14:39:54 SNSI1BA Up Ramp Aperture =1.E‑5

n:\\raw\\SNSinj\\SNSI1BA.001 Smoothed Linear

Time down: 1.00 ms

Imax = +1260.76 A Imin = +13.13 A

Integration time = .00498 s

Idiff = ‑1247.431 Max Idot = 670840 A/s

Ips2 = +1260.60 A Ips1 = +13.17 A

Idiff2 = +1247.473 Max Idot = 670840 A/s

Theta is ‑0.29 CW dB = (775,776)(776,776)(774,774)

Vc1 Max +1.94389 V Min ‑0.01032 V

Integral: +3.49207 mV‑s Aperture: 2.97 ms +3.48611 mV‑s

Offset Corrected Integral: +3.49440 mV‑s

Vc2 Max +0.00810 V Min ‑1.92859 V

Integral: ‑3.46458 mV‑s Aperture: 2.97 ms ‑3.45700 mV‑s

Offset Corrected Integral: ‑3.46216 mV‑s

\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_

14:39:54 SNSI1BA Down Ramp Aperture =1.E‑5

n:\\raw\\SNSinj\\SNSI1BA.001 Smoothed Linear

Time down: 1.00 ms

Imax = +1260.76 A Imin = +13.13 A

Integration time = .00498 s

Idiff = +1242.983 Max Idot = ‑1.3184E+6 A/s

Ips2 = +17.71 A Ips1 = +1260.69 A

Idiff2 = ‑1243.142 Max Idot = ‑1.3184E+6 A/s

Theta is ‑0.30 CW dB = (771,771)(760,760)(775,775)

Vc1 Max +0.02173 V Min ‑3.80728 V

Integral: ‑3.47483 mV‑s Aperture: 2.97 ms ‑3.49067 mV‑s

Offset Corrected Integral: ‑3.42492 mV‑s

Vc2 Max +3.77528 V Min ‑0.02185 V

Integral: +3.44377 mV‑s Aperture: 2.97 ms +3.46175 mV‑s

Offset Corrected Integral: +3.39270 mV‑s

\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_

14:44:14 SNSI1BA Up Ramp Aperture =1.E‑5

n:\\raw\\SNSinj\\SNSI1BA.002 Exponential Drop

Time down: 1.40 ms tau = 0.20

Imax = +1260.81 A Imin = +13.13 A

Integration time = .00538 s

Idiff = ‑1247.500 Max Idot = 667840 A/s

Ips2 = +1260.69 A Ips1 = +13.19 A

Idiff2 = +1247.551 Max Idot = 667840 A/s

Theta is ‑0.29 CW dB = (775,776)(779,778)(774,774)

Vc1 Max +1.90954 V Min ‑0.00934 V

Integral: +3.49136 mV‑s Aperture: 2.97 ms +3.48639 mV‑s

Offset Corrected Integral: +3.50738 mV‑s

Vc2 Max +0.00716 V Min ‑1.89380 V

Integral: ‑3.46379 mV‑s Aperture: 2.97 ms ‑3.45722 mV‑s

Offset Corrected Integral: ‑3.47531 mV‑s

\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_

14:44:14 SNSI1BA Down Ramp Aperture =1.E‑5

n:\\raw\\SNSinj\\SNSI1BA.002 Exponential Drop

Time down: 1.40 ms tau = 0.20

Imax = +1260.81 A Imin = +13.13 A

Integration time = .00538 s

Idiff = +1247.023 Max Idot = ‑559260 A/s

Ips2 = +13.73 A Ips1 = +1260.75 A

Idiff2 = ‑1247.065 Max Idot = ‑559260 A/s

Theta is ‑0.24 CW dB = (775,775)(776,776)(776,775)

Vc1 Max +0.01856 V Min ‑8.58445 V

Integral: ‑3.49034 mV‑s Aperture: 2.97 ms ‑3.49130 mV‑s

Offset Corrected Integral: ‑3.49416 mV‑s

Vc2 Max +8.49675 V Min ‑0.02142 V

Integral: +3.46070 mV‑s Aperture: 2.97 ms +3.46252 mV‑s

Offset Corrected Integral: +3.46748 mV‑s

\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_

14:46:26 SNSI1BA Up Ramp Aperture =1.E‑5

n:\\raw\\SNSinj\\SNSI1BA.003 1 ‑ Sqrt(t/T)

