<!-- Title: Day shift. Mon Mar  5, 2007 -->
<!-- SciCo: Lucio Cerrito -->
<!-- DAQAce: Andy Kubik -->
<!-- MonAce: &nbsp; -->
<!-- CO: Diego Cauz -->
<!-- OpManager: Sasha Pronko -->
<!-- Notes: -->
<pre>Store ending. 3h controlled access to follow</pre>


<!-- Date: Mon Mar 5 08:22:22.166 2007 -->
We're starting controlled access
<!-- Author: Lucio -->
<!-- Comment: Mon Mar  5 08;29;31 comment by...Lucio -->
The access is to replace a COT power supply that failed last night
<!-- Date: Mon Mar 5 08:24:30.031 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235667>235667</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 08:25:12.464 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235667>235667</A>
TERMINATE: done
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 08:26:04.909 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235667>235667</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Mon Mar  5 10;09;49 comment by...diego -->
<pre>there are a few hot WHA channels (15W, 16W, 17W), rms a bit higher than 5
for the rest is OK</pre>
<!-- Date: Mon Mar 5 08:26:05.999 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235668>235668</A>
ACTIVATE: RunSet: CALIB_CLC_QIE 
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 08:27:19.509 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235668>235668</A>
CALIBRATION: <U><B>clcqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#clcqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Mon Mar  5 10;21;30 comment by...diego -->
<pre>calib's are fine</pre>
<!-- Date: Mon Mar 5 08:27:20.311 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235668>235668</A>
TERMINATE: done
<!-- Author: Andy -->



<!-- Date: Mon Mar 5 08:33:11.727 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235669>235669</A>
ACTIVATE: RunSet: CALIB_QIE_FWD 
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 08:34:51.091 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235669>235669</A>
CALIBRATION: <U><B>bscqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#bscqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Mon Mar  5 10;29;56 comment by...diego -->
<pre>calib's are OK</pre>
<!-- Date: Mon Mar 5 08:34:52.621 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235669>235669</A>
TERMINATE: done
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 08:46:06.671 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235671>235671</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL 
<!-- Author: Andy -->



<!-- Date: Mon Mar 5 08:47:32.483 2007 -->
Several SVX alarms. SVX people is working on it from the control room
<!-- Author: Lucio -->

<!-- Date: Mon Mar 5 08:52:52.879 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235671>235671</A>
TERMINATE: done
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 08:54:09.656 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235673>235673</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 08:56:16.116 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235673>235673</A>
TERMINATE: done
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 08:56:34.346 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235673>235673</A>
CALIBRATION: <U><B>pluglaser</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#pluglaser target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Mon Mar  5 09;42;35 comment by...diego -->
<pre>since the East Plug laser has shifted over 580 nsec, I pull it back by 10 nsec. Changed DDGDelay0 parameter from 790 to 780</pre>
<!-- Date: Mon Mar 5 08:57:27.057 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235674>235674</A>
ACTIVATE: RunSet: BOTHPLUGS_MM 
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 08:59:17.715 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235674>235674</A>
TERMINATE: done
<!-- Author: Andy -->



<!-- Date: Mon Mar 5 09:03:29.431 2007 -->
PSM alarm, Arch PS CANWT_2, +5V Dig read 2V, cleared after 15 sec.
<!-- Author: Andy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235674' target=_top>235674</a>) -->

<!-- Date: Mon Mar 5 09:05:21.555 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235675>235675</A>
 Activated at 2007.03.05 09:05:08
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 09:05:22.085 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235675>235675</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [7,792,632] 
<!-- Author: Andy -->
<!-- Comment: Mon Mar  5 09;17;57 comment by...Andy -->
L1 Accept rate for this PMT spike run is 43.7 Hz
<!-- Date: Mon Mar 5 09:06:05.870 2007 -->
Another PSM alarm, CONET_1 (East Endwall)
<!-- Author: Andy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235675' target=_top>235675</a>) -->
<!-- Comment: Mon Mar  5 12;48;05 comment by...Dervin -->
Roberto trued off PS CONET_1 so he could replace CONET_2
<!-- Date: Mon Mar 5 09:16:56.545 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235675>235675</A>
 Terminated at 2007.03.05 09:16:42
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 09:16:57.508 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235675>235675</A>
TERMINATE: done
<!-- Author: Andy -->



