<!-- Title: Day shift. Thu Sep 27, 2001 -->
<!-- SciCo: Allan Clark -->
<!-- DAQAce: Moohyun Ahn -->
<!-- MonAce: Carsten Rott -->
<!-- CO: Maurizio Loreti -->
<!-- OpManager: Dave Ambrose -->
<!-- Notes: -->
Start shift


<!-- Date: Thu Sep 27 08:03:14 2001 -->
Cryo are doing work on S1, S3 in coming few minutes. Ignore alarms. They will inform when completed.
<!-- Author: Allan -->
<!-- Comment: Thu Sep 27 08;31;49 comment by...Allan -->
Now completed


<!-- Date: Thu Sep 27 08:33:05 2001 -->
starting to put detector components into standby
<!-- Author: Carsten -->


<!-- Date: Thu Sep 27 08:33:49 2001 -->
MCR announce Controlled Access
<!-- Author: Allan -->


<!-- Date: Thu Sep 27 08:38:04 2001 -->
COT,CMU,CMP,CMX,BMU,CES,CPR,CCR are in stdby. now going down to turn GAM on
<!-- Author: Carsten -->


<!-- Date: Thu Sep 27 08:38:34 2001 -->
Cryo call. They will start a purge. 
<p> 
Solenoid currently off for Watt can vacuum pumping 
</p>
<!-- Author: Allan -->


<!-- Date: Thu Sep 27 09:03:24 2001 -->
turned on the PS in 1RR06H and 1RR06I with Konstantin. He restarts monitoring program now.
<!-- Author: Carsten -->


<!-- Date: Thu Sep 27 09:06:12 2001 -->
BMU alarm caused by NW24
<!-- Author: Carsten -->
<!-- Comment: Thu Sep 27 09;12;42 comment by...Carsten -->
again
<!-- Comment: Thu Sep 27 09;21;03 comment by...Carsten -->
again ... paging BMU guys
<!-- Comment: Thu Sep 27 09;32;46 comment by...Allan -->
Call Dan Cyr at Wisconsin house ... he will come in
<!-- Comment: Thu Sep 27 09;58;51 comment by...Allan -->
BMU page called in ... he will consult with Dan Cyr
<!-- Date: Thu Sep 27 09:08:35 2001 -->
Karen requested to turn CES on ... is on now
<!-- Author: Carsten -->


<!-- Date: Thu Sep 27 09:11:50 2001 -->
I tried to make the CLC LED gain calibration. b0clc00 crate was reporting mistake all time. 
<!-- Author: Alex Pronko :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=127716' target=_top>127716</a>) -->




<!-- Date: Thu Sep 27 09:22:17 2001 -->
Turned on PEM,PSH,PHA
<!-- Author: Carsten -->


<!-- Date: Thu Sep 27 09:25:52 2001 -->
Opened West IMU steel all the way back to see whether this change in shielding helps cmp HV trips.
<!-- Author: RR -->


<!-- Date: Thu Sep 27 09:28:56 2001 -->
<b>SVX Controlled Access:</b> 
<p>We removed two SVX power supplies that had been being used for L00.  They will now be re-adjusted back for their original purpose.
<!-- Author: Napora and Slaughter -->


<!-- Date: Thu Sep 27 09:30:12 2001 -->
MCR has been informed that we have completed the Controlled Access
<!-- Author: Allan -->


<!-- Date: Thu Sep 27 09:30:22 2001 -->
CEM is in warning state ... Nikolay and Konstantin are already working on it.
<!-- Author: Carsten -->



<!-- Date: Thu Sep 27 09:36:28 2001 -->
Heartbeat problems on S2 ... Cryo will reboot it. There may be alarms
<!-- Author: Allan -->


