<!-- Title: 05:20:52  Tue Nov  9 2004 -->
<!-- Crewchief: Bill Noe -->
<!-- Op1: &nbsp; -->
<!-- Op2: &nbsp; -->
<!-- Op3: &nbsp; -->
<!-- Op4: &nbsp; -->
<!-- Op5: &nbsp; -->
<!-- Op6: &nbsp; -->
<!-- Notes: -->
No alarms this morning.


<!-- Date: Tue Nov 9 05:21:32 2004 -->
<table border=0><td><tr><td bgcolor=#e0e030>Operators on duty beginning Monday 11/15/05 at 0600.</td></tr></td></table>
<!-- Author: Bill Noe -->


<!-- Date: Tue Nov 9 06:21:06 2004 -->
Steve Gordon and Bruce Vollmer will not be in today.
<!-- Author: Bill Noe -->


<!-- Date: Tue Nov 9 06:22:53 2004 -->
Some kind of automatic update has occurred forcing a reboot upon completion. We need to track down this process and kill it. THIS IS UNACCEPTABLE!
<!-- Author: Bill Noe -->
<!-- Comment: Tue Nov  9 06;27;56 comment by...Jim L -->
This thing caused ALL NODES to reboot by them selves??????
<!-- Date: Tue Nov 9 06:23:56 2004 -->
Dean and KC will be testing the IMU for preparation to start flowing argon/ethane later this morning.
<!-- Author: Bill Noe -->
<!-- Comment: Tue Nov  9 09;22;39 comment by...KC -->
Dean and I are doing a argon ethane flast flush following procedure 13 as asked for by DEl.
<!-- Date: Tue Nov 9 07:50:16 2004 -->
Gas alarm is TOPSUP high. I asked KC to readjust all the flows on the platform.
<!-- Author: Bill Noe -->
<!-- Comment: Tue Nov  9 08;12;59 comment by...KC -->
Jim H. and I set all of the flows to their correct level.


<!-- Date: Tue Nov 9 08:39:49 2004 -->
Shut down both engines so Chuck (the Electrician), can wire in the back up power source plug-in for the engines. Should take about a half hour or so.
<!-- Author: Jim L -->
<!-- Comment: Tue Nov  9 09;02;36 comment by...Jim L -->
Chuck and company are done.--->restarting the engines and reloading the compressor.

<!-- Comment: Tue Nov  9 09;09;16 comment by...Jim L -->
Cryo recovering nicely----> all cryo alarms are active and dewar only needs to refill(at 46%).


<!-- Comment: Tue Nov  9 10;12;39 comment by...Jim L -->
Going good --->dewar is at 52.7% full.


<!-- Date: Tue Nov 9 11:01:24 2004 -->
Changed flow rate(Dean) on CES supply from 15 to 10 scfh also changed the hi/lo limits to 11 and 7 as are the others.
<!-- Author: Jim L -->


<!-- Date: Tue Nov 9 11:30:22 2004 -->
Gamma voltage alarm WHA 1 low. Contacted 2nd floor to let them know. Silenced the alarm.
<!-- Author: Bill Noe -->


<!-- Date: Tue Nov 9 12:24:04 2004 -->
Dewar level is 60.9%. Put the dry engine and PVMF back in loop.
<!-- Author: Jim H. -->


<!-- Date: Tue Nov 9 12:51:18 2004 -->
Dean, Del and I completed procedure 13 for COT. Gas is flowing and the cold trap is pre-cooling.
<!-- Author: KC -->
<!-- Comment: Tue Nov  9 13;57;16 comment by...DA -->
Cold Trap is operating.
<!-- Date: Tue Nov 9 13:19:25 2004 -->
Craig contacted FESS about the noisy PF-1 HVAC fan. <p> 
I contacted them about the leaking LCW pump #2. Tickets in on both.
<!-- Author: Bill Noe -->



<!-- Date: Tue Nov 9 14:00:03 2004 -->
We will not be adding the Ar/O2 to the Ar/Eth during this testing period for COT. Also, we will wait to start the Recirc system until tomorrow morning. Chuck is doing some electrical work with the panel at the 710 level in preparation for using the new bellows pump.
<!-- Author: D. Allspach -->


<!-- Date: Tue Nov 9 15:10:52 2004 -->
Shut down PF1 so mechanics could look at it.
<!-- Author: Craig -->


<!-- Date: Tue Nov 9 15:16:39 2004 -->
Restarted PF1 Mechanics will come back tomorrow when the whole system can be shutdown.
<!-- Author: Craig -->


<!-- Date: Tue Nov 9 15:44:49 2004 -->
With the help of Warren B. and Del.A. the flammable gas heads and the O2 heads in the Gas Mixing Shed have been Zeroed and Calibrated.
<!-- Author: George Wyatt -->