AndreaBozzo/ceres-open-data-index huggingface dataset:
COSTOF2, data from Airmar WS200 weather station on BOREL_V1 node, 2022-2023

[Lien vers la fiche source] This dataset contains data form the AIRMAX WS200 weather station including GPS and pitch/roll informations. This log is done by the COSTOF2 (COmmunication and STOrage Front-end). It is the Electronic core of BOREL the buoy that retrieves the data via satellite communication.
Autonomous temperature probes, data from the EMSO-Azores observatory, 2012-2013

[Lien vers la fiche source] This dataset contains temperature data acquired between July 2012 and August 2013 using 17 temperature probes (HTW, HTNKE and LTW) installed in smoker orifices or cracks. High-temperature WHOI probes (HTW) are composed of 2 J-type thermistors in the rod tip connected to 2 temperature data-loggers (named A & B), 2 data recordings are available for 1 HTW probe. They are able to record temperatures up to 415°C with an accuracy of ± 1.1°C, a resolution of 0.25°C at 375°C and a clock drift of 1min per month. High-temperature NKE probes (HTNKE) are able to record temperatures up to 450°C, with an accuracy of ±0.5°C, a resolution of 0.1°C and a clock drift of 1min/month. Low-temperature WHOI probes (LTW) are able to record temperatures up to 125°C. with an accuracy of ±0.22°C, a resolution of 0.025°C and a clock drift of ±1min per month. The array is not connected to an energy node. All probes are calibrated in time before deployment, probe time adjusted to UTC time. Temperature of probes is checked after recovery and before deployment. Clock drift after recovery is not implemented in data.
COSTOF2, data from AIS (MC MURDO KANATON3) on BOREL_V1 node, 2020-2021

[Lien vers la fiche source] This dataset contains data from the AIS ((KANATON3). It provides GPS informations (time, poisition, speed, direction) . This log starts on septembre 14 2020, stops on may 29 2021 and is done by the COSTOF2 (COmmunication and STOrage Front-end). It is the Electronic core of BOREL the buoy that retrieve the data via satellite communication.
DEAFS – Deep-sea Autonomous and automated Fluid sampler data from the EMSO-Azores observatory, 2019

[Lien vers la fiche source] High temperature hydrothermal fluids are collected by the Deep-sea Autonomous and automated Fluid Sampler. The instrument is deployed at the footstep of the Montsegur edifice. The site, named Montserrat, expels hydrothermal fluid at 275°C which were collected beforehand by gas-tight fluid sampler manipulated and triggered by the hydraulic arm of the submarine vehicle (HOV Nautile). Several chemical analyses are performed aboard the research vessel: such as salinity and d20/20 by refractometer, pH and Eh by electrode (pH standard 4.01 and 7; Eh standard solution at -220mV), H2S and total S concentrations measured by electrode (AquaMS, amperometric method) at measured pH value. Unfortunately, the DEAFS felt down shortly after its set-up. Only one fluid has been collected. Regional Node: EMSO-Azores, Lucky Strike Hydrothermal Field Vent field : Montserrat at the base of Montsegur Latitude : 37º17.2744’N Longitude : 32º16.5461’W 1699 m depth
Geochemistry of hydrothermal fluids at the Lucky Strike Hydrothermal Field data from the EMSO-Azores observatory, 2015

[Lien vers la fiche source] High temperature hydrothermal fluids are collected during the MOMARSAT 2015 cruise by gas-tight fluid sampler manipulated and triggered by the hydraulic arm of the submarine vehicle (ROV Victor6000). Several hydrothermal vents are sampled, namely : Aisics, Montsegur, Isabel, Sintra, Y3, White Castle, South Crystal, Crystal, Capelinhos, Cimendeff, Tour Eiffel, Cypress, Sapin. Several chemical analyses are performed aboard the research vessel: such as salinity and d20/20 by refractometer, Conductivity, salinity and total dissolved solid (TDS) by electrode (standard solution at 1413 µS/cm), pH and Eh by electrode (pH standard 4.01 and 7; Eh standard solution at -220mV), iron concentration by photometer (up to 5 ppm), H2S and total S concentrations measured by electrode (AquaMS, amperometric method) at measured pH value… Regional Node: EMSO-Azores Vent field : Lucky Strike Latitude : 37º17’N Longitude : 32º16’W 1600 – 1750 m depth
Geochemistry of hydrothermal fluids at the Lucky Strike Hydrothermal Field data from the EMSO-Azores observatory, 2013

