<!-- Title: 14:31:54  Wed Jun 26 2002 -->
<!-- Crewchief: COT Crew -->
<!-- Op1: &nbsp; -->
<!-- Op2: &nbsp; -->
<!-- Op3: &nbsp; -->
<!-- Op4: &nbsp; -->
<!-- Op5: &nbsp; -->
<!-- Op6: &nbsp; -->
<!-- Notes: -->
COT validation, etc. Post-shutdown


<!-- Date: Wed Jun 26 14:33:49 2002 -->
<pre>Before starting on validation of recent data, here is Bob's e-mail 
about validation with YMon: 
 
================================================================== 
Date: Tue, 18 Jun 2002 13:24:59 -0500 
From: Bob Wagner &lt;wagner@fnal.gov&gt; 
Subject: COT YMon status after shutdown 
 
Hi, 
 I looked at YMon for one of the first runs after the access (146759).  The 
only really new obvious problem is SL6 cell 250 chs 0-7 in quad C.  These 
have roughly normal occupancy, but the multiplicity is about a factor of 2 
high (mult &gt; 2.2) and the average time is about 40 ns high, indicating 
afterpulsing.  This is likely due to a Faraday cage being knocked loose 
during the silicon work (a SL5 Faraday cage nearby was completely knocked 
off).  An alternative explanation is a poorly seated cable at the output of 
the Repeater or input to the TDC.  If one gets the chance, the cable 
seatings could be checked during an appropriate access. 
 
 Hot channels that stick out in YMon and were there (at least some of the 
time) before the access: 
SL1-103-6: Occ = 1, Mult = about normal 
SL2-31-3: Occ=1, Mult = 8 
SL2-46-11: Occ =1, Mult = 8 
SL3-55-10: Occ=1, Mult = 8 
SL8-275-most channels: Occ = x2ish normal, Mult = about normal. 
                                                                 Bob 
 
 
</pre>
<!-- Author: K Burkett -->


<!-- Date: Wed Jun 26 16:52:37 2002 -->
<H2> Checks of data from store 1464 </H2> 
<br> 
The following runs were checked: 
<br> 
<TABLE BORDER=2 CELLSPACING=2 CELLPADDING=2 BGCOLOR="#FFFFFF" ALIGN=CENTER NOSAVE> 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> Run Number </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> Luminosity </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> Stage0 Evts </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 147160 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1>  2.3 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 1142 </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 147165 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 29.4 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 12738 </FONT></TH> 
</TR> 
 
</TABLE></BR> 
 
 
<H4> Residuals: </H4> 
 
The mean of the GoodHits residuals and the RMS of both AllHits and GoodHits 
residuals are listed below for run 147165. 
<br><br> 
 
<TABLE BORDER=2 CELLSPACING=2 CELLPADDING=2 BGCOLOR="#FFFFFF" ALIGN=CENTER NOSAVE> 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> Superlayer </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> All Hits RMS </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> Good Hits RMS </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> Good Hits Mean </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 1 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 281 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 231 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 65.3 </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 2 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 354 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 269 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 63.9 </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 3 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 260 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 215 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 61.5 </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 4 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 293 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 217 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 72.4 </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 5 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 248 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 186 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 68.3 </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 6 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 255 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 184 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 58.7 </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 7 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 232 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 169 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 68.2 </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 8 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 247 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 172 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 64.6 </FONT></TH> 
</TR> 
 
</TABLE></BR> 
The fact that the mean of the GoodHits residuals is 60-70 um 
indicates that t0 shifted by slightly more than 1 ns during 
the shutdown.  This should have been fixed by the fit that was 
done at the end of run 147165 and written to the database.<br> 
 
<H4> I won't compare to any reference run, but instead insert here 
the HitCells and UsedCells canvases </H4> 
 
<a href='http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=cot&action=view&page=-67&button=yes' target='xxx' onclick='jswindow()'><img 
             src='http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=cot&action=view&page=-67' width='300'></a> 
        </li>  
 
