<!-- Title: Eve shift. Mon Jul 21, 2008 -->
<!-- SciCo: Bo Jayatilaka -->
<!-- DAQAce: Jan Morlock -->
<!-- MonAce: &nbsp; -->
<!-- CO: Satyajit Behari -->
<!-- OpManager: Ankush Mitra -->
<!-- Notes: -->
<pre>Access underway. </pre>


<!-- Date: Mon Jul 21 16:10:51.555 2008 -->
<p>Work done this access with my buddy Sebastian: first tried to 
figure out problem with CEM XEF for wedges 00E-02E, presumably 
in calib card in b0ccal08.  Did not succeed, for more details 
see new access request  
<a href=":http://www-cdfonline.fnal.gov/java/cdfdb/servlet/ApproveAccessRequest?ACCESS_REQUEST_ID=0323"> 
here</a>.</p> 
 
<p>Also, attempted to fix problem with HV for CEM PMT 07W T8L, 
which has been drifting +/-100 V or more.  Simply power cycled 
Pisabox in this wedge, and pulled and checked AC receptacle 
with circuit tester (it was OK).  Now seems to be working much 
better, but still drifting about +/-5 V from the correct value 
after I reset the channel.  But it seems to be settling down. 
Will keep an eye on it.  If it gets bad again, will have to 
try swapping HV distribution board.</p> 
settling down.
<!-- Author: Steve Hahn :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265066' target=_top>265066</a>) -->
<!-- Comment: Wed Jul 23 14;07;04 comment by...Steve Hahn -->
I should also note I have now unmasked the CEM channel 07W T8L
which has, so far, remained stable.
<!-- Date: Mon Jul 21 16:14:56.188 2008 -->
Done TO for b0l2pu07. Shepherded it.
<!-- Author: Jan :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265066' target=_top>265066</a>) -->
<!-- Comment: Mon Jul 21 17;50;29 comment by...Jan -->
Again
<!-- Date: Mon Jul 21 16:23:24.734 2008 -->
CMX trip during stand-by in channels 4/5 NW. I turned power back to stand-by. This fixed the problem.
<!-- Author: Jan :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265066' target=_top>265066</a>) -->


<!-- Date: Mon Jul 21 16:31:18.332 2008 -->
<pre>BMU access report:  
 
We were trying to fix HV noise problem on NW54-55 and oscillation problem on NW55. As soon as the access started the oscillation on NW55 went away (we have seen this before many times). Our initial guess on HV noise on NW54-55 was the HV board trouble in either one of 54 or 55, but when we disconnected Xend jumper in order to identify which one is responsible, the noise disappeared. The problem might have been in the Xend. HV sector NW48 which contains NW54-55 was restored to normal configuration.  
We also swapped three preamps to fix single dead channels on NW127, SE100 and SE116. One dead channel on NW127 was fixed, but SE100 and 116 stayed the same. Those channels are not completely dead, but show very small signals (both look the same). This might be caused by blocked socket for preamp pins on HV boards. </pre>
<!-- Author: Woo-Hyun & Yeongdae & Varsha -->


<!-- Date: Mon Jul 21 16:50:13.105 2008 -->
PSM turned red on iFix display. Details showed that End Plugs is responsible for this. The details of End Plug showed SMD 08E red. Again the details of this crate showed 
<pre> 
A/D Channel: Channel 7 
Returned Value: 0.00 
Function: -7.5 V 
</pre> 
We called Mike Lindgren. He said that he will check on it.
<!-- Author: Jan :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265066' target=_top>265066</a>) -->


<!-- Date: Mon Jul 21 17:07:05.946 2008 -->
<p>PPR Access work. Investigated Dead towers in 05-06W: 1, 3, 5, 7, 9, 11, 13, 15. HV cable to MAPMT of these towers was ripped 
during the previous plug pull and shorted to ground. The CAEN 
crate end of the cable was an open circuit. Jamie, along with  
George, John, and Lew repaired the HV cable.  
</p>
<!-- Author: Willis: For Britney and Mike L :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265066' target=_top>265066</a>) -->

<!-- Date: Mon Jul 21 17:09:44.842 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265066>265066</A>
 Terminated at 2008.07.21 17:09:38
<!-- Author: RunControl -->


<!-- Date: Mon Jul 21 17:10:12.988 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265066>265066</A>
TERMINATE: Done
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 17:11:54.314 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265067>265067</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 17:12:42.683 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265067>265067</A>
TERMINATE: Done
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 17:13:58.934 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265068>265068</A>
ACTIVATE: RunSet: COT_ACE after closing plugs
<!-- Author: Eiko x4953 -->


