<!-- Title: Eve shift. Sun Sep  9, 2007 -->
<!-- SciCo: Wei-Ming Yao -->
<!-- DAQAce: Miguel Mondragon -->
<!-- MonAce: Glenn Strycker -->
<!-- CO: &nbsp; -->
<!-- OpManager: Steve Hahn -->
<!-- Notes: -->
<pre>Eve Shift Plan:   
   In Collison Hall:   
        WHA PMT repair finished for the day at 4:00 pm   
   In Control Room:   
        Do ACE training, calibration, L2 torture and taking  
        cosmics </pre>


<!-- Date: Sun Sep 9 16:13:42.690 2007 -->
<pre>Ignacio called back in answer to our silicon page -- he told us how to get the dewpoint monitor back up. 
 
1) Go to File/Open in the IFix menu 
2) Open the S: drive (C: is default, and we didn't find the file there) 
3) Go to the S:\dynamics\PIC\CDF\SVX directory 
4) Open realtimeplotter_dewpoint 
 
There is no need to click anything or set anything up, the program begins automatically. 
 
We followed these instructions and everything worked correctly.  Thanks, Ignacio!</pre>
<!-- Author: Glenn Strycker / Miguel M. -->
<!-- Comment: Sun Sep  9 16;27;11 comment by...Miguel M. -->
The temperatures are flat and stable
<!-- Date: Sun Sep 9 16:14:12.742 2007 -->
WHA work for today:<br> 
We replaced: <br> 
11W - 10L PM (it was drawing a lot of current) <br> 
20E - 6R  PM (it showed discharges during the last source run and <br> was drawing a lot of current). On inspection we did not see <br> the discharges but it was drawing more current than other pms. <br> To be on the safe side we decided to replace it. We also <br> inspect channel 7L on the same wedge (20E).  <br> 
<br> 
The work on the West WHA is finished for now. When the plug is <br> closed we need to do some work on the top wedges (3-8).  <br>
<!-- Author: Paolo and Fotis -->


<!-- Date: Sun Sep 9 16:20:24.442 2007 -->
We paged the TDC experts regarding the post earlier today during the morning shift (<a href= 
'/cgi/elog/elog.pl?nb=2007&action=view&page=726&frame=2&anchor=10:42:23.050'>Sun Sep 9 10:42:23</a>):<br> 
 
<pre> 
Something is strange for this board (b0etdc00): 
TDC moduleId = 1  
*** Wrong TDC boardType: found 8, expected 9 in slot 17 *** 
TDC moduleId = 0 6 12 18 32 38 44 50 71 83 103 115  
*** Wrong TDC boardType: found 8, expected 9 in slot 17 *** 
*** Wrong TDC boardType: found 9, expected 8 in slot 15 *** 
Cannot configure readout - check readout list. 
Some slot are swapped? 
</pre> 
 
We've been running without the EM_TIMING_00/etdc00 crate so far without any more problems, so the problem can wait until Monday if necessary.
<!-- Author: Glenn Strycker / Miguel M. -->
<!-- Comment: Sun Sep  9 16;28;01 comment by...Glenn Strycker / Miguel M. -->
It's Sunday -- we really shouldn't bother anyone about this today.  We decided to email Eric James (the TDC expert) and wait for his response rather than page again.


<!-- Date: Sun Sep 9 16:37:28.646 2007 -->
<pre>Short status summary  
-------------------------  
 
HV Status:      
 
 HV ON (green+bars): CEM, CHA (with warning), GAM (partially -- WHAs are all turned off, I believe because Fotis and Paolo were working on it) 
 Disabled/Alarm (grey  or red): PSM is red  
 
 Out-of-range (cyan): PEM, PHA, PSH (expected, since plug maintence is occurring) -- heartbeat status for all of these is pink, meaning "local PC unaccessable" 
 OFF (green, no bars): L00, SVX, ISL, COT, TOF, CLC, CPR, BSC, MNP, CEM, PTM, CMU, CMP, CMX, BMU, CSP, DSW, CSX, MSK, BSU, TSU 
 Monitors (green=ok): ICI, HBM, INH, VOL 
 
 Warning (yellow, no bars):  
          CHA: (-7|22L), (-2|22L), (+0|2R)      
          WHA: (channnels are blinking, HV bars are missing)  
          CES: (off, but yellow, max voltage 0.2%)   
 
