<!-- Title: Day shift. Wed Feb  8, 2006 -->
<!-- SciCo: DongHee Kim -->
<!-- DAQAce: Ai Nagano -->
<!-- MonAce: Jennifer Gimmell -->
<!-- CO: Charles Plager -->
<!-- OpManager: Greg Feild -->
<!-- Notes: -->
<pre>Day shift start in the middle of run 211397</pre>


<!-- Date: Wed Feb 8 08:28:28.300 2006 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2006&action=view&page=-1995&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-1995  width='300' align='top'></a></td><td>A recurring PSM Alarm; I catch Steve Hahn in the control room, and ask him if he knows about this problem: 
<p> 
<b>PSM Alarm for B0 crate SVT_08 2RR32C_2</b>: looks like a voltage problem on the +3.3 V Digital Channel 3. Steve then shows me how to plot the voltage summary, and here's what we produce. 
</p> 
<p> 
Looks like some voltage spikes (this plot is over about a day) every several hours. Steve is headed to talk to Dervin to ask him about this crate. Until then, it seems that after a few seconds, this crate stops alarming (this is what was referred to as "recovering itself" <a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2006&action=view&page=115&frame=2&anchor=01:58:32.862'> 
last shift</a>. Hopefully more to come... 
</p></td></tr></table>
<!-- Author: Jennifer :-) -->
<!-- Comment: Wed Feb  8 08;58;36 comment by...Jennifer :-) -->
One interesting thing to note... this was the crate that used to be on the whiteboard that needed to be masked. Now we are running with additional SVT crates (upgraded from 6→9), now this crate is included. Note the change on the board; this crate is now <b>excluded</b> from being masked.
<!-- Date: Wed Feb 8 08:45:59.158 2006 -->
flying wire
<!-- Author: ai :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211397' target=_top>211397</a>) -->
<!-- Comment: Wed Feb  8 09;45;52 comment by...ai -->
again
<!-- Comment: Wed Feb  8 10;46;05 comment by...ai -->
again
<!-- Date: Wed Feb 8 08:53:07.130 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211397">211397</a>
muons: <tt>TrigMon has seen over 17k events (out of over 200k) and the CMP canvas still says low statistics.  I think there's something not quite kosher here...</tt>
<!-- Author: Consumer Operator( Charles Plager) -->
<!-- Comment: Wed Feb  8 09;24;17 comment by...phil -->
you would be wrong
<!-- Comment: Wed Feb  8 15;12;38 comment by...Charles Plager -->
Wrong about what?  17k events isn't enough to get enough statistics to show anything?  I don't believe that.
<!-- Comment: Wed Feb  8 22;06;00 comment by...phil -->
now which of us do you think has looked at that plot a zillion times in the past
<!-- Date: Wed Feb 8 08:53:08.672 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211397">211397</a>
trigger: <tt>TrigMon SVT canvas 2.3.4 is red on the bottom but does <b>not</b> meet the standards for paging SVT.</tt>
<!-- Author: Consumer Operator( Charles Plager) -->




<!-- Date: Wed Feb 8 08:58:51.922 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-1997&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-1997' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-1998&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-1998' width='300'></a></center>
<!-- Author: Charles Plager -->
<!-- Comment: Wed Feb  8 08;59;57 comment by...Charles Plager -->
I believe the YMon Calorimeter references needs to be updated.  I talked to Steve Hahn and he agrees (right, Steve?).


<!-- Comment: Wed Feb  8 12;22;32 comment by...Jiyeon -->
I'll talk with Larry about the good run number for reference.

