<!-- Title: Day shift. Sat Aug 11, 2007 -->
<!-- SciCo: Peter Wittich -->
<!-- DAQAce: Petteri Mehtala -->
<!-- MonAce: &nbsp; -->
<!-- CO: &nbsp; -->
<!-- OpManager: Steve Hahn &nbsp; -->
<!-- Notes: -->
<pre>Quiet shift expected.  No access, no beam, no nothin'.</pre>





<!-- Date: Sat Aug 11 08:30:25.458 2007 -->
<pre>Running L2TORTURE. 
 
HVs off: COT, TOF, L00, SVX, ISL, clc, cpr, mnp, bcs 
HVs standby: CMU, CMP, CMX, BMU, CES, PEM, PHA, PSH 
HVs on: CSW, MSK, CSX, CSP, BSU, TSU, CEM, CHA, WHA, GAM 
 
 
Alarms and warnings from iFix: 
 
Plug temperature shows that most of the west plug temperatures are between 70 and 73 degrees. Eastplug PE09 and PE11 are low too: 73 and 71 degrees, respectively. (Normally 77 degrees.) 
 
L00 yellow in iFix due to West Vacuum Jacket Temp, it is 16.2 degC, bit lower than the warning lowerlimit 17 degC. 
 
SVX yellow in iFix due to Vacuum Jacket Temp for BHW1, BHW2, BHW3 and PCW, temperatures are 16.6 to 16.7 degC (17degC is the lower limit for warning). 
 
CHA PMT 22_02_LNW has HV Set 1523.5 and HV Diff 331. 
 
WHA PMT East Wedges 6-8 have HV Set to 900 to 1000 and HV Diff is around -30 to -40. WHA_03_09_RE has Set value 851 and HV Diff -5. 
 
Many of the racks having HB alarm or red, seems that VOLTMAN was not running properly. Hit the RUN button on it to see if it helps (at least the crates started to change colors now on VOLTMAN..) 
 
</pre>
<!-- Author: Petteri :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246823' target=_top>246823</a>) -->
<!-- Comment: Sat Aug 11 08;35;32 comment by...peter -->
paged SVX pager about slight low temp issue.

<!-- Comment: Sat Aug 11 08;50;59 comment by...Steve Hahn -->
<p>Nice summary, Petteri.</p>

<p>I suspect the low plug temperatures are due to the attempt by the process
system techs to get the humidity as low as possible in the collision hall this
weekend.  This in turn implies the temperatures will probably be lower than
usual, and the heating resistors in the plug can only add so much heat per unit
time.  You could check the temperatures and humidity by looking at the HVAC 
historical plots.  This is all just a guess on my part.</p>

<p>The CHA PMT is a known problem; I have notified Fotis.</p>

<p>The WHA PMTs problem sounds like the TSW-U Pisabox has decided to
lose its mind again.  I power cycled it once the IMUs were pulled Friday morning,
and it seemed to be behaving then, but this sounds like exactly the same 
problem.  It may be related to the COSWT-3 work.  I'll fix it again Monday.</p>

<p>VOLTMAN is running exceedingly slowly, due to the fact there are so many
racks with problems.  For each such rack, VOLTMAN tries an extremely large 
number of reads which bogs down the readout.  Dervin has decided not to fix
these problems given that we are replacing the readout cables in this shutdown
and his crew has jobs in the collision hall to stay on schedule.</p>

<!-- Comment: Sat Aug 11 09;57;53 comment by...Ulrich -->
I guess Steve is right about the collision hall temperature. I've never seen going vacuum jacket temperatures going out of range before. So either their are broken (in which case I would expect a much larger change in temperature) or the environment has changed. So as long as it is only a slight deviation and it is correlated with nearby temperatures (plug), just keep an eye on it.


<!-- Date: Sat Aug 11 10:07:39.371 2007 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2007&action=view&page=-6409&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2007&action=view&page=-6410  width='300' align='top'></a></td><td>The plug temprature plot for PW12. 
 
I checked the historical plots for collision hall temperature and humidity: the change was done at 5pm yesterday and they have been stable soon after the change. Comparing with the PTM plot, they seem not to be related to each other. There is a visible change in the temprature behaviour of the plug at last night around 2am-3am.</td></tr></table>
<!-- Author: Petteri -->
<!-- Comment: Sat Aug 11 10;33;47 comment by...Ulrich -->
So maybe the Silicon and the plug temperature problems are not related. You can find a historical plot of the vacuum jacket temperatures in the 
<a href=
'http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=silicon&action=view&page=365&frame=2&anchor=10:31:44'>
silicon e-log</a>. They are in line with the humidity reduction attempt yesterday around 17:00.

