<!-- Title: Day shift. Wed Nov 27, 2002 -->
<!-- SciCo: Stefano Belforte -->
<!-- DAQAce: Bernd Stelzer -->
<!-- MonAce: Lucio Cerrito -->
<!-- CO: Juan Fernandez -->
<!-- OpManager: Eric James -->
<!-- Notes: -->
<pre>store 2001 in, L=21.7E30 Run 154677 in progress without one ShowerMax card (run bad for calorimeter)</pre>
<!-- Date: Wed Nov 27 08:19:38 2002 -->
<pre>ProcMon died at ~7am - restarted it.</pre>
<!-- Author: Bernd :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154677' target=_top>154677</a>) -->


<!-- Date: Wed Nov 27 08:31:50 2002 -->
<pre>SCPU_BAD_VRB_BYTE_COUNT Error - HRR recovered 
--------------------------------------------- 
Attention!!!. SCPU_BAD_VRB_BYTE_COUNT Error !!!  
Hardware EVB has detected a problem with data quality.  
SCPU b0eb12 reports problems with a VRB header.   
</pre>
<!-- Author: Bernd :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154677' target=_top>154677</a>) -->


<!-- Date: Wed Nov 27 08:34:06 2002 -->
<pre>8:24:23 AM->Done Timeout: WCAL_00 </pre>
<!-- Author: Bernd :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154677' target=_top>154677</a>) -->


<!-- Date: Wed Nov 27 08:42:35 2002 -->
D0 is moving thier pots
<!-- Author: Stefano :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154677' target=_top>154677</a>) -->

<!-- Date: Wed Nov 27 08:48:51 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154677">154677</A>
 Terminated at 2002.11.27 08:47:36
<!-- processed by 131.225.236.162 -->
<!-- Author: RunControl -->


<!-- Date: Wed Nov 27 08:50:22 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154677">154677</A>
TERMINATE: End the run to power cycle a SMX board
<!-- processed by 131.225.235.217 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 09:00:22 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154678">154678</A>
 Activated at 2002.11.27 08:58:03
<!-- processed by 131.225.236.229 -->
<!-- Author: RunControl -->


<!-- Date: Wed Nov 27 09:00:22 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154678">154678</A>
ACTIVATE: Try to run with full detector
<!-- processed by 131.225.236.229 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 09:00:22 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154678">154678</A>
 Terminated at 2002.11.27 08:59:16
<!-- processed by 131.225.236.229 -->
<!-- Author: RunControl -->


<!-- Date: Wed Nov 27 09:00:52 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154678">154678</A>
TERMINATE: Still the same SMX errors
<!-- processed by 131.225.127.7 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 09:10:03 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154679">154679</A>
 Activated at 2002.11.27 09:08:48
<!-- processed by 131.225.127.7 -->
<!-- Author: RunControl -->


<!-- Date: Wed Nov 27 09:10:41 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154679">154679</A>
ACTIVATE: Try running after power cycle of plug
<!-- processed by 128.174.118.249 -->
<!-- Author: Bernd -->
<!-- Comment: Wed Nov 27 09;12;48 comment by...Bernd -->
<pre>Seems to work - Issue remains if we need to do plug calibration</pre>

<!-- Comment: Wed Nov 27 09;16;33 comment by...JImmy Proudfoot -->
Calorimeter SPLs take note - this would be easier if we collected 
together all of the procedures for recovery from power cycling.