<!-- Comment: Wed Feb  8 12;37;55 comment by...Larry   -->
Sure would be nice if the one hot channel did not blow the scale! 
<!-- Comment: Wed Feb  8 15;13;46 comment by...Charles Plager -->
You could set the scale by hand so that anything that is yellow or orange is a problem and red just means REALLY hot.  That way, one bad channel woudln't blow the scale.
<!-- Date: Wed Feb 8 09:01:59.045 2006 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2006&action=view&page=-1999&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-1999  width='300' align='top'></a></td><td>BSU East Real Barrel (lower right-hand plot) seems to have a single dead channel (next to a half-dead channel) near 80 that is not in the reference.  Instructions are to page if there are 4 or more, so it doesn't meet that criteria.  If this is a known problem, the reference plot should be updated.</td></tr></table>
<!-- Author: Charles Plager -->
<!-- Comment: Wed Feb  8 09;25;08 comment by...Phil -->
that does seem to be new - at least in the last few days
<!-- Date: Wed Feb 8 09:14:33.902 2006 -->
Scaler Monitor error for high dead time. 
L2 total DT 68%
<!-- Author: ai :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211397' target=_top>211397</a>) -->
<!-- Comment: Wed Feb  8 09;18;48 comment by...ai -->
HRRed but it happened again.
<!-- Comment: Wed Feb  8 09;21;21 comment by...ai -->
I hit HRR again. It seems to be OK now.
<!-- Date: Wed Feb 8 09:14:48.997 2006 -->
SVX B2W9L1 has tripped... yet again. See the <a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=287&frame=2&anchor=08:45:57'> 
Silicon e-log</a> where we can see that the currents are rising high above the software protection limit, thus giving us the trips. Future plans for this wedge include a new power supply. Ankush is increasing the software protection limit, and hopefuilly we won't have this alarm again.
<!-- Author: Jennifer :-) :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211397' target=_top>211397</a>) -->







<!-- Date: Wed Feb 8 10:13:45.377 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2000&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2000' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2001&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2001' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2002&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2002' width='300'></a></center>
<!-- Author: Hourly Plots! 08:00 - 09:00 Jennifer :-) -->


<!-- Date: Wed Feb 8 10:16:14.130 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2003&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2003' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2004&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2004' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2005&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2005' width='300'></a></center>
<!-- Author: Hourly Plots! 09:00 - 10:00 Jennifer :-) -->


<!-- Date: Wed Feb 8 10:18:00.231 2006 -->
busy TO
<!-- Author: ai :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211397' target=_top>211397</a>) -->


<!-- Date: Wed Feb 8 10:20:17.755 2006 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2006&action=view&page=-2006&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-2006  width='300' align='top'></a></td><td>TrigMon 2.3.4: SVT Table is red and now has an error rate greater than the limit on the reference page.  Will page SVT.</td></tr></table>
<!-- Author: Charles Plager -->
<!-- Comment: Wed Feb  8 10;21;10 comment by...ai -->
HRRed.


<!-- Date: Wed Feb 8 10:28:21.651 2006 -->
<pre>SVT people arrived after paged.  </pre>
<!-- Author: DongHee Kim :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211397' target=_top>211397</a>) -->



<!-- Date: Wed Feb 8 11:12:18.530 2006 -->
Plug HV trip for crate 3 (west plug). Note that there was also a HV trip for crate 1 in the <a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2006&action=view&page=115&frame=2&anchor=04:38:11.549'> 
wee hours of the morning</a>. Ai ran upstairs to the 3rd floor and restarted the HV monitoring, per ACE instructions. I restarted the run once HV Monitor gave me the green go.
<!-- Author: Jennifer :-) :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211397' target=_top>211397</a>) -->
<!-- Comment: Wed Feb  8 11;44;17 comment by...Jennifer :-) -->
<b>Since we've had two plug HV trips so close to one another, if an expert is reading: is this something to look into?</b>
<!-- Date: Wed Feb 8 11:12:42.365 2006 -->
RXPT error for b0svx06. SRC fatal error. 
Busy TO three times in a raw
<!-- Author: ai :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211397' target=_top>211397</a>) -->

<!-- Date: Wed Feb 8 11:16:12.836 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211397">211397</a>
 Terminated at 2006.02.08 11:16:02
<!-- Author: RunControl -->


<!-- Date: Wed Feb 8 11:17:05.271 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211397">211397</a>
TERMINATE: fix evb
<!-- Author: Ai x2080 -->
<!-- Comment: Wed Feb  8 11;41;50 comment by...Conor and Khaldoun -->
evb scpu 25 had crashed. We needed to end the run and cleanup evb. Things are fine now.
<!-- Comment: Wed Feb  8 11;57;23 comment by...markus -->
<pre>SCPU in EVB crate 25 failed to do a DMA transfer and as a consequence restarted itself. Si expert claimed they were not working on the crate which caused this problem in the past. Very rare failure ...</pre>
<!-- Date: Wed Feb 8 11:20:59.671 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2007&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2007' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2008&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2008' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2009&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2009' width='300'></a></center>
<!-- Author: Hourly Plots! 10:00 - 11:00 Jennifer :-) -->

<!-- Date: Wed Feb 8 11:22:35.885 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211398">211398</a>
 Activated at 2006.02.08 11:21:51
<!-- Author: RunControl -->


