<!-- Title: Day shift. Tue Oct 17, 2000 -->
<!-- Leader: Mel Shochet -->
<!-- Ace: Dan Cyr -->
<!-- Co1: Rob Snihur -->
<!-- Co2: Carla Pilcher -->
<!-- Co3: David Saltzberg(SOS) -->
<!-- Co4: &nbsp; -->
<!-- Co5: &nbsp; -->
<!-- Notes: -->
<pre>Use the integration database for calibration.

Frank and Jim will continue studying COT readout problem.
The SVX group will continue their studies with the SVX crates. </pre>


<!-- Date: Tue Oct 17 09:57:39 2000 -->
Doing calibrations using the calibration database.	 
 
1st up : QIE (had to reset wcal06 to run) 
<!-- Author: Dan :: (run 101167) -->


<!-- Date: Tue Oct 17 09:58:06 2000 -->
<pre>LED calibration</pre>
<!-- Author: Dan :: (run 101168) -->


<!-- Date: Tue Oct 17 09:59:00 2000 -->
COT (had to remove 10, 19 from bill's cot_calib_ore config) 
 
again using the integration db  
<!-- Author: Dan :: (run 101169) -->


<!-- Date: Tue Oct 17 10:01:20 2000 -->
XEF calib. Note : SOMEONE tampered with the voltages on 10W, 20W for the xenon flasher downstairs. They were set to 0 and 800v instead of the 2850. Fortunately P. Gris was present and able to reset the 2 channels. The 1st cal run (101170) was with the 2 channels low, 101173 is after they were brought up to proper voltage.
<!-- Author: Dan :: (run 101173) -->


<!-- Date: Tue Oct 17 10:03:20 2000 -->
Frank is using COT and evb to continues yesterdays studies. P. Gris took CCAL for expert studies. I will look into yesterdays muon logs and bring the muons up to voltage. I will then take a run using the muon trigger w/ wcal+pcal+muons 
  
<!-- Author: Dan -->


<!-- Date: Tue Oct 17 10:22:41 2000 -->
<pre>Took the central calorimeter to perform: 
- a LED calibration in the Xmode: run number 101176 
- a LED calibration in the Dmode: no run number;file: 
 /logger/disk1/csl/04/A20001017100142.test 
- a XEF calibration in the Dmode: no run number; file: 
/logger/disk1/csl/04/A20001017100431.test 
These last two files are on b0dau32.</pre>
<!-- Author: Ph.Gris -->


<!-- Date: Tue Oct 17 10:56:52 2000 -->
<pre>Resource Manager now running frop production again.  You  
should setup fer, but do not source the int file.</pre>
<!-- Author: W.Badgett,RStD -->


<!-- Date: Tue Oct 17 11:18:44 2000 -->
Fred Lewis is looking into the rack protection / fan failures 
in 2RR29C and 2RR30C.  2RR29C was fixed by turning on a  
circuit breaker.  Still investigating 2RR30C.  Meanwhile, CDF 
clock is back to SVT crates.  
<!-- Author: Bill Ashmanskas -->


<!-- Date: Tue Oct 17 11:19:24 2000 -->
<pre><pre>We have converted everyone back to the production database. 
We will NOT implement replication of the database at this time. 
 
We have learned 
1. The facility oracle uses to keep track of transactions before
they are written on the database (rollback) is not properly
configured in the integration instance of the database. We need
to understand if this is a problem already in the production
database or something that came about due to trying to replicate
the database to the offline.   
 
Request: we wish to take calibrations and monitor the performance
of the production database 
 
2. The time it took to copy the tables was long and can be
improved by having more processes running in the offline. We want
to try this out and then come back. 
 
3. The offline database had activity that is independent of this
test and aimed at studying the behaviour of the database in 2
years.   This was stressing the system for recovery (see 1) even
more. 
 
Actions 
1. resize the rollback in integration 
 
2. increase the number of processes used in replication on the
integration databases 
 
3. study the performance of the production database under a
calibration load and with normal running 
 
4. Do a new integration test with these new parameters and
without the offline integration being used for testing 
 
5. If successful, upgrade production. 
 
I should note that we may wish also to add the rest of the tables
to the test. 
 
Also, we learned that people do want to use things recently
entered in production (like triggers) so it would be best to do a
refresh of those  tables into integration before running 
</pre></pre>
<!-- Author: r. stdenis -->


<!-- Date: Tue Oct 17 11:48:26 2000 -->
Found that the first set of qie calibrations still ended up on integration. 
 
we have not done anything (we think) but it is now appearing on 
production as it should....  
<!-- Author: r. st. denis -->


<!-- Date: Tue Oct 17 12:26:26 2000 -->
<pre>BMU tripped.  Single channel problem; voltage lowered on that 
channel.</pre>
<!-- Author: M. Shochet -->


<!-- Date: Tue Oct 17 12:37:06 2000 -->
<pre>CMP trip.  Two channels tripped.  Reset them.</pre>
<!-- Author: M. Shochet -->
<!-- Comment: Tue Oct 17 12;38;45 comment by...M. Shochet -->
The two channels came up and held voltage OK>


<!-- Date: Tue Oct 17 12:47:08 2000 -->
<pre>Starting a 2_Jet_2_Test run (tag set 57) with the
calorimeters, muons, and COT crates 1-3 and 6-9.  Resetting of 
TDCs failed in crate 7, slot 14 and in crate 3, slot 17.  Those 
two crates are removed, and try again.</pre>
<!-- Author: M. Shochet :: (run 101646) -->
<!-- Comment: Tue Oct 17 13;09;25 comment by...M. Shochet -->
Run ended with TS suspended.  Frank rebooted it.  There were
approximately 6000 events taken.