Time down: 0.11 ms tau = 0.10

Imax = +1260.86 A Imin = +13.14 A

Integration time = .00408 s

Idiff = ‑1247.476 Max Idot = 689620 A/s

Ips2 = +1260.69 A Ips1 = +13.22 A

Idiff2 = +1247.689 Max Idot = 689620 A/s

Theta is ‑0.35 CW dB = (773,777)(770,770)(773,775)

Vc1 Max +2.02185 V Min ‑0.02953 V

Integral: +3.48640 mV‑s Aperture: 2.97 ms +3.48578 mV‑s

Offset Corrected Integral: +3.47223 mV‑s

Vc2 Max +0.03411 V Min ‑2.01416 V

Integral: ‑3.46482 mV‑s Aperture: 2.97 ms ‑3.45801 mV‑s

Offset Corrected Integral: ‑3.43422 mV‑s

\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_

14:46:26 SNSI1BA Down Ramp Aperture =1.E‑5

n:\\raw\\SNSinj\\SNSI1BA.003 1 ‑ Sqrt(t/T)

Time down: 0.11 ms tau = 0.10

Imax = +1260.86 A Imin = +13.14 A

Integration time = .00408 s

Idiff = +1234.516 Max Idot = ‑2.3356E+6 A/s

Ips2 = +26.25 A Ips1 = +1260.77 A

Idiff2 = ‑1235.453 Max Idot = ‑2.3356E+6 A/s

Theta is ‑0.28 CW dB = (765,765)(746,746)(775,776)

Vc1 Max +0.01477 V Min ‑12.94626 V

Integral: ‑3.44527 mV‑s Aperture: 2.97 ms ‑3.49060 mV‑s

Offset Corrected Integral: ‑3.36151 mV‑s

Vc2 Max +12.83122 V Min ‑0.02274 V

Integral: +3.41400 mV‑s Aperture: 2.97 ms +3.46362 mV‑s

Offset Corrected Integral: +3.33213 mV‑s

\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_

14:49:28 SNSI1BA Up Ramp Aperture =1.E‑5

n:\\raw\\SNSinj\\SNSI1BA.004 1 ‑ Sqrt(t/T)

Time down: 1.21 ms tau = 1.10

Imax = +1260.75 A Imin = +13.14 A

Integration time = .00518 s

Idiff = ‑1247.350 Max Idot = 659280 A/s

Ips2 = +1260.54 A Ips1 = +13.19 A

Idiff2 = +1247.293 Max Idot = 659280 A/s

Theta is ‑0.27 CW dB = (775,776)(787,787)(774,774)

Vc1 Max +1.99248 V Min ‑0.02975 V

Integral: +3.49170 mV‑s Aperture: 2.97 ms +3.48602 mV‑s

Offset Corrected Integral: +3.54404 mV‑s

Vc2 Max +0.02678 V Min ‑1.97385 V

Integral: ‑3.46319 mV‑s Aperture: 2.97 ms ‑3.45699 mV‑s

Offset Corrected Integral: ‑3.51330 mV‑s

\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_

14:49:28 SNSI1BA Down Ramp Aperture =1.E‑5

n:\\raw\\SNSinj\\SNSI1BA.004 1 ‑ Sqrt(t/T)

Time down: 1.21 ms tau = 1.10

Imax = +1260.75 A Imin = +13.14 A

Integration time = .00518 s

Idiff = +1245.674 Max Idot = ‑743080 A/s

Ips2 = +15.06 A Ips1 = +1260.74 A

Idiff2 = ‑1245.809 Max Idot = ‑743080 A/s

Theta is ‑0.24 CW dB = (774,773)(770,770)(775,775)

