<!-- Title: Owl shift. Thu Jul 20, 2006 -->
<!-- SciCo: Marco Rescigno  -->
<!-- DAQAce: Enrique Palencia -->
<!-- MonAce: Larry Tarini -->
<!-- CO: Alexander Paramonov -->
<!-- OpManager: Tom Wright -->
<!-- Notes: -->
<pre>Store 4841 continuing BOLum 117 E30 
Keep going.</pre>


<!-- Date: Thu Jul 20 00:16:34.585 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-6870&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6870' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-6871&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6871' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-6872&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6872' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-6873&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6873' width='300'></a></center>
<!-- Author: Dale -->
<!-- Comment: Thu Jul 20 00;21;11 comment by...Dale Stentz -->
First 2 plots are the TSU - Tom and Eric took care of the high occupancy so these (as desired) come up flat empty on the west TSU plots in Ymon and Trigmon, respectively.<br><br>
The last 2 plots are the BMU - looks like the whole NW is gone due to the PS problem noted in the previous shift.<br><br>Needless to say, both should be marked bad (IMU and ISU) and DQMon should spot these quick if it hasn't already.

<!-- Comment: Thu Jul 20 00;24;27 comment by...Kike -->
Yes, DQMon already complaint about IMU and ISU.

<!-- Date: Thu Jul 20 00:26:30.157 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220134>220134</A>
trigger: <tt>TrigMOn #139 MuonTriggerMonitor 1.4.6.1: West TSU TrigData/SimComparison<br>
is empty. This is cause by the TSU problem.</tt>
<!-- Author: Consumer Operator( Alexander Paramonov) -->



<!-- Date: Thu Jul 20 00:27:46.027 2006 -->
<TABLE bgcolor=#cdc9a5> 
<TR><TH>Date</th><th>Time</th><th>BLM</th><th colspan=2>Dose</th></tr> 
<tr><td>2006.07.20</td><td>00:25:40</td><td>W Inner BLM</td><td>53.16</td><td>RADS</td></tr> 
<tr><td>2006.07.20</td><td>00:25:40</td><td>W Outer BLM</td><td>26.58</td><td>RADS</td></tr> 
<tr><td>2006.07.20</td><td>00:25:40</td><td>E Inner BLM</td><td>53.16</td><td>RADS</td></tr> 
<tr><td>2006.07.20</td><td>00:25:40</td><td>E Outer BLM</td><td>53.16</td><td>RADS</td></tr> 
</TABLE> 
<strong>Integrated dosage</strong>
<!-- Author: Marco  :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220134' target=_top>220134</a>) -->


<!-- Date: Thu Jul 20 00:34:29.363 2006 -->
<pre>Trigger Inhibit is set from the components: 
 CPR,CCR: PC  		[033] Central Preradiator PC 
 CLC: PC  		[034] CLC PC 
 MP: PC  		[036] Miniplug PC 
 BSC: PC  		[037] Beam Shower Counter PC 
</pre>
<!-- Author: Kike :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220134' target=_top>220134</a>) -->
<!-- Comment: Thu Jul 20 00;48;54 comment by...Larry -->
Followed instructions to restart monitor on CLC PC.  CPR does not start automatically, east and west must both be set to ON.  They are VERY slow responding!


<!-- Comment: Thu Jul 20 01;32;07 comment by...Larry -->
iFIX may need to be rebooted: the iFIX Background Server screen is blank, is that normal?  Also, there are multiple copies of several of the displays, can they be eliminated without rebooting? 
<!-- Date: Thu Jul 20 00:54:28.315 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-6874&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6874' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-6875&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6875' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-6876&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6876' width='300'></a></center>
<!-- Author: Larry 23:00 -&gt; 0:00 -->


<!-- Date: Thu Jul 20 01:03:26.882 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-6877&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6877' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-6878&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6878' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-6879&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6879' width='300'></a></center>
<!-- Author: Larry 0:00 -&gt; 1:00 -->


<!-- Date: Thu Jul 20 01:15:00.048 2006 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2006&action=view&page=-6881&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-6881  width='300' align='top'></a></td><td>CSX occupancy. There are 4 ot more contiguous dead channels</td></tr></table>
<!-- Author: Kike -->
<!-- Comment: Thu Jul 20 01;29;27 comment by...Marco -->
CSX pager suggested to wait some more time to see if entries
appear, and page him again in case not.
<!-- Comment: Thu Jul 20 01;58;02 comment by...Marco  -->
Expert called again and told us they are going to look after this during the day. The fix may require access. We will mark 
the run BAD for CSX.
<!-- Comment: Thu Jul 20 06;39;18 comment by...jdl -->
I'm no expert, but that looks like a dead discriminator.
<!-- Comment: Thu Jul 20 06;51;06 comment by...Dale -->
A sad day for muons; we will have to mark CMX bad to safe guard the trigger.
<!-- Comment: Thu Jul 20 10;40;58 comment by...Phil Schlabach -->
I think it maps to crate 2, slot 4.
<!-- Comment: Thu Jul 20 15;27;57 comment by...Guram -->
Discriminator is replaced.
<!-- Date: Thu Jul 20 01:18:16.523 2006 -->
<pre>b0pcal08:Messenger:Jul 20, 2006 01:15:36 AM->Heap is corrupt, event 2101404; bytes free 24219424, bytes alloc 5809968; errors -1 -->   
 Additional Information:  
 
 Attention !!!. FERML_HEAP_CORRUPT ERROR !!!  
  