<!-- Date: Wed Nov 27 09:17:51 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154679">154679</A>
 Terminated at 2002.11.27 09:16:39
<!-- processed by 131.225.45.171 -->
<!-- Author: RunControl -->


<!-- Date: Wed Nov 27 09:18:23 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154679">154679</A>
TERMINATE: Ending run to do SMX calibration
<!-- processed by 131.225.234.39 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 09:18:40 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154679">154679</A>
TERMINATE: Ending run to do SMX calibration
<!-- processed by 146.139.52.54 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 09:22:44 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154680">154680</A>
ACTIVATE: Do SMQIE Central+Plug callibration after Plug crate power cycle
<!-- processed by 131.225.233.157 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 09:23:49 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154680">154680</A>
TERMINATE: Done calibration
<!-- processed by 131.225.236.161 -->
<!-- Author: Bernd -->



<!-- Date: Wed Nov 27 09:25:05 2002 -->
CCU 15, CH 10 gave a wrong threshold warning. Recovered OK according to instructions from 1st floor PC
<!-- Author: Lucio -->

<!-- Date: Wed Nov 27 09:32:14 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154681">154681</A>
 Activated at 2002.11.27 09:30:18
<!-- processed by 131.225.236.161 -->
<!-- Author: RunControl -->


<!-- Date: Wed Nov 27 09:32:30 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154681">154681</A>
ACTIVATE: New Run with full detector
<!-- processed by 131.225.236.162 -->
<!-- Author: Bernd -->



<!-- Date: Wed Nov 27 09:33:10 2002 -->
<pre>We re-cycled the 400Hz power on the east plug to see if we  
could recover the bad SMX board.  It looks like we were in  
fact able to recover this board.  Need to re-calibrate the  
SMX after power-cycle so quickly perform a SMX calibration.   
This calib run looks good so we go back to normal data 
collection. </pre>
<!-- Author: Eric James -->
<!-- Comment: Wed Nov 27 09;36;18 comment by...Eric -->
<pre>Note that the short run 154679 is still bad for SMX even 
though the problematic board had been recovered since the 
new calibration had not yet been performed.</pre>


<!-- Date: Wed Nov 27 09:40:48 2002 -->
<center><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-10969&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-10969' width='300'></a><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-10970&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-10970' width='300'></a><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-10971&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-10971' width='300'></a></center>
<!-- Author: Lucio -->
<!-- Comment: Wed Nov 27 09;42;52 comment by...Lucio -->
Hourly plots: Losses low and stable, abort gap stable, e lens ON,
RF stable, luminosity at 20E30 coming down smoothly


<!-- Date: Wed Nov 27 09:57:51 2002 -->
Silicon Cooling alarm, followed after 1 min by silicon trip and trigger inhibit. 
Paged Silicon General and silicon cooling expert.
<!-- Author: Stefano -->
<!-- Comment: Wed Nov 27 09;58;46 comment by...Stefano -->
Silicon cooling expert calls back. Ifix complains about low flow.

<!-- Comment: Wed Nov 27 10;00;36 comment by...Stefano -->
Silicon export (Steve Nahn) walks in

<!-- Comment: Wed Nov 27 10;03;15 comment by...Stefano -->
Steve Nahn confirms that silicn trip is due to cooling alarm.
Rob Roser talked to cooling expert and is now investigating downstairs. We have nothing to do but wait for Rob's results.

<!-- Comment: Wed Nov 27 10;07;51 comment by...Stefano -->
Steve Nahn was surprised that L00 did not trip, now it does ! Circle is closed and SVX,ISL and L00 are all tripped

<!-- Comment: Wed Nov 27 10;10;26 comment by...Stefano -->
R.Roser says he'll be done in 5 minutes
<!-- Comment: Wed Nov 27 10;16;38 comment by...Stefano -->
Silicon cooling is OK now. Steve Nahn bringing it up
<!-- Date: Wed Nov 27 10:10:25 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154681">154681</A>
 Terminated at 2002.11.27 10:08:47
<!-- processed by 131.225.236.229 -->
<!-- Author: RunControl -->



<!-- Date: Wed Nov 27 10:11:44 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154681">154681</A>
TERMINATE: Ending run due to silicon cooling problems
<!-- processed by 141.211.101.94 -->
<!-- Author: Bernd -->