<!-- Date: Wed Feb 8 11:22:36.394 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211398">211398</a>
ACTIVATE: RunSet: AAA_CURRENT  Trigger: PHYSICS_4_00 [1,781,616] 
<!-- Author: Ai x2080 -->




<!-- Date: Wed Feb 8 11:23:46.787 2006 -->
L2 decision TO
<!-- Author: ai :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211398' target=_top>211398</a>) -->



<!-- Date: Wed Feb 8 11:42:32.756 2006 -->
A *very* quick TOF HV trip, coupled with a trigger inhibit. By the time I clicked on TOF on the global alarm GUI, the trip receovered itself. Now, we're happily running again.
<!-- Author: Jennifer :-) :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211398' target=_top>211398</a>) -->



<!-- Date: Wed Feb 8 11:46:33.170 2006 -->
Consumer error.Stuck cellid S/B4/W6/L3/CO-5.HRRed.
<!-- Author: ai :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211398' target=_top>211398</a>) -->


<!-- Date: Wed Feb 8 11:46:47.465 2006 -->
flying wire
<!-- Author: ai :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211398' target=_top>211398</a>) -->


<!-- Date: Wed Feb 8 12:03:56.847 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211398">211398</a>
trigger: <tt>TrigMon 2.3.4 SVT Table is red again and the rate is again below the 0.005 limit.</tt>
<!-- Author: Consumer Operator( Charles Plager) -->


<!-- Date: Wed Feb 8 12:03:57.968 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211398">211398</a>
monitors: <tt>First three silimon canvases have a lot of red in them.  Low stats? Note: No problem with silicon.</tt>
<!-- Author: Consumer Operator( Charles Plager) -->



<!-- Date: Wed Feb 8 12:05:27.608 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2010&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2010' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2011&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2011' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2012&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2012' width='300'></a></center>
<!-- Author: Charles Plager -->
<!-- Comment: Wed Feb  8 12;05;50 comment by...Charles Plager -->
The red silimon canvases.

<!-- Date: Wed Feb 8 12:13:16.793 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211398">211398</a>
 Terminated at 2006.02.08 12:13:02
<!-- Author: RunControl -->


<!-- Date: Wed Feb 8 12:13:39.830 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211398">211398</a>
TERMINATE: svt setting
<!-- Author: Ai x2080 -->


<!-- Date: Wed Feb 8 12:18:25.468 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211399">211399</a>
 Activated at 2006.02.08 12:18:20
<!-- Author: RunControl -->


<!-- Date: Wed Feb 8 12:18:46.038 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211399">211399</a>
ACTIVATE: RunSet: AAA_CURRENT  Trigger: PHYSICS_4_00 [1,781,616] 
<!-- Author: Ai x2080 -->



<!-- Date: Wed Feb 8 12:19:43.641 2006 -->
Consumer error CER_SVXMON_HALT_RECOVER_RUN_ERROR
<!-- Author: ai :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211399' target=_top>211399</a>) -->



<!-- Date: Wed Feb 8 12:30:40.020 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2013&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2013' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2014&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2014' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2015&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2015' width='300'></a></center>
<!-- Author: Hourly Plots! 11:00 - 12:00 Jennifer :-) -->



<!-- Date: Wed Feb 8 12:38:18.211 2006 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2006&action=view&page=-2016&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-2016  width='300' align='top'></a></td><td>A little curious:<p> 
The error rates are <b>negative</b> for the last 5 minutes. Hmmmmmm....</td></tr></table>
<!-- Author: Charles Plager -->
<!-- Comment: Wed Feb  8 14;49;44 comment by...badgett -->
Thanks, Charles.  I've put in a protection against that 
happening, whenever the next rebuild is;  generally it will clear up anyway after a few 
minutes.   
<!-- Date: Wed Feb 8 12:39:52.198 2006 -->
<pre>We got paged by the CO for failures over threshold in TrigMon. 
As soon as we arived the DAQ ACE issued the HRR, following instructions from last night problem. 
Looking at data in run 211397 produced offline we observe the Lost of Synctronization. Thanks to the HRR the range of this lost of sync is negligible. 
We have still to unserstand the source of this problem.  
In the meanwhile we enable the boards in critical points of the SVT chain to issue the CDF_ERROR in case of lost sync. This would force  HRR and at least avoid that we take bad data.</pre>
<!-- Author: Taichi :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211397' target=_top>211397</a>) -->