 - Heap Corrupt for crate  b0pcal08  
 - Please SHEPHERD crate b0pcal08.  
[MLE] 
(MLE) b0svx08:Messenger:Jul 20, 2006 01:16:11 AM->Silicon Resonator: Sl 16 Ch 3 -> f843: Bytes: 00780 
 
(MLE) b0svx07:Messenger:Jul 20, 2006 01:16:11 AM->Silicon Resonator: Sl 13 Ch 3 -> fa23: Bytes: 00780 
 
(MLE) b0felix.fnal.gov:TEDCpp:Jul 20, 2006 01:16:14 AM->{ 1:16:14, 7/20/2006 }: L2-Pulsar-Linux: Halted while  processing happily 
(MLE) b0svx03:Messenger:Jul 20, 2006 01:16:26 AM->Silicon Resonator: Sl 10 Ch 9 -> e334: Bytes: 00064 
unction (S_errno_ETIMEDOUT) 
b0control21.fnal.gov:Thread-26:Jul 20, 2006 01:17:00 AM->Successfully shepherded client PCAL_08 (b0pcal08) -->   
 Additional Information:  
 
 Attention!!!. FERML_SHEPHERD_SUCCESS !!!  
  
 Successfully Shepherded b0pcal08  
 an additional AUTO HRR will be issued.   
[MLE] 
(RC)   1:17:00 Halt -> HALTED 
(RC)   1:17:26 Recover -> RECOVERED 
(RC)   1:17:55 Run -> ACTIVE 
</pre>
<!-- Author: Kike :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220134' target=_top>220134</a>) -->





<!-- Date: Thu Jul 20 02:06:28.125 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-6882&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6882' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-6883&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6883' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-6884&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6884' width='300'></a></center>
<!-- Author: Larry 1:00 -&gt; 2:00 -->


<!-- Date: Thu Jul 20 02:28:54.905 2006 -->
<table><tr><td><a href=/elog/notebooks/2006/Thu_Jul_20_2006_Owl.022852.ps align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-6886  align='top'></a></td><td>Obj. Mon. is not happy :(</td></tr></table>
<!-- Author: Alexander Paramonov -->
<!-- Comment: Thu Jul 20 12;32;31 comment by...peter -->
All these red plots point to a change in the trigger tables. So, nothing to worry about. I updated the ObjectMon reference histograms. Please let me know if there are problems or if the histograms are still red in the next days.
<!-- Date: Thu Jul 20 03:05:53.761 2006 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2006&action=view&page=-6887&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-6887  width='300' align='top'></a></td><td>pinkie! BIAS B2 W7 L0</td></tr></table>
<!-- Author: Larry -->
<!-- Comment: Thu Jul 20 03;09;43 comment by...Larry -->
Turned green, but went back to pink almost immediately.  Judging from the history, looks more like "just above the limit" than "large jump".  No page for now, will monitor closely.


<!-- Date: Thu Jul 20 03:19:57.225 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-6892&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6892' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-6893&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6893' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-6894&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6894' width='300'></a></center>
<!-- Author: Larry 2:00 -&gt; 3:00 -->


<!-- Date: Thu Jul 20 03:32:42.591 2006 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2006&action=view&page=-6895&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-6895  width='300' align='top'></a></td><td>B2 W7 L0 appears to have stabilized at new level.</td></tr></table>
<!-- Author: Larry -->


<!-- Date: Thu Jul 20 04:07:50.774 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-6896&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6896' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-6897&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6897' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-6898&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6898' width='300'></a></center>
<!-- Author: Larry 3:00 -&gt; 4:00 -->

<!-- Date: Thu Jul 20 04:19:10.775 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220134>220134</A>
 Terminated at 2006.07.20 04:18:37
<!-- Author: RunControl -->