<a href='http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=cot&action=view&page=-68&button=yes' target='xxx' onclick='jswindow()'><img 
             src='http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=cot&action=view&page=-68' width='300'></a> 
        </li>  
 
<br><br><br> 
 
Hit cells that are 10% above or below the average: 
<br><br> 
 
 
<TABLE BORDER=2 CELLSPACING=2 CELLPADDING=2 BGCOLOR="#FFFFFF" ALIGN=CENTER width="75%" NOSAVE> 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> Superlayer </TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> High HitCells</TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> Low HitCells</TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 1 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 103, 119 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> None </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 2 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 31, 46, 74 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 158 </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 3 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 55, 142 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 2, 126, 150 </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 4 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 94 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 227, 254 </FONT></TH> 
</TR>  
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 5 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 17, 100, 219, 236, 246 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> None </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 6 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 9, 165 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 69 </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 7 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 122, 123, 136, 190, 191, 273, 340, 369 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 286, 370, 371 </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 8 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1>  
 120, 129, 138, 148, 149, 152, 153, 154, 155, 203, 205, 207, 217, 275, 284,  
285, 290, 372, 373, 
 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 59, 72, 322, 445, 479 </FONT></TH> 
</TR> 
 
</TABLE> 
<br><br> 
In the UsedCells plots, the dead cell at SL3-2 shows up, with some 
effect on its neighbors.  <br> SL4-227 also shows up at ~50%.  
<!-- Author: K Burkett -->


<!-- Date: Mon Jul 1 00:48:23 2002 -->
<pre>Changes to HV control (on COT2): 
 
The "Local" check-box has been removed from the "Hv Control" window -- local (i.e., at COT2) control is always enabled.  As far as I know, nobody un-checked it. 
 
A "Solenoid Interlock" check-box has been added.  If checked (default), everything works as before.  Un-checking is similar to un-plugging the interlock cable (which is what we've done in the past to go above 50% without the solenoid).  Changing the state of this box requires a password, same as for iFix control. 
 
The check-box updating has been fixed.  (Previously, if you didn't enter the right password, the check-box display would change state even though in fact the setting had not changed.) 
 
The current readout for each channel is preceded by  pseudo-readout of ground.  This is to discharge the S&H amp (the ADC value is never actually read).  So we no longer have the odd pattern of potential currents being artificially high.  And hopefully now we can actually correlate current to luminosity.  (We finally got a store over the weekend to see how it works, and it seems pretty good.)</pre>
<!-- Author: Aseet Mukherjee -->


<!-- Date: Mon Jul 1 10:51:19 2002 -->
 
<H2> Checks of data from store 1482 </H2> 
<br> 
The following runs were checked: 
<br> 
<TABLE BORDER=2 CELLSPACING=2 CELLPADDING=2 BGCOLOR="#FFFFFF" ALIGN=CENTER NOSAVE> 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> Run Number </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> Luminosity </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> Stage0 Evts </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> Comments </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 147483 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 23.1 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 8217 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> - </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 147496 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1>  8.7 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 1900 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> - </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 147507 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1>  6.9 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 4659 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> XFT test run </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 147509 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1>  6.8 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 4882 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> XFT test run </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 147511 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1>  5.1 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 4104 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> XFT test run </FONT></TH> 
</TR> 
 
</TABLE></BR> 
 
 
<H4> Residuals: </H4> 
 
The mean of the GoodHits residuals is shifted back to close to zero, as 
it should be if t0 is correct.  Below are the RMS of both AllHits and GoodHits 
residuals and the mean of the GoodHits for run 147483.  Other than the 
expected shift of the mean, the values are consistent with past runs. 
<br><br> 
 
<TABLE BORDER=2 CELLSPACING=2 CELLPADDING=2 BGCOLOR="#FFFFFF" ALIGN=CENTER NOSAVE> 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> Superlayer </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> All Hits RMS </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> Good Hits RMS </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> Good Hits Mean </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 1 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 277 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 229 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 3.5 </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 2 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 351 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 270 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 8.6 </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 3 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 258 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 214 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 5.4 </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 4 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 291 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 219 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> -0.8 </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 5 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 246 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 184 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 0.6 </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 6 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 257 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 185 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 3.8 </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 7 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 230 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 166 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 4.0 </FONT></TH> 
</TR> 
 