Vc1 Max +0.02764 V Min ‑5.52135 V

Integral: ‑3.48412 mV‑s Aperture: 2.97 ms ‑3.49081 mV‑s

Offset Corrected Integral: ‑3.46537 mV‑s

Vc2 Max +5.46727 V Min ‑0.02626 V

Integral: +3.45443 mV‑s Aperture: 2.97 ms +3.46197 mV‑s

Offset Corrected Integral: +3.43869 mV‑s

\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_

14:52:03 SNSI1BA Up Ramp Aperture =1.E‑5

n:\\raw\\SNSinj\\SNSI1BA.005 Smoothed Linear

Time down: 0.50 ms

Imax = +1260.76 A Imin = +13.14 A

Integration time = .00448 s

Idiff = ‑1247.464 Max Idot = 678500 A/s

Ips2 = +1260.67 A Ips1 = +13.20 A

Idiff2 = +1247.491 Max Idot = 678500 A/s

Theta is ‑0.29 CW dB = (775,776)(760,760)(774,774)

Vc1 Max +1.99953 V Min ‑0.03170 V

Integral: +3.49229 mV‑s Aperture: 2.97 ms +3.48653 mV‑s

Offset Corrected Integral: +3.42488 mV‑s

Vc2 Max +0.02915 V Min ‑1.98551 V

Integral: ‑3.46420 mV‑s Aperture: 2.97 ms ‑3.45737 mV‑s

Offset Corrected Integral: ‑3.39330 mV‑s

\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_

14:52:03 SNSI1BA Down Ramp Aperture =1.E‑5

n:\\raw\\SNSinj\\SNSI1BA.005 Smoothed Linear

Time down: 0.50 ms

Imax = +1260.76 A Imin = +13.14 A

Integration time = .00448 s

Idiff = +1240.892 Max Idot = ‑2.312E+6 A/s

Ips2 = +19.61 A Ips1 = +1260.50 A

Idiff2 = ‑1241.220 Max Idot = ‑2.312E+6 A/s

Theta is ‑0.31 CW dB = (769,770)(755,755)(775,776)

Vc1 Max +0.02515 V Min ‑7.38363 V

Integral: ‑3.46663 mV‑s Aperture: 2.97 ms ‑3.49176 mV‑s

Offset Corrected Integral: ‑3.40294 mV‑s

Vc2 Max +7.31723 V Min ‑0.02408 V

Integral: +3.43529 mV‑s Aperture: 2.97 ms +3.46299 mV‑s

Offset Corrected Integral: +3.36959 mV‑s

\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_

14:53:39 SNSI1BA Up Ramp Aperture =1.E‑5

n:\\raw\\SNSinj\\SNSI1BA.006 Exponential Drop

Time down: 1.05 ms tau = 0.15

Imax = +1260.69 A Imin = +13.14 A

Integration time = .00502 s

Idiff = ‑1247.398 Max Idot = 682110 A/s

Ips2 = +1260.57 A Ips1 = +13.18 A

Idiff2 = +1247.419 Max Idot = 682110 A/s

Theta is ‑0.27 CW dB = (775,776)(778,778)(774,774)

Vc1 Max +1.97436 V Min ‑0.01894 V

Integral: +3.49217 mV‑s Aperture: 2.97 ms +3.48644 mV‑s

Offset Corrected Integral: +3.50244 mV‑s

Vc2 Max +0.01820 V Min ‑1.95612 V

Integral: ‑3.46451 mV‑s Aperture: 2.97 ms ‑3.45734 mV‑s

Offset Corrected Integral: ‑3.47297 mV‑s

\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_

14:53:39 SNSI1BA Down Ramp Aperture =1.E‑5

n:\\raw\\SNSinj\\SNSI1BA.006 Exponential Drop

Time down: 1.05 ms tau = 0.15

Imax = +1260.69 A Imin = +13.14 A

Integration time = .00502 s

Idiff = +1246.491 Max Idot = ‑749090 A/s

Ips2 = +14.14 A Ips1 = +1260.63 A

Idiff2 = ‑1246.524 Max Idot = ‑749090 A/s

Theta is ‑0.28 CW dB = (774,774)(772,772)(775,775)

Vc1 Max +0.01383 V Min ‑10.01577 V

Integral: ‑3.48697 mV‑s Aperture: 2.97 ms ‑3.49143 mV‑s

Offset Corrected Integral: ‑3.47815 mV‑s

Vc2 Max +9.91504 V Min ‑0.01289 V

Integral: +3.45741 mV‑s Aperture: 2.97 ms +3.46274 mV‑s

Offset Corrected Integral: +3.44728 mV‑s

\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_\_