<!-- Date: Wed Nov 27 10:22:52 2002 -->
Silicon Cooling -- we were in the process of training tech's to change filters and resin in silicon cooling.  In the process of this training(normally benign), a tech must have leaned against one of our manual valves and closed the main flow valve from the SVX chiller by 50-60%.  This reduced the pressure and hence the flow to SVX.  We ended up tripping everything in coming back up because the main bypass valve was closed when I opened the valve, the flows were instantly too high and tripped off.
<!-- Author: Rob+Eva -->


<!-- Date: Wed Nov 27 10:26:25 2002 -->
Beam is lost  
<!-- Author: Stefano -->


<!-- Date: Wed Nov 27 10:26:28 2002 -->
CCU06 has bad response - fixed. CCU23 is unreadable. Power cycling doesn't work. We need access to swap this CCU because 
36 TSU/BSU counters are dead now.
<!-- Author: Oleg Poukhov -->


<!-- Date: Wed Nov 27 10:32:35 2002 -->
We called MCR, it was not a quech. Power glitch likely. 
MCR does not a plan for what to do yet, will call back. 
We start quiet time calibration
<!-- Author: Stefani -->
<!-- Comment: Wed Nov 27 10;48;14 comment by...Stefano -->
From MCR e-log:<br>
ComEd lost a line in Lombard (line 10808). We got glitched as a result (probably on the re-close).
<!-- Date: Wed Nov 27 10:35:16 2002 -->
All HV at "pre-shot setup" configuration (STDBY/OFF), except PLUG, left ON for plug laser calibration
<!-- Author: Lucio -->

<!-- Date: Wed Nov 27 10:36:47 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154682">154682</A>
ACTIVATE: Do QIE CAL Calibration
<!-- processed by 131.225.236.229 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 10:36:47 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154682">154682</A>
TERMINATE: Done
<!-- processed by 131.225.236.229 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 10:42:09 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154683">154683</A>
ACTIVATE: CLC QIE Calibration
<!-- processed by 66.32.33.147 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 10:43:47 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154683">154683</A>
TERMINATE: Done
<!-- processed by 131.225.127.7 -->
<!-- Author: Bernd -->



<!-- Date: Wed Nov 27 10:44:58 2002 -->
<center><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-10972&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-10972' width='300'></a><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-10973&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-10973' width='300'></a><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-10974&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-10974' width='300'></a><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-10975&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-10975' width='300'></a></center>
<!-- Author: Lucio -->
<!-- Comment: Wed Nov 27 10;47;12 comment by...Lucio -->
TeV and SVX monitoring at the time of the loss of beam. It all looks clean, there was no HV trip in any subdetector.


<!-- Date: Wed Nov 27 10:47:13 2002 -->
Hand SVT to experts for one hour or so
<!-- Author: Stefano -->

<!-- Date: Wed Nov 27 10:48:02 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154684">154684</A>
ACTIVATE: Plug laser calibration (Normal Mode)
<!-- processed by 131.225.234.39 -->
<!-- Author: Bernd -->




<!-- Date: Wed Nov 27 10:52:55 2002 -->
<pre>MCR says: 
quite time until noon 
Next shot setup likely at 4pm</pre>
<!-- Author: Stefano -->

<!-- Date: Wed Nov 27 10:54:57 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154684">154684</A>
TERMINATE: Done
<!-- processed by 131.225.234.39 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 10:56:45 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154685">154685</A>
ACTIVATE: Plug Laser Calibration (MyronMode)
<!-- processed by 131.225.45.171 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 10:58:57 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154685">154685</A>
TERMINATE: Done
<!-- processed by 131.225.235.29 -->
<!-- Author: Bernd -->



<!-- Date: Wed Nov 27 11:06:24 2002 -->
Plug HV going to STDBY now
<!-- Author: Lucio -->

<!-- Date: Wed Nov 27 11:07:38 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154686">154686</A>
ACTIVATE: BSC/Miniplug calibration
<!-- processed by 131.225.45.171 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 11:08:30 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154686">154686</A>
TERMINATE: Done
<!-- processed by 131.225.127.7 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 11:11:48 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154687">154687</A>
ACTIVATE: CAL LED Calibration
<!-- processed by 131.225.45.171 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 11:12:54 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154687">154687</A>
TERMINATE: Done

