<!-- Title: Day shift. Thu Nov 14, 2002 -->
<!-- SciCo: Rick Snider -->
<!-- DAQAce: Lin Zhang -->
<!-- MonAce: JJ Schmidt -->
<!-- CO: Guram Chlachidze -->
<!-- OpManager: Matt Jones -->
<!-- Notes: -->
<pre>Store 1961 just dropped. Stack at 172E10. 
BSU/TSU red. 
Plan for shift.... 
* calibrations 
* Steve Tether event builder work 
* shot setup (run with Silicon if losses acceptable)</pre>

<!-- Date: Thu Nov 14 08:36:09 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154131">154131</A>
ACTIVATE: CAL QIE calibration. No beam during dry squeeze between stores.
<!-- processed by 64.233.196.88 -->
<!-- Author: Lin 2080 -->


<!-- Date: Thu Nov 14 08:37:23 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154131">154131</A>
TERMINATE: End CAL QIE calibration.
<!-- processed by 131.225.45.171 -->
<!-- Author: Lin 2080 -->



<!-- Date: Thu Nov 14 08:37:44 2002 -->
VxMon reports that FCAL crate is unreadable.
<!-- Author: Rick Snider -->

<!-- Date: Thu Nov 14 08:44:01 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154132">154132</A>
ACTIVATE: COT calibration. No beam. Partition 0.
<!-- processed by 131.225.235.234 -->
<!-- Author: Lin 2080 -->
<!-- Comment: Thu Nov 14 09;10;43 comment by...Guram -->
Checked, OK! - no new COMPLETE run written, no new calibration run written, numBad channels = 402. <br> 
<font color=red>Caution: num drifted good channels, crate 7 is 35
</font>
<!-- Date: Thu Nov 14 08:44:01 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154132">154132</A>
TERMINATE: Finish COT calibration.
<!-- processed by 131.225.235.234 -->
<!-- Author: Lin 2080 -->


<!-- Date: Thu Nov 14 08:48:44 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154133">154133</A>
ACTIVATE: TOF TAC calibration No beam. Partition 0.
<!-- processed by 131.225.235.234 -->
<!-- Author: Lin 2080 -->


<!-- Date: Thu Nov 14 08:50:38 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154133">154133</A>
TERMINATE: Finish TOF TAC calibration.
<!-- processed by 131.225.233.214 -->
<!-- Author: Lin 2080 -->


<!-- Date: Thu Nov 14 08:52:45 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154134">154134</A>
ACTIVATE: TOF QIE calibration. No beam. Patition 8.
<!-- processed by 131.225.233.51 -->
<!-- Author: Lin 2080 -->


<!-- Date: Thu Nov 14 08:53:36 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154134">154134</A>
TERMINATE: Finish TOF QIE calibration.
<!-- processed by 131.225.232.167 -->
<!-- Author: Lin 2080 -->



<!-- Date: Thu Nov 14 08:58:44 2002 -->
<pre>COT audible alarm in IFIX. We can find not evidence of 
any problem. I talked to Morris this morning and he does 
not understand. I will send email to JC Yun and Steve Hahn 
and ask them to investigate.</pre>
<!-- Author: JJ -->

<!-- Date: Thu Nov 14 09:03:43 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154135">154135</A>
ACTIVATE: Muon Calibration. TevKE: 149GeV, TevPR: 43.1E09. Partition 0.
<!-- processed by 131.225.127.7 -->
<!-- Author: Lin 2080 -->


<!-- Date: Thu Nov 14 09:03:43 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154135">154135</A>
TERMINATE: Finish MUON calibration.
<!-- processed by 131.225.127.7 -->
<!-- Author: Lin 2080 -->



<!-- Date: Thu Nov 14 09:03:43 2002 -->
<pre>COT heartbeat alarm in IFIX. Clears itself in 30 seconds.</pre>
<!-- Author: JJ -->