 Alarm (red): PSM (and PEM, PHA, PSH are cyan right now)  
 
 
The heart beat is working (green color) for all systems, save PEM, PHA, PSH where it is pink 
 
 
Deadtime:  4.46% (good, below 6%)   
L1 Accept: 39438 Hz (normal for L2 torture mode)   
L2 Accept: 245 Hz   (nomral)</pre>
<!-- Author: Glenn Strycker / Miguel M. -->

<!-- Date: Sun Sep 9 16:49:31.670 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249160>249160</A>
TERMINATE: We tried to manually remove etdc slots 8-9 instead of the entire etdc00 create -- however we noticed we should have removed slots 15 and 17.  We'll try again.
<!-- Author: Glenn Strycker and Miguel Mondragon -->


<!-- Date: Sun Sep 9 16:55:31.950 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249161>249161</A>
TERMINATE: We still can't get past the "config" step in the RunControl program.  We get this error message:

Strange (but not necessarily fatal) Run Configuration selections:
Crate EM_TIMING_00 is missing card TDC_EM_12 in slot 15
Crate EM_TIMING_00 is missing card TDC_TMS_00 in slot 17
Crate PCAL_07 is missing card SMXREADOUT_60 in slot 4

It looks like we cannot manually remove slots 15+17 (nor 8+9), so we're going to remove the entire crate EM_TIMING_00/b0etdc00 from the next run.
<!-- Author: Glenn Strycker and Miguel Mondragon -->



<!-- Date: Sun Sep 9 16:55:51.083 2007 -->
<pre>XTRP expert Kevin Pitts have sent me a e-mail about the XTRD errors that we observe in TrigMon since yseterday. The error is not a problem. The mail from Kevin is as follows:  
 
Hi Takuo, 
 
These trigmon errors are a feature of L2 torture.  We load up 
fake tracks in the XTRP during L2 torture runs, so the monitoring 
gets confused because the tracks originating in the XTRP don't 
match the XFT.   This has always been a feature of L2 torture 
runs, so there is no problem. 
 
cheers, 
 Kevin 
</pre>
<!-- Author: Takuo Yoshida :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249160' target=_top>249160</a>) -->

<!-- Date: Sun Sep 9 16:58:58.142 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249162>249162</A>
 Activated at 2007.09.09 16:58:49
<!-- Author: RunControl -->


<!-- Date: Sun Sep 9 17:01:04.437 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249162>249162</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,954,639]

We got this error message when we partitioned and configured:
"Strange (but not necessarily fatal) Run Configuration selections:
Crate PCAL_07 is missing card SMXREADOUT_60 in slot 4"

and then we got a SVT_PULSAR_09 timeout error, but that went away when we HRR'd.  I think the timeout error may have been random and unrelated.

COSMICS run seems to be going ok now.
<!-- Author: Glenn Strycker and Miguel Mondragon -->



<!-- Date: Sun Sep 9 17:26:08.448 2007 -->
We heard a Cryo alarm, but we didn't see anything wrong on the cryo monitor.  Everything looked in bounds -- the only things that looked a little off were already set to "alarm silenced", so these must be known problems (or shutdown-related standby status) and the alarm couldn't have been caused by these. 
<br><br> 
Flow Rate Leads A and B seem low (0.15 g/s -- 40 is the lower limit), but the text color is green and the value is remaining stable, so we're not going to call anyone. 
<br><br> 
System appears fine and stable.
<!-- Author: Glenn Strycker / Miguel M. -->

<!-- Date: Sun Sep 9 19:10:56.719 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249162>249162</A>
 Terminated at 2007.09.09 19:10:33
<!-- Author: RunControl -->


<!-- Date: Sun Sep 9 19:10:57.929 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249162>249162</A>
TERMINATE: COSMICS went fine... we'll stop and do some calibrations now.
<!-- Author: Glenn Strycker and Miguel Mondragon -->


<!-- Date: Sun Sep 9 19:14:30.275 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249163>249163</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Glenn Strycker and Miguel Mondragon -->