<!-- processed by 12.250.196.25 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 11:22:21 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154688">154688</A>
ACTIVATE: DPS OFF D Mode Calibration
<!-- processed by 131.225.122.33 -->
<!-- Author: nahn -->


<!-- Date: Wed Nov 27 11:23:26 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154688">154688</A>
TERMINATE: Reformatter throws it all away, so try again.
<!-- processed by 131.225.236.229 -->
<!-- Author: nahn -->



<!-- Date: Wed Nov 27 11:24:35 2002 -->
CMX expert is taking control of HV from 1st floor
<!-- Author: Lucio -->

<!-- Date: Wed Nov 27 11:28:40 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154689">154689</A>
ACTIVATE: not working
<!-- processed by 131.225.235.251 -->
<!-- Author: nahn -->


<!-- Date: Wed Nov 27 11:28:40 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154689">154689</A>
TERMINATE: doesn't work...
<!-- processed by 131.225.235.251 -->
<!-- Author: nahn -->


<!-- Date: Wed Nov 27 11:28:46 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154690">154690</A>
ACTIVATE: CAL Xenon (XEF) Calibration
<!-- processed by 131.225.127.7 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 11:30:44 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154690">154690</A>
TERMINATE: Done
<!-- processed by 131.225.232.152 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 11:40:33 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154691">154691</A>
TERMINATE: D Mode DPS OFF run done
<!-- processed by 131.225.233.160 -->
<!-- Author: nahn -->


<!-- Date: Wed Nov 27 11:43:51 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154692">154692</A>
ACTIVATE: DPS ON D Mode calibration
<!-- processed by 131.225.232.160 -->
<!-- Author: nahn -->


<!-- Date: Wed Nov 27 11:45:19 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154693">154693</A>
ACTIVATE: COT Calibration
<!-- processed by 131.225.235.166 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 11:45:19 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154693">154693</A>
ACTIVATE: COT Calibration
<!-- processed by 131.225.235.166 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 11:45:19 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154693">154693</A>
ACTIVATE: COT Calibration
<!-- processed by 131.225.235.166 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 11:45:19 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154693">154693</A>
ACTIVATE: COT Calibration
<!-- processed by 131.225.235.166 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 11:45:46 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154693">154693</A>
TERMINATE: Done
<!-- processed by 131.225.235.39 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 11:48:36 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154692">154692</A>
TERMINATE: DPS On Run finished

<!-- processed by 128.2.2.89 -->
<!-- Author: nahn -->


<!-- Date: Wed Nov 27 11:48:45 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154694">154694</A>
ACTIVATE: Muon Calibration
<!-- processed by 128.2.2.89 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 11:50:42 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154694">154694</A>
TERMINATE: Done
<!-- processed by 128.2.2.89 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 11:53:25 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154696">154696</A>
ACTIVATE: TOF TAC calibration
<!-- processed by 131.225.235.3 -->
<!-- Author: Bernd -->



<!-- Date: Wed Nov 27 11:53:55 2002 -->
<pre>Tested CMX miniskirt HV for SE 16,17, after changing ramp  
voltage yesterday. I cycled the wedges from off-&gt;on, stdby-&gt;on, 
and off-&gt;on again, with no problems. I have turned the wedges 
to stdby now. It takes about 4 minutes to ramp up with these 
slower settings, so please take that into account. The currents 
are stable at around 600 uA. </pre>
<!-- Author: Sal Rappoccio -->

<!-- Date: Wed Nov 27 11:55:00 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154696">154696</A>
TERMINATE: Done
<!-- processed by 131.225.236.162 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 12:00:07 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154697">154697</A>
ACTIVATE: TOF QIE Calibration
<!-- processed by 131.225.236.163 -->
<!-- Author: Bernd -->