<TR NOSAVE> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 8 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 246 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> 169 </FONT></TH> 
<TH ALIGN=CENTER ><FONT SIZE=-1> -0.3 </FONT></TH> 
</TR> 
 
</TABLE></BR> 
 
<H4> Run 147165 was used as a reference run </H4> 
Other than the shift in the mean of the residuals noted above, 
the runs are generally consistent.<br> 
There is a strange asymmetry near z=0 in the track Z0 distribution for  
run 147511 that did not appear in earlier runs.<br> 
<a href='http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=cot&action=view&page=-69&button=yes' target='xxx' onclick='jswindow()'><img 
             src='http://www-cdfonline.fnal.gov/cgi/elog/det-system-elog.pl?nb=cot&action=view&page=-69' width='300'></a> 
        </li>  
 
<br><br><br> 
 
Run 147483 could be a new reference 
run until we get a longer good run.<br><br>  
<!-- Author: K Burkett -->


<!-- Date: Tue Jul 2 03:29:59 2002 -->
<pre>Called for low voltage alarms.  All power to LSW of chamber is off, but VICORs are on.  In particular, M5_ILIM_LSW and P5_ILIM_LSW are off.  These run the "active fuse" circuit, and (I think) without it everything else shuts off.  D4_BUFFBD_LSW is also off, not sure what this means, thought this was an independent supply. 
 
Can't think of anything else to do without an access so had the cryo tech cycle power on the West endwall.  Didn't help, LSW is still off.  Asked psycho to request an access ASAP.</pre>
<!-- Author: Aseet Mukherjee -->


<!-- Date: Tue Jul 2 13:58:17 2002 -->
<pre>Conclusion to previous ... 120VAC fuse to the Condor 5V supplies was blown, as expected.  Replaced it (with a 4A fuse, same as the spare upstairs).  But the -5 ILIM supply still didn't work (the other three did).  Replaced the supply, now all is well. 
 
Replacing the supply is a lot harder than perhaps it should be.  If we get time and people for it, might be worth while to modify these boxes to use connections which do not require soldering.</pre>
<!-- Author: Aseet Mukherjee -->


<!-- Date: Sat Jul 6 13:09:32 2002 -->
<pre>  
        Today I was paged because some plots "turned red" in 
        Ymon - the affected sl's were 3, 6, 7 and 8. 
 
        I don't understand why they suddenly got high occupancies 
        around 11:45, since the luminosity had been dropping 
        since the morning, and the losses were not high. 
 
        I can't tell the exact cell number from looking at 
        the plots, but each high number seems to be very close 
        to cell numbers that were marked as "high hit cells" 
        by Kevin in the COT e-log. (Jun 26, 16:52) 
 
        The plots which prompted the page today are in the 
        consumer e-log (Sat July 6, 11:45) 
 
        From looking at these plots (as I said,it is difficult to 
        see the exact channel number, it looks the the high 
        occupancies are in 
 
        sl3 cell (~50) 
 
        sl6 cell (~10) 
 
        sl7 cell (~120, ~370) 
 
        sl8 cell (~275) 
 
        (My error in reading the number is probably +-5) These are close 
        to the cells marked as high occupancy by kevin (though he did 
        mark other ones that I did not see here) 
 
        sl3 cell 55 
        sl6 cell 9 
        sl7 cell 122, 123 and 369 
        sl8 cell 275 
 
        
</pre>
<!-- Author: Robyn Madrak -->


<!-- Date: Fri Jul 12 18:41:07 2002 -->
<h3>COT validation for the week of July 7 - July 14, 2002</h3> 
 The runs I am checking are below. I selected the runs by going to the run  
summary page and requiring a minium of 10nb^-1 of luminosity, and good physics  
running as determined by the "good run bit" from RC. All runs have the physics  
trigger table, are physics runs, and data type "beamdata." The slightly edited  
run summary is below.<br> 
 <br> 
 <br> 
      