<!-- Date: Thu Sep 27 09:40:17 2001 -->
Hyunsoo adviced us to keep an eye on the CMP North section. 
<!-- Author: Carsten -->
<!-- Comment: Thu Sep 27 09;46;15 comment by...Allan -->
Request is to monitor CMP(top) rates and B0lum plots when we have stable beams
<!-- Date: Thu Sep 27 09:43:18 2001 -->
During SVX calib, CSL is dead. 
Restart CSL and cleanup EVB and L3. 
reastart calib run but it fail. 
Call CSL expert. 
<!-- Author: MooHyun -->
<!-- Comment: Thu Sep 27 09;47;45 comment by...Allan -->
This is presently preventing D-mode SVX calibrations, and L2 Isolist work by Steve Kuhlmann



<!-- Date: Thu Sep 27 10:16:05 2001 -->
Dan is here to look at the BMU trips
<!-- Author: Carsten -->


<!-- Date: Thu Sep 27 10:25:35 2001 -->
Yesterday's update of the L3 monitor GUI was incomplete - should 
be fine after the next restart.  
<!-- Author: Arnd -->


<!-- Date: Thu Sep 27 10:29:32 2001 -->
MCR starts to test TeV magnet ramping
<!-- Author: Allan -->


<!-- Date: Thu Sep 27 10:30:23 2001 -->
Nikolay is still working on GAM,CHA,CEM,WHA and restarted monitoring program again ... should be up in 20min
<!-- Author: Carsten -->


<!-- Date: Thu Sep 27 10:31:05 2001 -->
<pre>BMU24 cathode trips at standby voltage. Lowered to 2000v. This is unusual, wondering if it is related to the toroids being pushed around?</pre>
<!-- Author: dan -->


<!-- Date: Thu Sep 27 10:42:37 2001 -->
<pre>Isolation tests finished.  Data transfer with latest 
firmware looked fine,  but there are bursts of  
data with zero isolation energy that we don't understand. 
Tests will continue later with the spare Isolist 
card. </pre>
<!-- Author: Steve Kuhlmann -->


<!-- Date: Thu Sep 27 10:50:59 2001 -->
MCR call. They will continue low-beta tests for about 15 minutes, then start shot setup
<!-- Author: Allan -->


<!-- Date: Thu Sep 27 10:53:42 2001 -->
CSL had a problem on the monitoring program, and stopped functioning probably because the shared memory could not be accessed from other CSL programs. We had to restart the CSL.
<!-- Author: Hiro, Tony -->


<!-- Date: Thu Sep 27 10:55:46 2001 -->
Request from SVX: Because of shot setup, they will not make calibrations before the store. However, after stable beams are declared, they wish to make a full calibration before integrating the SVX
<!-- Author: Allan -->


<!-- Date: Thu Sep 27 11:02:27 2001 -->
Monica took calorimeters, L1, and L2. She is doing L2 study.
<!-- Author: MooHyun -->
<!-- Comment: Thu Sep 27 11;51;35 comment by...Monica -->
I took a run (127727) with L2_TORTURE_V1 to make sure that the CLIST hardware is working properly after changing the socket of an FPGA PROM yesterday. Waiting for fcdfsgi2 to be back to run my validation code.
<!-- Comment: Thu Sep 27 13;08;17 comment by...Allan -->
Apologies by (P)scycho for not paging Heather - he thought that Steve and Monica had it in hand.
<!-- Date: Thu Sep 27 11:23:47 2001 -->
<pre>Proton injection tuneup has started ... see test bunches in the Tevatron. 
putting CES back in stdby</pre>
<!-- Author: Carsten -->


<!-- Date: Thu Sep 27 11:29:09 2001 -->
Call Cryo. They will now start ramping up the solenoid.
<!-- Author: Allan -->
<!-- Comment: Thu Sep 27 12;31;19 comment by...Allan -->
Solenoid now at full field
<!-- Date: Thu Sep 27 11:29:11 2001 -->
Start cosmic run 127728 with all components. 
COSMIC_NOTRACKS[3,205]
<!-- Author: MooHyun :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=127728' target=_top>127728</a>) -->
<!-- Comment: Thu Sep 27 11;36;19 comment by...MooHyun -->
End the run to give the CLC crates.