<!-- Date: Mon Jul 21 17:15:52.639 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265067>265067</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Mon Jul 21 17;21;28 comment by...Jan -->
I accidentally did this calibration with PEM/PHA/PSH on and CES in stand-by. I just repeated it.
<!-- Date: Mon Jul 21 17:17:05.938 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265069>265069</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Jan -->



<!-- Date: Mon Jul 21 17:17:08.155 2008 -->
<pre>Muon scintillator access summary: 
 
Bad cable connection tested at CCU for the CSP bottom east counter (ch.8 in YMon plots). 
Visual observation of the faulty Miniskirt counter was performed on the south west. Access to the counter is quite restricted - no way to reach it from the top side. Nothing was done, we still have signals from this counter - PMT on another side is ok. </pre>
<!-- Author: Sasha Simonenko :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265067' target=_top>265067</a>) -->

<!-- Date: Mon Jul 21 17:20:07.483 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265070>265070</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 17:20:42.045 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265070>265070</A>
TERMINATE: Done
<!-- Author: Jan -->



<!-- Date: Mon Jul 21 17:22:53.425 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265070>265070</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Mon Jul 21 17;27;17 comment by...Lindgren -->
<pre>Looks like one card in smd box 8e failed when the connector
got bumped.  I am swapping in a new one.</pre>
<!-- Date: Mon Jul 21 17:25:38.175 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265071>265071</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 17:26:47.058 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265071>265071</A>
TERMINATE: Done
<!-- Author: Jan -->



<!-- Date: Mon Jul 21 17:28:37.308 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265071>265071</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Mon Jul 21 17;29;30 comment by...Lindgren -->
<pre>Ok, the board swap fixed it, we should be good to go.
</pre>


<!-- Date: Mon Jul 21 17:38:08.375 2008 -->
<pre>COT electrons checkout after closing plugs: 
 
 Hot cells SL6-6,7 are gone now. Now we have two additional  
intermitten warm cells SL2-46,47, which had been hot and quiet  
with the same pattern (ie. the wires that are hot). Will observe it. 
 Otherwise, didnt see broken connections caused by closing plugs.</pre>
<!-- Author: Shin-Shan Yu (Eiko) -->


<!-- Date: Mon Jul 21 17:39:28.380 2008 -->
<table border=0><td><tr><td bgcolor=#e0e030>Dear shift crew: 
 
  Please take a normal COTCALIB_ACE calibration after the muon walls are closed (or anytime before taking data). </td></tr></td></table>
<!-- Author: Shin-Shan Yu -->

<!-- Date: Mon Jul 21 17:40:30.363 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265073>265073</A>
 Activated at 2008.07.21 17:40:08
<!-- Author: RunControl -->


<!-- Date: Mon Jul 21 17:40:31.116 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265073>265073</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,1101,706] 
<!-- Author: Jan -->



<!-- Date: Mon Jul 21 17:44:41.730 2008 -->
<center><a href='/cgi/elog/elog.pl?nb=2008&action=view&page=-3713&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2008&action=view&page=-3713' width='300'></a><a href='/cgi/elog/elog.pl?nb=2008&action=view&page=-3712&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2008&action=view&page=-3712' width='300'></a></center>
<!-- Author: Satyajit (CO) -->
<!-- Comment: Mon Jul 21 17;45;35 comment by...Satyajit (CO) -->
Plots of interest (to Mike L) from the latest SMX calibration run.



<!-- Date: Mon Jul 21 17:55:23.332 2008 -->
Plug 4E - Drive 7 was stuck on channel 2, vedge 18.0 75964. 
The source was returned to the garage by hand. No big friction 
was discovered. No investigation was done. 
Plug 1W - Drive 0 was stuck on channel 47, vedge 2.0 54021. 
The source was return to the garage by hand. No big friction 
was discovered. Any way the central washers were lubricated, 
the top washer was polished, the reel gear was replaced with other 
one which has bigger central hole. The stuck problem was connected 
with broken control panel which could extend wire but not retract. 
After the panel was replaced with spare one the action restored. 
<!-- Author: Nikolay Luzhetskiy -->

<!-- Date: Mon Jul 21 18:01:20.122 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265073>265073</A>
 Terminated at 2008.07.21 18:00:59
<!-- Author: RunControl -->


