**Sec. 2.22. Automatic Bulk Weighing Systems**[^1]

# A. Application

**A.1. General.** - This code applies to automatic bulk weighing
sys­tems, that is, weighing systems adapted to the automatic weighing of
a commodity in successive drafts of predetermined amounts auto­matically
record­ing the no-load and loaded weight values and accu­mulating the net
weight of each draft.

(Amended 1987)

**A.2. Also see General Code Requirements.**

# S. Specifications

**S.1. Design of Indicating and Recording Elements and Recorded
Repre­sentations.**

> **S.1.1. Zero Indication.** - Provisions shall be made to indi­cate and
> record a no-load reference value and, if the no-load reference value
> is a zero value indica­tion, to indicate and record an out-of-balance
> condition on both sides of zero.
>
> **S.1.1.1. Digital Zero Indication.** - A digital zero indi­cation
> shall represent a balance condition that is within ± 1/2 the value of
> the scale division.
>
> ***S.1.2. Value of Scale Division (d).** - The value of the scale
> division (d), expressed in a unit of weight, shall be equal to:*
>
> *(a) 1, 2, or 5; or*
>
> *(b) a decimal multiple or submultiple of 1, 2, or 5; or*
>
> *(c) a binary submultiple of a unit of weight.*

Examples: Scale divisions may be 0.01, 0.02, or 0.05; 0.1, 0.2, or 0.5;
1, 2, or 5; 10, 20, or 50; or 1/2, 1/4, 1/8, 1/16, etc.

> *\[Nonretroactive as of January 1, 1986\]*
>
> (Amended 1987)
>
> **S.1.3. Capacity Indication and Recorded Representation.** - ­An
> indicating or recording element shall not indicate or record any
> values when the gross load is in excess of 105 % of the capacity of
> the system.
>
> **S.1.4. Weighing Sequence.** - For systems used to receive (weigh
> in), the no-load reference value shall be determined and recorded only
> at the beginning of each weighing cycle. For systems used to deliver
> (weigh out), the no-load reference value shall be determined and
> recorded only after the gross load reference value for each weighing
> cycle has been indicated and recorded.
>
> **S.1.5. Recording Sequence.** - Provision shall be made so that all
> weight values are indicated until the completion of the recording of
> the indicated value.
>
> **S.1.6. Provision for Sealing Adjustable Components on Elec­tronic
> Devices.** - Provision shall be made for applying a security seal in a
> manner that requires the security seal to be broken before an
> adjustment can be made to any component affecting the performance of
> the device.

**S.2. Design of Balance and Damping Mechanism.**

> **S.2.1. Zero-Load Adjustment.** - The weighing system shall be
> equipped with manual or semiautomatic means by which the zero-­load
> balance or no-load refer­ence value indication may be adjusted. An
> automatic zero setting mechanism is prohibited.
>
> **S.2.1.1. Manual.** - A manual zero-load or no-load reference value
> setting mechanism shall be operable or accessible only by a tool
> outside of or entirely separate from this mechanism or enclosed in a
> cabinet.
>
> **S.2.1.2. Semiautomatic.** - A semiautomatic zero-load or no-load
> reference value setting mechanism shall meet the provisions of
> S.2.1.1. or shall be operable only when:
>
> \(a\) the indication is stable within ± 3 scale divisions, and
>
> \(b\) cannot be operated during a weighing operation.
>
> **S.2.2. Damping Means.** - A system shall be equipped with effective
> means necessary to bring the indications quickly to a readable, stable
> equilibrium. Effective means shall also be provided to permit the
> recording of weight values only when the indication is stable within
> plus or minus three scale divisions for devices with 10 000 scale
> divisions, or plus or minus one division for devices with less than
> 10 000 scale divi­sions.

**S.3. Interlocks and Gate Control.**

> **S.3.1. Gate Position.** - Provision shall be made to clearly
> indicate to the operator the position of the gates leading directly to
> and from the weigh hopper.
>
> **S.3.2. Interlocks.** - Each automatic bulk weighing system shall
> have operating interlocks to provide for the following:

\(a\) Product cannot be cycled and weighed if the weight record­ing
element is disconnected or subjected to a power loss.

\(b\) The recording element cannot print a weight if either of the gates
leading directly to or from the weigh hopper is open.