<!-- Comment: Sat Aug 11 10;49;46 comment by...Petteri -->
Yes, seems to be like that. 
<br><br>
<a href='/cgi/elog/elog.pl?nb=2007&action=view&page=-6411&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2007&action=view&page=-6412' ></a><br><br>

Looking the variable Pct_Heat for PW12 it seems that after 17:00 yesterday, it have been in quite high values and around 3am today, it just dropped to zero. Maybe it is somekind of heater problem. <br><br>

Tried to call David Clark, but he didn't answer, I guess it is too early..

<!-- Comment: Sat Aug 11 13;02;26 comment by...Petteri -->
Sent email to experts.
<!-- Comment: Sun Aug 12 11;35;45 comment by...Petteri -->
The temperatures came back to ok when rebooted the monitoring PC. See <a href=
'/cgi/elog/elog.pl?nb=2007&action=view&page=669&frame=2&anchor=11:13:11.104'>
Sunday shift elog. </a> 
<!-- Date: Sat Aug 11 11:51:14.573 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246823>246823</A>
TERMINATE: to do calibrations
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 11:54:29.789 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246824>246824</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 11:54:31.836 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246824>246824</A>
TERMINATE: done
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 11:54:32.437 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246824>246824</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 11 12;55;37 comment by...Petteri -->
bad channels on WHA
<!-- Date: Sat Aug 11 11:55:20.023 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246825>246825</A>
ACTIVATE: RunSet: CALIB_CLC_QIE 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 11:56:38.803 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246825>246825</A>
CALIBRATION: <U><B>clcqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#clcqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 11 12;56;30 comment by...Petteri -->
ok
<!-- Date: Sat Aug 11 11:56:39.210 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246825>246825</A>
TERMINATE: done
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 11:58:18.492 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246826>246826</A>
ACTIVATE: RunSet: CALIB_QIE_FWD 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 12:00:24.580 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246826>246826</A>
CALIBRATION: <U><B>bscqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#bscqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 11 12;57;09 comment by...Petteri -->
ok (east is out)
<!-- Date: Sat Aug 11 12:00:25.354 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246826>246826</A>
TERMINATE: done
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 12:00:41.742 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246827>246827</A>
 Activated at 2007.08.11 12:00:27
<!-- Author: RunControl -->


<!-- Date: Sat Aug 11 12:00:42.786 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246827>246827</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [7,792,632] 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 12:04:54.997 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246827>246827</A>
 Terminated at 2007.08.11 12:03:38
<!-- Author: RunControl -->


<!-- Date: Sat Aug 11 12:04:55.499 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246827>246827</A>
TERMINATE: done, L1 rate 162Hz.
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 12:11:11.283 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246828>246828</A>
ACTIVATE: RunSet: TOF_TAC_CALIB 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 12:12:24.120 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246828>246828</A>
TERMINATE: done
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 12:12:25.970 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246828>246828</A>
CALIBRATION: <U><B>toftac</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#toftac target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 11 12;58;13 comment by...Petteri -->
dead channels
<!-- Date: Sat Aug 11 12:13:25.561 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246829>246829</A>
ACTIVATE: RunSet: ACE_TOF_QIE_NEW 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 12:20:57.853 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246829>246829</A>
CALIBRATION: <U><B>tofqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#tofqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 11 12;59;15 comment by...Petteri -->
empty channels
<!-- Date: Sat Aug 11 12:20:58.264 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246829>246829</A>
TERMINATE: done
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 12:20:59.669 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246830>246830</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 12:21:00.874 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246830>246830</A>
TERMINATE: done
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 12:21:01.872 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246831>246831</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 12:21:02.220 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246831>246831</A>
TERMINATE: done
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 12:23:12.378 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246831>246831</A>
CALIBRATION: <U><B>cemled</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cemled target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 11 13;00;46 comment by...Petteri -->
dead channels
<!-- Date: Sat Aug 11 12:23:13.053 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246830>246830</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 11 13;01;13 comment by...Petteri -->
ok
<!-- Date: Sat Aug 11 12:23:14.292 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246832>246832</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 12:23:15.553 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246832>246832</A>
TERMINATE: done
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 12:25:31.945 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246832>246832</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 11 13;01;56 comment by...Petteri -->
ok
<!-- Date: Sat Aug 11 12:29:26.684 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246833>246833</A>
ACTIVATE: RunSet: XEF_ALL
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 12:29:27.845 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246833>246833</A>
CALIBRATION: <U><B>calxef</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calxef target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 11 12;54;22 comment by...Petteri -->
<pre>To me it is seems that the CAEN crates for the flashers does not turn on.
I pressed the start button, busy light comes on. The CAEN crates show 2850 immediately on the top of the crates. The value in the box after VM stays at 0000. The crates show channels 23 and 27, and have led on the point saying channel off. 