<!-- Date: Thu Nov 14 09:12:43 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154136">154136</A>
ACTIVATE: CAL LED calibration during proton injection. Partion 0.
<!-- processed by 131.225.127.7 -->
<!-- Author: Lin 2080 -->
<!-- Comment: Thu Nov 14 09;30;54 comment by...Guram -->
Checked, OK: No new bad channels!
<!-- Date: Thu Nov 14 09:12:44 2002 -->
<pre>Tests being done on PowerSupplyMonitoring. We should ignore 
alarms until work is done.</pre>
<!-- Author: JJ -->
<!-- Comment: Thu Nov 14 09;21;31 comment by...jj -->
<font size=+1>PAGE ATTENTION TO PSM ALARMS AGAIN.</font>
<!-- Date: Thu Nov 14 09:12:49 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154136">154136</A>
TERMINATE: Finish CAL LED calibration.
<!-- processed by 131.225.236.161 -->
<!-- Author: Lin 2080 -->





<!-- Date: Thu Nov 14 09:37:29 2002 -->
<pre>CEM, CHA, WHA boxes on global alarms are grey. Lin and Kaori 
have turned off monitoring while turning on Xenon HV for 
calibrations.</pre>
<!-- Author: JJ -->
<!-- Comment: Thu Nov 14 10;39;33 comment by...JJ -->
<pre>These systems are back in active monitor state.</pre>
<!-- Date: Thu Nov 14 09:44:34 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154137">154137</A>
ACTIVATE: CAL Xenon calibration. Proton Injection. Partition 0. 
<!-- processed by 131.225.127.7 -->
<!-- Author: Lin 2080 -->


<!-- Date: Thu Nov 14 09:45:40 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154137">154137</A>
TERMINATE: Finish CAL Xenon calibration.
<!-- processed by 128.220.233.172 -->
<!-- Author: Lin 2080 -->
<!-- Comment: Thu Nov 14 09;51;38 comment by...Lin/JJ -->
<pre>Run seemed to be stuck going into DONE state.
ABORT and try again.</pre>

<!-- Date: Thu Nov 14 09:53:25 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154138">154138</A>
ACTIVATE: Try CAL Xenon calibration again.
<!-- processed by 131.225.233.93 -->
<!-- Author: Lin 2080 -->
<!-- Comment: Thu Nov 14 10;34;19 comment by...Guram -->
CEM-Xen Calibration Checked: New horizontal white bands in the normalised PMT plots,
one wedge with only zero marker in PIN diode plots. <br>
According to the CO instructions <font color=blue> expert was e-mailed (kovalev@hep.upenn.edu</font>)
<!-- Comment: Thu Nov 14 10;37;23 comment by...Lin -->
The wedges 17 at eastsouth keep red during turning Xenon HV on. 
<!-- Date: Thu Nov 14 09:55:52 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154138">154138</A>
TERMINATE: Really finish CAL Xenon calibration this time.
<!-- processed by 131.225.242.188 -->
<!-- Author: Lin 2080 -->
<!-- Comment: Thu Nov 14 10;23;48 comment by...Lin -->
After this calibration finished, I turned the Xenon HV off. Wedge 17 in the eastsouth of CEM still kept red while the others turned black when turned off "All Flashers".
<!-- Comment: Thu Nov 14 10;39;11 comment by...Lin -->
It might take up to half an houre for the CEM CHA WHA HV bars to reappear on the HV after Xenon HV was turned off. Now they all are there.
<!-- Date: Thu Nov 14 10:07:14 2002 -->
<center><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-10371&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-10371' width='300'></a><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-10372&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-10372' width='300'></a><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-10373&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-10373' width='300'></a></center>
<!-- Author: JJ -->


<!-- Date: Thu Nov 14 10:14:49 2002 -->
Loading final protons.
<!-- Author: Rick Snider -->

<!-- Date: Thu Nov 14 10:19:58 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154139">154139</A>
 Activated at 2002.11.14 10:18:24
<!-- processed by 131.225.45.171 -->
<!-- Author: RunControl -->