<!-- Date: Thu Jul 20 04:19:11.452 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220134>220134</A>
TERMINATE: end run. Tevatron abort.
<!-- Author: Kike -->
<!-- Comment: Thu Jul 20 04;22;30 comment by...Kike -->
<pre>38 crate/s: b0imu00(80), b0imu01(81), b0cmu01(82), b0cmu00(83), b0cmp00(84), b0musc00(85), b0cmx00(86), b0cmx01(87), b0xtrp00(96), b0xft01(98), b0xft03(100), b0xft05(102), b0xft06(103), b0xft07(106), b0svt09(116), b0svt08(117), b0clc00(118), b0svt00(120), b0svt01(121), b0svt07(122), b0svt02(123), b0svt06(124), b0svt03(125), b0svt05(126), b0svt04(127), b0tsi03(138), b0puls00(139), b0l2cl00(144), b0l2cl01(145), b0l2cl02(146), b0l2cl03(147), b0l2cl04(148), b0l2cl05(149), b0l2pu01(150), b0l2pu02(151), b0l2pu00(152), b0htdc00(153), b0etdc00(154),  in error. [RXPT]
(MLW) b0control21.fnal.gov:Thread-26:Jul 20, 2006 04:18:59 AM->Inhibit flag CLEAR b0eproxy02.fnal.gov:main:Jul 20, 2006 04:20:53 AM->Error on L3 node b0eproxy02.fnal.gov (partition 0) Thu Jul 20 04:20:53 2006 l3proxy 
  in End (L3ProxyPartition.cc:3043)
  @L3_RELAY_FAILURE Level3 Relay failure detected!
The following subfarms need to be removed from the next runs:
      None. You can use all the same subfarms.
The following Level3 converter nodes are marked bad for now:
      None.
The full list of Level3 nodes giving Relay trouble is printed below

 b0l3s09p12

 ************************************************
 These Level3 nodes will be marked OFFLINE!!!
 Abort/End/Reset and Partition again!!!
 ************************************************

 -->  
 Additional Information: 

 Attention!!!. L3_RELAY_FAILURE Error !!!. 
 Level3 internal communications failure. One or more L3 nodes is likely
 to have died.
   Do not panic! Level3 will try to repair itself. We recommend:
     - do End/Abort+Reset ON ALL DEFINED PARTITIONS   
     - exclude all affected subfarms from your run configuration
       (find out list of bad subfarms from Error Logger: 
          if None, exclude nothing!)
     - start another run and page L3 farm pager at closest day/evening shift
     - if >2 subfarms affected page L3 farm IMMEDIATELY
[MLE]
(MLE) b0eproxy02.fnal.gov:main:Jul 20, 2006 04:20:54 AM->Error on L3 node b0eproxy02.fnal.gov (partition 0) Thu Jul 20 04:20:53 2006 l3proxy 
  in End (L3ProxyPartition.cc:2234)
  @L3_TRANS_FAIL - transition actions failed:      b0l3s09p12(Unrecognized response type for request)

38 crate/s: b0imu00(80), b0imu01(81), b0cmu01(82), b0cmu00(83), b0cmp00(84), b0musc00(85), b0cmx00(86), b0cmx01(87), b0xtrp00(96), b0xft01(98), b0xft03(100), b0xft05(102), b0xft06(103), b0xft07(106), b0svt09(116), b0svt08(117), b0clc00(118), b0svt00(120), b0svt01(121), b0svt07(122), b0svt02(123), b0svt06(124), b0svt03(125), b0svt05(126), b0svt04(127), b0tsi03(138), b0puls00(139), b0l2cl00(144), b0l2cl01(145), b0l2cl02(146), b0l2cl03(147), b0l2cl04(148), b0l2cl05(149), b0l2pu01(150), b0l2pu02(151), b0l2pu00(152), b0htdc00(153), b0etdc00(154),  in error. [RXPT]
(RC)   4:21:25 Reset -> START
</pre>
<!-- Date: Thu Jul 20 04:20:28.657 2006 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2006&action=view&page=-6899&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-6899  width='300' align='top'></a></td><td>Plot Abort data.</td></tr></table>
<!-- Author: Larry -->
<!-- Comment: Thu Jul 20 04;21;23 comment by...Larry -->
No new pinkies due to abort.


<!-- Comment: Thu Jul 20 04;29;30 comment by...Larry -->
CMX wedge 21 got stuck ON after End of Store -- ramped to STBY manually.
<!-- Comment: Thu Jul 20 04;45;06 comment by...Larry -->
PEM yellow warning in iFIX, HV summary has Crate 12 red. Followed online instructions for PEM-PHA-PSH.  Nothing amiss with switches or displays on 3rd floor (other than alarm state for crate). Instructions say to cycle HV: ramped HV to ON. Crate 6 went red, then cleared. Crate 12 remained red. Ramped to STBY.
<!-- Comment: Thu Jul 20 04;47;02 comment by...Larry -->
Red Crate 12 did not clear.  Ramped to OFF.