<!-- Date: Mon Mar 5 09:21:15.432 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235676>235676</A>
ACTIVATE: RunSet: ACE_TOF_QIE_NEW 
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 09:21:16.315 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235676>235676</A>
CALIBRATION: <U><B>tofqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#tofqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Mon Mar  5 10;36;11 comment by...diego -->
<pre>calib's are fine</pre>
<!-- Date: Mon Mar 5 09:21:17.540 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235676>235676</A>
TERMINATE: done
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 09:24:29.875 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235679>235679</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 09:25:00.508 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235679>235679</A>
TERMINATE: done
<!-- Author: Andy -->



<!-- Date: Mon Mar 5 09:26:40.194 2007 -->
I've sent an email to Greg Feild, asking to comment about the L1 input to L2_JPSI_TWO_CEM2 cross section vs. lumi seen in <a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2007&action=view&page=190&frame=2&anchor=02:20:43.847'> 
here </a>
<!-- Author: Lucio -->

<!-- Date: Mon Mar 5 09:28:48.048 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235679>235679</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Mon Mar  5 10;39;43 comment by...diego -->
<pre>calib's are fine</pre>
<!-- Date: Mon Mar 5 09:28:49.845 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235680>235680</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 09:30:50.420 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235680>235680</A>
TERMINATE: done
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 09:37:11.176 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235664>235664</A>
RUNSTATUS: <PRE>Marked Bad, explanation:
COT, COT11 missing, power supply failure 
</PRE>
<!-- Author: cdfscico -->


<!-- Date: Mon Mar 5 09:39:56.061 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235682>235682</A>
ACTIVATE: RunSet: BOTHPLUGS_NORMAL 
<!-- Author: Andy -->



<!-- Date: Mon Mar 5 09:46:25.458 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235682>235682</A>
TERMINATE: done
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 09:51:41.866 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235683>235683</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 09:52:45.158 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235683>235683</A>
TERMINATE: done
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 09:53:02.138 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235683>235683</A>
CALIBRATION: <U><B>pluglaser</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#pluglaser target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Mon Mar  5 10;12;17 comment by...diego -->
<pre>oops! the timing is still bad (did nit save the change in the config file). Try again</pre>
<!-- Date: Mon Mar 5 09:54:07.397 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235684>235684</A>
ACTIVATE: RunSet: BOTHPLUGS_MM 
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 09:55:53.828 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235684>235684</A>
TERMINATE: done
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 10:04:37.470 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235686>235686</A>
ACTIVATE: RunSet: BOTHPLUGS_MM_C 
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 10:06:42.749 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235686>235686</A>
TERMINATE: done
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 10:06:49.331 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235686>235686</A>
CALIBRATION: <U><B>pluglaser</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#pluglaser target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Mon Mar  5 10;11;33 comment by...diego -->
<pre>now the timing is fine</pre>
<!-- Date: Mon Mar 5 10:07:54.519 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235688>235688</A>
ACTIVATE: RunSet: BOTHPLUGS_MM 
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 10:09:36.115 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235688>235688</A>
TERMINATE: done
<!-- Author: Andy -->





<!-- Date: Mon Mar 5 10:15:31.928 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235690>235690</A>
ACTIVATE: RunSet: DAQTEST  Trigger: COSMICS [15,730,674] 
<!-- Author: badgett -->


<!-- Date: Mon Mar 5 10:16:41.506 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235691>235691</A>
ACTIVATE: RunSet: DAQTEST  Trigger: COSMICS [15,730,674] 
<!-- Author: badgett -->



<!-- Date: Mon Mar 5 10:19:00.181 2007 -->
<pre>Testing the PSGUI on the test bed. The test is isolated from the production environment.</pre>
<!-- Author: Renzo Borgatti -->



<!-- Date: Mon Mar 5 10:34:35.655 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235696>235696</A>
 Activated at 2007.03.05 10:34:21
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 10:34:36.466 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235696>235696</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,730,639] 
<!-- Author: Andy -->




<!-- Date: Mon Mar 5 10:44:43.307 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235696>235696</A>
 Terminated at 2007.03.05 10:43:27
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 10:44:44.754 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235696>235696</A>
TERMINATE: done
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 10:51:02.490 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235698>235698</A>
 Activated at 2007.03.05 10:50:44
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 10:51:03.310 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235698>235698</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,730,639] 
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 11:00:04.495 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235698>235698</A>
 Terminated at 2007.03.05 10:59:39
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 11:00:05.687 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235698>235698</A>
TERMINATE: tsi02 stuck
<!-- Author: Andy -->