<!-- Comment: Tue Oct 17 15;53;39 comment by...Ting Miao -->
Logged into the two crate b0cot03 and b0cot07. After the bus
resetting, TDCs in slot 17 (b0cot03) and slot 14  (b0cot07) 
came back. 
<!-- Date: Tue Oct 17 13:17:59 2000 -->
<pre>Start a run just like the last one.</pre>
<!-- Author: M. Shochet :: (run 101648) -->
<!-- Comment: Tue Oct 17 14;00;16 comment by...M. Shochet -->
Run ends with ~12000 events on trigger supervisor error.
<!-- Date: Tue Oct 17 13:49:58 2000 -->
<pre>Turned on MG-3.  It has been off since power trips yesterday.  MG-3 supplies power to CMP/CMX/IMU.  SVX doesn't use it yet.  </pre>
<!-- Author: Jonathan Lewis -->


<!-- Date: Tue Oct 17 13:58:26 2000 -->
<pre>I am now finished using b0cot10 and b0cot19. 
 
XFT crates continue to be used by experts.</pre>
<!-- Author: Jay Dittmann -->



<!-- Date: Tue Oct 17 14:34:17 2000 -->
<pre>Start another run like the other two runs.</pre>
<!-- Author: M. Shochet :: (run 101659) -->
<!-- Comment: Tue Oct 17 14;49;38 comment by...M. Shochet -->
End run after ~ 10K events to test Greg's new FRED prescale code.


<!-- Date: Tue Oct 17 15:01:41 2000 -->
<pre><strong>Greg fixed the FRED code so that a prescale of 1 is interpreted 
as 1 rather than 2.  (Thus all trigger rates will be a factor 
of 2 larger than before.)</strong>
 
A run is started just like the previous 3 (except, of course,  
that the prescales are as desired). </pre>
<!-- Author: M. Shochet :: (run 101668) -->
<!-- Comment: Tue Oct 17 15;14;51 comment by...M. Shochet -->
The run ended on a clock error.  There are an unknown number
of bad events at the end of the run.
<!-- Date: Tue Oct 17 15:04:35 2000 -->
<pre>The muon MG set is back on and the IMU is sending data up again!</pre>
<!-- Author: M. Shochet -->



<!-- Date: Tue Oct 17 15:18:04 2000 -->
<pre>Start another run just like the other ones.</pre>
<!-- Author: M. Shochet :: (run 101673) -->
<!-- Comment: Tue Oct 17 15;24;14 comment by...M. Shochet -->
End after 2K events to run calibrations.


<!-- Date: Tue Oct 17 15:37:10 2000 -->
<pre>Runninig CLC LED calibration since the beam is off at present.</pre>
<!-- Author: M. Shochet :: (run 101676) -->
<!-- Comment: Tue Oct 17 15;48;04 comment by...M. Shochet -->
The calibration run failed.  (No ack /frontend/clc/all/00)



<!-- Date: Tue Oct 17 15:54:45 2000 -->
<pre>The clock has gone into error mode several times.  It has been 
reset.</pre>
<!-- Author: M. Shochet -->


<!-- Date: Tue Oct 17 15:58:34 2000 -->
<pre>SUMMARY OF USEFUL RUNS: 
 
RUN		TRIGGER		EVENTS	  COMMENTS 
___		_______		______	  ________ 
 
 
101646		2_Jet_2_Test	~6000	  cal, muons, some COT 
 
101648		     "		~12 K	   "    "      "    " 
 
101659		     "		~10 K	   "    "      "    " 
 
101668		     "		~ 5 K	  same, with FRED prescale fix 
		This run ended with a clock failure. 
 
101673		     "          ~ 2 K      "    "     "     " 
 
 
</pre>
<!-- Author: M. Shochet -->


<!-- Date: Tue Oct 17 15:59:03 2000 -->
<font color=red>Shift Summary: </font><pre>Please use caution if you choose to put HTML into an entry.  Forgetting to use a close tag for each open tag will certainly cause problems. 
 
If you choose to use plain text, PLEASE USE CR's since this text window automatically wraps the text and gives the illusion of CR's.  (click to clear this message) 
</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   
Delivered Luminosity   0  pb-1
Acquired Luminosity    0  pb-1
Efficiency             
   
</td></tr></table></pre>
<!-- Author: M. Shochet -->


<!-- Date: Tue Oct 17 16:12:34 2000 -->
<pre>While working with the cot crates I noticed 
 
b0cot03 slot 17 dropped out but can be recovered using bus reset. 
b0cot07 slot 14 dropped out but can be recovered using bus reset. 
 
b0cot12 slot 16 dropped out could NOT be recovered with bus 
reset. For this crate you will have to use local done mode. 
 
 
</pre> <font color=red><cite>(Entry started on Tue Oct 17 16:12:31 2000)</cite></font>
<!-- Author: Frank Chlebana -->


<!-- Date: Tue Oct 17 16:53:47 2000 -->
<pre><br><b>Masked Calorimeter Towers:</b> 
<br>Masked the following calorimeter trigger towers for the 
reasons given below: 
<pre> 
eta  phi  Cal     Reason 
--------------------------------------------------- 
 16   23  HAD   WHA PMTs for tower 23E-9(both) is hot 
 15   20  HAD   Trigger Tower is hot, but PMTs not (??) 
 15   12  HAD   WHA PMT for tower 12E-7, PMT=1 is hot 
</pre></pre><font color=red><cite>(Entry started on Tue Oct 17 16:53:43 2000)</cite></font>
<!-- Author: R.J. Tesarek -->