<!-- Date: Mon Jul 21 18:01:21.709 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265073>265073</A>
TERMINATE: Done
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 18:07:17.934 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265075>265075</A>
TERMINATE: Trying to test the SILI-BIAS table... forgot to ignore inhibits (Oscar)
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 18:07:18.796 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=264813>264813</A>
<HTML>Run <A HREF=http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN=264813 target=_runSummary>264813</A> has not yet been checked for goodrun status; please do so.</HTML>
<!-- Author: RunControl -->


<!-- Date: Mon Jul 21 18:10:13.126 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265076>265076</A>
 Activated at 2008.07.21 18:09:51
<!-- Author: RunControl -->


<!-- Date: Mon Jul 21 18:10:14.656 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265076>265076</A>
ACTIVATE: RunSet: AAA_CURRENT  Trigger: SILI_BIAS_SCAN [4,1107,706] Test of a trigger table (Oscar)
<!-- Author: Jan -->



<!-- Date: Mon Jul 21 18:11:14.743 2008 -->
Ricardo and Dervin installed the new BLM/diamond mount on the east side, which will make future BLM moves much easier.  After the access, we checked out the diamonds and found that three of them were not getting voltage.  Ricardo went down and redid the connections and the voltage was restored.  All diamonds and BLMs are now running and enabled in the abort loop.
<!-- Author: Peter and Ricardo -->

<!-- Date: Mon Jul 21 18:13:15.448 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265076>265076</A>
 Terminated at 2008.07.21 18:12:01
<!-- Author: RunControl -->


<!-- Date: Mon Jul 21 18:13:16.263 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265076>265076</A>
TERMINATE: Test was successful for this trigger table (Oscar).
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 18:13:17.842 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=264813>264813</A>
RUNSTATUS: <PRE>Marked Bad, explanation:Bias scan, not physics run
</PRE>
<!-- Author: cdfscico -->



<!-- Date: Mon Jul 21 18:15:47.592 2008 -->
<pre>As indicated in the last trigger meeting, I tested the new SILI_BIAS_SCAN table that has been produced for us. It worked fine and events were collected with no problems observed. I want to remark that this version is using L2TAG 1107 that seemss to be obsolete, but it seems fine for this (we do not use anything special at L2).</pre>
<!-- Author: Oscar (for the silicon) -->

<!-- Date: Mon Jul 21 18:17:05.822 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265077>265077</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 18:17:06.084 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265077>265077</A>
TERMINATE: Done
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 18:17:07.725 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265077>265077</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Jan -->
<!-- Comment: Mon Jul 21 19;02;19 comment by...Satyajit (CO) -->
Calibration is successful
<!-- Date: Mon Jul 21 18:17:08.316 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265077>265077</A>
TERMINATE: Done
<!-- Author: Jan -->



<!-- Date: Mon Jul 21 18:18:54.082 2008 -->
All keys are back. Waiting for search and secure.
<!-- Author: Bo -->

<!-- Date: Mon Jul 21 18:19:17.856 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265077>265077</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->


<!-- Date: Mon Jul 21 18:20:55.227 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265078>265078</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Jan -->
<!-- Comment: Mon Jul 21 19;11;22 comment by...Satyajit (CO) -->
Calibration looks good, except for PHA wedge 2E channel 19 whose
RMS are between 17-18.
<!-- Date: Mon Jul 21 18:21:56.215 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265078>265078</A>
TERMINATE: Done
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 18:22:17.331 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265078>265078</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->


<!-- Date: Mon Jul 21 18:23:00.323 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265079>265079</A>
ACTIVATE: RunSet: CALIB_CLC_QIE 
<!-- Author: Jan -->
<!-- Comment: Mon Jul 21 19;13;29 comment by...Satyajit (CO) -->
Calibration is successful.
<!-- Date: Mon Jul 21 18:24:05.267 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265079>265079</A>
TERMINATE: Done
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 18:24:06.539 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265079>265079</A>
CALIBRATION: <U><B>clcqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#clcqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->


<!-- Date: Mon Jul 21 18:26:35.683 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265080>265080</A>
ACTIVATE: RunSet: CALIB_QIE_FWD 
<!-- Author: Jan -->
<!-- Comment: Mon Jul 21 19;17;19 comment by...Satyajit (CO) -->
Calibration successful.
<!-- Date: Mon Jul 21 18:26:59.540 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265080>265080</A>
TERMINATE: Done
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 18:27:17.842 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265080>265080</A>
CALIBRATION: <U><B>bscqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#bscqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->