<!-- Date: Thu Nov 14 10:21:32 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154139">154139</A>
ACTIVATE: New L3 EXEtest using cosmics[8,200,337] TriggerTable.
<!-- processed by 131.225.82.102 -->
<!-- Author: Lin 2080 -->



<!-- Date: Thu Nov 14 10:25:33 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154139">154139</A>
 Terminated at 2002.11.14 10:24:17
<!-- processed by 131.225.235.50 -->
<!-- Author: RunControl -->


<!-- Date: Thu Nov 14 10:25:33 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154139">154139</A>
 Terminated at 2002.11.14 10:24:17
<!-- processed by 131.225.235.50 -->
<!-- Author: RunControl -->


<!-- Date: Thu Nov 14 10:26:11 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154139">154139</A>
TERMINATE: Abort it. Seems sth. is stuck going to active.
<!-- processed by 131.225.235.50 -->
<!-- Author: Lin 2080 -->
<!-- Comment: Thu Nov 14 10;46;46 comment by...Lin -->
L3 error. Followed the insturction of "clear up L3 farms". 
<!-- Date: Thu Nov 14 10:29:39 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154140">154140</A>
TERMINATE: Try again the test.
<!-- processed by 131.225.235.50 -->
<!-- Author: Lin 2080 -->



<!-- Date: Thu Nov 14 10:36:25 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154141">154141</A>
 Activated at 2002.11.14 10:35:00
<!-- processed by 131.225.236.161 -->
<!-- Author: RunControl -->


<!-- Date: Thu Nov 14 10:37:23 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154141">154141</A>
ACTIVATE: really begin taking date for this test.
<!-- processed by 131.225.236.169 -->
<!-- Author: Lin 2080 -->






<!-- Date: Thu Nov 14 10:40:20 2002 -->
Final pbar's going in.
<!-- Author: Rick Snider -->
<!-- Comment: Thu Nov 14 10;42;46 comment by...Rick Snider -->
Paged silicon
<!-- Date: Thu Nov 14 10:40:47 2002 -->
<TABLE bgcolor=#cdc9a5> 
<TR><TH>Date</th><th>Time</th><th>BLM</th><th colspan=2>Dose</th></tr> 
<tr><td>2002.11.14</td><td>10:35:47</td><td>W Inner BLM</td><td>0.63</td><td>RADS</td></tr> 
Anti-protons being loaded.  
 
<tr><td>2002.11.14</td><td>10:35:47</td><td>W Outer BLM</td><td>0.06</td><td>RADS</td></tr> 
<tr><td>2002.11.14</td><td>10:35:47</td><td>E Inner BLM</td><td>1.60</td><td>RADS</td></tr> 
<tr><td>2002.11.14</td><td>10:35:47</td><td>E Outer BLM</td><td>2.11</td><td>RADS</td></tr> 
</TABLE> 
<strong>Integrated dosage</strong> 
  
<!-- Author: JJ -->
<!-- Comment: Thu Nov 14 11;07;09 comment by...JJ -->
<TABLE bgcolor=#cdc9a5>
<TR><TH>Date</th><th>Time</th><th>BLM</th><th colspan=2>Dose</th></tr>
<tr><td>2002.11.14</td><td>11:02:27</td><td>W Inner BLM</td><td>1.81</td><td>RADS</td></tr>
TEV QUENCH . 
<tr><td>2002.11.14</td><td>11:02:27</td><td>W Outer BLM</td><td>0.55</td><td>RADS</td></tr>
<tr><td>2002.11.14</td><td>11:02:27</td><td>E Inner BLM</td><td>2.88</td><td>RADS</td></tr>
<tr><td>2002.11.14</td><td>11:02:27</td><td>E Outer BLM</td><td>3.44</td><td>RADS</td></tr>
</TABLE>
<strong>Integrated dosage</strong>


<!-- Date: Thu Nov 14 10:43:08 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154141">154141</A>
 Terminated at 2002.11.14 10:41:25
<!-- processed by 131.225.127.7 -->
<!-- Author: RunControl -->