<table border="2" cellpadding="5" width="50%" align="center"> 
  <tbody> 
     <tr> 
  <td><b>N_RUNS</b></td> 
  <td width="50%"><b>SUM_LUMI</b></td> 
  <td><b>AVG_LUMI</b></td> 
  <td><b>MAX_LUMI</b></td> 
  <td><b>MIN_LUMI</b></td> 
     </tr> 
  <tr> 
  <td>21</td> 
  <td>       1,348.30 nb-1</td> 
  <td>          64.20 nb-1</td> 
  <td>         288.38 nb-1</td> 
  <td>          10.97 nb-1</td> 
     </tr> 
      
  </tbody>  
</table> 
 <br> 
 The complete list of runs is below, with the links to the run summaries. 
&nbsp;Somewhat annoyingly, these are ordered in inverse chronological order. 
Go figure. &nbsp;I also added a Stage0 column, which indicates whether or 
not we found the Stage0 root file. Why don't we <i>always</i> find that file?<br> 
 <br> 
      
<table border="2" cellpadding="5" width="85%" bgcolor="#ffffff" 
 align="center"> 
  <tbody> 
     <tr> 
  <td><b>RUN</b></td> 
  <td><b>DAY</b></td> 
  <td><b>ACTIVATE</b></td> 
        <td><b>LIVELUMI</b></td> 
         <td><b>LIVETIME</b></td> 
       <td valign="middle" align="center" width="5%">Stage0?<br> 
       </td> 
  </tr> 
  <tr> 
  <td><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148157">148157  
(0x242BD)</a></td> 
  <td>2002.07.12</td> 
  <td>05:44:44</td> 
        <td>            47 nb-1</td> 
         <td align="right"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/PlotTime?TABLE=LEVEL1_CONDITIONS&amp;X=FRONTENDTIME&amp;Y=LIVETIME&amp;WHERE=%27RC_RUNNUMBER=148157%27&amp;TITLE=LIVETIME_Run_148157&amp;XLABEL=%27seconds%27&amp;YLABEL=%27counts%27"> 
 02:48:13.3</a></td> 
       <td valign="middle" align="center" width="5%"><br> 
       </td> 
  </tr> 
  <tr> 
  <td><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148154">148154  
(0x242BA)</a></td> 
  <td>2002.07.11</td> 
  <td>18:17:55</td> 
        <td>           288 nb-1</td> 
         <td align="right"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/PlotTime?TABLE=LEVEL1_CONDITIONS&amp;X=FRONTENDTIME&amp;Y=LIVETIME&amp;WHERE=%27RC_RUNNUMBER=148154%27&amp;TITLE=LIVETIME_Run_148154&amp;XLABEL=%27seconds%27&amp;YLABEL=%27counts%27"> 
 10:32:06.9</a></td> 
       <td valign="middle" align="center" width="5%">1<br> 
       </td> 
  </tr> 
  <tr> 
  <td><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148153">148153  
(0x242B9)</a></td> 
  <td>2002.07.11</td> 
  <td>16:04:21</td> 
        <td>            94 nb-1</td> 
         <td align="right"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/PlotTime?TABLE=LEVEL1_CONDITIONS&amp;X=FRONTENDTIME&amp;Y=LIVETIME&amp;WHERE=%27RC_RUNNUMBER=148153%27&amp;TITLE=LIVETIME_Run_148153&amp;XLABEL=%27seconds%27&amp;YLABEL=%27counts%27"> 
 02:06:12.1</a></td> 
       <td valign="middle" align="center" width="5%">1<br> 
       </td> 
  </tr> 
  <tr> 
  <td><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN_NUMBER=148152">148152  
(0x242B8)</a></td> 
  <td>2002.07.11</td> 
  <td>15:18:40</td> 
        <td>            24 nb-1</td> 
         <td align="right"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/PlotTime?TABLE=LEVEL1_CONDITIONS&amp;X=FRONTENDTIME&amp;Y=LIVETIME&amp;WHERE=%27RC_RUNNUMBER=148152%27&amp;TITLE=LIVETIME_Run_148152&amp;XLABEL=%27seconds%27&amp;YLABEL=%27counts%27"> 
 00:26:36.8</a></td> 