<!-- Comment: Thu Jul 20 04;48;29 comment by...Larry -->
Red crate 12 cleared. Ramped back to STBY to be ready for calibrations.
<!-- Comment: Thu Jul 20 06;53;18 comment by...Dale -->
[conserning CMX 21 issue] I believe that something like this happened durning the last abort in the previous store.
<!-- Date: Thu Jul 20 04:21:25.223 2006 -->
Tevatron beam lost. MCR will call when they have a plan
<!-- Author: Marco  :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220134' target=_top>220134</a>) -->


<!-- Date: Thu Jul 20 04:25:51.105 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220135>220135</A>
TERMINATE: did the coldstart through config.
<!-- Author: Kike -->



<!-- Date: Thu Jul 20 04:26:59.380 2006 -->
<table><tr><td><a href=/cgi/elog/elog.pl?nb=2006&action=view&page=-6900&button=yes target='xxx' onclick='jswindow()'; align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-6900  width='300' align='top'></a></td><td>Abort data -- no shadow. Sorry!</td></tr></table>
<!-- Author: Larry -->




<!-- Date: Thu Jul 20 04:34:13.383 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220136>220136</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Kike -->


<!-- Date: Thu Jul 20 04:34:31.738 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220136>220136</A>
TERMINATE: end CAL QIE calib.
<!-- Author: Kike -->


<!-- Date: Thu Jul 20 04:35:22.650 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220136>220136</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Thu Jul 20 05;48;00 comment by...Sasha -->
<pre>OK</pre>
<!-- Date: Thu Jul 20 04:35:44.025 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220137>220137</A>
ACTIVATE: RunSet: CALIB_CLC_QIE 
<!-- Author: Kike -->


<!-- Date: Thu Jul 20 04:36:38.543 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220137>220137</A>
CALIBRATION: <U><B>clcqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#clcqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Thu Jul 20 05;46;23 comment by...Sasha -->
<pre>OK</pre>
<!-- Date: Thu Jul 20 04:36:39.728 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220137>220137</A>
TERMINATE: end CLC QIE calib,
<!-- Author: Kike -->
<!-- Comment: Thu Jul 20 04;39;56 comment by...Kike -->
<pre>Cannot reset the CLC ADMENs since I cannot connect to b0clc00:

b0control21:136:~&gt; setup fer
setup-ppc: assuming PowerPC 603 processor
b0control21:137:~&gt; vxlogin b0clc00
spawn rsh b0clc00
b0clc00.fnal.gov: Connection refused
trying normal rlogin (/usr/bin/rlogin)  WARNING: NO ENCRYPTION!

Sorry, this system is engaged by user: cdfdaq.
rlogin: connection closed.
spawn_id: spawn id exp4 not open
    while executing
"interact
"
    (file "/cdf/onln/code/cdfprod/fer/v4_96/src//vxlogin" line 23)
</pre>

<!-- Date: Thu Jul 20 04:41:27.268 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220138>220138</A>
ACTIVATE: RunSet: CALIB_QIE_FWD 
<!-- Author: Kike -->


<!-- Date: Thu Jul 20 04:42:57.723 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220138>220138</A>
CALIBRATION: <U><B>bscqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#bscqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Thu Jul 20 05;13;26 comment by...Kike -->
<pre>Calibration does not look good. Following the recovery procedure, I powercycled fcal crates. Will try another calibration.</pre>
<!-- Date: Thu Jul 20 04:42:58.171 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220138>220138</A>
TERMINATE: end BSC QIE calib.
<!-- Author: Kike -->





<!-- Date: Thu Jul 20 04:52:14.260 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220139>220139</A>
 Activated at 2006.07.20 04:51:52
<!-- Author: RunControl -->


<!-- Date: Thu Jul 20 04:52:15.985 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220139>220139</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [7,792,632] 
<!-- Author: Kike -->


<!-- Date: Thu Jul 20 04:58:59.948 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220139>220139</A>
 Terminated at 2006.07.20 04:58:37
<!-- Author: RunControl -->



<!-- Date: Thu Jul 20 04:58:59.427 2006 -->
PEM hung up again (crate 12) while ramping to ON for calibration. Ramping to STBY.
<!-- Author: Larry :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220139' target=_top>220139</a>) -->
<!-- Comment: Thu Jul 20 05;04;50 comment by...Larry -->
Still red--ramping to OFF. Red crate 12 cleared but HV Stat Val in iFIX says ON for PEM PHA and PSH.
<!-- Comment: Thu Jul 20 05;14;49 comment by...Larry -->
Ramped to STDBY. Crate 12 turned red again. Ramped to ON. Crate 12 cleared.  We're in business.
<!-- Date: Thu Jul 20 04:59:49.246 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220139>220139</A>
TERMINATE: end PMT Spike run. Average L1A rate 75.2 Hz
<!-- Author: Kike -->