The calibration plot is also showing almost no counts at all except one hot tower.

I could turn the channels on on CAEN crates, but since I don't know if they are turned off on purpose, I leave them like that. </pre>
<!-- Date: Sat Aug 11 12:29:28.921 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246833>246833</A>
TERMINATE: done
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 12:35:32.764 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246834>246834</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 12:35:33.334 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246834>246834</A>
TERMINATE: done
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 12:41:38.224 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246835>246835</A>
 Activated at 2007.08.11 12:41:08
<!-- Author: RunControl -->


<!-- Date: Sat Aug 11 12:41:39.837 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246835>246835</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,927,674] 
<!-- Author: Petteri -->













<!-- Date: Sat Aug 11 14:14:38.304 2007 -->
still here, all is quiet, running cosmics...
<!-- Author: peter :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246835' target=_top>246835</a>) -->


<!-- Date: Sat Aug 11 14:57:45.815 2007 -->
<font color=red>Shift Summary: </font><pre>Slow shift today. 
 
Came in to find run from last night ongoing. Were running L2Torture, 
 
Petteri gave a nice summary at the beginning of the shift as to what 
state we found the detector in.  Some weirdness with temperature 
shifts during the night at 3 am -not sure where that came from. 
 
No excitement during the shift. Did a full suite of calibrations,  
then ran cosmics.  
</pre> 
<!-- Author: Peter Wittich -->

<!-- Date: Sat Aug 11 15:01:13.391 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246835>246835</A>
 Terminated at 2007.08.11 15:00:49
<!-- Author: RunControl -->


<!-- Date: Sat Aug 11 15:01:14.812 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246835>246835</A>
TERMINATE: End for Calibration
<!-- Author: Petteri -->



<!-- Date: Sat Aug 11 15:01:49.768 2007 -->
<font color=red>Shift Summary: </font><pre>Try to end shift early?</pre> 
<!-- Author: Kevin -->
<!-- Comment: Sat Aug 11 15;02;22 comment by...Kevin Lannon -->
Apparently, I can't start the evening shift until the appointed time?  Hmmmm . . . .

<!-- Comment: Sat Aug 11 15;07;07 comment by...K.Lannon -->
Ah-ha!  From the e-log help page:
<p>Presents the user with a form to start the current shift. The application is aware of the time of day and the operations schedule. This button is dimmed until the current shift expires. Shift personnel's names are required.
<p> Well, Ye Li and I have started our shift at 3pm.  I'll start the new shift log in one hour.
<!-- Date: Sat Aug 11 15:04:32.205 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246836>246836</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 15:04:33.566 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246836>246836</A>
TERMINATE: done
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 15:05:16.536 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246836>246836</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 11 15;08;04 comment by...Ye Li -->
WHA all dead, I think it's because they have been blocked to readout by experts. Everything should be okay, although I'll mark bad on the board
<!-- Date: Sat Aug 11 15:06:04.722 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246837>246837</A>
ACTIVATE: RunSet: CALIB_CLC_QIE 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 15:06:35.617 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246837>246837</A>
CALIBRATION: <U><B>clcqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#clcqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 11 15;12;13 comment by...Ye Li -->
Generally the same as yesterday: Slope/SloErr fluctuates quite obviously for RMS and Mean for different channels. Marked okay.
<!-- Date: Sat Aug 11 15:09:14.376 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246837>246837</A>
TERMINATE: done
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 15:12:10.815 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246838>246838</A>
CALIBRATION: <U><B>bscqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#bscqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 11 15;13;14 comment by...Ye Li -->
The same as yesterday: east tower all off. I will mark it bad
<!-- Date: Sat Aug 11 15:12:48.721 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246838>246838</A>
ACTIVATE: RunSet: CALIB_QIE_FWD 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 15:12:49.092 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246838>246838</A>
TERMINATE: done
<!-- Author: Petteri -->



<!-- Date: Sat Aug 11 15:15:23.391 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246839>246839</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 15:25:37.856 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246839>246839</A>
TERMINATE: done
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 15:25:38.674 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246840>246840</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 15:25:39.239 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246840>246840</A>
TERMINATE: done
<!-- Author: Petteri -->


<!-- Date: Sat Aug 11 15:25:42.972 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246840>246840</A>
CALIBRATION: <U><B>pluglaser</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#pluglaser target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->