\(c\) A \"low paper\" sensor, when provided, is activated.

\(d\) The system will operate only in the proper sequence in all modes
of operation.

\(e\) When an overfill alarm is activated, the system shall indicate and
record an overfill condition.

> (Amended 1993)
>
> **S.3.3. Overfill Sensor.**

\(a\) The weigh hopper shall be equipped with an overfill sensor which
will cause the feed gate to close, activate an alarm, and inhibit
weighing until the overfill condition has been corrected.

> (Added 1993)
>
> *(b) If the system is equipped with a lower garner or surge bin, that
> garner shall also be equipped with an overfill sensor which will cause
> the gate of the weigh hopper to remain open, activate an alarm, and
> inhibit weighing until the overfill condition has been corrected.*
>
> *\[Nonretroactive as of January 1, 1998\]*
>
> (Amended 1997)

**S.4. Design of Weighing Elements.**

> **S.4.1. Antifriction Means.** - At all points at which a live part of
> the mecha­nism may come into contact with another part in the course of
> normal usage, frictional effects shall be reduced to a minimum by
> means of suitable antifriction means, opposing surfaces and points
> being properly shaped, finished, and hardened.
>
> **S.4.2. Adjustable Components.** - An adjustable component, such as a
> potenti­ometer, shall be held securely in adjustment and, except for a
> component for adjusting level or a no-load reference value, shall not
> be adjustable from the outside of the device.
>
> **S.4.3. Multiple Load-Receiving Elements.** - A system with a single
> indicating or recording element, or a combination indicating recording
> element, that is coupled to 2 or more load-receiving elements with
> inde­pendent weighing systems, shall be provided with means to prohibit
> the activation of any load-­receiving element (or elements) not in use,
> and shall be provided with automatic means to indicate clearly and
> definitely which load-receiving element (or ele­ments) is in use.
>
> **S.4.4. Venting.** - All weighing systems shall be vented so that any
> internal or external pressure will not affect the accuracy or
> operation of the system.

**S.5. Marking Requirements. (See also G-S.1.)**

> **S.5.1. Capacity and Value of the Scale Division.** - The capa­city of
> the weighing system and the value of the scale division shall be
> clearly and conspicuously marked on the indicating element near the
> weight value indications.
>
> **S.5.2. Weighing Elements.** - On a weighing element not permanently
> attached to an indicating element, there shall be clearly and
> permanently marked for the purposes of identification, the name,
> initials, or trademark of the manu­facturer, the manufacturer\'s
> designation that positively identifies the pattern or design, and the
> nominal capacity.
>
> ***S.5.3. Temperature Limits.** - Unless the temperature range is -10
> °C to +40 °C (14 °F to 104 °F), the temperature range shall be marked
> on the device.*
>
> *\[Non­retroactive as of January 1, 1986\]*
>
> (Added 1985)
>
> ***S.5.4. Accuracy Class.***
>
> *(a) All systems used to weigh grain shall be marked Class III\*.*

(b) *All other systems shall be marked either Class III or III L\*.*

> *(\*See Section 2.20 Scales Code for the parameters for these accuracy
> classes for scales. The specific requirements for automatic bulk
> weighing systems apply to these devices when there is a conflict
> between the Scales Code and the Automatic Bulk Weighing Systems
> Code.)*
>
> *\[Nonretroactive as of January 1, 1986\]*
>
> (Added 1985) (Amended 1992)

# N. Notes

**N.1. Testing Procedures.**

> **N.1.1. Test Weights.** - The increasing load test shall be conducted
> using test weights equal to at least 10 % of the capacity of the
> system:
>
> \(a\) on automatic grain bulk-weighing systems installed after January
> 1, 1984, and

(b) on other automatic bulk-weighing systems installed after January 1,
    1986.