<!-- Date: Sun Sep 9 19:14:46.224 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249163>249163</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep  9 19;31;07 comment by...Glenn Strycker -->
status "incomplete" -- see elog entries below.
<!-- Date: Sun Sep 9 19:18:35.078 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249163>249163</A>
TERMINATE: Something hung and we were stuck in the ready state with a yellow icon.  I'll re-run it to make sure the calibration was ok.
<!-- Author: Glenn Strycker and Miguel Mondragon -->


<!-- Date: Sun Sep 9 19:20:36.680 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249164>249164</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Glenn Strycker and Miguel Mondragon -->


<!-- Date: Sun Sep 9 19:21:44.032 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249164>249164</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep  9 19;31;29 comment by...Glenn Strycker -->
Statuts "incomplete" -- see elog entries below
<!-- Date: Sun Sep 9 19:25:19.683 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249164>249164</A>
TERMINATE: "activate" transition stuck
<!-- Author: Glenn Strycker and Miguel Mondragon -->



<!-- Date: Sun Sep 9 19:28:03.802 2007 -->
The sevb crate is giving us problems (as noted in the previous shift for the TOF_TAC calibration).  We restarted the software event builder through the SoftEvb Proxy Viewer, reset RunControl, re-partitioned, ran config, and now will retry calibration again.
<!-- Author: Glenn Strycker / Miguel M. -->

<!-- Date: Sun Sep 9 19:28:43.713 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249165>249165</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Glenn Strycker and Miguel Mondragon -->


<!-- Date: Sun Sep 9 19:29:52.740 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249165>249165</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep  9 19;32;09 comment by...Glenn Strycker -->
Status "incomplete"

<!-- Date: Sun Sep 9 19:33:40.445 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249165>249165</A>
TERMINATE: Falied "activate"
<!-- Author: Glenn Strycker and Miguel Mondragon -->



<!-- Date: Sun Sep 9 19:34:12.199 2007 -->
Miguel tried selecting a b0ccal?? crate during the calibration and manually hit "activate" and got an error.  It looks like the calibration is really hanging and sevb is broken.  We're trying another calibration to see if it is just a problem with CALQIE, or a crate, or something else.
<!-- Author: Glenn Strycker / Miguel M. -->

<!-- Date: Sun Sep 9 19:35:24.617 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249166>249166</A>
TERMINATE: forgot to remove bad crates
<!-- Author: Glenn Strycker and Miguel Mondragon -->


<!-- Date: Sun Sep 9 19:37:01.429 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249167>249167</A>
 Activated at 2007.09.09 19:36:45
<!-- Author: RunControl -->


<!-- Date: Sun Sep 9 19:37:32.808 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249167>249167</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,954,639] pcal, etdc00 out of partition
<!-- Author: Glenn Strycker and Miguel Mondragon -->


<!-- Date: Sun Sep 9 19:37:33.347 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249167>249167</A>
 Terminated at 2007.09.09 19:37:05
<!-- Author: RunControl -->


<!-- Date: Sun Sep 9 19:37:34.604 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249167>249167</A>
TERMINATE: activate works
<!-- Author: Glenn Strycker and Miguel Mondragon -->


<!-- Date: Sun Sep 9 19:40:36.211 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249168>249168</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Glenn Strycker and Miguel Mondragon -->


<!-- Date: Sun Sep 9 19:41:26.858 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249168>249168</A>
TERMINATE: smx qie done. 
<!-- Author: Glenn Strycker and Miguel Mondragon -->


<!-- Date: Sun Sep 9 19:43:05.363 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249169>249169</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Glenn Strycker and Miguel Mondragon -->


<!-- Date: Sun Sep 9 19:46:24.252 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249168>249168</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep  9 19;47;46 comment by...Glenn Strycker -->
TEST -- do NOT use
<!-- Date: Sun Sep 9 19:46:25.215 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249169>249169</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep  9 19;47;58 comment by...Glenn Strycker -->
TEST -- do NOT use
<!-- Date: Sun Sep 9 19:46:26.067 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249168>249168</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep  9 19;48;05 comment by...Glenn Strycker -->
TEST -- do NOT use
<!-- Date: Sun Sep 9 19:46:27.027 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249169>249169</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep  9 19;48;14 comment by...Glenn Strycker -->
TEST -- do NOT use
<!-- Date: Sun Sep 9 19:46:28.929 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249169>249169</A>
TERMINATE: cal qie done; sucessful run.
<!-- Author: Glenn Strycker and Miguel Mondragon -->