<!-- Date: Thu Jul 20 05:12:43.834 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220140>220140</A>
ACTIVATE: RunSet: CALIB_QIE_FWD 
<!-- Author: Kike -->



<!-- Date: Thu Jul 20 05:13:53.272 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220140>220140</A>
CALIBRATION: <U><B>bscqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#bscqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Thu Jul 20 05;32;01 comment by...Sasha -->
<pre>Now bscqie looks OK.</pre>
<!-- Date: Thu Jul 20 05:13:54.690 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220140>220140</A>
TERMINATE: end BSC QIE calib.
<!-- Author: Kike -->



<!-- Date: Thu Jul 20 05:16:56.010 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220141>220141</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL 
<!-- Author: Kike -->



<!-- Date: Thu Jul 20 05:18:12.947 2006 -->
<center><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-6902&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6902' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-6903&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6903' width='300'></a><a href='/cgi/elog/elog.pl?nb=2006&action=view&page=-6904&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6904' width='300'></a></center>
<!-- Author: Larry 4:00 -&gt; 5:00 -->

<!-- Date: Thu Jul 20 05:24:09.932 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220141>220141</A>
TERMINATE: end BOTHPLUGS_NORMAL calib.
<!-- Author: Kike -->


<!-- Date: Thu Jul 20 05:25:07.659 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220142>220142</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: Kike -->



<!-- Date: Thu Jul 20 05:25:10.845 2006 -->
<table><tr><td><a href=/elog/notebooks/2006/Thu_Jul_20_2006_Owl.052507.ps align=top><img src=/cgi/elog/elog.pl?nb=2006&action=view&page=-6906  align='top'></a></td><td>New hot towers.</td></tr></table>
<!-- Author: Sasha -->
<!-- Comment: Thu Jul 20 08;34;42 comment by...m mattson (ADMEM) -->
On the reference page, we define "hot" as anything with a Sigma over 10.  Once you ignore PHA ch 0 and 1, all of the other channels in this list are okay (abnormal enough to keep tabs, but good for data taking).
<!-- Date: Thu Jul 20 05:27:46.822 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220142>220142</A>
TERMINATE: end BOTHPLUGS_MM_C calib.
<!-- Author: Kike -->


<!-- Date: Thu Jul 20 05:27:47.642 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220142>220142</A>
CALIBRATION: <U><B>pluglaser</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#pluglaser target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Thu Jul 20 05;54;09 comment by...Sasha -->
<pre>OK</pre>
<!-- Date: Thu Jul 20 05:29:05.668 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220143>220143</A>
ACTIVATE: RunSet: BOTHPLUGS_MM 
<!-- Author: Kike -->


<!-- Date: Thu Jul 20 05:30:25.165 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220143>220143</A>
TERMINATE: end BOTHPLUGS_MM calib.
<!-- Author: Kike -->
<!-- Comment: Thu Jul 20 05;35;40 comment by...Larry -->
After Plug calibration PEM gave yellow iFIX warning for crate 12 when ramped to STDBY. Ramped to OFF; yellow warning went away but HB Stat Val indicates ON for whole plug.  Time to proceed to Recovery Step 1?
<!-- Comment: Thu Jul 20 06;21;58 comment by...Larry -->
Proceeded to Recovery Step 1. Crate 12 reset. Check Passed LED did not come on.  Killed Plug HV monitor program. Switched OFF crate 12.  Waited 10 minutes. Switched crate ON.  LED came back on. Restarted GUIs.  Everything Okay and green in Detector On/Off Status GUI except "Trigger Inhibit Write Error", WTF? Downstairs, yellow warning for PEM cleared in iFIX.  Going to Recovery Step 4. 
<!-- Comment: Thu Jul 20 06;29;19 comment by...Larry -->
Check that -- paging Expert instead.
<!-- Comment: Thu Jul 20 06;49;29 comment by...Larry -->
BTW--please change documentation to state that no login or password is required for this procedure.  If you try to do so, user cdf-cap-cdfephv is NOT recognized by security, even with correct password.
<!-- Comment: Thu Jul 20 06;50;29 comment by...Larry -->
Willis thought the "Trigger Inhibit Write Error" was a communication problem with iFIX and he will look at it around 8 am.
<!-- Date: Thu Jul 20 05:33:31.666 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220144>220144</A>
CALIBRATION: <U><B>tofqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#tofqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Thu Jul 20 06;08;51 comment by...Sasha -->
<pre>OK</pre>
<!-- Date: Thu Jul 20 05:33:32.682 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220144>220144</A>
ACTIVATE: RunSet: ACE_TOF_QIE_NEW 
<!-- Author: Kike -->


<!-- Date: Thu Jul 20 05:33:52.214 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220144>220144</A>
TERMINATE: end TOF QIE calib.
<!-- Author: Kike -->



<!-- Date: Thu Jul 20 05:39:37.488 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220145>220145</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Kike -->