<!-- Date: Mon Mar 5 11:01:37.144 2007 -->
I replaced the DCAS board in b0l2cl03 slot 10 (SNO 48) with one of the spares (SNO 25).  We are trying to test it with COSMICS but Greg has L1 in some funny state now so all we are seeing is empty events.
<!-- Author: twright -->

<!-- Date: Mon Mar 5 11:03:04.648 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235703>235703</A>
 Activated at 2007.03.05 11:02:20
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 11:03:05.858 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235703>235703</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,730,639] 
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 11:03:06.287 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235703>235703</A>
 Terminated at 2007.03.05 11:02:25
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 11:04:32.776 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235704>235704</A>
 Activated at 2007.03.05 11:04:21
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 11:04:33.307 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235704>235704</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,730,639] 
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 11:06:42.753 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235704>235704</A>
 Terminated at 2007.03.05 11:06:39
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 11:06:51.068 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235704>235704</A>
TERMINATE: tsi02 stuck again
<!-- Author: Andy -->



<!-- Date: Mon Mar 5 11:09:22.283 2007 -->
The change I made to the CMU HV on <a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2007&action=view&page=182&frame=2&anchor=12:05:46.470'> 
Friday morning</a> got wiped out when HVmon was <a href= 
'http://www-cdfonline.fnal.gov/cgi/elog/elog.pl?nb=2007&action=view&page=183&frame=2&anchor=20:43:03.373'> 
restarted on evening shift</a> so I got no data. I've restored the change.
<!-- Author: Phil -->

<!-- Date: Mon Mar 5 11:11:06.344 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235705>235705</A>
 Activated at 2007.03.05 11:10:49
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 11:11:07.777 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235705>235705</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,730,639] 
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 11:19:05.947 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235705>235705</A>
 Terminated at 2007.03.05 11:19:02
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 11:22:39.710 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235711>235711</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [18,791,633] 
<!-- Author: Andy -->
<!-- Comment: Mon Mar  5 11;23;06 comment by...Andy -->
COT taken out

<!-- Date: Mon Mar 5 11:36:26.421 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235715>235715</A>
 Activated at 2007.03.05 11:35:28
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 11:36:27.394 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235715>235715</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,730,674] RXPD test run
<!-- Author: Andy x2080 -->


<!-- Date: Mon Mar 5 11:36:28.487 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235715>235715</A>
 Terminated at 2007.03.05 11:36:12
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 11:40:54.607 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235717>235717</A>
 Activated at 2007.03.05 11:39:10
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 11:40:55.204 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235717>235717</A>
 Terminated at 2007.03.05 11:39:33
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 11:43:25.025 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235719>235719</A>
 Activated at 2007.03.05 11:43:15
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 11:43:26.236 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235719>235719</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,730,674] 
<!-- Author: Andy x2080 -->


<!-- Date: Mon Mar 5 11:44:55.529 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235719>235719</A>
 Terminated at 2007.03.05 11:44:41
<!-- Author: RunControl -->



<!-- Date: Mon Mar 5 11:45:33.669 2007 -->
<pre>PSGUI testing session ends.</pre>
<!-- Author: Renzo Borgatti -->

<!-- Date: Mon Mar 5 11:50:38.975 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235720>235720</A>
 Activated at 2007.03.05 11:50:21
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 11:50:39.674 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235720>235720</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,730,674] 
<!-- Author: Andy x2080 -->


<!-- Date: Mon Mar 5 11:51:22.192 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235720>235720</A>
 Terminated at 2007.03.05 11:51:05
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 11:54:03.339 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235721>235721</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Andy x2080 -->


<!-- Date: Mon Mar 5 11:54:42.772 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235721>235721</A>
TERMINATE: done
<!-- Author: Andy x2080 -->


<!-- Date: Mon Mar 5 11:56:14.653 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235721>235721</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->


<!-- Date: Mon Mar 5 11:59:01.804 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235722>235722</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->


<!-- Date: Mon Mar 5 12:00:13.548 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235723>235723</A>
 Activated at 2007.03.05 12:00:04
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 12:00:14.802 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235723>235723</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,730,674] 
<!-- Author: Andy x2080 -->


<!-- Date: Mon Mar 5 12:00:37.087 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235723>235723</A>
 Terminated at 2007.03.05 12:00:32
<!-- Author: RunControl -->



<!-- Date: Mon Mar 5 12:00:37.193 2007 -->
<pre>After Dervin fixed power supply for crate b0cot10-12, I ran an ASDQ  
debugging calibration and found TDCs in b0cot00 slot3, b0cot01 slot17 not responding. A reset of b0cot00 brought back slot3  
for 10 mins only. 
 