<!-- Date: Thu Nov 14 10:43:08 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154141">154141</A>
TERMINATE: End the test run.
<!-- processed by 131.225.127.7 -->
<!-- Author: Lin 2080 -->



<!-- Date: Thu Nov 14 10:43:09 2002 -->
<pre>Technician was working on HVAC unit in third floor computing 
room. Unit is back on-line (and temperature in that room 
is no longer rising!).  Rob Roser reports that unit needs  
two new humidity sensors but this replacement will not be 
done until next extended access or shutdown.</pre>
<!-- Author: JJ -->


<!-- Date: Thu Nov 14 10:54:30 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154142">154142</A>
ACTIVATE: Running AAA_SHOTSETUP run.
<!-- processed by 131.225.236.229 -->
<!-- Author: Lin 2080 -->



<!-- Date: Thu Nov 14 11:05:34 2002 -->
Lost the shot. B0 quad quench, possibly among others.	
<!-- Author: Rick Snider -->
<!-- Comment: Thu Nov 14 11;23;25 comment by...Rick Snider -->
Paged people with outstanding access requests:  BMU/TSU to address CCU5 problem. Need less than an hour.
<!-- Date: Thu Nov 14 11:07:39 2002 -->
Icon MEAN(C:LOSTP), MEAN(C:LOSTPB), RMS(C:LOSTP), RMS(C:LOSTPb), C:B0PAGC on TevMon keep red. But it's time for proton and antiproton ramping and scraping, we don't need to worry about them. Just wait for scraping finishs.
<!-- Author: Lin :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154142' target=_top>154142</a>) -->
<!-- Comment: Thu Nov 14 11;08;43 comment by...Lin -->
Sorry, shot has been lost.



<!-- Date: Thu Nov 14 11:24:17 2002 -->
MCR is setting up for a CDF access.  They will call us when they 
are ready.  Oleg was paged, Camille is comming in to replace CCU05.
<!-- Author: M Jones -->
<!-- Comment: Thu Nov 14 11;25;37 comment by...M Jones -->
They are ready now.


<!-- Date: Thu Nov 14 11:26:01 2002 -->
MCR grants immediate access.
<!-- Author: Rick Snider -->

<!-- Date: Thu Nov 14 11:26:24 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154142">154142</A>
TERMINATE: End run for CLC QIE calibration.
<!-- processed by 131.225.236.169 -->
<!-- Author: Lin 2080 -->


<!-- Date: Thu Nov 14 11:30:35 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154143">154143</A>
ACTIVATE: CLC QIE calibration. No beam, partition 0.
<!-- processed by 131.225.236.169 -->
<!-- Author: Lin 2080 -->


<!-- Date: Thu Nov 14 11:31:41 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154143">154143</A>
TERMINATE: Finish CLC QIE calibration.
<!-- processed by 131.225.236.229 -->
<!-- Author: Lin 2080 -->



<!-- Date: Thu Nov 14 11:43:54 2002 -->
Under guidance of expert, we did a <tt>reset_crate b0fcal00</tt><br> 
to try and clear up problem intermittent red state in <br> 
VXWORKS System Monitor. Will watch and see if we have success.
<!-- Author: JJ -->


<!-- Date: Thu Nov 14 12:13:53 2002 -->
Turned CES/CPR/CCR from STANDBY to OFF to do 
SMQIE calibration.
<!-- Author: JJ -->

<!-- Date: Thu Nov 14 12:24:11 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154144">154144</A>
ACTIVATE: SMX QIE calibration. No beam. Partition 0.
<!-- processed by 131.225.236.31 -->
<!-- Author: Lin 2080 -->


<!-- Date: Thu Nov 14 12:25:02 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154144">154144</A>
TERMINATE: Finish SMX QIE calibration.
<!-- processed by 131.225.232.173 -->
<!-- Author: Lin 2080 -->