<!-- Date: Thu Jul 20 05:41:43.158 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220145>220145</A>
TERMINATE: end SMX QIE calib.
<!-- Author: Kike -->


<!-- Date: Thu Jul 20 05:42:37.229 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220145>220145</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Thu Jul 20 06;03;07 comment by...Kike -->
Looks bad. Will do it again.
<!-- Date: Thu Jul 20 05:43:22.500 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220146>220146</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: Kike -->


<!-- Date: Thu Jul 20 05:45:30.956 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220146>220146</A>
CALIBRATION: <U><B>cemled</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cemled target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Thu Jul 20 05;51;25 comment by...Sasha -->
<pre>OK.</pre>
<!-- Date: Thu Jul 20 05:45:41.956 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220146>220146</A>
TERMINATE: end CEM LED calib.
<!-- Author: Kike -->



<!-- Date: Thu Jul 20 05:47:51.741 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220147>220147</A>
TERMINATE: end COT calib.
<!-- Author: Kike -->


<!-- Date: Thu Jul 20 05:47:52.594 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220147>220147</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Kike -->



<!-- Date: Thu Jul 20 05:50:06.836 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220147>220147</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Thu Jul 20 06;20;06 comment by...Sasha -->
<pre>OK</pre>
<!-- Date: Thu Jul 20 05:50:07.455 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220148>220148</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: Kike -->


<!-- Date: Thu Jul 20 05:51:24.827 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220148>220148</A>
CALIBRATION: <U><B>calxef</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calxef target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->


<!-- Date: Thu Jul 20 05:51:25.172 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220148>220148</A>
TERMINATE: end CEM XEF calib.
<!-- Author: Kike -->



<!-- Date: Thu Jul 20 05:53:33.824 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220149>220149</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1 
<!-- Author: Kike -->



<!-- Date: Thu Jul 20 05:54:11.084 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220149>220149</A>
TERMINATE: end Muon calib.
<!-- Author: Kike -->


<!-- Date: Thu Jul 20 05:56:37.060 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220150>220150</A>
ACTIVATE: RunSet: TOF_TAC_CALIB 
<!-- Author: Kike -->


<!-- Date: Thu Jul 20 06:00:51.233 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220150>220150</A>
TERMINATE: end TOF TAC calib.
<!-- Author: Kike -->


<!-- Date: Thu Jul 20 06:00:52.301 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220150>220150</A>
CALIBRATION: <U><B>toftac</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#toftac target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Thu Jul 20 06;07;37 comment by...Sasha -->
<pre>OK</pre>
<!-- Date: Thu Jul 20 06:02:08.535 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220151>220151</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Kike -->



<!-- Date: Thu Jul 20 06:03:35.207 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220151>220151</A>
TERMINATE: end SMX QIE calib.
<!-- Author: Kike -->


<!-- Date: Thu Jul 20 06:05:17.688 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220151>220151</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Thu Jul 20 06;09;32 comment by...Sasha -->
<pre>ok</pre>

<!-- Date: Thu Jul 20 06:11:38.101 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220152>220152</A>
 Activated at 2006.07.20 06:11:03
<!-- Author: RunControl -->


<!-- Date: Thu Jul 20 06:11:39.194 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220152>220152</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,730,639] 
<!-- Author: Kike -->


<!-- Date: Thu Jul 20 06:18:42.375 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220152>220152</A>
 Terminated at 2006.07.20 06:18:21
<!-- Author: RunControl -->


<!-- Date: Thu Jul 20 06:18:43.586 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220152>220152</A>
TERMINATE: end run to redo CEM XEF cali.
<!-- Author: Kike -->



<!-- Date: Thu Jul 20 06:21:09.355 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220153>220153</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: Kike -->



<!-- Date: Thu Jul 20 06:23:03.576 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220153>220153</A>
CALIBRATION: <U><B>calxef</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calxef target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->


<!-- Date: Thu Jul 20 06:23:09.485 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220153>220153</A>
TERMINATE: end CEM XEF calib.
<!-- Author: Kike -->


<!-- Date: Thu Jul 20 06:27:28.587 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220154>220154</A>
 Activated at 2006.07.20 06:25:48
<!-- Author: RunControl -->


<!-- Date: Thu Jul 20 06:27:29.765 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220154>220154</A>
ACTIVATE: RunSet: COSMICS_NOSILICON_TAPE  Trigger: COSMICS_3D [2,799,639] 
<!-- Author: Kike -->




<!-- Date: Thu Jul 20 06:35:33.026 2006 -->
<center><a href='/elog/notebooks/2006/Thu_Jul_20_2006_Owl.063440.ps'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6908' ></a><a href='/elog/notebooks/2006/Thu_Jul_20_2006_Owl.063459.ps'><img src='/cgi/elog/elog.pl?nb=2006&action=view&page=-6910' ></a></center>
<!-- Author: Sasha -->
<!-- Comment: Thu Jul 20 06;36;11 comment by...Sasha -->
<pre>calxef calibration looks funny. Too many dead channels.</pre>







