Mawson Tide Gauge Data 1992-2016

Over time there have been a number of tide gauges deployed at Mawson Station, Antarctica.  The data download files contain further information about the gauges, but some of the information has been summarised here.  Note that this metadata record only describes tide gauge data from 1992 to 2016.  More recent data are described elsewhere.
Tide Gauge 1 (TG001)
1992-03-05 - 1992-05-13
This folder contains monthly download files from the first deployment of a submerged tide gauge at Mawson in March 1992.
These files are ASCII hexadecimal files. They need to be converted to decimal.
The resultant values are absolute seawater pressures in mbar.
Tide Gauge 4 (TG004)
1993-03-22 - 1999-12-29
This folder contains the following folders:-
old_tidedata  
    monthly download files from the second deployment of a submerged tide gauge at Mawson in March 1993.
    These files are ASCII hexadecimal files. They need to be converted to decimal.
    The resultant values are absolute seawater pressures in mbar.
raw
    memory images from submerged tide gauge. file extension is memory bank number. 
    These files are processed by a utility called tgxtract.exe which creates files in same format as those in old_tidedata folder.
    These file have extension .srt. They are then converted to decimal pressure values.
interim
    files produced during processing of .raw files.
output
    output file (.srt) which have been sent to BoM.
Tide Gauge 13 (TG013)
2014-06-04 - 2016-11-04
Tide Gauge 20 (TG020)
1999-11-05 - 2009-12-21
This folder contains the following folders:-
raw
    memory images from submerged tide gauge. file extension is memory bank number. 
    These files are processed by a utility called tgxtract.exe which creates files in same format as original download format.
    These file have extension .srt. 
    These files are ASCII hexadecimal files. They need to be converted to decimal.
    The resultant values are absolute seawater pressures in mbar.
interim
    files produced during processing of .raw files.
output
    output file (.srt) which have been sent to BoM.
Tide Gauge 41 (TG041)
2008-03-02 - 2010-11-16
This folder contains the following folders:-
raw
    memory images from submerged tide gauge. file extension is memory bank number. 
    These files are processed by a utility called tgxtract.exe which creates files in same format as original download format.
    These file have extension .srt. 
    These files are ASCII hexadecimal files. They need to be converted to decimal.
    The resultant values are absolute seawater pressures in mbar.
interim
    files produced during processing of .raw files.
output
    output file (.srt) which have been sent to BoM.
Documentation from older metadata record:
Documentation dated 2001-03-26
Mawson Submerged Tide Gauge
The gauge used at Mawson was designed in 1991/2 by Platypus Engineering, Hobart, Tasmania.
It was intended to be submerged in about 7 metres of water in a purpose made concrete mooring in the shape of a truncated pyramid.
The gauge measures pressure using a Paroscientific Digiquartz Pressure Transducer with a full scale pressure of 30 psi absolute. The accuracy of the transducer is 1 in 10,000 of full scale over the calibrated temperature range.
The overall accuracy of the system is better than +/- 3 mm for a known water density.
Data is retrieved from the gauges by lowering a coil assembly on the end of a cable over a projecting knob on the top of the gauge and by use of an interface unit ,a serial connection can be established to the gauge. Time setting and data retrieval can be then achieved.
The first of these gauges were first deployed Mawson in early 1992 in a a mooring in Horseshoe Harbour. The gauge was found to have some communications problems and was removed in May 1992. Tidal records from 6/3/92 to present have been retrieved from it.
A new gauge was deployed at Mawson in March 1993.
Data has been retrieved from these gauges irregularly since then. The records are complete since deployment except for a few days in late 1995. The loss was caused by a fault in the software which allows directory entries to overwrites data when the directory memory has been filled.
The first gauge used at Mawson in 1992 was refitted with a higher pressure transducer and was later deployed at Heard Island in Atlas Cove.
Conversion of raw data to tidal records is done as detailed in document DATAFORMAT1.DOC .
As the current gauge is expected to require a new battery soon, a new mooring has been placed close to the original and a  new gauge has been deployed. 
Levelling
Several attempts have been made at precise levelling of the gauge.
The first was in the Summer of 1995/6. Roger Handsworth, Tom Gordon and Natasha Adams physically measured the level of the top of the gauge in its mooring and derived a reading when a known column of water was over the gauge. 
The next attempt was in the Summer of 1996/7 when Roger Handsworth and Paul Delaney made timed water level measurements close to the gauge and the tide gauge benchmark. From this work, and from tidal records, a value for MSL for Mawson was derived.
Permanent Gauge
In the summer of 1995/6 two possible sites for a permanent Aquatrak type tide gauge were identified.
As neither of these sites were approved, a survey in the Summer of 1996/7 identified two more suitable sites. One of these, the site at the base of East arm, near the Variometer Building, was approved and a bore hole was drilled to exit about 6 metres below MSL.  A power cable was run from the variometer building to provide two phase 240V power to the site.
A heated borehole liner containing an Aquatrak wave guide and a Druck pressure transducer was inserted into the bore hole. Two datalogger will be added to the installation in 2001 to complete the installation.
A radio modem will be used to link the dataloggers to the AAD network.
Documentation dated 2008-10-17
Mawson
A new submerged gauge ,TG41, was deployed at Mawson on 2008-03-03.
Submerged Tide gauge TG20 was removed on 2008-08-26.
There is a useful overlap of data between the gauges of about 104 days.
The dataloggers used in the shored based tide gauge installation have been replaced with Campbell Scientific CR1000 dataloggers.
The aquatrak shore based gauge at Mawson has not been operating since march 2008.
The shore base pressure gauge is still operating.

Tags: AMD, AMD/AU, CEOS, CONTINENT > ANTARCTICA, EARTH SCIENCE > OCEANS > TIDES, EARTH SCIENCE > OCEANS > TIDES > TIDAL HEIGHT, EARTH SCIENCE > OCEANS > TIDES > TIDAL RANGE, FIXED OBSERVATION STATIONS, GEOGRAPHIC REGION > POLAR, OCEAN > SOUTHERN OCEAN, TIDE GAUGES
Organization: Australian Ocean Data Network