Dervin reseated these two TDCs. Now results of both ASDQ and ACE COT calibrations look fine.</pre>
<!-- Author: Shin-Shan Yu -->

<!-- Date: Mon Mar 5 12:02:34.180 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235664>235664</A>
trigger: <tt>L1: XFT Track Canvas shows large spikes in the left plots<br>
    Xtc Slots Efficiency has the bottom plot red</tt>
<!-- Author: Consumer Operator( diego cauz) -->


<!-- Date: Mon Mar 5 12:02:35.174 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235664>235664</A>
monitors: <tt>ObjectMon slide 93 has the "Track Phi" plot red</tt>
<!-- Author: Consumer Operator( diego cauz) -->



<!-- Date: Mon Mar 5 12:03:59.333 2007 -->
I am working on debugging RXPD readout in a multi- 
partition environment.
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235724' target=_top>235724</a>) -->

<!-- Date: Mon Mar 5 12:05:16.026 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235725>235725</A>
 Activated at 2007.03.05 12:05:01
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 12:05:17.271 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235725>235725</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,730,674] 
<!-- Author: Andy x2080 -->


<!-- Date: Mon Mar 5 12:07:57.236 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235725>235725</A>
 Terminated at 2007.03.05 12:07:16
<!-- Author: RunControl -->



<!-- Date: Mon Mar 5 12:14:46.381 2007 -->
We're bringing up muon HV, and going to take a cosmic run.
<!-- Author: Andy -->


<!-- Date: Mon Mar 5 12:15:14.801 2007 -->
<p>A long and frustrating time with iFIX and PSM.  The first 
problem was PSM went red on the global alarms.  Checking the 
"D" button revealed that all portions of iFIX were red.  I went 
upstairs to check if VOLT.EXE was running properly only to find 
that I could not even change focus between windows properly. 
So, I killed VOLT.EXE and iFIX, and rebooted VOLTMAN.</p> 
 
<p>When VOLTMAN and iFIX came back, I restarted VOLT.EXE which 
seemed to work.  But when I came downstairs, VNODE1 still showed 
PSM red in the global alarms.  Going back upstairs, I found that 
some east plug shower max crates were reading back the -7.5 and 
-15 as -25 and were red.  I pointed this out to Dervin, who 
went downstairs with Roberto and company, but could find  
nothing wrong with these crates.  At this point, I tried just 
restarting VOLT.EXE.  At first, all crates read out as good, 
but after a few minutes, crates on the first floor starting  
reading out in error or without communications; the original 
problem with the plug shower max crates did not reappear,  
however.</p> 
 
<p>At this point, after consulting with Mark, we tried killing 
VOLT.EXE and iFIX and rebooting a second time.  After this, all 
crates came back green in VOLT.EXE, and PSM is green in the 
global alarms on VNODE1.</p> 
 
<p>So, this is what Mark is hypothesizing happened: he had been 
updating iFIX on cdf-fs2 for the iFIX webserver.  Part of the 
update also updated Java to Java 10 SDK on cdf-fs2.  VOLTMAN is 
unique amongst the iFIX PCs in that it write info to an Access 
DB on cdf-fs2.  There is a Java Oracle interface on cdf-fs2 to 
read the info from the Access DB and write it into an Oracle 
DB.  The suspicion is that while the updates to cdf-fs2 were 
being done and until VOLTMAN was rebooted (or at least iFIX 
was restarted on VOLTMAN), there were problems with the  
Access DB file on cdf-fs2.  Whew!  My apologies to Dervin for 
making he and his crew wait while we straightened this out.</p>
<!-- Author: Steve Hahn :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235721' target=_top>235721</a>) -->
<!-- Comment: Mon Mar  5 13;04;36 comment by...jj -->
There had been some concern that iFix problems might have been
related to offsite network access problems reported this morning.  Steve says this was NOT the case...   (Good thing since
 slow controls should NOT be dependent on network access to
offsite.)
<!-- Date: Mon Mar 5 12:16:14.324 2007 -->
Everybody is out of the collision hall. 
COT calibration checked up ok, calorimeter calibrations were also ok. 
We'll check muons with HV up and cosmics until MCR calls to tell us they're injecting some beam.
<!-- Author: Lucio -->


<!-- Date: Mon Mar 5 12:19:05.690 2007 -->
Controlled access ended
<!-- Author: Lucio -->