[Lien vers la fiche source] High temperature hydrothermal fluids are collected during the MOMARSAT 2013 by gas-tight fluid sampler manipulated and triggered by the hydraulic arm of the submarine vehicle (ROV Victor6000). Several hydrothermal vents are sampled, namely : Aisics, Montsegur, Isabel, Sintra, Y3, White Castle, South Crystal, Crystal, Capelinhos, Cimendeff, Tour Eiffel, Cypress, Sapin. Several chemical analyses are performed aboard the research vessel: such as salinity and d20/20 by refractometer, Conductivity, salinity and total dissolved solid (TDS) by electrode (standard solution at 1413 µS/cm), pH and Eh by electrode (pH standard 4.01 and 7; Eh standard solution at -220mV), iron concentration by photometer (up to 5 ppm). Regional Node: EMSO-Azores Vent field : Lucky Strike Latitude : 37º17’N Longitude : 32º16’W 1600 – 1750 m depth
COSTOF2, technical data from SeaMoN East, node of the EMSO-Azores observatory, 2016-2017

[Lien vers la fiche source] This dataset contains technical parameters (Voltage in V, current in mA, energy in Ah, internal pressure in mbar, water leak detection) acquired since September 2016 (n=332) by the COSTOF (COmmunication and STOrage Front-end), which is the Electronic core of SeaMoN East, the bottom station of EMSO-Azores dedicated to geophysical observations (data acquired every 6 hours).
Autonomous temperature probes data from the EMSO-Azores observatory, 2023-2024

[Lien vers la fiche source] This dataset contains temperature data acquired between july 2023 and august 2024 with an array of 29 autonomous temperature probes installed in smoker orifices, cracks or patches of diffuse hydrothermal venting. We used 5 types of probes (HTW, HTWN, HTNKE, LTW, and LTGrad). Probes locations, acquisition parameters and clock drift values (UTC-timeprobe) are listed in the metadata file. Data are provided for each deployed instrument as one or two text files, uncorrected for clock drift: temperature data
Hydrographic mooring data from EMSO-Azores observatory, 2010-2011

[Lien vers la fiche source] This dataset contains the data acquired with the hydrographic mooring of the EMSO-Azores observatory during the years 2010 and 2011. The data consist of temperature, salinity and pressure. There is one file per instrument. Instruments include: 3 solos (RBR) and 2 microcats (Seabird).
Autonomous temperature probes data from the EMSO-Azores observatory, 2021-2022

[Lien vers la fiche source] This dataset contains temperature data acquired between june 2021 and june 2022 with an array of 32 autonomous temperature probes installed in smoker orifices, cracks or patches of diffuse hydrothermal venting. We used 7 types of probes (HTW, HTWN, HTNKE, LTNKE, LTW, LTI and LTGrad). Probes locations, acquisition parameters and clock drift values (UTC-timeprobe) are listed in the metadata file. Data are provided for each deployed instrument as one or two text files, uncorrected for clock drift: temperature data.
Geochemistry of hydrothermal fluids at the Lucky Strike Hydrothermal Field data from the EMSO-Azores observatory, 2022

[Lien vers la fiche source] High temperature hydrothermal fluids are collected during the MOMARSAT 2022 cruise by gas-tight fluid sampler manipulated and triggered by the hydraulic arm of the submarine vehicle (HOV Nautile). Several hydrothermal vents are sampled, namely : Aisics, Montsegur, Montserrat, Isabel, Sintra, Y3, White Castle, Crystal, South Crystal, Capelinhos. Several chemical analyses are performed aboard the research vessel: such as salinity and d20/20 by refractometer, Conductivity, salinity and total dissolved solid (TDS) by electrode (standard solution at 1413 µS/cm), pH and Eh by electrode (pH standard 4.01 and 7; Eh standard solution at -220mV), iron concentration by photometer (up to 5 ppm), H2S and total S concentrations measured by electrode (AquaMS, amperometric method) at measured pH value… Regional Node: EMSO-Azores Vent field : Lucky Strike Latitude : 37º17’N Longitude : 32º16’W 1600 – 1750 m depth
Lucky Strike, EMSO-Azores observatory : mooring 2023 dataset