       <td valign="middle" align="center" width="5%">1<br> 
       </td> 
  </tr> 
  <tr> 
  <td><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN_NUMBER=147874">147874  
(0x241A2)</a></td> 
  <td>2002.07.08</td> 
  <td>18:04:23</td> 
        <td>            32 nb-1</td> 
         <td align="right"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/PlotTime?TABLE=LEVEL1_CONDITIONS&amp;X=FRONTENDTIME&amp;Y=LIVETIME&amp;WHERE=%27RC_RUNNUMBER=147874%27&amp;TITLE=LIVETIME_Run_147874&amp;XLABEL=%27seconds%27&amp;YLABEL=%27counts%27"> 
 01:30:48.5</a></td> 
       <td valign="middle" align="center" width="5%">1<br> 
       </td> 
  </tr> 
  <tr> 
  <td><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN_NUMBER=147873">147873  
(0x241A1)</a></td> 
  <td>2002.07.08</td> 
  <td>16:57:31</td> 
        <td>            22 nb-1</td> 
         <td align="right"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/PlotTime?TABLE=LEVEL1_CONDITIONS&amp;X=FRONTENDTIME&amp;Y=LIVETIME&amp;WHERE=%27RC_RUNNUMBER=147873%27&amp;TITLE=LIVETIME_Run_147873&amp;XLABEL=%27seconds%27&amp;YLABEL=%27counts%27"> 
 00:57:10.1</a></td> 
       <td valign="middle" align="center" width="5%">0<br> 
       </td> 
  </tr> 
  <tr> 
  <td><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN_NUMBER=147869">147869  
(0x2419D)</a></td> 
  <td>2002.07.08</td> 
  <td>06:59:03</td> 
        <td>           272 nb-1</td> 
         <td align="right"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/PlotTime?TABLE=LEVEL1_CONDITIONS&amp;X=FRONTENDTIME&amp;Y=LIVETIME&amp;WHERE=%27RC_RUNNUMBER=147869%27&amp;TITLE=LIVETIME_Run_147869&amp;XLABEL=%27seconds%27&amp;YLABEL=%27counts%27"> 
 08:38:48.7</a></td> 
       <td valign="middle" align="center" width="5%">1<br> 
       </td> 
  </tr> 
  <tr> 
  <td><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN_NUMBER=147867">147867  
(0x2419B)</a></td> 
  <td>2002.07.08</td> 
  <td>06:24:28</td> 
        <td>            11 nb-1</td> 
         <td align="right"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/PlotTime?TABLE=LEVEL1_CONDITIONS&amp;X=FRONTENDTIME&amp;Y=LIVETIME&amp;WHERE=%27RC_RUNNUMBER=147867%27&amp;TITLE=LIVETIME_Run_147867&amp;XLABEL=%27seconds%27&amp;YLABEL=%27counts%27"> 
 00:16:37.9</a></td> 
       <td valign="middle" align="center" width="5%">1<br> 
       </td> 
  </tr> 
  <tr> 
  <td><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN_NUMBER=147866">147866  
(0x2419A)</a></td> 
  <td>2002.07.08</td> 
  <td>04:18:13</td> 
        <td>            74 nb-1</td> 
         <td align="right"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/PlotTime?TABLE=LEVEL1_CONDITIONS&amp;X=FRONTENDTIME&amp;Y=LIVETIME&amp;WHERE=%27RC_RUNNUMBER=147866%27&amp;TITLE=LIVETIME_Run_147866&amp;XLABEL=%27seconds%27&amp;YLABEL=%27counts%27"> 
 01:37:10.2</a></td> 
       <td valign="middle" align="center" width="5%">1<br> 
       </td> 
  </tr> 
  <tr> 
  <td><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN_NUMBER=147865">147865  
(0x24199)</a></td> 
  <td>2002.07.08</td> 
  <td>01:29:28</td> 
        <td>           108 nb-1</td> 
         <td align="right"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/PlotTime?TABLE=LEVEL1_CONDITIONS&amp;X=FRONTENDTIME&amp;Y=LIVETIME&amp;WHERE=%27RC_RUNNUMBER=147865%27&amp;TITLE=LIVETIME_Run_147865&amp;XLABEL=%27seconds%27&amp;YLABEL=%27counts%27"> 