<!-- Date: Wed Feb 8 12:43:04.354 2006 -->
front end crate error from SVT_PULSER_08 SVT_FANOUT_07. 
RXPT error for b0svt07. 
<!-- Author: ai :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211399' target=_top>211399</a>) -->


<!-- Date: Wed Feb 8 12:45:53.329 2006 -->
flying wire
<!-- Author: ai :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211399' target=_top>211399</a>) -->
<!-- Comment: Wed Feb  8 13;45;58 comment by...ai -->
again
<!-- Comment: Wed Feb  8 14;46;05 comment by...ai -->
again
<!-- Comment: Wed Feb  8 15;49;14 comment by...ai -->
again
<!-- Date: Wed Feb 8 13:11:06.598 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211399">211399</a>
monitors: <tt>ObjectMon Muon Monitor has a red plot (CMP phi stub).  It is low statistics, but since it's red, I'm making this entry.  Plot to follow.</tt>
<!-- Author: Consumer Operator( Charles Plager) -->



<!-- Date: Wed Feb 8 13:11:25.260 2006 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2006&action=view&page=-2017&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-2017  width='300' align='top'></a></td><td>ObjectMon and it's red plot</td></tr></table>
<!-- Author: Charles Plager -->
<!-- Comment: Wed Feb  8 13;12;37 comment by...Charles Plager -->
As per instructions, ObjectMon experts were notified by email.


<!-- Comment: Fri Feb 10 17;48;47 comment by...Vadim -->
looks more like a statistics effect here
<!-- Date: Wed Feb 8 13:18:33.565 2006 -->
frontend crate error from SVT_FILTER_06. RXPT error for b0svt06. 
<!-- Author: ai :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211399' target=_top>211399</a>) -->


<!-- Date: Wed Feb 8 13:23:12.977 2006 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2006&action=view&page=-2018&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-2018  width='300' align='top'></a></td><td>AGIGI2 seems to be fluctuating a bit; I decided to look at it over the past 12 hours of the store, and it seems like it's only rising slowly. Now, I realize that in my fast-time plotter, I'm only looking at a 0 to 3 E09 scale, so the small fluctuations look gigantic, but I would like to note that there has been a slight change in the losses.</td></tr></table>
<!-- Author: Jennifer :-) -->


<!-- Date: Wed Feb 8 13:25:52.334 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2019&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2019' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2020&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2020' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2021&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2021' width='300'></a></center>
<!-- Author: Hourly Plots! 12:00 - 13:00 Jennifer :-) -->



<!-- Date: Wed Feb 8 14:17:02.784 2006 -->
I got a pop up message on Run Control. It said Trigger Inhibit is set from TOF HV but iFix did not alarmed. 
<!-- Author: ai :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211399' target=_top>211399</a>) -->
<!-- Comment: Wed Feb  8 15;41;23 comment by...Yuri -->
happened before <a href=
'http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2006&action=view&page=111&frame=2&anchor=21:04:29.330'>
user link text </a>
<!-- Date: Wed Feb 8 14:30:21.689 2006 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2006&action=view&page=-2022&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-2022  width='300' align='top'></a></td><td>O.k.  The plot that was red is now green.  Of course, the plot next to it that was green is now red.</td></tr></table>
<!-- Author: Charles Plager -->
<!-- Comment: Fri Feb 10 17;52;05 comment by...Vadim -->
this distribution is very trigger-sensitive
<!-- Date: Wed Feb 8 14:41:07.791 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2023&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2023' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2024&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2024' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2025&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2025' width='300'></a></center>
<!-- Author: Hourly Plots! 13:00 - 14:00 Jennifer :-) -->

<!-- Date: Wed Feb 8 14:43:11.839 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211399">211399</A>
cal: <tt>WHAPMT still has some holes, but it is probably low statistics.</tt>
<!-- Author: Consumer Operator( Charles Plager) -->


<!-- Date: Wed Feb 8 14:43:12.550 2006 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211399">211399</A>
monitors: <tt>Different plot now red in ObjectMon (although fit looks good by eye). &nbsp;Plot in elog.</tt>
<!-- Author: Consumer Operator( Charles Plager) -->