[Lien vers la fiche source] This dataset compiles mooring data collected at the Lucky Strike hydrothermal site, deployed from 24/06/2022 to 17/07/2023 as part of the EMSO-Azores observatory during the MomarSat Cruise. This experiment has been operated to provide long time-series of bottom currents and water column properties (position : 37°17.747' N 32°16.732' W). The dataset spans 815 meters from the seafloor and contains recordings of temperature, pressure, salinity and 3D velocity. The sampling intervals of the sensors range from 15 seconds to 10 minutes. The data was acquired with an Acoustic Doppler Current Profiler (ADCP Aquadopp V2.00.02), two SBE 37 Microcat probes (SBE37SM-RS232), nine RBRsolo, and one RBRduet T.D. The RBR sensors were quality controlled through an inter-comparison exercise using a SBE 19plus Seacat CTD. An overview of the experimental setup, a map of the mooring location, and a script to read the data are provided. The data is available in .cnv, .csv and .xlsx file formats, for compatibility with the pandas library in Python.
CHEMINI total dissolved iron concentrations from the EMSO-Azores observatory, 2011-2012

[Lien vers la fiche source] This dataset contains real-time total dissolved iron concentrations ((Fe(II) + Fe(III); µmol/l) acquired from the EMSO-Azores seafloor observatory TEMPO from July 2011 to March 2012 (n=1780). Data are purchased by the CHEMINI Fe (CF0), a CHEmical MINIaturized analyser dedicated to in situ determination of iron concentrations (every 12 hours, without in situ calibration, Vuillemin et al., 2009). The instrument is deployed on TEMPO part of Seamon East environmental monitoring node. The sample inlet is located on a mussel bed at the base of the Eiffel Edifice at 1695m depth.
Buoy management unit 2 data from the EMSO-Azores observatory, 2010-2011

[Lien vers la fiche source] This dataset contains technical parameters (Voltage in V, internal pressure in mbar, water intrusion detection, tilt in °) acquired between October 2010 and December 2010 by the buoy management unit 2, which is the Electronic core of BOREL, the transmission buoy of the EMSO-Azores observatory (data acquired every 6 hours).
Autonomous temperature probes, data from the EMSO-Azores observatory, 2016-2017

[Lien vers la fiche source] This dataset contains temperature data acquired between August 2016 and July 2017 using 22 temperature probes (HTW, HTWN, HTNKE and LTW) installed in smoker orifices or cracks. High-temperature WHOI probes (HTW) are composed of 2 J-type thermistors in the rod tip connected to 2 temperature data-loggers (named A & B), 2 data recordings are available for 1 HTW probe. They are able to record temperatures up to 415°C with an accuracy of ± 1.1°C, a resolution of 0.25°C at 375°C and a clock drift of 1min per month. High-temperature WHOI new probes (HTWN) are composed of 1 temperature data-logger and 1 attitude logger to measure absolute orientation (tilt) and acceleration. The thermocouple of their rod tip can be J or T-Type. Mostly used J-Type thermocouple is able to record temperatures up to 450°C, with an accuracy of ± 0.6°C, a resolution of 0,03°C and a clock drift of 1min/year. High-temperature NKE probes (HTNKE) are able to record temperatures up to 450°C, with an accuracy of ±0.5°C, a resolution of 0.1°C and a clock drift of 1min/month. Low-temperature WHOI probes (LTW) are able to record temperatures up to 125°C. with an accuracy of ±0.22°C, a resolution of 0.025°C and a clock drift of ±1min per month. Low-temperature NKE probes (LTNKE) are able to record temperatures up to 100°C, with an accuracy of ±0.5°C, a resolution of 0.12°C at 60°C and a clock drift of 1min per month. The array is not connected to an energy node. All probes are calibrated in time before deployment, probe time adjusted to UTC time. Temperature of probes is checked after recovery and before deployment. Clock drift after recovery is not implemented in data.
Buoy management unit 1 data from the EMSO-Azores observatory, 2010-2011