 01:59:14.3</a></td> 
       <td valign="middle" align="center" width="5%">1<br> 
       </td> 
  </tr> 
  <tr> 
  <td bgcolor="#ffcccc"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN_NUMBER=147844">147844  
(0x24184)</a></td> 
  <td>2002.07.07</td> 
  <td>12:45:31</td> 
        <td>            41 nb-1</td> 
         <td align="right"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/PlotTime?TABLE=LEVEL1_CONDITIONS&amp;X=FRONTENDTIME&amp;Y=LIVETIME&amp;WHERE=%27RC_RUNNUMBER=147844%27&amp;TITLE=LIVETIME_Run_147844&amp;XLABEL=%27seconds%27&amp;YLABEL=%27counts%27"> 
 01:26:08.5</a></td> 
       <td valign="middle" align="center" width="5%">0<br> 
       </td> 
  </tr> 
  <tr> 
  <td bgcolor="#ffcccc"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN_NUMBER=147843">147843  
(0x24183)</a></td> 
  <td>2002.07.07</td> 
  <td>12:00:54</td> 
        <td>            14 nb-1</td> 
         <td align="right"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/PlotTime?TABLE=LEVEL1_CONDITIONS&amp;X=FRONTENDTIME&amp;Y=LIVETIME&amp;WHERE=%27RC_RUNNUMBER=147843%27&amp;TITLE=LIVETIME_Run_147843&amp;XLABEL=%27seconds%27&amp;YLABEL=%27counts%27"> 
 00:28:56.9</a></td> 
       <td valign="middle" align="center" width="5%">0<br> 
       </td> 
  </tr> 
  <tr> 
  <td bgcolor="#ffcccc"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN_NUMBER=147842">147842  
(0x24182)</a></td> 
  <td>2002.07.07</td> 
  <td>11:22:27</td> 
        <td>            15 nb-1</td> 
         <td align="right"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/PlotTime?TABLE=LEVEL1_CONDITIONS&amp;X=FRONTENDTIME&amp;Y=LIVETIME&amp;WHERE=%27RC_RUNNUMBER=147842%27&amp;TITLE=LIVETIME_Run_147842&amp;XLABEL=%27seconds%27&amp;YLABEL=%27counts%27"> 
 00:29:59.5</a></td> 
       <td valign="middle" align="center" width="5%">1<br> 
       </td> 
  </tr> 
  <tr> 
  <td><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN_NUMBER=147838">147838  
(0x2417E)</a></td> 
  <td>2002.07.07</td> 
  <td>10:41:23</td> 
        <td>            16 nb-1</td> 
         <td align="right"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/PlotTime?TABLE=LEVEL1_CONDITIONS&amp;X=FRONTENDTIME&amp;Y=LIVETIME&amp;WHERE=%27RC_RUNNUMBER=147838%27&amp;TITLE=LIVETIME_Run_147838&amp;XLABEL=%27seconds%27&amp;YLABEL=%27counts%27"> 
 00:29:41.1</a></td> 
       <td valign="middle" align="center" width="5%">0<br> 
       </td> 
  </tr> 
  <tr> 
  <td><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN_NUMBER=147837">147837  
(0x2417D)</a></td> 
  <td>2002.07.07</td> 
  <td>09:49:32</td> 
        <td>            15 nb-1</td> 
         <td align="right"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/PlotTime?TABLE=LEVEL1_CONDITIONS&amp;X=FRONTENDTIME&amp;Y=LIVETIME&amp;WHERE=%27RC_RUNNUMBER=147837%27&amp;TITLE=LIVETIME_Run_147837&amp;XLABEL=%27seconds%27&amp;YLABEL=%27counts%27"> 
 00:25:41.4</a></td> 
       <td valign="middle" align="center" width="5%">1<br> 
       </td> 
  </tr> 
  <tr> 
  <td><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN_NUMBER=147836">147836  
(0x2417C)</a></td> 
  <td>2002.07.07</td> 
  <td>07:59:00</td> 
