<!-- Title: 22:40:09  Mon Feb  3 2003 -->
<!-- Crewchief: V Tiwari -->
<!-- Op1: M Mattson -->
<!-- Op2: Z Huang -->
<!-- Op3: &nbsp; -->
<!-- Op4: &nbsp; -->
<!-- Op5: &nbsp; -->
<!-- Op6: &nbsp; -->
<!-- Notes: -->
New Week


<!-- Date: Tue Feb 4 15:05:49 2003 -->
qietest checkout<br> 
1) On 01Feb03, when I tried to calibrate the pcal crates and 
   download LER=1 with qietest (rboot_plug, calib_plug), 
   the calibration 
   part failed because there were lots of channels were the 
   slope/sigma errors. Because of this, I requested Bill 
   Badgett and Frank Chlebana to rebuild the fer package to 
   include all the latest code changes, including Mario 
   Spezziga's FER_SmartCal package (timer).<br> 
2) This was done by 02Feb03, with the test and current versions 
   having all the changes<br> 
3) Frank Chlebana and I tested qietest around 0900hrs on 03Feb03. 
   I ran rboot_plug/calib_plug (test) and all pcal crates 
   were calibrated, 
   except the usual pcal08. I also ran calib_plug (current) 
   and got identical results. While we are not exactly sure 
   why this calibration and download worked on Monday and 
   failed on Sunday, we suspect it could be due to package 
   mismatches in the VME+qietest system. I calibrated and  
   downloaded pcal08 with LER=1 manually with qietest.<br> 
4) For these tests, I have modified cdf_cal/admem/bin's 
   login.xpt, ckcrate.xtp, and chcrate_src.xpt to pick up 
   qietest using the FER_DIR from the fer setup: 
   send "cd \"$env(FER_DIR)/VxWorks-5.3/PPC\"\r" ;<br> 
   Frank Chelebana recommends that this be used henceforth.
<!-- Author: Willis -->


<!-- Date: Wed Feb 5 10:08:00 2003 -->
<b>Weekly Calorimeter Checkout</b><br> 
<br> 
<b>Plug Calorimeter</b> 
<pre> 
tmp/b0pcal00.tmp: ### Crate: b0pcal00 -- All ok! 
tmp/b0pcal01.tmp: ### Crate: b0pcal01 -- All ok! 
tmp/b0pcal02.tmp: ### Crate: b0pcal02 -- All ok! 
tmp/b0pcal03.tmp: ### Crate: b0pcal03 -- All ok! 
tmp/b0pcal04.tmp: ### Crate: b0pcal04 -- All ok! 
tmp/b0pcal05.tmp: ### Crate: b0pcal05 -- All ok! 
tmp/b0pcal06.tmp: ### Crate: b0pcal06 -- All ok! 
tmp/b0pcal07.tmp: ### Crate: b0pcal07 -- All ok! 
tmp/b0pcal08.tmp: ### Crate: b0pcal08 -- Pedestal problem(s)  
tmp/b0pcal09.tmp: ### Crate: b0pcal09 -- All ok! 
tmp/b0pcal10.tmp: ### Crate: b0pcal10 -- All ok! 
tmp/b0pcal11.tmp: ### Crate: b0pcal11 -- All ok! 
</pre> 
This is the known PIN diode problem, which unfortunately was not addressed during the shutdown 
<pre>*******************   Start of FAILURE messages   ************** VALUE    CAP RANGE** 
* ADMEM=17 Chan= 0 Failure  6: Raw pedestal RMS TOO big       (  125.3809) ( 3) ( 0) * 
*******************   End of FAILURE messages   *************************************</pre> 
<b>Source Monitoring</b> 
<pre> 
ped/tmp/sped_b0pcal00.tmp: ### Crate: b0pcal00 -- All ok! 
ped/tmp/sped_b0pcal01.tmp: ### Crate: b0pcal01 -- SMON ped problem   
ped/tmp/sped_b0pcal02.tmp: ### Crate: b0pcal02 -- All ok! 
ped/tmp/sped_b0pcal03.tmp: ### Crate: b0pcal03 -- All ok! 
ped/tmp/sped_b0pcal04.tmp: ### Crate: b0pcal04 -- SMON ped problem   
ped/tmp/sped_b0pcal05.tmp: ### Crate: b0pcal05 -- All ok! 
ped/tmp/sped_b0pcal06.tmp: ### Crate: b0pcal06 -- SMON ped problem   
ped/tmp/sped_b0pcal07.tmp: ### Crate: b0pcal07 -- SMON ped problem   
ped/tmp/sped_b0pcal08.tmp: ### Crate: b0pcal08 -- All ok! 
ped/tmp/sped_b0pcal09.tmp: ### Crate: b0pcal09 -- All ok! 
ped/tmp/sped_b0pcal10.tmp: ### Crate: b0pcal10 -- All ok! 
ped/tmp/sped_b0pcal11.tmp: ### Crate: b0pcal11 -- All ok! 
</pre>  
<!-- Author: m mattson -->


<!-- Date: Wed Feb 5 12:10:18 2003 -->
<table><tr><td><a href=/elog/notebooks/plug/Wed_Feb__5_2003_121003.ps align=top><img src=/cgi/elog/det-system-elog.pl?nb=plug&action=view&page=-191  align='top'></td><td>EndPlug Mean and RMS Plots</a></td></tr></table>
<!-- Author: z huang -->