<!-- Date: Thu Nov 14 12:30:36 2002 -->
Access to fix CCU5 problem
<!-- Author: Rick Snider -->
<!-- Comment: Thu Nov 14 13;03;51 comment by...Rick Snider -->
Done
<!-- Date: Thu Nov 14 12:34:09 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154145">154145</A>
TERMINATE: This was just a test that the silicon would pass ColdStart.
<!-- processed by 128.97.23.249 -->
<!-- Author: Lin 2080 -->


<!-- Date: Thu Nov 14 12:41:14 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154146">154146</A>
ACTIVATE: BSC/Miniplug calibration. No beam. Quite time. Partition 0.
<!-- processed by 131.225.242.220 -->
<!-- Author: Lin 2080 -->


<!-- Date: Thu Nov 14 12:41:33 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154146">154146</A>
TERMINATE: End of BSC/Miniplug calibration.
<!-- processed by 155.178.180.5 -->
<!-- Author: Lin 2080 -->



<!-- Date: Thu Nov 14 12:46:50 2002 -->
Silicon systems check out OK after the quench.  More details in  
<a href= 
'/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=130&frame=2&anchor=12:40:46'> 
this silicon e-log entry</a>
<!-- Author: Jason Nielsen -->


<!-- Date: Thu Nov 14 12:48:04 2002 -->
<pre>PELCO camera controls (for cameras in Collision Hall) are 
not working. Tim Cunneen is coming to check out problem.</pre>
<!-- Author: JJ -->
<!-- Comment: Thu Nov 14 13;25;06 comment by...JJ -->
<pre>Steve Hahn and Tim showed up at same time. It turns out Steve
knows how to reset/login the camera control box. Everything
working again.</pre>
<!-- Date: Thu Nov 14 12:56:28 2002 -->
<pre>Removed non-longer-used voice scheduler entry, which I think was the source of spurious COT audibe alarms.</pre>
<!-- Author: Aseet Mukherjee -->



<!-- Date: Thu Nov 14 13:11:27 2002 -->
<pre>Camille doing test run to check out IMU.</pre>
<!-- Author: JJ :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154148' target=_top>154148</a>) -->


<!-- Date: Thu Nov 14 13:21:54 2002 -->
CCU05 (TSU&BSU) was replaced during controlled access. 
Now CCU07 is there.
<!-- Author: Akram Artikov, Oleg Poukhov -->


<!-- Date: Thu Nov 14 13:22:35 2002 -->
VxMon display for b0imu00 is red -- unreadable.
<!-- Author: Rick Snider -->



<!-- Date: Thu Nov 14 13:28:54 2002 -->
<pre>CSP/BSU/TSU/CSW were all grey in Global Alarms.  Closed and 
restarted Global Alarms and these systems are no longer grey. 
However CCU replacement still shows up as red. 95% of all 
people associated with muon systems are grouping on 1st floor 
investigating. 
 
Camille reports that CCU07 put in as replacement for CCU05 is 
also bad. We are paging OPS Manager to discuss plan.</pre>
<!-- Author: JJ -->
<!-- Comment: Thu Nov 14 13;53;20 comment by...JJ -->
<pre>Note that we have strange behavior in Global Alarms. When we
hit the "D" button for PSM and then click the "Muon/SVX/FCAL"
button, there is suppose to be another popup window. Instead
the Power Supply Monitoring window closes and we get no popup.

Dervin is looking for Marcin Wolters who perhaps can help.</pre>
<!-- Comment: Thu Nov 14 14;51;31 comment by...JJ -->
<br><tt>Oopsss</tt><br>
Camille had minimized popup window. It was there all along<br>
on the toolbar. This is non-problem.<br>
<b>Note that another replacement CCU is being tested for the next<br>
time we get an access.</b></a>
<!-- Comment: Thu Nov 14 15;30;11 comment by...JJ -->
<pre>TEV quenched an hour ago and a crew is replacing the
CCU again.</pre>
<!-- Date: Thu Nov 14 13:33:04 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154148">154148</A>
TERMINATE: Close test run.
<!-- processed by 131.225.127.7 -->
<!-- Author: Lin 2080 -->