<!-- Date: Mon Mar 5 12:19:54.502 2007 -->
From MCR: "We're starting a low beta checkout now. Shot setup will follow."
<!-- Author: Lucio -->


<!-- Date: Mon Mar 5 12:22:30.812 2007 -->
CMX TRIP Northeast Wedge 20
<!-- Author: Andy -->
<!-- Comment: Mon Mar  5 12;23;00 comment by...Andy -->
tripped again on recovery

<!-- Comment: Mon Mar  5 12;25;47 comment by...Andy -->
third times a charm, we recovered to standby and then to full voltage


<!-- Comment: Mon Mar  5 12;40;08 comment by...Andy -->
just tripped again, recovered to standby 
<!-- Date: Mon Mar 5 12:29:31.919 2007 -->
CMX TRIP, different wedge this time, SW Wedge 15
<!-- Author: Andy -->
<!-- Comment: Mon Mar  5 12;31;58 comment by...Andy -->
tripped on first and second recovery attempt

<!-- Comment: Mon Mar  5 12;37;45 comment by...Lucio -->
paged CMX



<!-- Comment: Mon Mar  5 12;53;11 comment by...Lucio -->
Dale is checking it
<!-- Comment: Mon Mar  5 13;34;18 comment by...Dale and Andy -->
Andy manged to get the both SW 15 and NE 20 up and holding at stand-by.  I promply turn the wedges back up to full voltage - everything held fine without additional trips.  <br><br>
There was a lot of activity in the HVAC via the control access.  It appears the rapid change set them off.  We've seen this before but the wedges seems particularly upset until things settled down.<br><br>
Wedges are at stand-by now and should be good to go for the next shot.  No special instructions but be mindful of these two suspect wedges if you start getting CMX trips.
<!-- Date: Mon Mar 5 12:41:36.658 2007 -->
<table border=0><td><tr><td bgcolor=#e0e030>The updated ScalerMon that we used over the weekend ("test" version)  has been incorporated into the current <b><i>fer </i></b>package. So there is  NO longer  need to start a special version. Thanks.</td></tr></td></table>
<!-- Author: Donatella T. -->

<!-- Date: Mon Mar 5 12:42:16.029 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235736>235736</A>
ACTIVATE: RunSet: DAQTEST  Trigger: COSMICS [15,730,674] 
<!-- Author: BB x2080 -->




<!-- Date: Mon Mar 5 12:57:28.089 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235737>235737</A>
 Activated at 2007.03.05 12:57:15
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 12:57:29.152 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235737>235737</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,730,674] 
<!-- Author: Andy x2080 -->




<!-- Date: Mon Mar 5 13:10:23.756 2007 -->
I believe I have fixed the various problems related to RXPD  
bank readout in crate RETURN_XPT_00 b0tsi02 − the  
last problems were parallel partitions clobbering the  
setting of the readout partition.  Several global variables  
in the <I>fer</I> code are now arrays by partition, and the  
other ones, especially SHMEM_partition, are set properly  
now for the various front-end cases.  The  <I>fer</I>  
version is <i>v5_37m</I> and is current.  
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235737' target=_top>235737</a>) -->
<!-- Comment: Mon Mar  5 14;06;16 comment by...greg -->
<pre>If there are problems with tsi02 (the RXPT crate)
please page Bill/DAQ -- not Greg!</pre>
<!-- Date: Mon Mar 5 13:25:32.610 2007 -->
CMX TRIP NW Wedge 21
<!-- Author: Andy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235737' target=_top>235737</a>) -->
<!-- Comment: Mon Mar  5 13;36;40 comment by...Dale and Andy -->
This might have been related to the <a href=
'/cgi/elog/elog.pl?nb=2007&action=view&page=191&frame=2&anchor=12:29:31.919'>
above</a> but we are incline to think the problem was of a slight differnt nature.  The anode on this went (this was a very rare problem we had at the end of the last long shutdown).  It recovered fine so we hope that this was truely a rare event.
<!-- Date: Mon Mar 5 13:27:07.512 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235737>235737</A>
 Terminated at 2007.03.05 13:26:52
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 13:27:11.285 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235737>235737</A>
TERMINATE: beam soon
<!-- Author: Andy x2080 -->



<!-- Date: Mon Mar 5 13:28:52.958 2007 -->
Muons also checked ok. MCR is about to inject beam
<!-- Author: Lucio -->

<!-- Date: Mon Mar 5 13:31:00.905 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235739>235739</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: Andy x2080 -->