> (Amended 1987)
>
> **N.1.2. Increasing-Load Test.** - An increasing-load test consisting
> of substitution and strain-load tests shall be conducted up to the
> used capacity of the weighing system.
>
> (Amended 1987)
>
> **N.1.3. Decreasing-Load Test.** - A decreasing-load test shall be
> conducted on devices used to weigh out.
>
> (Added 1986)
>
> **N.1.4. Zero Balance or No-Load Reference Value Change Test.** - A
> test for change of zero-balance or no-load reference value shall be
> con­ducted on all scales after the removal of any test load. The change
> shall not be more than the minimum tolerance applicable.
>
> ***N.1.5. Discrimination Test.** - A discrimination test shall be
> conducted on all automatic indicating scales with the weighing device
> in equilibrium at zero-load and at maximum test load, and under
> controlled conditions in which environmental factors are reduced to
> the extent that they will not affect the results obtained.*
>
> *\[Nonretroactive as of January 1, 1986\]*
>
> **N.1.5.1. Digital Device.** - On a digital device, this test is
> conducted from just below the lower edge of the zone of uncertainty
> for increasing-load tests, or from just above the upper edge of the
> zone of uncertainty for decreasing-load tests.

(Added 1987)

**N.2. Verification (Testing) Standards.** - Standard weights and masses
used in verifying weighing devices shall comply with requirements of
NIST Handbook 105-1 (Class F) or the tolerances expressed in Appendix A,
Funda­mental Considerations, paragraph 3.2. (i.e., one-third of the
smallest tolerance applied).­

# T. Tolerances

**T.1. Tolerance Application.** - Tolerance values shall be applied to
all indications and recorded representations of a weighing system.

**T.1.1.** **To Errors of Underregistration and Overregistration.** -
The tolerances hereinafter prescribed shall be applied equally to errors
of under­registration and errors of overregistration.

> **T.1.2. To Increasing-Load Tests.** - Basic tolerances shall be
> applied.
>
> **T.1.3. To Decreasing-Load Tests.** - Basic tolerances shall be
> applied to systems used to weigh out.
>
> (Added 1986)
>
> **T.1.4. To Tests Involving Digital Indica­tions or
> Representa­tions.**­ - To the tolerances that would otherwise be
> applied, there shall be added an amount equal to one-half the value of
> the scale division. This does not apply to digital indications or
> recorded representations that have been corrected for rounding using
> error weights.
>
> (Added 1986)

**T.2. Minimum Tolerance Values.** - The minimum tolerance value shall
not be less than half the value of the scale division.

> **T.2.1. For Systems used to Weigh Construction Materials.** - The
> minimum maintenance and acceptance tolerance shall be 0.1 % of the
> weigh­ing capacity of the system, or the value of the scale division,
> whichever is less.
>
> (Added 1986)

**T.3. Basic Tolerance Values.**

> **T.3.1. Acceptance Tolerance.** - The basic acceptance tolerance
> shall be one-half the basic maintenance tolerance.
>
> **T.3.2. For Systems used to Weigh Grain.** - The basic mainte­nance
> toler­ance shall be 0.1 % of test load.
>
> **T.3.3. For­ all­ Other Systems.** - The basic maintenance tolerance
> shall be 0.2 % of test load.
>
> (Amended 1986)

***T.4. Time Dependence.** - At constant test conditions, the indication
20 seconds after the application of a load and the indication after 1
hour shall not differ by more than the absolute value of the applicable
tolerance for the applied load.*

*\[Non­retroactive and enforceable as of January 1, 1987\]*

(Added 1986)

**T.5. Repeatability.** - The results obtained by several weighings of
the same load under reasonably static test conditions shall agree within
the absolute value of the maintenance tolerance for that load, and shall
be within applicable tolerances.

(Added 1986)

**T.6. Discrimination, Digital Automatic Indicating Scales.** - A test
load equivalent to 1.4 d shall cause a change in the indicated or
recorded value of at least 2.0 d. This requires the zone of uncertainty
to be not greater than 0.3 times the value of the scale division.