<!-- Date: Thu Nov 14 13:35:40 2002 -->
<pre>Took Level3 and hEVB to test a new software from Steve T.  
</pre>
<!-- Author: Boris/Steve -->


<!-- Date: Thu Nov 14 13:47:37 2002 -->
<pre>Use IMU00 and IMU01 for test runs: 
 
Crate IMU_00 is missing card TDC_BMU_01 in slot 12 
Crate IMU_00 is missing card TDC_BMU_02 in slot 13 
Crate IMU_00 is missing card TDC_BMU_03 in slot 14 
Crate IMU_00 is missing card TDC_BMU_04 in slot 15 
........ 
Crate IMU_01 is missing card TDC_BMU_18 in slot 19 
Crate IMU_01 is missing card TDC_BMU_19 in slot 20 
 
Just wandering if this is Ok.  
</pre>
<!-- Author: Boris.  -->
<!-- Comment: Thu Nov 14 14;53;05 comment by...JJ -->
<pre>Camille did a test run with the two IMU crates and did not get
any errors. IMU crates are probably OK.</pre>
<!-- Date: Thu Nov 14 13:48:07 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154149">154149</A>
 Activated at 2002.11.14 13:46:38
<!-- processed by 131.225.236.136 -->
<!-- Author: RunControl -->


<!-- Date: Thu Nov 14 13:49:16 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154149">154149</A>
 Terminated at 2002.11.14 13:47:44
<!-- processed by 131.225.236.136 -->
<!-- Author: RunControl -->




<!-- Date: Thu Nov 14 13:53:40 2002 -->
The long and the short of it is that we swapped a bad CCU (5) for one that was no better (7). We are mostly back to the situation we were in at the beginning of the day. Oleg still needs to modify some files to account for the 5-7 swap. 
 
Oleg and Akram are looking for something to eat; Camille is looking for someone to kill.
<!-- Author: Phil Schlabach -->

<!-- Date: Thu Nov 14 14:09:01 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154150">154150</A>
 Activated at 2002.11.14 14:07:01
<!-- processed by 131.225.235.18 -->
<!-- Author: RunControl -->


<!-- Date: Thu Nov 14 14:15:16 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154150">154150</A>
 Terminated at 2002.11.14 14:13:59
<!-- processed by 131.225.235.18 -->
<!-- Author: RunControl -->


<!-- Date: Thu Nov 14 14:28:43 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154151">154151</A>
ACTIVATE: fake svt delay tests
<!-- processed by 131.225.236.161 -->
<!-- Author: matt/greg -->


<!-- Date: Thu Nov 14 14:34:13 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154152">154152</A>
 Activated at 2002.11.14 14:31:55
<!-- processed by 131.225.127.7 -->
<!-- Author: RunControl -->


<!-- Date: Thu Nov 14 14:42:33 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154152">154152</A>
 Terminated at 2002.11.14 14:41:07
<!-- processed by 131.225.127.7 -->
<!-- Author: RunControl -->



<!-- Date: Thu Nov 14 14:53:05 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154154">154154</A>
 Activated at 2002.11.14 14:51:26
<!-- processed by 131.225.127.7 -->
<!-- Author: RunControl -->



<!-- Date: Thu Nov 14 14:58:07 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154154">154154</A>
 Terminated at 2002.11.14 14:53:50
<!-- processed by 131.225.127.7 -->
<!-- Author: RunControl -->



<!-- Date: Thu Nov 14 14:59:50 2002 -->
Another quench. Try another access during quench recovery?
<!-- Author: Rick Snider -->

<!-- Date: Thu Nov 14 15:08:31 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154156">154156</A>
 Activated at 2002.11.14 15:07:04
<!-- processed by 131.225.45.171 -->
<!-- Author: RunControl -->


<!-- Date: Thu Nov 14 15:09:49 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154156">154156</A>
 Terminated at 2002.11.14 15:08:27
<!-- processed by 131.225.235.217 -->
<!-- Author: RunControl -->


