Specifications/Requirements for Purchase Request RBD.1183

+-----+---------------------------------------------------------+------+
| **  | **Description**                                         | **Q  |
| Ite |                                                         | ty** |
| m** |                                                         |      |
+-----+---------------------------------------------------------+------+
| 1   | Digital Pressure Indicator                              | 1    |
|     |                                                         |      |
|     | -   Range: 1 psi differential or lower                  |      |
|     |                                                         |      |
|     | -   Precision: 0.004% of FS                             |      |
|     |                                                         |      |
|     | -   Resolution: 0.001 inH20 or better                   |      |
|     |                                                         |      |
|     | -   Stability: 0.01% per year                           |      |
|     |                                                         |      |
|     | -   Zero Drift: 0.005% of full scale per 24 hours       |      |
|     |                                                         |      |
|     | -   Re-zeroing capability                               |      |
|     |                                                         |      |
|     | -   Digital Display                                     |      |
|     |                                                         |      |
|     | -   Display Resolution: 0.000001                        |      |
|     |                                                         |      |
|     | -   Pressure Medium: air or nitrogen                    |      |
|     |                                                         |      |
|     | ```{=html}                                              |      |
|     | <!-- -->                                                |      |
|     | ```                                                     |      |
|     | -   Programmable via IEEE-488 Interface                 |      |
|     |                                                         |      |
|     | -   LabVIEW driver                                      |      |
|     |                                                         |      |
|     | -   NIST traceable calibration                          |      |
|     |                                                         |      |
|     | -   Rack mount                                          |      |
+-----+---------------------------------------------------------+------+
| 2   | Digital Pressure Indicator                              | 1    |
|     |                                                         |      |
|     | -   Range: 1 psi differential                           |      |
|     |                                                         |      |
|     | -   Precision: 0.004% of FS                             |      |
|     |                                                         |      |
|     | -   Resolution: 0.001 inH20 or better                   |      |
|     |                                                         |      |
|     | -   Stability: 0.01% per year                           |      |
|     |                                                         |      |
|     | -   Zero Drift: 0.005% of full scale per 24 hours       |      |
|     |                                                         |      |
|     | -   Re-zeroing capability                               |      |
|     |                                                         |      |
|     | -   Digital Display                                     |      |
|     |                                                         |      |
|     | -   Display Resolution: 0.000001                        |      |
|     |                                                         |      |
|     | -   Pressure Medium: air or nitrogen                    |      |
|     |                                                         |      |
|     | ```{=html}                                              |      |
|     | <!-- -->                                                |      |
|     | ```                                                     |      |
|     | -   Programmable via IEEE-488 Interface                 |      |
|     |                                                         |      |
|     | -   LabVIEW driver                                      |      |
|     |                                                         |      |
|     | -   NIST traceable calibration                          |      |
|     |                                                         |      |
|     | -   Rack Mount                                          |      |
+-----+---------------------------------------------------------+------+
| 3   | Digital Pressure Indicator                              | 1    |
|     |                                                         |      |
|     | -   Range: 5 psi differential                           |      |
|     |                                                         |      |
|     | -   Precision: 0.003% of FS                             |      |
|     |                                                         |      |
|     | -   Resolution: 0.001 inH20 or better                   |      |
|     |                                                         |      |
|     | -   Stability: 0.01% per year                           |      |
|     |                                                         |      |
|     | -   Zero Drift: 0.005% of full scale per 24 hours       |      |
|     |                                                         |      |
|     | -   Re-zeroing capability                               |      |
|     |                                                         |      |
|     | -   Digital Display                                     |      |
|     |                                                         |      |
|     | -   Display Resolution: 0.000001                        |      |
|     |                                                         |      |
|     | -   Pressure Medium: air or nitrogen                    |      |
|     |                                                         |      |
|     | ```{=html}                                              |      |
|     | <!-- -->                                                |      |
|     | ```                                                     |      |
|     | -   Programmable via IEEE-488 Interface                 |      |
|     |                                                         |      |
|     | -   LabVIEW driver                                      |      |
|     |                                                         |      |
|     | -   NIST traceable calibration                          |      |
|     |                                                         |      |
|     | -   Rack Mount                                          |      |
+-----+---------------------------------------------------------+------+