<!-- Date: Thu Sep 27 11:44:20 2001 -->
Start new cosmic run 127729 w/o clc.
<!-- Author: MooHyun :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=127729' target=_top>127729</a>) -->
<!-- Comment: Thu Sep 27 12;00;07 comment by...MooHyun -->
End the run to integrate clc crates in.
<!-- Date: Thu Sep 27 11:44:30 2001 -->
final proton bunches are being loaded now
<!-- Author: Carsten -->



<!-- Date: Thu Sep 27 11:56:47 2001 -->
Nikolay is still trying to restore CEM,CHA,WHA,GAM. He restarted the monitoring program multiple times and CEM,CHA,WHA come up fine but GAM keeps changing from offline to warning present state.
<!-- Author: Carsten -->


<!-- Date: Thu Sep 27 11:58:06 2001 -->
<p>Have done CLC gain calibration.<br>
<!-- Author: Alexander Pronko,Alexander Sukhanov :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=127730-127734' target=_top>127730-127734</a>) -->



<!-- Date: Thu Sep 27 12:01:59 2001 -->
Jean and Andy warn me that MCR should attempt to tilt the beam when they have a store. They are in a Si meeting (Pump Room) and we should warn them.
<!-- Author: Allan -->
<!-- Comment: Thu Sep 27 13;09;31 comment by...Allan -->
Requested tilt in MCR unite is +500 in x(horizontal), and +150 in y(vertical)
<!-- Date: Thu Sep 27 12:06:08 2001 -->
Start new cosmic run 127735 with all components.
<!-- Author: MooHyun :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=127735' target=_top>127735</a>) -->
<!-- Comment: Thu Sep 27 12;12;43 comment by...MooHyun -->
End the run to reboot b0puls01.


<!-- Date: Thu Sep 27 12:20:06 2001 -->
Start cosmic run 127736 w/o CAL_PULSER_01.
<!-- Author: MooHyun :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=127736' target=_top>127736</a>) -->
<!-- Comment: Thu Sep 27 12;20;33 comment by...MooHyun -->
BTO from VRB_SVX_02 -> HRR
<!-- Comment: Thu Sep 27 12;28;18 comment by...MooHyun -->
End the run to put CAL_PULSER_01 back in.
<!-- Date: Thu Sep 27 12:20:27 2001 -->
TOF alarm (it was turned off !!) paging expert.
<!-- Author: Carsten -->
<!-- Comment: Thu Sep 27 12;32;02 comment by...Stefano -->
Two TOF channels (crate 1 - board 4 - channels 11/14) with 
internal trip. Problems solved after switched off/on 
the channels.
<!-- Date: Thu Sep 27 12:27:58 2001 -->
Magnet field reaches 13750Gauss ... proton injection seem to be delayed, no final bunches in the Tevatron, yet. 
<!-- Author: Carsten -->



<!-- Date: Thu Sep 27 12:29:57 2001 -->
<pre>Attempted to reboot b0puls01 with some new code that should 
fix spurious error messages, caused by incorrect end-of-run. The 
crate didn't boot up properly, since the boot scripts were 
changed, but the new table file not correctly ups-declared. 
After some digging in the table files, it boots OK now. 
</pre>
<!-- Author: Arnd -->




<!-- Date: Thu Sep 27 12:39:56 2001 -->
decleared the new cdfsoftb0 release, 4.0.0d as a current 
default version.
<!-- Author: Kaori -->


<!-- Date: Thu Sep 27 12:49:26 2001 -->
Nikolay is done and GAM should be fine now
<!-- Author: Carsten -->




<!-- Date: Thu Sep 27 13:14:05 2001 -->
Reboot the front-end crates. b0mutr00 and b0svt02 were not reboot by the script "./reboot_shot b0". Reboot them via VxWorks Monitor.
<!-- Author: MooHyun -->


<!-- Date: Thu Sep 27 13:15:48 2001 -->
Now the final protns are really injected
<!-- Author: Carsten -->
<!-- Comment: Thu Sep 27 13;23;15 comment by...Carsten -->
Initial Proton Load =5.70E12