<!-- Date: Sun Sep 9 19:47:20.848 2007 -->
Miguel tried several things, and evenutally he re-loaded the run configuration.  We noticed that the flow chart displayed on the RunControl was different than the aborted CALQIE runs logged above.  It seems that the CALQIE calibration configuration wasn't loading correctly before -- maybe an error with RunControl?  I'm going to restart RunControl and see if that makes the rest of the calibrations go smoothly.
<!-- Author: Glenn Strycker -->





<!-- Date: Sun Sep 9 19:51:28.945 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249170>249170</A>
ACTIVATE: RunSet: ACE_QIE 
<!-- Author: Glenn Strycker and Miguel Mondragon -->


<!-- Date: Sun Sep 9 19:52:35.638 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249170>249170</A>
CALIBRATION: <U><B>calqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep  9 20;09;31 comment by...Glenn Strycker -->
BAD --

Same comments as before -- 11W CHA has mean=0; sig>5 in 21E WHA and 5W CEM.

***NEW** -> sig>5 in 14E CEM (channel 9)
Since it is low (just a little above 5) we will ignore reporting this for now.  WATCH this -- it is not a known problem -- and email if it gets bigger.
<!-- Date: Sun Sep 9 19:54:05.215 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249171>249171</A>
ACTIVATE: RunSet: CALIB_CLC_QIE 
<!-- Author: Glenn Strycker and Miguel Mondragon -->


<!-- Date: Sun Sep 9 19:56:02.183 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249171>249171</A>
CALIBRATION: <U><B>clcqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#clcqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep  9 20;11;09 comment by...Glenn Strycker -->
OK
<!-- Date: Sun Sep 9 19:56:24.044 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249172>249172</A>
ACTIVATE: RunSet: CALIB_QIE_FWD 
<!-- Author: Glenn Strycker and Miguel Mondragon -->


<!-- Date: Sun Sep 9 19:57:20.602 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249172>249172</A>
CALIBRATION: <U><B>bscqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#bscqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep  9 20;12;10 comment by...Glenn Strycker -->
OK
<!-- Date: Sun Sep 9 20:07:30.980 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249173>249173</A>
 Activated at 2007.09.09 20:06:07
<!-- Author: RunControl -->


<!-- Date: Sun Sep 9 20:07:31.424 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249173>249173</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [7,957,639] 
<!-- Author: Glenn Strycker and Miguel Mondragon -->


<!-- Date: Sun Sep 9 20:07:32.482 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249173>249173</A>
 Terminated at 2007.09.09 20:06:55
<!-- Author: RunControl -->


<!-- Date: Sun Sep 9 20:08:03.090 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249173>249173</A>
TERMINATE: I was trying to run the calib without removing the pcal07 crate -- I'll remove the smxreadout60 slot and try again.
<!-- Author: Glenn Strycker and Miguel Mondragon -->



<!-- Date: Sun Sep 9 20:09:58.359 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249174>249174</A>
 Activated at 2007.09.09 20:09:38
<!-- Author: RunControl -->


<!-- Date: Sun Sep 9 20:09:59.538 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249174>249174</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [7,957,639] 
<!-- Author: Glenn Strycker and Miguel Mondragon -->
<!-- Comment: Sun Sep  9 20;10;31 comment by...Glenn Strycker -->
We removed the smxreadout_60 slot from the PCAL07 crate during this run.



<!-- Date: Sun Sep 9 20:20:36.896 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249174>249174</A>
 Terminated at 2007.09.09 20:19:52
<!-- Author: RunControl -->


<!-- Date: Sun Sep 9 20:20:37.719 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249174>249174</A>
TERMINATE: DISREGARD THIS RUN -- The consumers crapped out during the middle of this calibration, so we'll re-run it.
<!-- Author: Glenn Strycker and Miguel Mondragon -->


<!-- Date: Sun Sep 9 20:24:31.478 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249175>249175</A>
ACTIVATE: RunSet: ACE_CALORIMETER_MINIMAL  Trigger: SUMET5 [7,957,639]
We removed the smxreadout_60 slot from the PCAL07 crate during this run.
<br>
Got this message (as previously):
Strange (but not necessarily fatal) Run Configuration selections:
Crate PCAL_07 is missing card SMXREADOUT_60 in slot 4
<!-- Author: Glenn Strycker and Miguel Mondragon -->