<!-- Date: Thu Nov 14 15:09:49 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154155">154155</A>
TERMINATE: svt delay test run
<!-- processed by 131.225.235.217 -->
<!-- Author: matt/greg -->


<!-- Date: Thu Nov 14 15:19:28 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154158">154158</A>
 Activated at 2002.11.14 15:17:24
<!-- processed by 131.225.127.7 -->
<!-- Author: RunControl -->


<!-- Date: Thu Nov 14 15:19:28 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154158">154158</A>
 Terminated at 2002.11.14 15:17:43
<!-- processed by 131.225.127.7 -->
<!-- Author: RunControl -->




<!-- Date: Thu Nov 14 15:30:57 2002 -->
<pre>PSM alarm. Source is work being done to replace CCU.</pre>
<!-- Author: JJ -->
<!-- Comment: Thu Nov 14 15;46;17 comment by...JJ -->
PSM green again. CCU work crew is leaving collision hall.
<!-- Date: Thu Nov 14 15:33:21 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154161">154161</A>
 Activated at 2002.11.14 15:31:55
<!-- processed by 131.225.235.38 -->
<!-- Author: RunControl -->


<!-- Date: Thu Nov 14 15:37:52 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154161">154161</A>
 Terminated at 2002.11.14 15:36:18
<!-- processed by 131.225.236.229 -->
<!-- Author: RunControl -->



<!-- Date: Thu Nov 14 15:50:24 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154165">154165</A>
 Activated at 2002.11.14 15:48:55
<!-- processed by 131.225.236.229 -->
<!-- Author: RunControl -->
<!-- Comment: Thu Nov 14 17;35;55 comment by...JJ -->
<a href=
'/cgi/elog/elog.pl?nb=2002&action=view&page=953&frame=2&anchor=17:33:05'>
Cracker Jack CCU Replacement Team in Action! </a> 

<!-- Date: Thu Nov 14 15:50:24 2002 -->
Completed access to replace CCU07.
<!-- Author: Rick Snider -->

<!-- Date: Thu Nov 14 15:54:32 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154165">154165</A>
 Terminated at 2002.11.14 15:52:40
<!-- processed by 131.225.122.33 -->
<!-- Author: RunControl -->


<!-- Date: Thu Nov 14 15:56:39 2002 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>15:55:02</TD>                                                       
<TD ALIGN=CENTER>::</TD>                                                
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- processed by 131.225.232.91 -->
<!-- Author: End of Shift Report -->



<!-- Date: Thu Nov 14 16:04:41 2002 -->
<font color=red>Shift Summary: </font><pre> 
08:00  Store 1961 dropped. Stack at 172 mA. 
 
08:15  Shot setup in progress. 
       Running calibrations. 
 
08:58  Audible COT HV alarm. No evidence of a problem. 
 
10:05  Run two CR test runs with new L3 executable. COSMICS [8,200,337] 
 
10:12  Loading final p's. 
 
10:39  Loading final pbars. Paged silicon 
 
11:05  Lost the shot. Stack at 13 mA. 
 
11:10  Start running calibrations. 
 
12:30  Access to fix CCU5. 
 
12:56  Aseet fixes spurious audible COT alarms. 
 
13:00  Access completed.  
       Replaced CCU05 with CCU07.  
        
13:28  The newly installed CCU07 is bad. 
        
13:35  New HEVB software testing. 
 
15:25  Access to replace CCU07. 
 
15:59  Evaluating CCU replacement. 
       HEVB testing in progress. 
 
No luminosity delivered. 
</pre>
<!-- Author: Rick Snider -->

<!-- Date: Thu Nov 14 16:11:01 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154168">154168</A>
 Activated at 2002.11.14 16:09:38
<!-- processed by 131.225.235.217 -->
<!-- Author: RunControl -->



<!-- Date: Thu Nov 14 16:12:27 2002 -->
Silicon checks out OK after the quench.  Please page the silicon general pager after <B>every</B> quench even if you think there was no beam.  We have a very simple sequence of checks we perform to ensure the safety of the detector.
<!-- Author: Jason Nielsen -->