<!-- Date: Mon Mar 5 13:32:38.343 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235739>235739</A>
CALIBRATION: <U><B>calxef</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calxef target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Mon Mar  5 13;46;47 comment by...diego -->
<pre>calib's are fine</pre>
<!-- Date: Mon Mar 5 13:32:40.718 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235739>235739</A>
TERMINATE: done
<!-- Author: Andy x2080 -->



<!-- Date: Mon Mar 5 13:33:55.948 2007 -->
Shot setup begins
<!-- Author: Lucio -->



<!-- Date: Mon Mar 5 13:40:50.346 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235740>235740</A>
CALIBRATION: <U><B>toftac</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#toftac target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Mon Mar  5 13;49;24 comment by...diego -->
<pre>calib's are fine</pre>
<!-- Date: Mon Mar 5 13:40:51.796 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235740>235740</A>
TERMINATE: done
<!-- Author: Andy x2080 -->


<!-- Date: Mon Mar 5 13:40:52.374 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235740>235740</A>
ACTIVATE: RunSet: TOF_TAC_CALIB 
<!-- Author: Andy x2080 -->


<!-- Date: Mon Mar 5 13:46:29.243 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235741>235741</A>
ACTIVATE: RunSet: L2TORTURE  Trigger: L2_TORTURE [18,791,633] 
<!-- Author: Andy x2080 -->






<!-- Date: Mon Mar 5 14:10:08.686 2007 -->
<pre>New online monitor release cdfsoftb0 6.1.4ar is ready to be 
installed in the control room (not yet installed). 
This release has updates in: 
-  TrigMon (XFT related) requested by Alison 
-  TrigMon (XTRP related) requested by Kevin Pitts 
-  ConsumerFramework ErrorReceiver (related to SVXMon EndRun)  
   by Donatella 
 
Plan (unless there are problems) is to start the next  
store as is, and after checking the performances, 
install the new release 6.1.4ar. The current default is 6.1.4aq 
and all the monitors are running with this version. 
</pre>
<!-- Author: kaori -->


<!-- Date: Mon Mar 5 14:11:29.394 2007 -->
<P> 
Have fixed the problem with the SVX good luminosity  
accounting for store  
<A HREF=/daq/stores/store5259.html>5259</A> − the  
luminosity from run 235593 had not been included in the  
total good SVX lumi.   There was a small problem with the  
RSF_STATUS bit − status changes were not triggering  
a recompute of the store numbers;  now they are. 
<P>
<!-- Author: badgett :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=234593' target=_top>234593</a>) -->


<!-- Date: Mon Mar 5 14:18:59.315 2007 -->
CO noticed an Error message in TrigMon 
<pre> 
Error found by Pulsar Monitor  Page L2, Issue HRR 
</pre> 
We're going to page L2 and see what's up.
<!-- Author: Andy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235741' target=_top>235741</a>) -->
<!-- Comment: Mon Mar  5 14;23;05 comment by...Lucio -->
L2 pulsar expert responded. He's taking a look into it


<!-- Comment: Mon Mar  5 14;33;51 comment by...Shawn -->
I looked in the L2 log files and I see bit level errors in the XTRP track list buffer.  This looks like the XTRP bit-level problem that we have seen before on the level 2 system.  I have told the shift crew that this should not be a problem with the upcoming data taking run.  In the past, it usually goes away with HRR (or two).
<p>If during the next store the Muon path turns red (meaning the problem has reoccured), then page L2 and issue and an HRR.  If it doens't go away then we need to contact the XTRP experts.
<!-- Date: Mon Mar 5 14:23:09.952 2007 -->
Done TO for cot17, accompanied by RXPT error for the same crate.
<!-- Author: Andy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235741' target=_top>235741</a>) -->
<!-- Comment: Mon Mar  5 14;23;51 comment by...Andy -->
another done TO, and another, shepherding

<!-- Comment: Mon Mar  5 14;25;41 comment by...Andy -->
shepherd successful, then done TO immediately after

<!-- Date: Mon Mar 5 14:26:44.884 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235741>235741</A>
TERMINATE: loading protons
<!-- Author: Andy x2080 -->


<!-- Date: Mon Mar 5 14:30:40.220 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235744>235744</A>
ACTIVATE: RunSet: AAA_SHOTSETUP  Trigger: PHYSICS_4_02 [2,865,677] 
<!-- Author: Andy x2080 -->