[Lien vers la fiche source] This dataset contains technical parameters (Voltage in V, internal pressure in mbar, water intrusion detection, tilt in °) acquired between October 2010 and December 2010 by the buoy management unit 1, which is the Electronic core of BOREL, the transmission buoy of the EMSO-Azores observatory (data acquired every 6 hours).
COSTOF2, processed data from high resolution GPS (LEICA GR25) on BOREL_V1 node, 2021-2022

[Lien vers la fiche source] This dataset contains processed data from the high resolution GPS (Leica GR25). This log is done by the COSTOF2 (COmmunication and STOrage Front-end). It is BOREL’s Electronic core that retrieves the data via satellite communication
COSTOF2, technical data from SeaMoN West, node of the EMSO-Azores observatory, 2017-2018

[Lien vers la fiche source] This dataset contains technical parameters (Voltage in V, current in mA, energy in Ah, internal pressure in mbar, water leak detection) acquired since September 2017 (n=332) by the COSTOF (COmmunication and STOrage Front-end), which is the Electronic core of SeaMoN West, the bottom station of EMSO-Azores dedicated to geophysical observations (data acquired every 6 hours).
Imposition Forfaitaire sur les Entreprises de Réseaux (IFER) par secteur d'activité - CA Saint-Quentin-en- Yvelines

Ventilation de l'Imposition Forfaitaire des Entreprises de Réseaux (IFER) perçue par la communauté d'agglomération de Saint-Quentin-en-Yvelines par secteur d'activité.
Buoy management unit 2 data from the EMSO-Azores observatory, 2014-2015

[Lien vers la fiche source] This dataset contains technical parameters (Voltage in V, internal pressure in mbar, water intrusion detection, tilt in °) acquired between July 2014 and April 2015 by the buoy management unit 2, which is the Electronic core of BOREL, the transmission buoy of the EMSO-Azores observatory (data acquired every 6 hours).
Hydrographic mooring data from EMSO-Azores observatory, 2014-2015

[Lien vers la fiche source] This dataset contains the data acquired with the hydrographic mooring of the EMSO-Azores observatory during the years 2014 and 2015. The data consist of temperature, salinity, pressureand velocity. There is one file per instrument. Instruments include: 10 solos (RBR), 2 microcats (Seabird) and 1 aquadopp (Nortek).
COSTOF2, data from AIS (MC MURDO KANATON3) on BOREL_V1 node, 2022-2023

[Lien vers la fiche source] This dataset contains data from the AIS ((KANATON3). It provides GPS informations (time, poisition, speed, direction) . This log is done by the COSTOF2 (COmmunication and STOrage Front-end). It is the Electronic core of BOREL the buoy that retrieve the data via satellite communication.
Autonomous temperature probes data from the EMSO-Azores observatory, 2020-2021

[Lien vers la fiche source] This dataset contains temperature data acquired between September 2020 and june 2021 with an array of 33 autonomous temperature probes installed in smoker orifices, cracks or patches of diffuse hydrothermal venting. We used 6 types of probes (HTW, HTWN, HTNKE, LTNKE, LTW and LTGrad). Probes locations, acquisition parameters and clock drift values (UTC-timeprobe) are listed in the metadata file.
Geochemistry of hydrothermal fluids at the Lucky Strike Hydrothermal Field data from the EMSO-Azores observatory, 2018