<!-- Date: Sun Sep 9 20:25:31.219 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249175>249175</A>
 Activated at 2007.09.09 20:23:37
<!-- Author: RunControl -->


<!-- Date: Sun Sep 9 20:34:12.899 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249175>249175</A>
 Terminated at 2007.09.09 20:33:59
<!-- Author: RunControl -->


<!-- Date: Sun Sep 9 20:35:10.426 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249175>249175</A>
TERMINATE: L1A Rate = 50 Hz (good, between 40 and 100 Hz as recommended)
<!-- Author: Glenn Strycker and Miguel Mondragon -->


<!-- Date: Sun Sep 9 20:37:19.387 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249176>249176</A>
CALIBRATION: <U><B>tofqie</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#tofqie target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep  9 21;00;34 comment by...Glenn Strycker -->
BAD -- WCAL04 still causing the missing stripe on plots.
<!-- Date: Sun Sep 9 20:41:55.069 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249177>249177</A>
ACTIVATE: RunSet: SMQIECALIB 
<!-- Author: Glenn Strycker and Miguel Mondragon -->


<!-- Date: Sun Sep 9 20:46:03.037 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249177>249177</A>
CALIBRATION: <U><B>smx</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#smx target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep  9 21;02;21 comment by...Glenn Strycker -->
BAD -- same as previous shift today and yesterday.   However, we ran with PCAL07, so it is somewhat expected.
<!-- Date: Sun Sep 9 20:46:04.866 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249178>249178</A>
ACTIVATE: RunSet: ACE_LED_ALL 
<!-- Author: Glenn Strycker and Miguel Mondragon -->


<!-- Date: Sun Sep 9 20:46:08.305 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249178>249178</A>
CALIBRATION: <U><B>cemled</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cemled target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep  9 21;03;30 comment by...Glenn Strycker -->
BAD -- ADC counts still all 0s.  See elog in previous shift for a more detailed report.
<!-- Date: Sun Sep 9 20:47:55.015 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249179>249179</A>
ACTIVATE: RunSet: COTCALIB_ACE 
<!-- Author: Glenn Strycker and Miguel Mondragon -->


<!-- Date: Sun Sep 9 20:47:56.221 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249179>249179</A>
CALIBRATION: <U><B>cot</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#cot target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep  9 21;05;46 comment by...Glenn Strycker -->
BAD --
<br>
known problem, see elog entries during the past few days -- too many bad channels.
<br>
<pre>
COT Calibration run 249179 taken at 20:45:35 on 2007-09-09

aveStart:                        209.40   
total numBad:                    540     - Too Many or Too Few Bad channels
   numBad  cells:                  1      
   numBad cables:                  0      
   numBad slots:                   2      

Number of missing channels =    0 

aveDrift:                       0.255    
max number drifted in crate:  1374       
total numDrifted:            10422 
   number of drifted  cells       65      
   number of drifted cables       85      
   number of drifted  slots       61      

Too many bad or missing channels in calibration.
Please page Shin-Shan Yu (Eiko) 
status=0
 
Out of the last 10 calibrations, constants have been updated 0 times
</pre>
<!-- Date: Sun Sep 9 20:53:12.068 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249180>249180</A>
ACTIVATE: RunSet: XEF_ALL 
<!-- Author: Glenn Strycker and Miguel Mondragon -->


<!-- Date: Sun Sep 9 20:55:01.302 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249180>249180</A>
CALIBRATION: <U><B>calxef</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#calxef target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep  9 21;06;43 comment by...Glenn Strycker -->
OK
<!-- Date: Sun Sep 9 20:58:47.011 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249181>249181</A>
ACTIVATE: RunSet: CALIB_MUON_ALL1
<br><br>
Got this message:
Strange (but not necessarily fatal) Run Configuration selections:
Crate IMU_00 is missing card TDC_BMU_03 in slot 14
<br><br>
Got a DAQ mode error warning, too, even though we're still in COSMICS mode on the consumers.
<!-- Author: Glenn Strycker and Miguel Mondragon -->