        <td>            46 nb-1</td> 
         <td align="right"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/PlotTime?TABLE=LEVEL1_CONDITIONS&amp;X=FRONTENDTIME&amp;Y=LIVETIME&amp;WHERE=%27RC_RUNNUMBER=147836%27&amp;TITLE=LIVETIME_Run_147836&amp;XLABEL=%27seconds%27&amp;YLABEL=%27counts%27"> 
 01:12:49.2</a></td> 
       <td valign="middle" align="center" width="5%">1<br> 
       </td> 
  </tr> 
  <tr> 
  <td><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN_NUMBER=147835">147835  
(0x2417B)</a></td> 
  <td>2002.07.07</td> 
  <td>06:05:12</td> 
        <td>            68 nb-1</td> 
         <td align="right"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/PlotTime?TABLE=LEVEL1_CONDITIONS&amp;X=FRONTENDTIME&amp;Y=LIVETIME&amp;WHERE=%27RC_RUNNUMBER=147835%27&amp;TITLE=LIVETIME_Run_147835&amp;XLABEL=%27seconds%27&amp;YLABEL=%27counts%27"> 
 01:36:57.3</a></td> 
       <td valign="middle" align="center" width="5%">1<br> 
       </td> 
  </tr> 
  <tr> 
  <td><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN_NUMBER=147834">147834  
(0x2417A)</a></td> 
  <td>2002.07.07</td> 
  <td>04:36:30</td> 
        <td>            54 nb-1</td> 
         <td align="right"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/PlotTime?TABLE=LEVEL1_CONDITIONS&amp;X=FRONTENDTIME&amp;Y=LIVETIME&amp;WHERE=%27RC_RUNNUMBER=147834%27&amp;TITLE=LIVETIME_Run_147834&amp;XLABEL=%27seconds%27&amp;YLABEL=%27counts%27"> 
 01:08:48.3</a></td> 
       <td valign="middle" align="center" width="5%">1<br> 
       </td> 
  </tr> 
  <tr> 
  <td><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN_NUMBER=147832">147832  
(0x24178)</a></td> 
  <td>2002.07.07</td> 
  <td>02:30:20</td> 
        <td>            57 nb-1</td> 
         <td align="right"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/PlotTime?TABLE=LEVEL1_CONDITIONS&amp;X=FRONTENDTIME&amp;Y=LIVETIME&amp;WHERE=%27RC_RUNNUMBER=147832%27&amp;TITLE=LIVETIME_Run_147832&amp;XLABEL=%27seconds%27&amp;YLABEL=%27counts%27"> 
 01:03:00.6</a></td> 
       <td valign="middle" align="center" width="5%">0<br> 
       </td> 
  </tr> 
  <tr> 
  <td><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN_NUMBER=147830">147830  
(0x24176)</a></td> 
  <td>2002.07.07</td> 
  <td>01:14:29</td> 
        <td>            37 nb-1</td> 
         <td align="right"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/PlotTime?TABLE=LEVEL1_CONDITIONS&amp;X=FRONTENDTIME&amp;Y=LIVETIME&amp;WHERE=%27RC_RUNNUMBER=147830%27&amp;TITLE=LIVETIME_Run_147830&amp;XLABEL=%27seconds%27&amp;YLABEL=%27counts%27"> 
 00:36:26.0</a></td> 
       <td valign="middle" align="center" width="5%">0<br> 
       </td> 
  </tr> 
  <tr> 
  <td><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/RunSummary?RUN_NUMBER=147827">147827  
(0x24173)</a></td> 
  <td>2002.07.07</td> 
  <td>00:22:48</td> 
        <td>            11 nb-1</td> 
         <td align="right"><a 
 href="http://www-cdfonline.fnal.gov/java/cdfdb/servlet/PlotTime?TABLE=LEVEL1_CONDITIONS&amp;X=FRONTENDTIME&amp;Y=LIVETIME&amp;WHERE=%27RC_RUNNUMBER=147827%27&amp;TITLE=LIVETIME_Run_147827&amp;XLABEL=%27seconds%27&amp;YLABEL=%27counts%27"> 
 00:09:51.6</a></td> 
       <td valign="middle" align="center" width="5%">1<br> 
       </td> 
  </tr> 
      