<!-- Date: Sat Aug 11 15:25:42.268 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246840>246840</A>
CALIBRATION: <U><B>pluglaser</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#pluglaser target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 11 15;27;50 comment by...Ye Li -->
the same as yesterday: TDC mean shifted to higher value. I'll mark it okay
<!-- Date: Sat Aug 11 15:27:02.761 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246841>246841</A>
ACTIVATE: RunSet: BOTHPLUGS_MM 
<!-- Author: Ye -->



<!-- Date: Sat Aug 11 15:29:56.442 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246841>246841</A>
TERMINATE: done
<!-- Author: Ye -->


<!-- Date: Sat Aug 11 15:31:57.990 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246842>246842</A>
ACTIVATE: RunSet: ACE_TOF_QIE_NEW 
<!-- Author: Ye -->


<!-- Date: Sat Aug 11 15:31:58.928 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246842>246842</A>
CALIBRATION: <U><B>tofqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#tofqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 11 15;34;17 comment by...Ye Li -->
Different from the reference for wcal03 for all plot: seems to be off. Marked bad
<!-- Date: Sat Aug 11 15:31:59.284 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246842>246842</A>
TERMINATE: done
<!-- Author: Ye -->


<!-- Date: Sat Aug 11 15:33:17.028 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246843>246843</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Ye -->



<!-- Date: Sat Aug 11 15:35:14.059 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246843>246843</A>
TERMINATE: done
<!-- Author: Ye -->


<!-- Date: Sat Aug 11 15:37:20.853 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246843>246843</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 11 15;43;56 comment by...Ye Li -->
okay
<!-- Date: Sat Aug 11 15:38:22.604 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246844>246844</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: Ye -->


<!-- Date: Sat Aug 11 15:41:32.391 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246844>246844</A>
TERMINATE: done
<!-- Author: Ye -->


<!-- Date: Sat Aug 11 15:41:33.463 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246844>246844</A>
CALIBRATION: <U><B>cemled</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cemled target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 11 15;46;50 comment by...Ye Li -->
West Wedge 11 and 12 ADC counts 0, possible dead channel (the same as yesterday)
<!-- Date: Sat Aug 11 15:41:34.935 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246845>246845</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Ye -->


<!-- Date: Sat Aug 11 15:41:35.004 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246845>246845</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 11 15;47;42 comment by...Ye Li -->
okay
<!-- Date: Sat Aug 11 15:41:36.288 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246845>246845</A>
TERMINATE: done
<!-- Author: Ye -->


<!-- Date: Sat Aug 11 15:43:05.340 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246846>246846</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Ye -->


<!-- Date: Sat Aug 11 15:43:20.722 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246846>246846</A>
TERMINATE: done
<!-- Author: Ye -->





<!-- Date: Sat Aug 11 15:51:46.097 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246847>246847</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: Ye -->


<!-- Date: Sat Aug 11 15:52:28.124 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246847>246847</A>
CALIBRATION: <U><B>calxef</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calxef target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 11 15;54;49 comment by...Ye Li -->
bad as previous calibration
<!-- Date: Sat Aug 11 15:53:40.860 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246847>246847</A>
TERMINATE: done
<!-- Author: Ye -->



<!-- Date: Sat Aug 11 15:56:55.303 2007 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>PhysicsTable</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L3Accepts</TH>
<TH>DelivLumi nb<SUP>-1</SUP></TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>15:55:01</TD>
<TD ALIGN=CENTER>::</TD>
<TD ALIGN=RIGHT></TD>               <TD ALIGN=RIGHT></TD>
<TD ALIGN=RIGHT></TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->


<!-- Date: Sat Aug 11 16:00:18.266 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246848>246848</A>
ACTIVATE: RunSet: TOF_TAC_CALIB 
<!-- Author: Ye -->



<!-- Date: Sat Aug 11 16:00:59.348 2007 -->
<font color=red>Shift Summary: </font><pre>Try to start the evening shift log again.</pre> 
<!-- Author: Kevin -->


<!-- Date: Sat Aug 11 16:02:17.485 2007 -->
<font color=red>Shift Summary: </font><pre>?</pre> 
<!-- Author: Kevin -->

<!-- Date: Sat Aug 11 16:03:04.874 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246848>246848</A>
TERMINATE: done
<!-- Author: Ye -->


<!-- Date: Sat Aug 11 16:03:54.660 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=246848>246848</A>
CALIBRATION: <U><B>toftac</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#toftac target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sat Aug 11 16;06;28 comment by...Ye Li -->
wcal00 and wcal03 seemed to be off. Everything else generally okay.