<!-- Date: Sun Sep 9 21:00:52.148 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249182>249182</A>
ACTIVATE: RunSet: TOF_TAC_CALIB 
<!-- Author: Glenn Strycker and Miguel Mondragon -->






<!-- Date: Sun Sep 9 21:06:53.202 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249182>249182</A>
CALIBRATION: <U><B>toftac</B></U> calibration <A HREF=http://www-cdfonline.fnal.gov/internal/daq/Calibrations/web.html#toftac target=_calibresults>results</A> ready for review
<!-- Author: CDFCALIB -->
<!-- Comment: Sun Sep  9 21;08;38 comment by...Glenn Strycker -->
--  BAD -- WCAL04 still causing the missing stripe on plots.  Same problem as for the TOFQIE calibration.

<!-- Date: Sun Sep 9 21:14:17.315 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249183>249183</A>
 Activated at 2007.09.09 21:13:52
<!-- Author: RunControl -->


<!-- Date: Sun Sep 9 21:14:59.352 2007 -->
Run <A HREF=/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249183>249183</A>
ACTIVATE: RunSet: ACE_COSMICS_ALL  Trigger: COSMICS [15,954,639] -- removed the EM_TIMER_00 crate as well as the smxreadout_60 slot from PCAL07 as before.  This should be a stable setup for an overnight run.
<!-- Author: Glenn Strycker and Miguel Mondragon -->



<!-- Date: Sun Sep 9 21:50:06.145 2007 -->
Silicon temperatures are behaving well; flat and stable
<!-- Author: Miguel M. -->


<!-- Date: Sun Sep 9 21:59:12.556 2007 -->
The current cosmics run has been erro-free for 45 minutes. We will leave it running overnight.
<!-- Author: Miguel M. / Glenn S. :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249183' target=_top>249183</a>) -->


<!-- Date: Sun Sep 9 22:01:37.489 2007 -->
We are leaving. Good night
<!-- Author: Miguel M. -->


<!-- Date: Sun Sep 9 22:02:01.204 2007 -->
<pre>The eve shift summary:  

 In control room: taken cosmic, done L2 torture, ACE training and  
  calibration. 
 In Collision Hall:  
   Paolo and Fotis finished WHA/PMA work for the day at 4:00 PM  
 
 There are some issues about TDC swap, the expert is notified.  
 There was a Cryo alarm, but we did not see anything wrong on  
 the cryo monitor. 
 
 We are leaving with running cosmic over night.</pre>    
 
<!-- Author: Scico :: (run <a href='/java/cdfdb/servlet/RunSummary?RUN_NUMBER=249183' target=_top>249183</a>) -->

<!-- Date: Sun Sep 9 23:56:00.995 2007 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>PhysicsTable</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L3Accepts</TH>
<TH>DelivLumi nb<SUP>-1</SUP></TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>23:55:01</TD>
<TD ALIGN=CENTER>::</TD>
<TD ALIGN=RIGHT></TD>               <TD ALIGN=RIGHT></TD>
<TD ALIGN=RIGHT></TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->


<!-- Date: Mon Sep 10 07:55:07.093 2007 -->
<TABLE BORDER=2 CELLSPACING=1 CELLPADDING=1 BGCOLOR=WHITE>
<TR BGCOLOR=YELLOW><TH>RunNumber</TH>
<TH>DataType</TH>
<TH>PhysicsTable</TH>
<TH>Begin</TH>
<TH>End</TH>
<TH>LiveTime</TH>
<TH>L3Accepts</TH>
<TH>DelivLumi nb<SUP>-1</SUP></TH>
<TH>LiveLumi nb<SUP>-1</SUP></TH>
<TH>GR</TH>
<TH>SC</TH>
<TH>RC</TH></TR>
<TR><TD>Totals</TD>
<TD> </TD>
<TD> </TD>
<TD> </TD>
<TD>07:55:00</TD>
<TD ALIGN=CENTER>::</TD>
<TD ALIGN=RIGHT></TD>               <TD ALIGN=RIGHT></TD>
<TD ALIGN=RIGHT></TD>
<TD> </TD>
<TD> </TD>
<TD> </TD></TR>

</TABLE>
<!-- Author: End of Shift Report -->