  </tbody>  
</table> 
 <br> 
 We check these runs against an earlier reference run, 147165. This is an  
early post-shutdown run. It probably isn't a good reference run as the means  
and rms's shift a lot during this time (changes in constants). So I switch  
to 147834, which is a long run, after comparing it to 147165 and finding that 
seem ok besides the shifts.<br> 
 <br> 
 Note that there are several COT occurances during this time.<br> 
  
<ul> 
   <li>COT18 failed with DTO's, causing a large number of runs to be marked  
bad by the shift crew. &nbsp;runs without B0COT18. I've marked them red in  
the table above; they are `bad' from a COT point of view as the detector is 
not complete.</li> 
    
  <ul> 
     <li>147842</li> 
     <li>147843</li> 
     <li>147844</li> 
    
  </ul> 
   <li>We get a fair number of errors of the following nature from the cotval  
script, in many runs, for SL 6, 7, and 8:</li> 
  
</ul> 
  
<pre>=================   Residuals versus distance from Stage0    =================<br>--------------------   Comparing ResVsDistSL5 Plots   -------------------<br>Bins that are HIGH by 8 sigma: <br>------------------------------ <br>Number of HIGH bins = 0<br><br>Bins that are LOW by 8 sigma: <br>------------------------------ <br>Number of LOW bins = 0<br><br>WARNING (ResVsDistSL5 plot): Some bins off by over 6 sigma.<br><br><br>--------------------   Comparing ResVsDistSL8 Plots   -------------------<br>Bins that are HIGH by 8 sigma: <br>------------------------------ <br>Dist(cm)= -1.1,  Residual= 0.00765846,   Dev = 1376.13 sigma.<br>Dist(cm)= -1.06,         Residual= 0,    Dev = 759.938 sigma.<br>Number of HIGH bins = 2<br><br>Bins that are LOW by 8 sigma: <br>------------------------------ <br>Dist(cm)= -1.08,         Residual= 0,    Dev = -823.578 sigma.<br>Dist(cm)= -1.02,         Residual= -0.124752,    Dev = -235.972 sigma.<br>Number of LOW bins = 2<br><br></pre> 
  
<blockquote>I'm not sure what to make of them. They seem to come and go for  
the outer SL's. They do appear to only be in the outer parts of the cell (distances 
greater than 0.7 cm). &nbsp;Below is a picture of the offending entry for 
run 147874, which is what the above pasting is from.<br> 
   <br> 
 </blockquote> 
  
<ul> 
  
</ul> 
<div align="center"><a 
 href="/cgi/elog/det-system-elog.pl?nb=cot&amp;action=view&amp;page=-70&amp;button=yes" 
 target="xxx" onclick="jswindow()"><img 
 src="/cgi/elog/det-system-elog.pl?nb=cot&amp;action=view&amp;page=-70" 
 width="400"> 
</a></div> 
All around, it looks pretty good. Except for those runs with the missing 
crate 18, I think they are all ok for this week.<br> 
    
<!-- Author: Peter Wittich -->


<!-- Date: Mon Jul 29 00:23:12 2002 -->
<pre>Paged right around midnight because all SLs tripped, then SL2 would not come on. 
 
The all-layers trip is another glitch on the solenoid interlock, this time captured by the scope.  A thunder storm was in progress at the time and is probably the source of the noise. 
 
SL2 is a known BiRa turn-on problem, I got it to go by first going off, then to 100%, just for SL2 from COT2. 
</pre>
<!-- Author: Aseet Mukherjee -->