<!-- Date: Wed Nov 27 12:00:08 2002 -->
<pre>Calibrations checked ok. 
New COMPLETE constants for COT calibration. Num. bad ch = 415. 
Crate 10 has large number of drifted channels (96).</pre>
<!-- Author: Juan Pablo -->

<!-- Date: Wed Nov 27 12:00:51 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154697">154697</A>
TERMINATE: Done
<!-- processed by 128.220.233.195 -->
<!-- Author: Bernd -->


<!-- Date: Wed Nov 27 12:10:15 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154698">154698</A>
 Activated at 2002.11.27 12:08:46
<!-- processed by 131.225.242.221 -->
<!-- Author: RunControl -->
<!-- Comment: Wed Nov 27 12;12;52 comment by...Bernd -->
<pre>Raw L1 rate: ~64-65Hz (no beam in machine)</pre>
<!-- Date: Wed Nov 27 12:10:15 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154698">154698</A>
ACTIVATE: Taking Rick's run
<!-- processed by 131.225.242.221 -->
<!-- Author: Bernd -->



<!-- Date: Wed Nov 27 12:10:46 2002 -->
JC called and he is going to restart HV control program.  He updated the program so that TOF icon will be highlighted when 
it is clicked.  This will take a few minutes.
<!-- Author: kaori -->
<!-- Comment: Wed Nov 27 12;23;06 comment by...Lucio -->
Done
<!-- Date: Wed Nov 27 12:18:04 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154698">154698</A>
 Terminated at 2002.11.27 12:16:37
<!-- processed by 131.225.236.132 -->
<!-- Author: RunControl -->


<!-- Date: Wed Nov 27 12:19:23 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154698">154698</A>
TERMINATE: Done
<!-- processed by 131.225.233.157 -->
<!-- Author: Bernd -->




<!-- Date: Wed Nov 27 12:37:21 2002 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2002&action=view&page=-10976&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2002&action=view&page=-10977  width='300' align='top'></td><td>diff between last two COMPLETE COTCHANT0 runs</a></td></tr></table>
<!-- Author: Juan Pablo -->
<!-- Comment: Wed Nov 27 12;39;00 comment by...Juan Pablo -->
<pre> On the latest COT calibration run,
crate 10 had large number of drifted channels (96).
 I could trace it down to a single slot (18)
for which the times have shifted by about 0.1-0.2 ns.

</pre>


<!-- Date: Wed Nov 27 13:11:12 2002 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2002&action=view&page=-10979&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2002&action=view&page=-10979  width='300' align='top'></td><td>Pilot protons being injected; spikes in the abort gap reaching 500kHz. Silicon is on STDBY, I'm keeping an eye on the rad doses</a></td></tr></table>
<!-- Author: Lucio -->


<!-- Date: Wed Nov 27 14:02:31 2002 -->
Tevatron quenched during studies at A1 and F3  
<!-- Author: Stefano -->
<!-- Comment: Wed Nov 27 14;23;40 comment by...Stefano -->
MCR expects fast recover (~30min). Shot still cheduled for 4pm-ish
<!-- Date: Wed Nov 27 14:04:23 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154699">154699</A>
 Activated at 2002.11.27 14:03:12
<!-- processed by 144.92.164.198 -->
<!-- Author: RunControl -->


<!-- Date: Wed Nov 27 14:05:03 2002 -->
Run <A HREF="/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154699">154699</A>
ACTIVATE: Take cosmics
<!-- processed by 65.144.129.100 -->
<!-- Author: Bernd -->



<!-- Date: Wed Nov 27 14:06:45 2002 -->
<pre>Done timeout from WCAL_00 - HRR recovered OK</pre>
<!-- Author: Bernd :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154699' target=_top>154699</a>) -->



<!-- Date: Wed Nov 27 15:45:40 2002 -->
<font color=red>Shift Summary: </font><pre>08:00 Taking data w silicon. Store=2001  L=2.2E31 Run=154677 
      One ShowrMax card is bad and removed from readout list 
08:50 Stop run to cycle 400Hz to East Plug to 
      recover the Shower Max card. After this ShMax calibration 
      have to be done again 
09:32 Start data taking with all detector. 
09:57 ISL cooling trips 
10:26 Beam is lost due to power glitch just before silicon is back 
14:00 Start cosmics with all detector waiting for beam 
 