<!-- Date: Thu Sep 27 13:27:18 2001 -->
Erik and Manuel are going to work on the Plug HV PC for 15min.
<!-- Author: Carsten -->


<!-- Date: Thu Sep 27 13:37:59 2001 -->
Now loading the antiprotons
<!-- Author: Allan -->


<!-- Date: Thu Sep 27 13:42:36 2001 -->
BSC alarm ... cleared after 3sec.
<!-- Author: Carsten -->


<!-- Date: Thu Sep 27 13:46:41 2001 -->
Antiprotons are in and Tevatron starts ramping
<!-- Author: Carsten -->


<!-- Date: Thu Sep 27 13:49:37 2001 -->
End of tests for CLIST; CLIST #1 ( now in the b0l2de00 crate) seems happy with L2_TORTURE_v3 for runs 127741 and 127742 (for this last one I masked off MB_XING and JET_0.5); CLIST #2 fix to EOE* logic seems to be working.
<!-- Author: monica -->


<!-- Date: Thu Sep 27 14:00:03 2001 -->
Beam is stable now. Initial luminosity is at 5.3E30 and proton losses are at 8kHz
<!-- Author: Carsten -->
<!-- Comment: Thu Sep 27 14;15;53 comment by...MooHyun -->
Reset CLC lum
<!-- Date: Thu Sep 27 14:07:55 2001 -->
Turning on COT,TOF,CMU,CMP,CMX,BMU,CES,CPR,CCR. TOF does not come up. going downstairs to turn TOF on.
<!-- Author: Carsten -->


<!-- Date: Thu Sep 27 14:09:21 2001 -->
<p> 
Stable beam. Store 725. Luminosity 5.34E+30 
</p><p> 
HV turned on 
</p><p> 
Silicon taken out for x-mode calibration 
<Request for beam tilt relayed to MCR, with the numbers (+500 MCR units in x, +150 units in y). They are waiting for D0 numbers, and the beam steer-er. 
</p>
<!-- Author: Allan -->


<!-- Date: Thu Sep 27 14:09:50 2001 -->
<pre>CMP tripped in Top South A1 and A4</pre>
<!-- Author: Carsten -->
<!-- Comment: Thu Sep 27 14;15;28 comment by...carsten -->
turned CMP back on. tripped again in A1 and A4. experts are here.



<!-- Comment: Thu Sep 27 14;20;29 comment by...Carsten -->
turning of South Wall, Top and North Wall
<!-- Comment: Thu Sep 27 14;38;26 comment by...Allan -->
Need the currents still 
<!-- Date: Thu Sep 27 14:16:59 2001 -->
turned TOF on downstairs but get an alarm (from PM0). paging expert.
<!-- Author: Carsten -->
<!-- Comment: Thu Sep 27 14;21;06 comment by...Carsten -->
Stefano is checking it.


<!-- Comment: Thu Sep 27 14;28;55 comment by...Stefano -->
TOF Channel 4 (crate 2, board 5) in overcurrent.
Tried to fix it without success.
Now the channel is out of monitoring.
<!-- Date: Thu Sep 27 14:23:42 2001 -->
*ALL* the consumer monitors have crashed at the same time; they have been restarted.
<!-- Author: M. Loreti :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=127738' target=_top>127738</a>) -->


<!-- Date: Thu Sep 27 14:25:11 2001 -->
<pre>Start run using new physics table PHYSICS_0_02[1,208] w/o silicon. Run 127744 
Total DT is about 8% : 9% from Busy, 48% from Readout,  
                       7% from L2, and 35% from tsi</pre>
<!-- Author: MooHyun :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=127744' target=_top>127744</a>) -->


<!-- Date: Thu Sep 27 14:28:08 2001 -->
New HV settings loaded for PES just before collisions. 
<ul><li>A few channels had previously used default HV values, which were far too high. These channels were set to more reasonable values. 
<li>Many other channels had corrections loaded from Martin v.d.Mey's analysis of a PES source run.</ul> 
See plug E-log for full details.
<!-- Author: Erik, Alon, Mike L. -->