<!-- Date: Mon Jul 21 18:28:32.797 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265081>265081</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL 
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 18:30:39.196 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265081>265081</A>
TERMINATE: Done T0: CAL_PULSER_01 -> Abort
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 18:31:17.840 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265082>265082</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL 
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 18:35:59.091 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265082>265082</A>
TERMINATE: Done TO CAL_PULSER_01
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 18:38:05.912 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=264816>264816</A>
<HTML>Run <A HREF=http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN=264816 target=_runSummary>264816</A> has not yet been checked for goodrun status; please do so.</HTML>
<!-- Author: RunControl -->


<!-- Date: Mon Jul 21 18:46:36.846 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265083>265083</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL 
<!-- Author: Jan -->



<!-- Date: Mon Jul 21 18:51:42.828 2008 -->
The red telephone rang. We seem to have an opportunity for better car insurance rates....
<!-- Author: Bo -->


<!-- Date: Mon Jul 21 18:52:16.185 2008 -->
During Laser calibration we always got CAL_PULSER_01 Done TO. The L2 expert Petro adviced me to just continue since it isn´t included in the current setting.
<!-- Author: Jan :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265083' target=_top>265083</a>) -->

<!-- Date: Mon Jul 21 19:01:39.435 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265083>265083</A>
TERMINATE: Done TO for CAL_PULSER_01 caused RC to stop
<!-- Author: Jan -->




<!-- Date: Mon Jul 21 19:04:17.649 2008 -->
Search and secure is done. Called cryo to turn AC up in hall.
<!-- Author: Bo -->

<!-- Date: Mon Jul 21 19:06:27.910 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265084>265084</A>
 Activated at 2008.07.21 19:05:37
<!-- Author: RunControl -->


<!-- Date: Mon Jul 21 19:06:28.184 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265084>265084</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,1101,706] 
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 19:07:42.216 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265084>265084</A>
 Terminated at 2008.07.21 19:07:32
<!-- Author: RunControl -->


<!-- Date: Mon Jul 21 19:09:23.511 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265086>265086</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL 
<!-- Author: Jan -->




<!-- Date: Mon Jul 21 19:12:48.789 2008 -->
Morris has checked out the COT and asked to page XFT to unmask Quadrant B with the exception of cell 27. Karen has called back and said this will be done. 
<!-- Author: Bo -->
<!-- Comment: Mon Jul 21 19;18;35 comment by...Bo -->
Miscommunication! It was indeed quadrant A that was masked before
<!-- Comment: Mon Jul 21 19;30;12 comment by...Karen -->
SL2 Quadrant A (0-90 in phi) has now been unmasked except for Cell 27 
<!-- Date: Mon Jul 21 19:16:28.683 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265086>265086</A>
TERMINATE: Done
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 19:17:19.375 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265087>265087</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: Jan -->



<!-- Date: Mon Jul 21 19:18:15.973 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265087>265087</A>
TERMINATE: Done TO for CAL_PULSER_01
<!-- Author: Jan -->



<!-- Date: Mon Jul 21 19:20:24.605 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265088>265088</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 19:20:25.325 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265088>265088</A>
TERMINATE: Done TO for CAL_PULSER_01
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 19:20:26.640 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265089>265089</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: Jan -->



<!-- Date: Mon Jul 21 19:43:00.294 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265089>265089</A>
TERMINATE: Done TO for CAL_PULSAR_01
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 19:47:03.513 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265090>265090</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 19:49:42.105 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265090>265090</A>
TERMINATE: Done
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 19:49:43.341 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265087>265087</A>
CALIBRATION: <U><B>pluglaser</B></U> calibration FAILED, please investigate
<!-- Author: CDFCALIB -->


<!-- Date: Mon Jul 21 19:50:34.306 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265091>265091</A>
ACTIVATE: RunSet: BOTHPLUGS_MM 
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 19:53:40.651 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265091>265091</A>
TERMINATE: Done
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 19:53:41.640 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265092>265092</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL 
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 19:54:38.428 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265092>265092</A>
TERMINATE: Done TO for CAL_PULSER_01
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 19:56:46.261 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265093>265093</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 19:57:09.233 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265093>265093</A>
TERMINATE: Done TO for CAL_PULSER_01
<!-- Author: Jan -->



<!-- Date: Mon Jul 21 20:00:03.656 2008 -->
I got the error message ¨Ethernet Event Builder reported error code -1¨. So I restarted RC.
<!-- Author: Jan :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265094' target=_top>265094</a>) -->