Shift highlights: 
- Start with one Plug ShowerMax board out of DAQ. OpMan and 
  calorimeter SPL decide to cycle power to the VME crate to 
  try restart the bad card. Have to toggle 400Hz power to all 
  East Plug. Trick worked. ShowerMax calibration had to be 
  redone. This is a general procedure for ShMax cards after 
  power cycling. Loose ~40min of data there 
- Loose ~30min of data due to silicon cooling, then loose beam 
- Do all calibration and some SVT timing studies 
- CMX HV turnon has a slower ramp now, will take 4min 
- There are 36 dead (no HV) photomultipliers in TSU/BSU waiting 
  for a 1hr access to repalce a failed CCU. Expect dead regions 
  in YMON.</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   154677,154679,154681
Delivered Luminosity   180  
Acquired Luminosity    96  
Efficiency             54%
   
</td></tr></table></pre>
<!-- Author: Stefano -->

<!-- Date: Wed Nov 27 15:56:21 2002 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154677>
154677</A> 
x25C35</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_1_03 [2,194,329]</TD>                                       
<TD ALIGN=CENTER>06:33:53</TD>                                          
<TD ALIGN=CENTER>08:47:36</TD>                                          
<TD ALIGN=CENTER>02:05:34</TD>                                          
<TD ALIGN=RIGHT>     75,586,856</TD>                                    
<TD ALIGN=RIGHT>     1,094,687</TD>                                     
<TD ALIGN=RIGHT>     300,878</TD>                                       
<TD ALIGN=RIGHT>  170.879</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154679>
154679</A> 
x25C37</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_1_03 [2,194,329]</TD>                                       
<TD ALIGN=CENTER>09:08:48</TD>                                          
<TD ALIGN=CENTER>09:16:39</TD>                                          
<TD ALIGN=CENTER>00:07:31</TD>                                          
<TD ALIGN=RIGHT>      4,148,846</TD>                                    
<TD ALIGN=RIGHT>        61,470</TD>                                     
<TD ALIGN=RIGHT>      16,990</TD>                                       
<TD ALIGN=RIGHT>    8.956</TD>                                          
<TD ALIGN=CENTER>0</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>0</TD></TR>                                            
                                                                        
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=154681>
154681</A> 
x25C39</TD>                                                             
<TD>BEAM</TD>                                                           
<TD>PHYSICS_1_03 [2,194,329]</TD>                                       
<TD ALIGN=CENTER>09:30:18</TD>                                          
<TD ALIGN=CENTER>10:08:47</TD>                                          
<TD ALIGN=CENTER>00:23:26</TD>                                          
<TD ALIGN=RIGHT>     13,977,741</TD>                                    
<TD ALIGN=RIGHT>       189,801</TD>                                     
<TD ALIGN=RIGHT>      53,760</TD>                                       
<TD ALIGN=RIGHT>   28.779</TD>                                          
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD>                                                 
<TD ALIGN=CENTER>1</TD></TR>                                            
                                                                        
<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>15:55:02</TD>                                                       
<TD ALIGN=CENTER>02:36:31</TD>                                          
<TD ALIGN=RIGHT>      93,713,443</TD>                                   
<TD ALIGN=RIGHT>      1,345,958</TD>                                    
<TD ALIGN=RIGHT>       371,628</TD>                                     
<TD ALIGN=RIGHT>  208.614</TD>                                          
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- processed by 131.225.127.7 -->
<!-- Author: End of Shift Report -->



<!-- Date: Wed Nov 27 15:57:53 2002 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2002&action=view&page=-10980&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2002&action=view&page=-10980  width='300' align='top'></td><td>Luminosity and datataking until beam loss</a></td></tr></table>
<!-- Author: Lucio -->