<!-- Date: Mon Mar 5 14:37:27.364 2007 -->
CCU trips. The ACE is downstairs trying to reset them
<!-- Author: Lucio -->
<!-- Comment: Mon Mar  5 14;55;42 comment by...Lucio -->
We've paged ISU... he's coming to check
<!-- Comment: Mon Mar  5 15;12;42 comment by...Andy -->
BSU/CSP LV bars have returned
<!-- Comment: Mon Mar  5 15;14;04 comment by...Andy -->
and there they go again
<!-- Comment: Mon Mar  5 15;55;48 comment by...Akram -->
During shot setup some communication problem happened with CCU 13 and CCU 25. These SSU's used for CSP. Min. HV is at 0 for these CCU's , but real HV is OK. I checked pulses (looks fine) from WEST TOP and Bottom counters . Occupancy plots should be OK. Ignore ALARM, please. We still afraid  to use power cycling because of -5 V problem.
<!-- Date: Mon Mar 5 14:54:23.057 2007 -->
Final protons loaded: 11040E9
<!-- Author: Lucio -->



<!-- Date: Mon Mar 5 15:01:08.647 2007 -->
Loading pbars; Silicon and opsman are here
<!-- Author: Lucio -->




<!-- Date: Mon Mar 5 15:19:18.831 2007 -->
Pbar loaded: 2556E09
<!-- Author: Lucio -->


<!-- Date: Mon Mar 5 15:20:43.850 2007 -->
<h3>Offsite Network Update</h3><br><br> 
<pre> 
From: JJ Schmidt 
To: Fabrizio SCURI 
Subject: Internet Connectivity Issues 
 
Offsite access to some (not all) sites has been a problem 
today.  I consulted with DataComm and we believe that if 
remote CO was able to connect last night, then remote CO 
should be able to connect tonight even if problems are 
not solved. 
 
However - DataComm says some progress is being made by 
ESNET on understanding the problems and perhaps solutions 
will be finished by this eveing. 
 
In any case, the earlier we test the link tonight the 
more time we will have to deal with problems. 
 
   JJ 
 
---------- Forwarded message ---------- 
Date: Mon, 05 Mar 2007 14:55:23 -0600 
From: bsshelpdesk 
To: EN-ALL_USERS 
Subject: Internet Connectivity Issues 
 
Please be advised that we have received the following communication from  
the CD Help Desk: 
 
There is a problem involving Fermilab's ISP EsNET that actually developed  
over the weekend. A fix was put into place and then earlier today, the CD  
Helpdesk started fielding calls reporting various off-site web-sites  
were'nt' accessible, namely Federal sites although this may have been  
attributed to where the reporter was trying to access. 
 
Unfortunately, we are unable to provide an estimated time when  
connectivity will be restored. You may contact the CD-HelpDesk seeking a  
status when you like. 
 
Hopefully, this matter will be resolved soon. 
</pre>
<!-- Author: jj -->


<!-- Date: Mon Mar 5 15:21:48.573 2007 -->
cot17 giving issues.  Done TO immediately on ACTIVE, shepherded once but didn't help, trying again
<!-- Author: Andy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235746' target=_top>235746</a>) -->
<!-- Comment: Mon Mar  5 15;28;11 comment by...Andy -->
second didn't help either.  Expert says problem happened last week and was not understood, but went away on its own.  We hope it goes away again.
<!-- Comment: Mon Mar  5 15;56;21 comment by...Andy -->
more... expert is going to power cycle
<!-- Date: Mon Mar 5 15:23:24.672 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235746>235746</A>
TERMINATE: getting ready for data
<!-- Author: Andy x2080 -->



<!-- Date: Mon Mar 5 15:24:48.261 2007 -->
Having problems with COT17 DTO
<!-- Author: Lucio -->



<!-- Date: Mon Mar 5 15:29:24.298 2007 -->
CPR alarm, Mary went to investigate
<!-- Author: Andy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235752' target=_top>235752</a>) -->
<!-- Comment: Mon Mar  5 15;31;11 comment by...Andy -->
seems to be fine, we'll be watching


<!-- Date: Mon Mar 5 15:35:18.496 2007 -->
Scraping complete at 15:35. Store # 52640
<!-- Author: Lucio -->


<!-- Date: Mon Mar 5 15:36:12.686 2007 -->
Date	Time	BLM	Dose 
2007.03.05	15:31:23	W Inner BLM	29.404	RADS 
2007.03.05	15:31:23	W Outer BLM	21.347	RADS 
2007.03.05	15:31:23	E Inner BLM	33.054	RADS 
2007.03.05	15:31:23	E Outer BLM	24.703	RADS 
Integrated dosage
<!-- Author: Lucio -->