<!-- Date: Mon Jul 21 20:00:52.642 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265094>265094</A>
 Activated at 2008.07.21 20:00:43
<!-- Author: RunControl -->


<!-- Date: Mon Jul 21 20:01:10.808 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265094>265094</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,1101,706] L1MET test - Laura
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 20:09:54.367 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265094>265094</A>
 Terminated at 2008.07.21 20:09:41
<!-- Author: RunControl -->


<!-- Date: Mon Jul 21 20:09:55.275 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265094>265094</A>
TERMINATE: Done
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 20:13:24.578 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265095>265095</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [19,1104,705] : Scaler Monitor test (Olga)
<!-- Author: Jan -->



<!-- Date: Mon Jul 21 20:15:00.036 2008 -->
<table border=0><td><tr><td bgcolor=#e0e030>Request to Aces: When we are taking data, please keep an eye on the L2 log file. If you see a lot of errors scrolling down the screen, please issue an HRR. These errors will include references about Bunch Counters and Buffer, along with other info. These are bunch counter and buffer number mismatches. So, if you see these scrolling fast down the L2 log file screen, please issue an HRR immediately. 
<br> 
Thanks!</td></tr></td></table>
<!-- Author: Heather -->

<!-- Date: Mon Jul 21 20:33:33.953 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265097>265097</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [19,1104,705] 
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 20:53:14.114 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265101>265101</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [19,1104,705] 
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 21:07:28.047 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265103>265103</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [19,1104,705] 
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 21:25:21.235 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265106>265106</A>
 Activated at 2008.07.21 21:24:17
<!-- Author: RunControl -->


<!-- Date: Mon Jul 21 21:25:22.295 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265106>265106</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [7,1108,706] 
<!-- Author: Jan -->
<!-- Comment: Mon Jul 21 21;35;42 comment by...Jan -->
Quiet-time; L1A instanteneous rate: 58.1
<!-- Date: Mon Jul 21 21:27:03.623 2008 -->
Laura, Pedro and I were trying to reproduce a problem observed for the Scaler Monitor a couple of weeks ago (a configuration command was received by the decision node in the middle of the run.). We tried different combinations but we were not able to send any configuration command to the decision node (from Run Control) once the run was started. 
<!-- Author: Pedro, Laura and Olga -->

<!-- Date: Mon Jul 21 21:33:19.569 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265106>265106</A>
 Terminated at 2008.07.21 21:32:57
<!-- Author: RunControl -->


<!-- Date: Mon Jul 21 21:33:29.568 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265106>265106</A>
TERMINATE: Done
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 21:35:42.239 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265107>265107</A>
ACTIVATE: RunSet: ACE_TOF_QIE_NEW 
<!-- Author: Jan -->
<!-- Comment: Mon Jul 21 21;48;51 comment by...Satyajit (CO) -->
Calibration successful.
<!-- Date: Mon Jul 21 21:37:51.027 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265107>265107</A>
TERMINATE: Done
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 21:37:52.597 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265093>265093</A>
CALIBRATION: <U><B>pluglaser</B></U> calibration FAILED, please investigate
<!-- Author: CDFCALIB -->


<!-- Date: Mon Jul 21 21:37:53.107 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265107>265107</A>
CALIBRATION: <U><B>tofqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#tofqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->


<!-- Date: Mon Jul 21 21:37:54.337 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265108>265108</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Jan -->
<!-- Comment: Mon Jul 21 22;02;40 comment by...Satyajit (CO) -->
Calibration successful.
<!-- Date: Mon Jul 21 21:44:10.475 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265108>265108</A>
TERMINATE: Done
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 21:44:12.598 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265109>265109</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: Jan -->
<!-- Comment: Mon Jul 21 21;51;48 comment by...Satyajit (CO) -->
Calibration successful.
<!-- Date: Mon Jul 21 21:44:13.193 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265108>265108</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->


<!-- Date: Mon Jul 21 21:44:14.459 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265109>265109</A>
TERMINATE: Done
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 21:44:15.884 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265109>265109</A>
CALIBRATION: <U><B>cemled</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cemled target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->


<!-- Date: Mon Jul 21 21:44:16.386 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265110>265110</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Jan -->
<!-- Comment: Mon Jul 21 22;01;09 comment by...Satyajit (CO) -->
Calibration is successful.
<!-- Date: Mon Jul 21 21:44:17.082 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265110>265110</A>
TERMINATE: Done
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 21:45:22.105 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265110>265110</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->