[Lien vers la fiche source] High temperature hydrothermal fluids are collected during the MOMARSAT 2018 cruise by gas-tight fluid sampler manipulated and triggered by the hydraulic arm of the submarine vehicle (HOV Nautile). Several hydrothermal vents are sampled, namely : Aisics, Montsegur, Montserrat, Isabel, Sintra, Y3, White Castle, Crystal, Capelinhos, Cimendeff, Tour Eiffel. Several chemical analyses are performed aboard the research vessel: such as salinity and d20/20 by refractometer, Conductivity, salinity and total dissolved solid (TDS) by electrode (standard solution at 1413 µS/cm), pH and Eh by electrode (pH standard 4.01 and 7; Eh standard solution at -220mV), iron concentration by photometer (up to 5 ppm), H2S and total S concentrations measured by electrode (AquaMS, amperometric method) at measured pH value. Regional Node: EMSO-Azores Vent field : Lucky Strike Latitude : 37º17’N Longitude : 32º16’W 1600 – 1750 m depth
COSTOF2, data from AIS (MC MURDO KANATON3) on BOREL_V1 node, 2021-2022

[Lien vers la fiche source] This dataset contains data from the AIS ((KANATON3). It provides GPS informations (time, poisition, speed, direction) . This log starts on june 07 2021, stops on june 11 2022 and is done by the COSTOF2 (COmmunication and STOrage Front-end). It is the Electronic core of BOREL the buoy that retrieve the data via satellite communication.
Geochemistry of hydrothermal fluids at the Lucky Strike Hydrothermal Field data from the EMSO-Azores observatory, 2017

[Lien vers la fiche source] High temperature hydrothermal fluids are collected during the MOMARSAT 2017 cruise by gas-tight fluid sampler manipulated and triggered by the hydraulic arm of the submarine vehicle (ROV Victor6000). Several hydrothermal vents are sampled, namely : Aisics, Montsegur, Montserrat, Isabel, Sintra, Y3, White Castle, South Crystal, Crystal, Capelinhos, Cimendeff, Tour Eiffel, Cypress, Sapin. Several chemical analyses are performed aboard the research vessel: such as salinity and d20/20 by refractometer, Conductivity, salinity and total dissolved solid (TDS) by electrode (standard solution at 1413 µS/cm), pH and Eh by electrode (pH standard 4.01 and 7; Eh standard solution at -220mV), iron concentration by photometer (up to 5 ppm), H2S and total S concentrations measured by electrode (AquaMS, amperometric method) at measured pH value. Regional Node: EMSO-Azores Vent field : Lucky Strike Latitude : 37º17’N Longitude : 32º16’W 1600 – 1750 m depth
Buoy management unit 1 data from the EMSO-Azores observatory, 2016-2017

[Lien vers la fiche source] This dataset contains technical parameters (Voltage in V, internal pressure in mbar, water intrusion detection, tilt in °) acquired since September 2016 by the buoy management unit 1, which is the Electronic core of BOREL, the transmission buoy of the EMSO-Azores observatory (data acquired every 6 hours).
Array of autonomous short period Ocean Bottom Seismometers from the EMSO-Azores observatory, 2011-2012

[Lien vers la fiche source] This dataset contains raw data, drift corrected data and SeisComP 3 Data Structure (SDS) from Ocean Bottom Seismometers (OBS) deployed at EMSO-Azores from July 2011 to July 2012. The raw data are initialy in Lcheapo format. They are fixed at first to remove bugs and they are transformed to miniSEED format. The drift corrected data are obtained from fixed raw data. The drift corrected data are obtained from fixed raw data. A value is put in the Time Correction eld (eld 16) of each data header. By default, the correction is applied to the Record Start Time (eld 8) and the Time Correction Applied ag (eld 12, bit 1) is activated. Alternatively, only the Time Correction Applied eld is changed. Seiscomp3 Data Structure (SDS) is the final product. It's the correct structure to be use in the Seiscomp 3 software.
CHEMINI total dissolved iron concentrations from the EMSO-Azores observatory, 2020-2021