<!-- Date: Mon Mar 5 15:38:04.310 2007 -->
<pre>Date	Time	BLM	Dose 
2007.03.05	15:31:23	W Inner BLM	29.404	RADS 
2007.03.05	15:31:23	W Outer BLM	21.347	RADS 
2007.03.05	15:31:23	E Inner BLM	33.054	RADS 
2007.03.05	15:31:23	E Outer BLM	24.703	RADS</pre>
<!-- Author: Lucio -->


<!-- Date: Mon Mar 5 15:40:06.608 2007 -->
Luminosity: 234.6E30
<!-- Author: Lucio -->

<!-- Date: Mon Mar 5 15:41:18.940 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235752>235752</A>
 Activated at 2007.03.05 15:41:11
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 15:41:19.319 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235752>235752</A>
ACTIVATE: RunSet: AAA_CURRENT  Trigger: PHYSICS_4_02 [2,865,677] 
<!-- Author: Andy x2080 -->


<!-- Date: Mon Mar 5 15:44:03.163 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235752>235752</A>
 Terminated at 2007.03.05 15:43:03
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 15:44:04.379 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235752>235752</A>
TERMINATE: trying again
<!-- Author: Andy x2080 -->



<!-- Date: Mon Mar 5 15:46:51.015 2007 -->
errors in cmp00, cmu00, cmu01, cmx00, cmx01, etdc00, and imu00.  We're paging DAQ and TDC pagers
<!-- Author: Andy :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235755' target=_top>235755</a>) -->
<!-- Comment: Mon Mar  5 15;52;02 comment by...Lucio -->
TDC says it's a DAQ issue. DAQ is here now

<!-- Comment: Mon Mar  5 15;54;51 comment by...Andy -->
Shepherded all crates successfully (eventually).  Taking data.
<!-- Date: Mon Mar 5 15:54:13.480 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235755>235755</A>
 Activated at 2007.03.05 15:54:06
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 15:54:14.833 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235755>235755</A>
ACTIVATE: RunSet: AAA_CURRENT  Trigger: PHYSICS_4_02 [2,865,677] 
<!-- Author: Andy x2080 -->



<!-- Date: Mon Mar 5 15:55:11.436 2007 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>PhysicsTable</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L3Accepts</TH>
<TH>DelivLumi nb<SUP>-1</SUP></TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>15:55:00</TD>
<TD ALIGN=CENTER>::</TD>
<TD ALIGN=RIGHT></TD>               <TD ALIGN=RIGHT></TD>
<TD ALIGN=RIGHT></TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->




<!-- Date: Mon Mar 5 16:01:36.803 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235755>235755</A>
 Terminated at 2007.03.05 16:01:14
<!-- Author: RunControl -->


<!-- Date: Mon Mar 5 16:01:37.729 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235755>235755</A>
TERMINATE: Having troubles with cot17
<!-- Author: Michal x2080 -->
<!-- Comment: Mon Mar  5 16;04;36 comment by...pjw, convery -->
Peter says Mousumi was involved last time and may have suggestions.  He also says that they put cot17 in a separate partition and then put it back in the main partition and it worked.
<!-- Date: Mon Mar 5 16:02:51.953 2007 -->
<font color=red>Shift Summary: </font><pre>8.00  store ending and controlled access to replace a COT PS 
12:19 access ends, taking calibration and cosmics, going into shot setup 
15:00 bumpy start for us, store 52640 starts with lumi 235E30, but we're down with a DAQ problem and power cycling COT17.</pre> 
<br><b>End of Shift Numbers</b>
<table><tr><td width=60% align=left>
<b>CDF Run II <p></b>                       
<pre>
Runs                   235752 235755
Delivered Luminosity   302.6 nb-1  
Acquired Luminosity    13.8 nb-1  
Efficiency             4.2%
   
</td></tr></table></pre>
<!-- Author: Lucio -->

<!-- Date: Mon Mar 5 16:04:24.375 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235757>235757</A>
ACTIVATE: RunSet: AAA_CURRENT  Trigger: PHYSICS_4_02 [2,865,677] 
<!-- Author: Michal x2080 -->


<!-- Date: Mon Mar 5 16:04:25.741 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=235757>235757</A>
 Activated at 2007.03.05 16:04:08
<!-- Author: RunControl -->