems to pass all tests.  We  
ask the shift crew to please take a close look at  
stage0 and XFT plots at start of the next store.  A 
good COT calibration was taken, although there were  
not enough changes to cause a new calibration file  
to be written out.</pre>
<!-- Author: Eric James -->

<!-- Date: Sun Oct 9 13:18:59.281 2005 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205347">205347</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,716,538] Running cosmics ...
<!-- Author: Sergo x2080 -->
<!-- Comment: Sun Oct  9 14;54;08 comment by...Sergo -->
Everything looks fine. After they replaced tdc board Cot02 crate doesn't bother us any more. 
<!-- Date: Sun Oct 9 13:53:59.942 2005 -->
Busy TO b0tsi03
<!-- Author: Sergo :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205347' target=_top>205347</a>) -->



<!-- Date: Sun Oct 9 14:09:06.431 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-18273&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-18273' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-18274&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-18274' width='300'></a></center>
<!-- Author: Anton -->
<!-- Comment: Sun Oct  9 14;13;14 comment by...Anton -->
calqie, calib on 10/09/05 (day)
chk ped for 23E, ch 11, cap 0-3;
chk slopes for 18E, ch 4 


<!-- Date: Sun Oct 9 14:14:07.206 2005 -->
<h4>Silicon Access Report</h4> 
 
<p>SIlicon went in to investigate IB2W2L0 as this ladder was not returning any data during Saturday morning; although it fixed itself later.  We were able to reconfirm the error and we went into the collision hall to replace an RX and FIB.  However this didn&339t remedy the problem.  It was eventually traced to a VRB which had to be reset.  So the Silicon was unnecessary.<p> 
 
<p>We did a COLD-START to put the Silicon detector back to its default state and it should now be ready. Furthur details can be found in the <a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=269&frame=2&anchor=14:09:49'> 
silicon e-log</a></p>. 
 
 
 
 
<!-- Author: Ankush, Sebastian,Sasha, Mark and Jared -->



<!-- Date: Sun Oct 9 14:42:38.639 2005 -->
<center><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-18276&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-18276' width='300'></a><a href='/cgi/elog/elog.pl?nb=2005&action=view&page=-18275&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2005&action=view&page=-18275' width='300'></a></center>
<!-- Author: Anton -->
<!-- Comment: Sun Oct  9 14;44;19 comment by...Anton -->
Wedge 15 not in the reference - pls add or put comments.


<!-- Date: Sun Oct 9 14:45:20.407 2005 -->
I hear three toots of an alarm. No visual indicators of a problem. 
<!-- Author: Freeman -->
<!-- Comment: Sun Oct  9 15;22;10 comment by...Freeman -->
Again
<!-- Date: Sun Oct 9 15:01:00.870 2005 -->
Silicon has told us they are done.
<!-- Author: Rick St. Denis -->
<!-- Comment: Sun Oct  9 15;51;09 comment by...Robin (80 degrees in Davis) -->
No news or photos of the CDF runners in the marathon yet?  I count on the e-logs to keep me informed about these things!
<!-- Date: Sun Oct 9 15:06:53.012 2005 -->
b0tsi00:Messenger:Oct 9, 2005 3:05:25 PM->TM2 Timeout waiting for pending events from HEVB2b, partition 0, lastEvent 180911 -->   
 Additional Information:  
 
 Attention !!!. FERML_TM2_HEVB2B_EVENTS_PENDING_TIMEOUT !!!  
 - AUTO HRR will be issued.   
 - If problem persists, you may have to clean up the Event Builder.  
[MLE] 
 
 Hrred. 
 
<!-- Author: Sergo -->


<!-- Date: Sun Oct 9 15:17:48.639 2005 -->
Icicle's heartbeat alarms. I restart the program. 
<!-- Author: Freeman -->



<!-- Date: Sun Oct 9 15:55:23.870 2005 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>Physics able</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L1Accepts</TH>
<TH>L2Accepts</TH>
<TH>L3Accepts</TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205324 target=_blank>205324</A> x3220C</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_3_07 [2,713,594]</FONT></TD>
<TD ALIGN=CENTER>06:22:06</TD>
<TD ALIGN=CENTER>09:13:00</TD>
<TD ALIGN=CENTER>02:41:29</TD>
<TD ALIGN=RIGHT>    150,307,839</TD>
<TD ALIGN=RIGHT>     1,446,508</TD>
<TD ALIGN=RIGHT>     520,366</TD>
<TD ALIGN=RIGHT>  206.906</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=205325 target=_blank>205325</A> x3220D</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_3_07 [2,713,594]</FONT></TD>
<TD ALIGN=CENTER>09:16:34</TD>
<TD ALIGN=CENTER>10:52:11</TD>
<TD ALIGN=CENTER>01:26:19</TD>
<TD ALIGN=RIGHT>     71,028,033</TD>
<TD ALIGN=RIGHT>       706,089</TD>
<TD ALIGN=RIGHT>     257,414</TD>
<TD ALIGN=RIGHT>  101.466</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>15:55:00</TD>
<TD ALIGN=CENTER>04:07:48</TD>
<TD ALIGN=RIGHT>     221,335,872</TD>
<TD ALIGN=RIGHT>      2,152,597</TD>
<TD ALIGN=RIGHT>       777,780</TD>
<TD ALIGN=RIGHT>  308.372</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Sun Oct 9 15:58:48.689 2005 -->
<font color=red>Shift Summary: </font><pre><pre> 
Smooth data taking until Beam was dumped at 11 
Access for Silicon and TDC replacement. TDC replacement successful, Silicon not (but no loss either) 
 
Did calibrations. There is some question as to whether the Calorimeter QIE calibrations that are out of range should be included in the "known errors" log. 
 
Taking cosmics, waiting for beam 
 
Pending Issues 
-------------- 
o TOF Inhibit did not inhibit (owl shift 10-9) 
o Monday: DB Wants to do disk arrangement 
o Run 205070 W->Mu Nu 
o L3 Crashes 
 
Runs: 205324, 205325 
 
 
Begin shift or beam at 08:00:19 
End shift or beam at 10:52:25 
Delivered luminosity: 208.10 
Acquired luminosity: 189.39 
Totals 
Date:	2005.10.09 
Shift:	day 
Delivered luminosity:	208.1 nb-1 
Acquired luminosity:	189.4 nb-1 
Efficiency:	91.0 
</pre>
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   205324-5
Delivered Luminosity   209  
Acquired Luminosity    189  
Efficiency             91
   
</td></tr></table></pre>
<!-- Author: TJL  -->