<!-- Date: Thu Sep 27 14:29:23 2001 -->
<pre>All the monitor processes crashed again.  Alerted Matt, who in turn is contacting Kaori.</pre>
<!-- Author: M. Loreti -->


<!-- Date: Thu Sep 27 14:32:48 2001 -->
Silicon is also on now
<!-- Author: Carsten -->



<!-- Date: Thu Sep 27 14:42:17 2001 -->
<p> 
MCR call. The change in beam tilt has been completed. This resulted in increased lostp rates 
lostp = 11500 compared with 8000 previously,   
lostpb unchanged 
</p><p>
<!-- Author: Allan -->


<!-- Date: Thu Sep 27 14:46:54 2001 -->
Problems with monitoring of PEM,PHA,PSH. They suddenly turned grey after they were running fine. Erik went up to check it.
<!-- Author: Carsten -->


<!-- Date: Thu Sep 27 14:57:08 2001 -->
X-mode calibrations continuing. Apparent failure when writing into the database. Following Op-manager request, we transfer to next task: L2 cutting on jets (approx 1 hour run). Tasks are: 
<li> 2. SVX x-mode calibration 
<li> 3. Beam test of physics_0_02[1,208] with silicon in 
<li> 4. Test for L2 cutting on jets  
<li> 5. More L2 clustering (Monica, Heather) 
<li> 6. Cutting on SVT (Luciano, pager) 
  
<!-- Author: Allan -->
<!-- Comment: Thu Sep 27 15;12;48 comment by...Allan -->

SVX x-mode calibration now completed
<!-- Date: Thu Sep 27 14:59:30 2001 -->
When activate run , got the busy status - 100% busy DT and no event processed by L3.  
Reboot b0tsi00 and cleanup EVB.
<!-- Author: MooHyun -->


<!-- Date: Thu Sep 27 15:00:53 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-2449&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-2449  width='300' align='top'></td><td>SVX radiation</a></td></tr></table>
<!-- Author: Carsten -->
<!-- Comment: Thu Sep 27 15;05;04 comment by...Carsten -->
<pre>SVRAD during the shot setup (08:00-13:56):
SVRAD0=1.211
SVRAD1=0.219
SVRAD2=2.006
SVRAD3=1.930

SVRAD values before the reset at 13:56 for store 725:
SVRAD0=1.632
SVRAD1=0.219
SVRAD2=33.849
SVRAD3=27.555</pre>


<!-- Date: Thu Sep 27 15:11:26 2001 -->
Start new physics run using new physics table without silicon.
<!-- Author: MooHyun :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=127752' target=_top>127752</a>) -->
<!-- Comment: Thu Sep 27 15;21;11 comment by...MooHyun -->
DTO from CMX_01 -> HRR
<!-- Comment: Thu Sep 27 15;32;09 comment by...MooHyun -->
End the run by requset of L2 crew - they are doning L2 study.
<!-- Date: Thu Sep 27 15:20:30 2001 -->
Monitoring for PEM,PHA,PSH are fixed. Stefano is still working on the monitoring for TOF
<!-- Author: Carsten -->




<!-- Date: Thu Sep 27 15:36:50 2001 -->
CMP went into error state. Caused by bottom D4. It shows a voltage of 3021V that is 21V above set value. Talked with expert. He says it is likely that it is a read back problem, we should not worry to much about it. D4 will be on the CMP todo list, it is broken but not in a way that it corrupts that data...
<!-- Author: Carsten -->


<!-- Date: Thu Sep 27 15:54:09 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-2450&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-2450  width='300' align='top'></td><td>Shift Summary of Losses</a></td></tr></table>
<!-- Author: Carsten -->


<!-- Date: Thu Sep 27 15:55:37 2001 -->
Test of L2 cutting on Jets using Physics_0_02_v-1 table. 
At 14K events, changed the L2 prescales on L2_AUTO_L1JET5 
and L2_AUTO_L1JET10 from a value of 1 to a value of 100. 
At 37K events, reduced the prescale on L1_JET5_&_CLC_PS20 down 
to 1.  
 This is running new code with Error checks for Clist sync errors. See entry below for trigger rates. This is running 
without silicon.
<!-- Author: Stephen, Peter Wittich :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=127754' target=_top>127754</a>) -->


