<!-- Title: Owl shift. Wed Mar 20, 2002 -->
<!-- SciCo: Giorgio Chiarelli -->
<!-- DAQAce: Diego Tonelli -->
<!-- MonAce: Muge Unel -->
<!-- CO: &nbsp; -->
<!-- OpManager: Arnd Mayer -->
<!-- Notes: -->
<pre>Shift starts. Long night to go</pre>


<!-- Date: Wed Mar 20 00:28:06 2002 -->
the racks alarms is silenced for the 2nd floor. fwd detector hv pc is offline (pink) but i am accepting it as is, like the prev. shift did. there is a firus warning which Giorgio is checking on.
<!-- Author: muge -->


<!-- Date: Wed Mar 20 00:39:16 2002 -->
<pre> Trying to run Cosmics_all[2,233]. L3 failed coldstart, 
 cleaned up..then I got Done Time Out from all WCAL and PCAL 
 crates,and in the same bunch of errors, Busy TO from XFT  
crates, SCPU-error and RXPT error loop detected in b0xft00,02, 
04.Cleaning up of EVB solved all, now we are running cosmics 
 smoothly. For back tracking of the errors  some samples from 
 run 141280 errorfile: 
 
----------  WCAL/PCAL DTO ------------------------------ 
(MLE) b0pcal08:Messenger:12:07:37 AM->Event 1: Bunchcounters in slot 15 and slot 
 19 disagree (at least) 
(MLE) b0pcal09:Messenger:12:07:36 AM->Event 1: Bunchcounters in slot 14 and slot 
 18 disagree (at least) 
(MLE) b0pcal10:Messenger:12:07:38 AM->Event 1: Bunchcounters in slot 15 and slot 
 19 disagree (at least) 
(MLE) b0pcal08:Messenger:12:07:37 AM->Runtime Error 1, Event 1: Bunch counter mi 
smatch, mismatch count = 4 
(MLE) b0pcal09:Messenger:12:07:36 AM->Runtime Error 1, Event 1: Bunch counter mi 
smatch, mismatch count = 4 
(MLE) b0pcal10:Messenger:12:07:38 AM->Runtime Error 1, Event 1: Bunch counter mi 
smatch, mismatch count = 8 
 
------------------ XFT BUSY TIME OUT ---------------- 
8'33" 3 crate/s: b0xft00(97), b0xft02(99), b0xft04(101),  busy.[RXPT] 
(MLE) b0tsi02:Messenger:12:08:04 AM->BUSY timeout 
(MLE) b0tsi02:Messenger:12:08:08 AM->BUSY timeout 
</pre>
<!-- Author: Diego :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141281' target=_top>141281</a>) -->


<!-- Date: Wed Mar 20 00:44:04 2002 -->
We have a message on the MAC running Firus. It says that "lost control, call software support personal". Message poped up at 9.26 PM. Talked to previous shift, haven't noticed. Talked to cryo guys, no idea. No name anywhere... started paging people. 
Fire tech (only number by the mac): no answer. talked to various people: J.C.Yun [General Ifix, slow controls etc] no idea, paged Rich Schmidt, Bill Noe, Mark Knapp (as for JC Yun)
<!-- Author: GiorgioC -->


<!-- Date: Wed Mar 20 00:55:16 2002 -->
Starting a new cosmic run with silicon inside as requested by Gino.
<!-- Author: Diego :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141282' target=_top>141282</a>) -->


<!-- Date: Wed Mar 20 02:25:44 2002 -->
Running on cosmics with Gino checking the effects of recent silicon fixes to Reformatter rate. 
L3 subfarm c10 stuck in ended (pink) state. 
<!-- Author: Diego :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141283' target=_top>141283</a>) -->


<!-- Date: Wed Mar 20 03:19:28 2002 -->
a few warnings (color code yellow) on cmp hv. i could not see nothing particularly drastic changes in rates and currents.. 
<!-- Author: muge -->


<!-- Date: Wed Mar 20 03:47:24 2002 -->
L3 Subfarm c15 stuck pink. The node 228 is dead.
<!-- Author: Diego :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141283' target=_top>141283</a>) -->


<!-- Date: Wed Mar 20 04:35:34 2002 -->
COT calibration done. No errors.
<!-- Author: Diego :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141284' target=_top>141284</a>) -->
<!-- Comment: Wed Mar 20 04;54;05 comment by...Diego -->
DBANA comparison with last calibration looks ok.
<!-- Date: Wed Mar 20 04:36:02 2002 -->
Muon calibration done. No errors.
<!-- Author: Diego :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141285' target=_top>141285</a>) -->


<!-- Date: Wed Mar 20 04:36:37 2002 -->
TOF TAC calibration done. No errors.
<!-- Author: Diego :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141286' target=_top>141286</a>) -->


<!-- Date: Wed Mar 20 04:40:07 2002 -->
TOF-QIE calibraton done. No errors
<!-- Author: Diego :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141287' target=_top>141287</a>) -->
<!-- Comment: Wed Mar 20 04;57;15 comment by...Diego -->
DBANA check  of TOFQIEPED looks OK.
<!-- Date: Wed Mar 20 04:46:03 2002 -->
Switched back to cosmics.Running.
<!-- Author: Diego :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141288' target=_top>141288</a>) -->
<!-- Comment: Wed Mar 20 05;22;11 comment by...Diego -->
L3 farms resumed.


<!-- Date: Wed Mar 20 05:30:07 2002 -->
Had a couple of orange pop-up about reformatter errors. 
The first HRR does'nt seem to helped, the second yes..no more orange pop-up now.
<!-- Author: Diego :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141288' target=_top>141288</a>) -->


<!-- Date: Wed Mar 20 06:01:15 2002 -->
B0dap30 and 31 brought down for scheduled work.
<!-- Author: GiorgioC -->


<!-- Date: Wed Mar 20 06:04:15 2002 -->
Run ended for work on b0dap30, I suppose I 'll loose RC communication
<!-- Author: Diego :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=141288' target=_top>141288</a>) -->
<!-- Comment: Wed Mar 20 06;37;54 comment by...Diego -->
pingmon and vxmon all red


<!-- Date: Wed Mar 20 06:49:49 2002 -->
Cryo have FIRUS restarted. There was some connectivity problem (even Dispatcher lost its own FIRUS...) and MCR now has fixed it.
<!-- Author: GiorgioC -->


<!-- Date: Wed Mar 20 07:50:45 2002 -->
<center><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-1472&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-1472' width='300'></a><a href='/cgi/elog/elog.pl?nb=2002&action=view&page=-1473&button=yes' target='xxx' onclick='jswindow()'><img src='/cgi/elog/elog.pl?nb=2002&action=view&page=-1473' width='300'></a></center>
<!-- Author: muge -->
<!-- Comment: Wed Mar 20 07;51;13 comment by...muge -->
losses during owl.

<!-- Date: Wed Mar 20 07:56:04 2002 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>Run Number</TH>
<TH>Data Type</TH>
<TH>Physics Table</TH>
<TH>Begin Time</TH>
<TH>End Time</TH>
<TH>Live Time</TH>
<TH>L1 Accepts</TH>
<TH>L2 Accepts</TH>
<TH>L3 Accepts</TH>
<TH>Live Lumi, nb-1</TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>

<TR><TD>Totals</TD>                                                     
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD>07:55:02</TD>                                                       
<TD ALIGN=CENTER>::</TD>                                                
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD ALIGN=RIGHT></TD>                                                   
<TD> </TD>                                                              
<TD> </TD>                                                              
<TD> </TD></TR>                                                         
                                                                        
</TABLE>
<!-- Author: End of Shift Report -->