<!-- Date: Mon Jul 21 21:47:21.339 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265111>265111</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: Jan -->
<!-- Comment: Mon Jul 21 21;58;33 comment by...Satyajit (CO) -->
Calibration is good, except for the already known dead region
in wedges 0-2 on the east side and hot blocks in wedges 3 and 5
in the normalized plot.
<!-- Date: Mon Jul 21 21:48:50.298 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265111>265111</A>
CALIBRATION: <U><B>calxef</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calxef target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->


<!-- Date: Mon Jul 21 21:48:51.658 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265111>265111</A>
TERMINATE: Done
<!-- Author: Jan -->



<!-- Date: Mon Jul 21 21:51:48.526 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265112>265112</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Jan -->



<!-- Date: Mon Jul 21 21:53:42.974 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265112>265112</A>
TERMINATE: Done
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 21:53:43.394 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265113>265113</A>
ACTIVATE: RunSet: TOF_TAC_CALIB 
<!-- Author: Jan -->




<!-- Date: Mon Jul 21 22:02:40.498 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265113>265113</A>
TERMINATE: Done
<!-- Author: Jan -->



<!-- Date: Mon Jul 21 22:12:32.004 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265114>265114</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_TEST_5_01 [22,1111,723] for Emily and for 5 min
<!-- Author: Jan -->
<!-- Comment: Mon Jul 21 22;20;15 comment by...Jan -->
After running for for 5 Minutes the CO told me, that TrigMon was´t running. So I will just start a new run with TrigMon after this one.
<!-- Date: Mon Jul 21 22:17:58.812 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265114>265114</A>
TERMINATE: Done
<!-- Author: Jan -->



<!-- Date: Mon Jul 21 22:17:59.873 2008 -->
MCR is at least an hour away from injecting beam.
<!-- Author: Bo -->


<!-- Date: Mon Jul 21 22:21:23.077 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265115>265115</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_TEST_5_01 [22,1111,723] for Emily and for 5 min and with TrigMon
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 22:28:01.561 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265115>265115</A>
TERMINATE: Done
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 22:47:00.219 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265116>265116</A>
 Activated at 2008.07.21 22:46:49
<!-- Author: RunControl -->


<!-- Date: Mon Jul 21 22:47:41.944 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265116>265116</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,1101,706] 
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 22:53:07.603 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265116>265116</A>
 Terminated at 2008.07.21 22:51:56
<!-- Author: RunControl -->


<!-- Date: Mon Jul 21 22:56:39.541 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265117>265117</A>
 Activated at 2008.07.21 22:55:44
<!-- Author: RunControl -->


<!-- Date: Mon Jul 21 22:57:18.940 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265117>265117</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,1101,706] 
<!-- Author: Jan -->


<!-- Date: Mon Jul 21 23:01:59.402 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265117>265117</A>
 Terminated at 2008.07.21 23:01:23
<!-- Author: RunControl -->


<!-- Date: Mon Jul 21 23:10:35.932 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265118>265118</A>
 Activated at 2008.07.21 23:10:10
<!-- Author: RunControl -->


<!-- Date: Mon Jul 21 23:10:36.467 2008 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=265118>265118</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,1101,706] 
<!-- Author: Jan -->



<!-- Date: Mon Jul 21 23:13:42.065 2008 -->
Aseet will be doing some tests and asked us to ignore COT HV trips for now. 
<!-- Author: Bo -->
<!-- Comment: Mon Jul 21 23;35;14 comment by...Bo -->
Done now.

<!-- Date: Mon Jul 21 23:55:59.271 2008 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>PhysicsTable</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L3Accepts</TH>
<TH>DelivLumi nb<SUP>-1</SUP></TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>23:55:01</TD>
<TD ALIGN=CENTER>::</TD>
<TD ALIGN=RIGHT></TD>               <TD ALIGN=RIGHT></TD>
<TD ALIGN=RIGHT></TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->



<!-- Date: Mon Jul 21 23:58:41.685 2008 -->
<font color=red>Shift Summary: </font>No beam for duration of shift. Access ended around 1900 after which some more tests were performed by experts (trigger table, COT, L2). Continued with cosmics/L2 Torture until next shot setup happens. 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   
Delivered Luminosity   0  
Acquired Luminosity    0  
Efficiency             0
   
</td></tr></table></pre>
<!-- Author: Bo -->