<!-- Date: Thu Sep 27 15:57:57 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-2451&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-2451  width='300' align='top'></td><td>Trigger Rates with cutting on L2 Jets</a></td></tr></table>
<!-- Author: Stephen, Peter -->


<!-- Date: Thu Sep 27 15:59:18 2001 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2001&action=view&page=-2452&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2001&action=view&page=-2452  width='300' align='top'></td><td>Shift summary luminosity</a></td></tr></table>
<!-- Author: Carsten -->


<!-- Date: Thu Sep 27 16:00:56 2001 -->
No radiation for store 725 so far.
<!-- Author: Carsten -->


<!-- Date: Thu Sep 27 16:11:43 2001 -->
<font color=red>Shift Summary: </font><br><font size=+1><strong>Day Shift Summary by SciCo</strong></font><br> 
<p> 
08:00	Shift start. Cryo is pumping the Watt can, and doing some work on S1, S3. 
	MCR have announced a Controlled Access 8:30-9:30 
</p><p> 
08:30	Controlled Access 
	Napora and Slaughter remove 2 SVX power supplies temporarily used for L00. 
	Roser et al. opened West IMU steel all the way back to see whether this  
	change in shielding helps cmp HV trips.The answer is NO. See below. 
</p><p> 
09:30	Controlled access completed, shot setup started. 
</p><p> 
09:45	Attempt an SVX calibration. This fails because od CSL. Experts come in. 
	CSL had a problem on the monitoring program, and stopped functioning  
	probably because the shared memory could not be accessed from other  
	CSL programs. CSL restarted. Work completed 10:55.  
	For this reason, silicon D-mode calibrations abandonned, and L2  
	Isolist/L2 clustering studies were made using software event builder. 
<li>	There is a request for SVX D-mode calibrations asap. 
</p><p> 
10:15	Brings voltages up after the flodd of previous 2 shifts, and the contolled 
	access: 
<li>	Long period to restrat monitoring of GAM,CHA,CEM,WHA, in particular GAM which remained out of commission until ~12:00. 
<li>	Repeated BMU trips: BMU24. Expert called. Voltage lowered to 2000V, an  
access is required to understand cause. 
</p><p> 
11:45	L2 isolation studies concluded (10:45),  
	further studies required. Further CLIST  
	studies by Monica and Heather (finished 13:50). 
	Shot setup starts with single bunch injection. Magnet ramped up (completed  
	12:30). 
</p><p> 
12:00	CLC gain calibration completed, followed by COSMIC runs to check. 
</p><p> 
12-13:30 Additional work included: 
<li>	 new b0puls01 code to fix error messages (Arnd) 
<li>	 reboot of ront-end crates. b0mutr00 and b0svt02 
<li>	 cdfsoftb0 release 4.0.0d dclared as  current default version for  
	 consumer tasks.  
<li>	 Plug HV values reset 
</p><p> 
15:00	Stable beam declared at 15:00 ... L=5.3E30, with Lostp = 8KHz 
	The following things have been done..... 
<li>	Beam tilted to +500 MCR units (x) and +150 MCR units (y) 
<li>	Silicon taken out of DAQ for X-mode calibrations. 
<li>	During this time run Physics_0_02[1,208] w/o SVX. Attempt to run monitor 
	tasks not successful, and at end of shift Kaori/Matt were working on it. 
<li>	Trigger table to Physics_0_02[1,208], again w/o silicon, is used to  
	test L2 jet cutting. This is in progress (127754) 
<li>	Next step is to integrate the SVX into this trigger table, and if necessary 
	to fine-tune the beam displacement after checks. 
</p>   
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   127728-127754
Delivered Luminosity   34.9  
Acquired Luminosity    15.4  
Efficiency             44%
   
</td></tr></table></pre>
<!-- Author: Allan -->