[Lien vers la fiche source] This dataset contains dissolved iron concentrations ((Fe(II) + Fe(III); µmol/l) acquired between Septembre 2020 and November 2020 (n=90) using the CHEMINI Fe, a CHEmical MINIaturized analyser (samples taken every 24 hours, with weekly in situ calibration using a 20 µmol/l iron standard, Vuillemin et al., 2009). The sample inlet was positioned on the mussel bed at the base of the Tour Eiffel edifice of the Lucky Strike hydrothermal vent (1695 depth). CHEMINI Fe (CF2) was mounted on the TEMPO module which was itself connected to the SeaMON East node of the EMSO-Azores observatory.
Geochemistry of hydrothermal fluids at the Lucky Strike Hydrothermal Field data from the EMSO-Azores observatory, 2016

[Lien vers la fiche source] High temperature hydrothermal fluids are collected during the MOMARSAT 2016 cruise by gas-tight fluid sampler manipulated and triggered by the hydraulic arm of the submarine vehicle (ROV Victor6000). Several hydrothermal vents are sampled, namely : Aisics, Montsegur, Isabel, Sintra, Y3, White Castle, South Crystal, Crystal, Capelinhos, Cimendeff, Tour Eiffel, Cypress, Sapin. Several chemical analyses are performed aboard the research vessel: such as salinity and d20/20 by refractometer, Conductivity, salinity and total dissolved solid (TDS) by electrode (standard solution at 1413 µS/cm), pH and Eh by electrode (pH standard 4.01 and 7; Eh standard solution at -220mV), H2S and total S concentrations measured by electrode (AquaMS, amperometric method) at measured pH value… Regional Node: EMSO-Azores Vent field : Lucky Strike Latitude : 37º17’N Longitude : 32º16’W 1600 – 1750 m depth
Hydrographic mooring data from EMSO-Azores observatory, 2012-2013

[Lien vers la fiche source] This dataset contains the data acquired with the hydrographic mooring of the EMSO-Azores observatory during the years 2012 and 2013. The data consist of temperature, salinity and pressure. There is one file per instrument. Instruments include: 9 solos (RBR) and 1 microcat (Seabird).
Array of autonomous short period Ocean Bottom Seismometers from the EMSO-Azores observatory, 2015-2016

[Lien vers la fiche source] This dataset contains raw data, drift corrected data and SeisComP 3 Data Structure (SDS) from Ocean Bottom Seismometers (OBS) deployed at EMSO-Azores from April 2015 to August 2016. The raw data are initialy in Lcheapo format. They are fixed at first to remove bugs and they are transformed to miniSEED format. The drift corrected data are obtained from fixed raw data. The drift corrected data are obtained from fixed raw data. A value is put in the Time Correction eld (eld 16) of each data header. By default, the correction is applied to the Record Start Time (eld 8) and the Time Correction Applied ag (eld 12, bit 1) is activated. Alternatively, only the Time Correction Applied eld is changed. Seiscomp3 Data Structure (SDS) is the final product. It's the correct structure to be use in the Seiscomp 3 software.
COSTOF2, technical data from SeaMoN West, node of the EMSO-Azores observatory, 2016-2017

[Lien vers la fiche source] This dataset contains technical parameters (Voltage in V, current in mA, energy in Ah, internal pressure in mbar, water leak detection) acquired since September 2016 (n=332) by the COSTOF (COmmunication and STOrage Front-end), which is the Electronic core of SeaMoN West, the bottom station of EMSO-Azores dedicated to geophysical observations (data acquired every 6 hours).
Buoy management unit 1 data from the EMSO-Azores observatory, 2011-2012

[Lien vers la fiche source] This dataset contains technical parameters (Voltage in V, internal pressure in mbar, water intrusion detection, tilt in °) acquired between Jannuary 2011 and December 2011 by the buoy management unit 1, which is the Electronic core of BOREL, the transmission buoy of the EMSO-Azores observatory (data acquired every 6 hours).
Buoy management unit 1 data from the EMSO-Azores observatory, 2013-2014

[Lien vers la fiche source] This dataset contains technical parameters (Voltage in V, internal pressure in mbar, water intrusion detection, tilt in °) acquired between September 2013 and February 2014 (n=332) by the buoy management unit 1, which is the Electronic core of BOREL, the transmission buoy of the EMSO-Azores observatory (data acquired every 6 hours).