<!-- Date: Wed Feb 8 14:47:12.403 2006 -->
Quick RPS alarm. Once I clicked on the global alarm page, it was cleared. 
<!-- Author: Jennifer :-) :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211399' target=_top>211399</a>) -->
<!-- Comment: Wed Feb  8 15;19;24 comment by...convery -->
Due to flying wire.  Sadly, there is no detailed information on the global alarms page available for any of the systems controlled by clc1.  One would have to look at the monitoring on the clc1 pc itself.
<!-- Comment: Wed Feb  8 15;20;59 comment by...Jennifer :-) -->
So, one way that I believe I can assume that this quick RPS alarm is nothing to worry about is the fact that it was <b>only</b> the RPS. Looking at the ACE instructions for an RPS HV trip, all 5 HV controls: CLC, BSC, RPS, MNP, CPR are all controlled by the same monitor. If this were a loss in HV monitoring (one which required a PC reboot or a restart of the HV monitoring) I would have gotten alarms / heartbeat alarms for all 5.
<!-- Date: Wed Feb 8 15:03:52.846 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2026&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2026' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2027&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2027' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-2028&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-2028' width='300'></a></center>
<!-- Author: PUNCTUAL Hourly Plots!! 14:00 - 15:00 Jennifer :-) -->


<!-- Date: Wed Feb 8 15:11:47.360 2006 -->
For those of you who have always wanted the consumers slideshows in your office, but didn't want to mess with <tt>HistoDisplay</tt>, you're now in luck.  From the makers of WebTalks and Consumer Slides on the Web, we introduce <a href="http://www-cdfonline.fnal.gov/cgi/internal/mon/cdfdoc/ConsumerGifs/rotateCanvases.pl">RotatingCanvases</a>.  <p>You can pick which consumers to look at, the refresh speed, and whether you want to go forwards or backwards.  Unless you enter a run number, you will automatically be looking at the run the C.O. is currently checking.  There is a cron job automatically updating the plots, so you'll be sure to be looking at the latest.  Try it out and let me know what  you think.
<!-- Author: The maker of WebTalks :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211399' target=_top>211399</a>) -->







<!-- Date: Wed Feb 8 15:55:50.496 2006 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>Physics able</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L1Accepts</TH>
<TH>L2Accepts</TH>
<TH>L3Accepts</TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211397 target=_blank>211397</A> x339C5</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_4_00 [1,781,616]</FONT></TD>
<TD ALIGN=CENTER>05:20:38</TD>
<TD ALIGN=CENTER>11:16:02</TD>
<TD ALIGN=CENTER>05:19:18</TD>
<TD ALIGN=RIGHT>    428,011,799</TD>
<TD ALIGN=RIGHT>     6,299,974</TD>
<TD ALIGN=RIGHT>   1,538,840</TD>
<TD ALIGN=RIGHT>  586.816</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211398 target=_blank>211398</A> x339C6</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_4_00 [1,781,616]</FONT></TD>
<TD ALIGN=CENTER>11:21:51</TD>
<TD ALIGN=CENTER>12:13:02</TD>
<TD ALIGN=CENTER>00:48:39</TD>
<TD ALIGN=RIGHT>     58,522,630</TD>
<TD ALIGN=RIGHT>       835,433</TD>
<TD ALIGN=RIGHT>     211,050</TD>
<TD ALIGN=RIGHT>   74.413</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=211399 target=_blank>211399</A> x339C7</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_4_00 [1,781,616]</FONT></TD>
<TD ALIGN=CENTER>12:18:20</TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>03:28:24</TD>
<TD ALIGN=RIGHT>    241,048,917</TD>
<TD ALIGN=RIGHT>     3,592,407</TD>
<TD ALIGN=RIGHT>     888,995</TD>
<TD ALIGN=RIGHT>  281.407</TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER></TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>15:55:01</TD>
<TD ALIGN=CENTER>09:36:22</TD>
<TD ALIGN=RIGHT>     727,583,346</TD>
<TD ALIGN=RIGHT>     10,727,814</TD>
<TD ALIGN=RIGHT>     2,638,885</TD>
<TD ALIGN=RIGHT>  942.637</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Wed Feb 8 15:58:19.649 2006 -->
<font color=red>Shift Summary: </font><pre>In overall, quiet day but take two new runs. 
 
08:00 Start with the run 211397 
 
 
11:22  new run 211398 start since scpu 25 had crashed. We needed to end the run and cleanup evb. 
 
12:18  211399 with SVT setting 
 
 
NOTE: Level2 Pulsar paged but never answered. later on, turned out to be minor proplem. but answer back to CCR important though.</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   211397-211399
Delivered Luminosity   719.3  
Acquired Luminosity    643.7  
Efficiency             89.5
   
</td></tr></table></pre>
<!-- Author: DongHee Kim -->