<!-- Date: Thu Jul 20 06:55:28.763 2006 -->
<table border=0><td><tr><td bgcolor=#e0e030>Attention !!!.  CER_XMON_SCALER_SYNCH_ERROR !!!.  
 CDF missed a B0 marker !. This suggest a serious clock problem. 
 We STRONGLY SUGGEST to: 
 - End the Run 
 - Check the clock for errors.  
[MLE] 
(MLE) b0felix.fnal.gov:TEDCpp:Jul 20, 2006 06:54:58 AM->{ 6:54:58, 7/20/2006 }: L2-Pulsar-Linux: Halted while  waiting for ClusterFinder data 
(RC)   6:54:59 Halt -> HALTED 
</td></tr></td></table>
<!-- Author: Kike :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220154' target=_top>220154</a>) -->
<!-- Comment: Thu Jul 20 07;09;23 comment by...Ankush -->
<pre>did anyone look at the clock crate ??</pre>
<!-- Comment: Thu Jul 20 07;23;10 comment by...Larry -->
Clock errors on RHS module (sequencer?) will not clear.
<!-- Comment: Thu Jul 20 07;23;48 comment by...Kike -->
Yes, we did. We are not sure though if it is ok... will call Steve. 
<!-- Comment: Thu Jul 20 07;57;59 comment by...Steve Hahn -->
<table border=0><td><tr><td bgcolor=#e0e030><p>There was no clock error.  According to the clock page (link at the top of
the main online page), there has not been a
clock error since July 3.  I have to conclude the above error is somehow internal
to the scaler crate; I wouldn't be surprised if it was somehow associated with a
lightning strike as this would explain the coincidental silicon problems.<p>

<p>There seems to be some confusion on the part of the shift crew.  The error
LEDs in one group on the sequencer marked 371, 742, 1113, and BC PH are 
indicators that there are either empty beam crossings or beam outside beam
crossings.  They should not be present when colliding beam is in the Tevatron, but
they <u>should</u> be present when there is not collisions.  This is all explained
on the DAQ ace clock page.</p></td></tr></td></table>
<!-- Date: Thu Jul 20 06:56:56.882 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220154>220154</A>
 Terminated at 2006.07.20 06:56:35
<!-- Author: RunControl -->


<!-- Date: Thu Jul 20 06:56:57.616 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220154>220154</A>
TERMINATE: end run.
<!-- Author: Kike -->


<!-- Date: Thu Jul 20 07:04:47.488 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220155>220155</A>
ACTIVATE: RunSet: COLDSTART  Trigger: COSMICS [15,730,598] 
<!-- Author: larry x2080 -->


<!-- Date: Thu Jul 20 07:05:20.920 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220156>220156</A>
 Activated at 2006.07.20 07:04:53
<!-- Author: RunControl -->


<!-- Date: Thu Jul 20 07:05:21.739 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220156>220156</A>
ACTIVATE: RunSet: COSMICS_NOSILICON_TAPE  Trigger: COSMICS_3D [2,799,639] 
<!-- Author: Kike -->


<!-- Date: Thu Jul 20 07:07:52.073 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220155>220155</A>
TERMINATE: ending run
<!-- Author: larry x2080 -->



<!-- Date: Thu Jul 20 07:12:59.807 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220157>220157</A>
TERMINATE: need another config
<!-- Author: larry x2080 -->


<!-- Date: Thu Jul 20 07:16:35.328 2006 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220158>220158</A>
TERMINATE: end to reconfig
<!-- Author: larry x2080 -->



<!-- Date: Thu Jul 20 07:19:08.422 2006 -->
MCR called in to tell they are going to inject some beam in the Tevatron in few minutes
<!-- Author: Marco  -->




<!-- Date: Thu Jul 20 07:25:57.067 2006 -->
At almost exact same time as RC Clock Error message, had iFIX alarm for SVC: PS trips for B0W1, B0W8, B1W7, B1W8, B2W4, B3W11, B5W0. Paged Silicon. Followed Ankush's instructions re: clock resync and Si coldstart; however trip B2W4 reappeared along with a single ladder in B0W8.  No further action till Ankush arrives.
<!-- Author: Larry :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220156' target=_top>220156</a>) -->


<!-- Date: Thu Jul 20 07:26:00.020 2006 -->
<h3>Clock glitch</h3> 
 
<p>The CDF clock had an errors and the Silicon Clock Cloning Card (CCC) switched to its internal mode.  I switched the CCC to lock to the CDF clock and instructed the ACE to recover the tripped ladders. However some of these ladders would trip again.  Also the CDF clock still appears to be in error.</p> 
 
<p>So I will wait for the CDF clock to be error free and then I will recover the still tripped silicon ladders.</p>
<!-- Author: Ankush -->
<!-- Comment: Thu Jul 20 07;32;38 comment by...Larry -->
Per Steve Hahn, CDF clock "errors" are due to lack of beam and not an actual error.  