(Added 1985)

***T.7. Influence Factors.** - The following factors are applicable to
tests conducted under controlled conditions only, provided that:*

> *(a) types of devices approved prior to January 1, 1986, and
> manufac­tured prior to January 1, 1988, need not meet the requirements
> of this section; and*
>
> *(b) new types of devices submitted for approval after January 1,
> 1986, shall comply with the requirements of this section; and*
>
> *(c) all devices manufactured after January 1, 1988, shall comply with
> the re­quirements of this section.*

*\[Nonretroactive as of January 1, 1986\]*

> ***T.7.1. Temperature.** - Devices shall satisfy the tolerance
> require­ments under the following temperature conditions:*
>
> ***T.7.1.1.** If not marked on the device, the temperature limits
> shall be:*
>
> *-10 ºC to 40 ºC* *(14 ºF to 104 ºF)*
>
> ***T.7.1.2.** If temperature limits are specified for the device, the
> range shall be at least 30 °C (54 °F).*
>
> ***T.7.1.3. Temperature Effect on Zero-Load Balance.** - The zero-­load
> indicator shall not vary by more than 1 division per 5 ºC (9 ºF)
> change in tempera­ture*.
>
> ***T.7.1.4. Operating Temperature.** - An indicating or recording
> element shall not display or record any usable values until the
> operating temperature necessary for accurate weighing and a stable
> zero‑balance condition has been attained.*
>
> *\[Nonretroactive as of January 1, 1986\]*
>
> ***T.7.2. Barometric Pressure.** - The zero indication shall not vary
> by more than one scale division for a change in barometric pressure of
> 1 kPa over the total barometric range of 95 kPa to 105 kPa (28 in to
> 31 in of mercury).*
>
> *\[Nonretroactive as of January 1, 1986\]*
>
> **T.7.3. Electric Power Supply.**

## T.7.3.1. Power Supply, Voltage and Frequency.

> \(a\) Weighing devices that operate using alternating current must
> perform within the conditions defined in paragraphs T.2. through T.7.
> inclusive over the line voltage range of 100 V to 130 V or 200 V
> to 250 V rms as appropriate and over the frequency range of 59.5 Hz to
> 60.5 Hz.
>
> \(b\) Battery-operated instruments shall not indicate nor record
> values outside the applicable tolerance limits when battery power
> output is excessive or deficient.
>
> ***T.7.3.2. Power Interruption.** - A power interruption shall not
> cause an indicating or recording element to display or record any
> values outside the appli­cable tolerance limits.*
>
> *\[Nonretroactive as of January 1, 1986\]*
>
> (Added 1985)

# UR. User Requirements

**UR.1. Selection Requirements.**

> ***UR.1.1. For Systems used to Weigh Grain.** - The­ number of scale
> divisions of a weighing system shall not be less than 2 000 nor
> greater than 10 000 divisions.*
>
> *\[Nonretroactive as of January 1, 1984\]*
>
> (Amended 1986 and 1992)
>
> ***U.R.1.2. For Systems used to Weigh Commodities other than
> Grain.** - The number of scale divisions shall not be less than 500
> nor greater than 10 000.*
>
> *\[­Nonretroactive as of January 1, 1987\]*
>
> (Added 1986)

**UR.2. Installation Requirements.**

> **UR.2.1. Protection From Environmental Factors.** - The indicat­ing
> elements, the lever system or load cells, the load-receiving element,
> and any perma­nently installed test weights shall be adequately
> protected from environ­mental factors such as wind, weather, and RFI
> that may ad­versely affect the operation or performance of the system.
>
> **UR.2.2. Foundation, Supports, and Clearance.** - The foundation and
> supports of any system shall be such as to provide strength, rigidity,
> and per­manence of all components, and clearance shall be provided
> around all live parts so that no contact can result before or during
> operation of the system.

**UR.3. Loading Requirements.**

> **UR.3.1. For Systems Used to Weigh Grain.** - A system shall not be
> used to weigh drafts less than 40 % of the weighing capacity of the
> system except for a final partial draft. Loads shall not normally be
> retained on the weighing element for a period longer than a normal
> weighing cycle.
>
> (Amended 1986)
>
> ***UR.3.2. For Systems Used to Weigh Commodities Other than Grain.** -
> A system shall not be used to weigh drafts less than 20 % of the
> weighing capacity of the system except for a final partial draft.
> Loads shall not normally be retained on the weighing element for a
> period longer than a normal weighing cycle.*
>
> *\[Nonretroactive as of January 1, 1987\]*
>
> (Added 1986)

**UR.4. System Modification.** - The weighing system shall not be
modi­fied except when the modification has been approved by a competent
engineering authority, preferably that of the engineering department of
the manufacturer of the scale, and the official with statutory authority
having jurisdiction over the scale.

(Amended 1991)

[^1]: ^1^(Title amended 1986)