<!-- Comment: Thu Jul 20 13;13;52 comment by...greg feild -->
<pre>XMON SCALER SYNC ERRORS

The scaler crate has a circuit that counts the number 
of clock ticks between B0 markers.  If this number is
different than 159 then an error is asserted.  
Experience says that this error is real and the BO can 
be misplaced.

Doubters can hook up a scope or LA (if there are any 
free ones left!) to a TRACER pod and trigger on this 
condition themselves!

 </pre>
<!-- Date: Thu Jul 20 07:50:45.382 2006 -->
<pre>(MLW) b0control21.fnal.gov:Thread-26:Jul 20, 2006 07:49:41 AM-&gt;Received error message: errmon Not a state HaltRecoverRun  Busy Timeout. [RXPT](MLE) b0control21.fnal.gov:Thread-26:Jul 20, 2006 07:49:41 AM-&gt;Requested Halt-Recover-Run issued (errmon) 
(MLE) b0control21.fnal.gov:Thread-26:Jul 20, 2006 07:49:41 AM-&gt;Busy Timeout: XFT_FINDER_07  
(MLE) b0felix.fnal.gov:TEDCpp:Jul 20, 2006 07:49:50 AM-&gt;{ 7:49:50, 7/20/2006 }: L2-Pulsar-Linux: Halted while  waiting for ClusterFinder data 
(RC)   7:49:50 Halt -&gt; HALTED 
(RC)   7:49:57 Recover -&gt; RECOVERED 
(RC)   7:49:59 Run -&gt; ACTIVE 
</pre>
<!-- Author: Kike :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220156' target=_top>220156</a>) -->

<!-- Date: Thu Jul 20 07:57:01.900 2006 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>PhysicsTable</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L3Accepts</TH>
<TH>DelivLumi nb<SUP>-1</SUP></TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD><A HREF=/java/cdfdb/servlet/RunSummary?RUN=220134 target=_runSummary>220134</A> x35BE6</TD>
<TD>BEAM</TD>
<TD><FONT SIZE=-1>PHYSICS_4_00 [4,816,640]</FONT></TD>
<TD ALIGN=CENTER>23:43:12</TD>
<TD ALIGN=CENTER>04:18:37</TD>
<TD ALIGN=CENTER>04:00:47</TD>
<TD ALIGN=RIGHT>   1,500,825</TD>   <TD ALIGN=RIGHT>  1,589.99</TD>
<TD ALIGN=RIGHT>  1,371.89</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD>
<TD ALIGN=CENTER>1</TD></TR>

<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>07:55:01</TD>
<TD ALIGN=CENTER>04:00:47</TD>
<TD ALIGN=RIGHT>     1,500,825</TD> <TD ALIGN=RIGHT>  1,589.99</TD>
<TD ALIGN=RIGHT>  1,371.89</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->




<!-- Date: Thu Jul 20 07:58:58.245 2006 -->
<font color=red>Shift Summary: </font><pre>Store 4841 colliding current BOLum 117 E30 
Two problems with forward muon detector/trigger: 
 
  TSU  off  
  BMU  Power Supply on NW side 
 
0 30 CPR:CCR PC Inhibit. Took ~10 minutes to clear. 
1 00 Paged CSX pager. Few channels missing in CSX occupancy  
     plot. Expert will look after this, during day time.  
     Affected channels IW 6-14, XW 0-4. 
1:18 PCAL08 Heap corrupt 
4:00 Severe Weather watch 
4:15 beam lost: TeV crate $F2 lost power 
4:30 Run 220134 has BMU/ISU marked bad and CSX too. 
5:00 Calibrations. SMX repeated twice. Many (likely all) the channels  
     in crate ccal03 failed. XEF also twice, but dead channel appearing: 
     wedge 2,10,20 East. All other calibrations fine.  
6:50 Big Thunder: Silicon power supply affected.  
7:15 Injecting beam into the Tevatron for tune checking at 150 GeV      
7:30 Some residual problem (ladder trip) with silicon. Silicon  
     expert coming.  
 
</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   220134
Delivered Luminosity   1455 nb-1  
Acquired Luminosity    1246 nb-1  
Efficiency             85.7%
   
</td></tr></table></pre>
<!-- Author: Marco  -->







<!-- Date: Thu Jul 20 16:06:00.978 2006 -->
another test
<!-- Author: Steve Hahn :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=220182' target=_top>220182</a>) -->